We wanted to implement an NGS strategy to globally analyze hereditary cancer with diagnostic quality while retaining the same degree of understanding and control we had in pre-NGS strategies. To do this, we developed the I2HCP panel, a custom bait library covering 122 hereditary cancer genes. We improved bait design, tested different NGS platforms and created a clinically driven custom data analysis pipeline. The I2HCP panel was developed using a training set of hereditary colorectal cancer, hereditary breast and ovarian cancer and neurofibromatosis patients and reached an accuracy, analytical sensitivity and specificity greater than 99%, which was maintained in a validation set. I2HCP changed our diagnostic approach, involving clinicians and a genetic diagnostics team from panel design to reporting. The new strategy improved diagnostic sensitivity, solved uncertain clinical diagnoses and identified mutations in new genes. We assessed the genetic variation in the complete set of hereditary cancer genes, revealing a complex variation landscape that coexists with the disease-causing mutation. We developed, validated and implemented a custom NGS-based strategy for hereditary cancer diagnostics that improved our previous workflows. Additionally, the existence of a rich genetic variation in hereditary cancer genes favors the use of this panel to investigate their role in cancer risk.
Genetic diagnostic laboratories testing for cancer predisposition have rapidly integrated next-generation sequencing (NGS) technologies into their diagnostic workflows1,2,3,4. More genes have gradually been included to solve problems like genetic heterogeneity or overlapping clinical manifestations among distinct cancer predisposition syndromes5,6,7,8,9,10. Today, gene panels represent a good compromise between testing just a few genes and obtaining information from the whole exome for routine hereditary cancer testing11,12,13. Comprehensive guidelines, recommendations and standards have been published to guide the development, validation and implementation of NGS-based clinical genetic testing15,16,17,18,19,20,21,22. This material has aided the adaptation of NGS to the standardized frameworks developed for clinical genetic tests23. Despite these advances, for a genetic diagnostic laboratory mastering pre-NGS techniques, the biggest challenge when adopting NGS is to maintain the same degree of understanding and control over the whole process of detecting and interpreting genetic changes22,24.
The diagnostic activity of the ICO-IMPPC Joint Program for Hereditary Cancer focuses on the detection and interpretation of all inherited genetic variants that confer a higher risk of developing cancer. This activity encompasses all types of hereditary cancer (HC) syndromes, although we work mainly with hereditary colorectal cancer (Familial Adenomatous Polyposis, FAP, and Hereditary Non-Polyposis Colorectal Cancer, HNPCC), hereditary breast and ovarian cancer (HBOC) and neurofibromatoses Type 1 and Type 2 (NF1, NF2) and related disorders such as RASopathies and Phakomatoses. Due to genetic heterogeneity, clinical heterogeneity and overlapping clinical manifestations, diagnostic activity in the field of hereditary cancer requires multiple gene testing12,25,26.
To cover diagnostic necessities, homogenize diagnostic procedures for different conditions and preserve understanding and control over the complete workflow, we developed a comprehensive custom NGS-based diagnostic strategy to be implemented in the routine diagnostics scenario applicable to most of the genes involved in hereditary cancer and related disorders, similarly to other initiatives (Mainstreaming Cancer Genetics)27. Our aim was to develop a tool and a diagnostic strategy that would ensure the desired sequencing quality, provide great flexibility in the bioinformatic analysis to enhance clinical utility, and enable us to gather global data on the genetic variation of hereditary cancer.
Development and validation of a hereditary cancer NGS panel
The I2HCP diagnostics NGS panel design and set up
We have developed the ICO-IMPPC Hereditary Cancer Panel (I2HCP), which comprises 122 genes associated with cancer predisposition syndromes and RASopathies (Supplementary Table S1) and a custom data analysis pipeline. All genes had been previously identified as germline-mutated in relation to hereditary cancer (Cancer Census, OMIM or Orphanet). Combining all genes of interest in a single panel simplifies and unifies laboratory procedures in a single workflow when testing for the different conditions. The sequencing results are then filtered during the bioinformatic analysis and only selected genes are analyzed on the basis of the clinical indications (Fig. 1a).
