This protocol outlines a procedure for collecting and analyzing point spread functions (PSFs). It describes how to prepare fluorescent microsphere samples, set up a confocal microscope to properly collect 3D confocal image data of the microspheres and perform PSF measurements. The analysis of the PSF is used to determine the resolution of the microscope and to identify any problems with the quality of the microscope's images. The PSF geometry is used as an indicator to identify problems with the objective lens, confocal laser scanning components and other relay optics. Identification of possible causes of PSF abnormalities and solutions to improve microscope performance are provided. The microsphere sample preparation requires 2–3 h plus an overnight drying period. The microscope setup requires 2 h (1 h for laser warm up), whereas collecting and analyzing the PSF images require an additional 2–3 h.
Subscribe to Journal
Get full journal access for 1 year
only $8.25 per issue
All prices are NET prices.
VAT will be added later in the checkout.
Tax calculation will be finalised during checkout.
Rent or Buy article
Get time limited or full article access on ReadCube.
All prices are NET prices.
Pawley, J.B. Handbook of Biological Confocal Microscopy 2nd ed. (Plenum Press, 1995).
Goodwin, P.C. Evaluating optical aberration using fluorescent microspheres: methods, analysis, and corrective actions. Methods Cell Biol. 81, 397–413 (2007).
Airy, G.B. On the diffraction of an object-glass with circular aperture. Trans. of the Cambridge Philosoph. Soc. 5, 283–291 (1835).
Wilhelm, S., Gröbler, B., Gluch, M. & Heinz, H. in Carl Zeiss Principles (Carl Zeiss, 1997).
Beyer, H. Handbuch der Mikroskopie 2nd ed. (VEB Verlag Technik, 1985).
Gibson, S.F. & Lanni, F. Experimental test of an analytical model of aberration in an oil-immersion objective lens used in three-dimensional light microscopy. J. Opt. Soc. Am. A 9, 154–166 (1992).
Hiraoka, Y., Sedat, J.W. & Agard, D.A. Determination of three-dimensional imaging properties of a light microscope system. Partial confocal behavior in epifluorescence microscopy. Biophys. J. 57, 325–333 (1990).
Goodwin, P.C. in Digital Microscopy 3rd ed., Vol. 81 (eds. Sluder, G. & Wolf, D.E.) 397–413 (Elsevier, 2007).
Brown, C.M. et al. Raster image correlation spectroscopy (RICS) for measuring fast protein dynamics and concentrations with a commercial laser scanning confocal microscope. J. Microsc. 229, 78–91 (2008).
Digman, M.A., Brown, C.M., Horwitz, A.R., Mantulin, W.W. & Gratton, E. Paxillin dynamics measured during adhesion assembly and disassembly by correlation spectroscopy. Biophys. J. 94, 2819–2831 (2008).
Digman, M.A. et al. Measuring fast dynamics in solutions and cells with a laser scanning microscope. Biophys. J. 89, 1317–1327 (2005).
Digman, M.A. et al. Fluctuation correlation spectroscopy with a laser-scanning microscope: exploiting the hidden time structure. Biophys. J. 88, L33–L36 (2005).
Wiseman, P.W. et al. Spatial mapping of integrin interactions and dynamics during cell migration by image correlation microscopy. J. Cell Sci. 117, 5521–5534 (2004).
Wallace, W., Schaefer, L.H. & Swedlow, J.R. A workingperson's guide to deconvolution in light microscopy. Biotechniques 31, 1076–1078, 1080, 1082 passim (2001).
Swedlow, J.R. Quantitative fluorescence microscopy and image deconvolution. Methods Cell Biol. 81, 447–465 (2007).
Swedlow, J.R. Quantitative fluorescence microscopy and image deconvolution. Methods Cell Biol. 72, 349–367 (2003).
Swedlow, J.R. & Platani, M. Live cell imaging using wide-field microscopy and deconvolution. Cell Struct. Funct. 27, 335–341 (2002).
Cole, R.W. & Turner, J.N. Light-emitting diodes are better illumination sources for biological microscopy than conventional sources. Microsc. Microanal. 14, 243–250 (2008).
Lacoste, J., Vining, C., Zuo, D., Spurmanis, A. & Brown, C.M. Optimal conditions for live-cell microscopy and raster image correlation spectroscopy (RICS). in Annual Reviews in Fluorescence 2010 (ed. Geddes, C.D.) (2011).
Koehler, A. New method of illumination for photomicrographical purposes. J. Royal Microscopical Soc. 14, 261–262 (1894).
Nyquist, H. Certain topics in telegraph transmission theory. Trans. AIEE 46, 617–644 (1928).
We acknowledge the use of the Wadsworth Center's Advanced Light Microscopy & Image Analysis Core Facility, as well as the McGill Life Sciences Complex Imaging Facility for portions of the work presented. We thank R. Stack for help with sample preparation, PSF image stack acquisition and data analysis. We thank B. Northan from Media Cybernetics for generating the theoretical PSF images for Figure 1, C. Glowinski for preparing 3D PSF figures using the Bitplane Imaris software for Figure 2, A. Spurmanis for providing objective lens cleaning images for Figure 3, as well as F. Waharte and M. Thibault for providing Supplementary Methods on Nikon and Olympus confocals, respectively. We thank B. Eason and K. Young for critical reading of the manuscript.
The authors declare no competing financial interests.
About this article
Cite this article
Cole, R., Jinadasa, T. & Brown, C. Measuring and interpreting point spread functions to determine confocal microscope resolution and ensure quality control. Nat Protoc 6, 1929–1941 (2011). https://doi.org/10.1038/nprot.2011.407
An Improved Target Method to Quantitatively Measure the Lateral Resolution of the Confocal Raman Microscope
Journal of Applied Spectroscopy (2021)
Scientific Reports (2021)
Optics Express (2021)
Technical considerations in the Verasonics research ultrasound platform for developing a photoacoustic imaging system
Biomedical Optics Express (2021)
Estimation of point spread function of an optical microscope using stochastic minimization of least square errors
Journal of Optics (2020)