To comply with the exhaustive analysis needed for diagnostics, the I2HCP regions of interest (ROIs) spanned all protein coding regions and intron-exon boundaries (−35/+20 bp), including all regions usually left untargeted for technical reasons such as repeats and homologous regions. Sufficient coverage was sought to ensure that all bases within ROIs were covered at a minimum of 30x. In a first approach, a training set (n = 23, see Materials and Methods and Supplementary Table S2) was used to evaluate the performance of a first version of the SureSelect bait library, consisting of 106 genes that were sequenced in an Ion Torrent PGM sequencer using a 318 chip, pooling 4 samples before enrichment. Data were analyzed using both CLC Genomics Workbench (Qiagen) and our custom NGS pipeline (Fig. 2). This approach showed a good capture yield and a low percentage of non-covered regions: 1.54 × 106 ± 0.5 × 106 SD reads per sample were produced, the mean depth of coverage was 490.7 ± 221.5 SD, and more than 86% ± 4% SD of the targeted ROI bases was covered ≥ 30x. 123 out of 136 pathogenic and non-pathogenic variants previously identified in the training set were correctly detected, providing a sensitivity of 90.4% (84.2–94.8%). However, this first approach failed to identify 7 out of 23 pathogenic variants, mostly due to their location within homopolymers (Supplementary Table S2).
On the basis of these initial results, two changes were made: a) To improve the percentage of well-covered regions, we studied coverage distribution and performed a systematic bait library redesign to increase the coverage of poorly covered regions (Supplementary Fig S1). In addition, we increased to 122 the number of genes in the panel (Supplementary Table S1); b) To improve variant calling quality, we analyzed all data using only our custom NGS pipeline, to ensure greater control over variant calling algorithms and parameters. We also decided to test the performance of a second sequencing platform: Illumina MiSeq.
Using the same training set of samples and the new bait library design, we set up a second approach based on the SureSelect XT protocol for Illumina, pooling 12 samples after enrichment and sequencing them in a Miseq 2 × 250 v2 cartridge (see Materials and Methods). Sequencing data were analyzed using our custom NGS pipeline (Fig. 2). 3.3 × 106 ± 0.83 × 106 SD paired reads were obtained per sample, the mean depth of coverage was 495 ± 170 SD and the coverage uniformity was of 0.31 ± 0.009 SD. More than 98.9% ± 0.7% SD of the targeted bases was covered ≥ 30× (Supplementary Fig S2). Detailed sequencing statistics are given in Supplementary Table S3. This new approach correctly identified 22 of the 23 pathogenic variants in the training set. The missing pathogenic mutation was located in the first exon of MSH6 gene and was not detected due to low coverage. As established in the I2HCP strategy (see implementation section), this exon was Sanger sequenced and the pathogenic variant identified (Supplementary Tables S2 and S4). In addition, all 113 previously identified non-pathogenic SNVs and indels were correctly detected (Supplementary Table S4). Taking into account all pathogenic and non-pathogenic variants previously identified in the training set, NGS data provided a sensitivity of 100% (97.3–100%). We also detected 78 additional SNV in the ROIs of the genes previously tested by different methodologies (CSCE, cDNA sequencing, etc.). Most of these variants were known SNPs and all of them were either homozygous or located in intronic regions and thus not detected by pre-NGS analysis (Supplementary Table S4). Overall, no false positive was identified, resulting in a specificity of 100% (98.3–100%) for the training set.
Data analysis pipeline for NGS-based genetic diagnostics
NGS data were processed using a custom diagnostics-oriented data analysis pipeline, based on standard tools (BWA, VarScan2 and Annovar) and an extensive set of custom R scripts (see Material and Methods and Supplementary Figure S4). By creating a custom pipeline we were able to include several specific quality controls and measures such as targeted bases not reaching diagnostic quality, global and exon coverage measures, specific measures of variant quality for difficult regions, etc. We also ensured full control over the software used: the algorithms, the software versions and their specific parameters, which are locked for each version of the pipeline, and the data sources used for the basic analysis (genome version, gene models…) and annotations (dbSNP, ClinVar…). All data were stored in a relational database and a custom pipeline management system was used to keep track of the analyses carried out. Finally, the full pipeline is under version control on our institution’s Git servers, enabling us to keep track of every modification and to revert to a previous version if necessary. In addition, in contrast to research-oriented NGS data analysis pipelines, our pipeline focuses on reducing false negatives at the expense of potentially increasing false positives, the latter of which would be uncovered in the Sanger variant validation step of the I2HCP strategy. Finally, we developed the first version of an exon-level copy-number calling algorithm based on successive steps of coverage depth normalization (see Material and methods). Although this is a preliminary version and not yet validated for diagnostics, it allowed us to correctly identify the true copy number changes in the training set (Supplementary Table S2) and therefore to increase the mutation detection yield.
After the I2HCP panel had been set up using the training set, an independent group of 40 samples was used as a validation set (Material and Methods; Fig. 2; Supplementary Table S5). These samples had previously been genetically characterized, with 36 found to have a known disease-causing mutation and 4 no pathogenic variant. We focused the variant analysis only on those genes tested in the previous genetic diagnostic workflows (Supplementary Table S1). The whole validation process was performed blindly. Each sample produced 3.1 × 106 ± 0.58 × 106 SD paired reads, the mean depth of coverage was 398 ± 138 SD, the coverage uniformity was of 0.31 ± 0.02 SD, and 98.4% ± 1.0% SD of the targeted bases was covered ≥ 30× (Fig. 3; Supplementary Figure S2; Supplementary Table S3), similar to the performance achieved in the training set. This new approach correctly identified all 36 pathogenic variants in the validation set, including substitutions, small insertions/deletions (up to 19 bp), and deletions of a single exon and a whole gene. No pathogenic variants were identified in the 4 samples that previously tested negative (see Supplementary Table S5). In addition, 180 out of 186 previously identified variants were detected (VUS and polymorphisms). Of the 6 missing variants, one was located in a low-coverage region of the MSH6 gene, and the remaining 5 in a highly complex region of PMS2. These regions are routinely re-analyzed by Sanger sequencing (see Implementation section; Supplementary Figure S3). Therefore, considering the 358 known variants in the training and validation sets, the analytical sensitivity of the panel was 98.4% (96.4–99.4%) and the analytical specificity was of 100% (99–100%). In addition, considering the whole set of 122 genes, within-run precision (repeatability) was 98% (97–98.7%) and between-run precision (reproducibility) was 95.7% (94.6–96.6%), both within the optimal range for diagnostic purposes (ref. 19; see Material and Methods for details).
Implementation into routine genetic testing: an I2HCP-based diagnostic strategy
Once the I2HCP had been validated, we modified the overall HC diagnostic strategy to unlock the potential of the new panel and integrated it into routine genetic testing (Fig. 1b). As in pre-NGS diagnostics, the clinical presentation of patients drives the diagnostic approach, thus a pre-test clinical evaluation of the hereditary cancer condition in question is required before initiating genetic testing. Next, sample preparation, NGS sequencing and data analysis are performed for the whole set of 122 genes; only genes with current clinical utility for each hereditary cancer condition are then further analyzed up to variant interpretation. The groups of genes with clinical utility for HBOC, FAP, HNPCC and neurofibromatosis have been pre-defined by clinicians and the genetic diagnostics team (Supplementary Table S1). The panel format provides the flexibility to select ad hoc groups of genes for other conditions or particular clinical presentations or to broaden the test, even after a first round of analysis. Any ROI of the analyzed gene set with a single base below 30x coverage is Sanger sequenced to ensure diagnostic quality. The total number of Sanger sequencing required is low and the regions involved are recurrent (Supplementary Table S6 and Figure S3). We also use Sanger sequencing as an independent technique to validate all reportable variants.
Since the I2HCP diagnostic strategy was implemented into our laboratory routine, more than 150 samples have been processed. In general, quality parameters have improved with respect to the validation set (3.2 × 106 ± 1.5 × 106 SD paired reads; mean depth 452 ± 225 SD; uniformity 0.31 ± 0.02 SD; C30 99.2 ± 0.7 SD). At the same time, the custom development of I2HCP has provided the plasticity required for continuous development and rapid updates. Since its implementation in routine diagnostics, the sequencing kit has changed from v2 (2 × 250 cycles) to v3 (2 × 300 cycles), the number of genes has increased to 126 (including the genes LZTR1, PIK3CA, RASA2 and GRB2), the number of samples per MiSeq run has increased from 12 to 16, and multiple upgrades in variant annotation have been made.
Despite the excellent analytical sensitivity and specificity it provides, the I2HCP NGS panel does not consistently detect some types of pathogenic variants present in the mutational spectrum of these genes and syndromes (e.g., deep intronic mutations, insertion of repetitive elements, etc.). Thus, other techniques such as mRNA analysis could be integrated to complement the I2HCP in particular cases. In addition, for some conditions, constitutional copy-number alterations (CNA) account for a considerable percentage of disease-causing mutations. A clear example is neurofibromatosis type 1, in which about 7% of mutations in the NF1 gene are due to total gene deletions or intragenic CNAs28. Analysis of the CNAs of specific genes is part of the I2HCP diagnostic strategy and is mainly carried out by MLPA, depending on the condition and clinical status of each patient. Although the I2HCP is not yet validated for routine genetic diagnostics, it can be used for CNA analysis. Since the I2HCP strategy was introduced, 100 samples have been complemented by MLPA analysis of different genes (171 MLPA tests). We have also analyzed the presence of CNAs in these 100 cases by applying an exon-level copy-number calling algorithm as part of the I2HCP pipeline. Ninety-nine samples have been analyzed (1 sample, corresponding to 3 MLPA analyses, did not reach the minimum quality criteria for analysis). Of the remaining 168 CNAs analyses, 159 were negative and 9 positive (8 whole gene deletions and one two-exon deletion). In all cases, MLPA and I2HCP results were concordant (Supplementary Table S7).
The overall analysis, including genetic variant interpretation and complementary tests, is summarized in a pre-report that is evaluated by a multidisciplinary team. Depending on the results and the patient’s clinical presentation, clinicians may request the analysis of additional genes in the I2HCP panel. Since NGS data are already available and processed, the analysis can start at the gene-filtering step of the pipeline. Finally, a report compliant with EuroGentest recommendations is generated and clinicians perform a post-test clinical evaluation concerning genetic counseling and clinical management.
The new genetic testing strategy improves HC diagnostics
One of the questions raised by the enhanced testing potential of the I2HCP-based strategy is whether retrospective re-analysis is warranted for particular cases that tested negative in previous diagnostic workflows or for uncertain clinical presentations that did not fully meet testing criteria. As a pilot study, we selected a group of 14 cases for re-analysis with the new strategy (Material and methods; Table 1). This group comprised 2 HBOC, 10 HNPCC positive for the Amsterdam criteria, 1 Schwannomatosis and 1 patient with clinical suspicion of NF1. In all cases, in addition to the respective clinically-driven gene lists (Supplementary Figure S1), we analyzed all 122 genes in the I2HCP. As seen in Table 1, the re-analysis enabled us to solve one case with an uncertain clinical diagnosis: a whole CDKN2A deletion causing melanoma and neural system tumor syndrome (OMIM: 155755) was identified in the patient with clinical suspicion of NF1 (Sample R2) (Supplementary Figure S5). The re-analysis also allowed us to increase the sensitivity of mutation detection by analyzing the pre-defined sets of genes (e.g.: Samples R1 and R4). Finally, the analysis of the whole I2HCP identified variants in new genes with potential impact on the clinical phenotype (e.g., samples R3, R5 and R6).
Variation landscape of hereditary cancer genes in hereditary cancer patients
I2HCP contains the majority of genes currently associated with hereditary cancer. The creation of a comprehensive panel was intended primarily to facilitate diagnostic activity, but it was also designed to facilitate the compilation of all genetic variants present in genes associated with cancer predisposition. Thus, it was possible to generate a global view of the constitutional variation present in these genes from the raw data produced for genetic testing. This opens the possibility of studying not only the disease-causing mutation but also the contribution of this variation to the presentation and evolution of the disease. Figure 4 shows the variation landscape of the genes in I2HCP for the 63 individuals in the training and validation sets. In particular, it contains the variants present in coding regions and canonical splice sites with a minor allele frequency (MAF) below 1%.
The global analysis revealed a complex variation landscape that coexists with the disease-causing mutation, including nonsense, frameshift, splicing, missense and synonymous variants. The number of variants per individual ranges from 1 to 9, with an average of 5, and presents no differences among HC conditions. Similarly, not taking into account the disease-causing mutation, there is diversity in the number of variants per gene, with the top 10% of genes accounting for around 40% of the genetic variation. In 27 cases a patient had more than one variant in the same gene, accounting for 55 out of 278 variants present. In 8 of the 27 cases these multiple variants accumulated in the disease-causing gene (data not shown).
Although this variation landscape represents only a small number of HC patients, some notable observations can be made. For instance, although tuberous sclerosis patients were not considered in this study, the TSC2 gene was among the genes that accumulated more variation, most of which had a very low MAF. Additionally, most nonsense, frameshift and splicing variants concentrate in DNA repair genes. Finally, there are a number of potentially pathogenic variants in genes not directly related to the phenotype that could be worth exploring. It would therefore be interesting to study the complexity of this variation landscape and its relationship with the presented cancer phenotypes in a larger sample.
The adoption of NGS technology by a routine genetic diagnostics laboratory presents various challenges29, among them the new competences required for both wet and dry labs and the complete reorganization of diagnostic activity. However, crucial to genetic testing is the ability to maintain understanding and control over the whole diagnostic workflow, to identify the potential limitations in the detection and interpretation of genetic variants and the implications for the diagnostic report. The prospective integration of new NGS methodologies and more complex bioinformatic analysis raised concerns about possible loss of overall control, so we decided to customize our NGS diagnostic strategy as far as possible. This was achieved by developing an NGS-based workflow comprising a panel of 122 HC genes and a clinically driven custom data analysis pipeline. We designed ROIs to comprehensively sequence all coding exons and desired intron-exon boundaries of hereditary cancer genes and RASopathies according to our diagnostic activity. The custom nature of I2HCP provides the plasticity for continuous and rapid updating. In the few months since the I2HCP strategy was implemented for routine diagnostics, the number of genes in the panel has increased, sequencing procedures and pipeline analysis have been updated and more samples per run can be tested.
According to Eurogentest NGS recommendations18 I2HCP can be considered a Type A test, given the high sensitivity and specificity achieved during the development and validation processes. However, it must be taken into account that the panel was set up by analyzing HBOC, FAP, HNPCC and neurofibromatosis patients. Although the quality parameters apply to all genes tested, for genes and conditions with particular mutational spectrums or testing complexities a specific analysis of I2HCP performance could be required.
The custom panel provides great flexibility, allowing for the analysis of pre-established gene sets for particular clinical conditions but also the testing of specific genes on demand (up to the entire set of 122 genes) in those cases with distinct clinical presentations. The customization of the panel and its comprehensive nature foster interplay and dialog between clinicians and the genetic diagnostics team for panel design, evaluation of results, the possible need to analyze additional genes, and reporting. In addition, the results of the re-analysis pilot study show the potential of applying the new strategy to previous negative tests and to patients who do not fully meet clinical criteria.
The I2HCP diagnostic strategy compensates for the limitations of the panel by integrating Sanger sequencing of complex and low-coverage regions, mRNA analysis when required and CNA analysis by MLPA. The I2HCP CNA detection algorithm has shown a 100% concordance with MLPA results in the 168 assessable tests performed so far. Robust development of this algorithm could substantially reduce the number of MLPAs required for diagnostics.
The global analysis of all genes in the panel revealed a complex variation landscape that coexists with the known disease-causing mutation. The number of variants identified per individual and the frequency of variants per gene were consistent with previous reports using other panels11. The study of this global variation and the systematic collection of genetic and phenotypic data in a higher number of patients could provide the evidence needed to establish additional genes as conferring cancer predisposition and to make reliable risk estimates for patient management and counseling30,31,32.
In summary, we developed and validated a custom NGS-based diagnostic strategy for hereditary cancer and implemented it into our routine diagnostic activity. The mutation detection rate increased, while maintaining control over the whole process. Complete analysis of the 122 hereditary cancer genes tested revealed a complex variation landscape that coexists with the disease-causing mutation. Analysis of this landscape in a higher number of patients could help to better estimate individual cancer risk.
This study was approved by the IMPPC scientific direction and carried out in accordance with IMPPC guidelines. Signed informed consent was obtained from all participants. Genomic DNA from 73 unrelated individuals clinically diagnosed with a cancer predisposition syndrome was obtained from blood lymphocytes using standard protocols. The 73 samples were grouped in three different sample sets: training, validation and re-analysis. The training set comprised 23 samples from patients who met the clinical criteria for HBOC (n = 7), FAP (n = 2), HNPCC (n = 7), NF1 (n = 6) and NF2 (n = 1). The validation set comprised 13 HBOC patients, 7 FAP, 11 HNPCC, 5 NF1, 2 NF2 and 1 patient who met the clinical criteria for Schwannomatosis, for a total of 40 samples. These samples contained 222 variants (36 pathogenic plus 186 VUS and polymorphisms), which was sufficient to determine specificity and sensitivity with a 95% CI of width <1.3%23. All samples had been genetically tested using pre-NGS workflows and methods (cDNA and DNA Sanger sequencing, conformation-sensitive capillary electrophoresis (CSCE), denaturing high-performance liquid chromatography (DHPLC) and multiplex ligation-dependent probe amplification (MLPA)). These samples presented a broad mutational spectrum, including single-nucleotide variants (SNVs), small insertions and deletions and multiple exon deletions, many of them located in complex sequence contexts (Supplementary Figure S6). Finally, the re-analysis set consisted of 14 individuals: 9 HNPCC positive for the Amsterdam criteria, 1 NF1 and 4 patients from the validation set with no pathogenic mutation as detected by pre-NGS testing workflows (Supplementary Table S1). Moreover, the global results for the 150 patients genetically diagnosed using I2HCP were compared to the training and validation sets. The testing criteria for these patients were based on current international clinical criteria guidelines33. Samples were codified and mutation analysis was performed blindly for the validation set.
We used Agilent eArray to design our SureSelect bait library V1 (Agilent, California, USA), covering 106 genes for a total of 0.45 Mb. For each gene, we defined the ROIs as all coding exons and intron/exon boundaries (−35/+20 bp) of all translated isoforms according to NCBI human genome build 37 (GRCh37) and the Ensembl release 67. All ROIs were exhaustively covered with capture baits. A second design was developed after studying the behavior of V1 baits to improve capture results. In short, we monitored the coverage of all exons, identified problematic regions, performed a systematic rebalancing of baits and increased bait tiling using custom algorithms to improve capture of poorly covered areas and to enhance coverage uniformity (Supplementary Figure S1). The bait library V2 included 122 genes responsible for most of the cancer predisposition syndromes and RASopathies (Supplementary Table S1), for a total of 0.5 Mb of targeted DNA.
Sample preparation and sequencing
DNA was sonicated using a Covaris S2 (Covaris, Woburn, MA, USA). Sample preparation was performed following the SureSelect XT protocol for Ion Torrent or MiSeq. In the first approach, samples were enriched with bait library V1 after combining 4 equimolar indexed samples (pre-capture pooling) and sequenced in a PGM (Ion Torrent) 318 chip with One Touch 200 DLv2 template reagents and Ion PGM 200 sequencing reagents. In the second approach, 12 samples were enriched with bait library V2 according to the manufacturer’s instructions (Agilent) with minor modifications34, pooled after capture and sequenced in a MiSeq (Illumina) with Reagent Kit v2, 2 × 250.
Validation by Sanger Sequencing
For the subset of genes analyzed for a patient, any ROI with at least one base below 30x was Sanger sequenced using standard protocols (primer sequences available upon request). 30x minimum coverage was established as per De Leener et al.35. Reportable variants (all pathogenic variants and VUS only for some syndromes) were also validated by Sanger sequencing. Human Genome Variation Society (www.hgvs.org) nomenclature guidelines were used to name the mutation at the DNA level and the predicted resulting protein.
NGS data were processed using a custom data analysis pipeline based on standard tools. In short, fastq files were mapped against the GRCh37 human genome assembly corresponding to Ensembl release 6736 using BWA mem37 and a sorted bam file was created with samtools38. Exhaustive coverage metrics were produced using a combination of bedtools39 and custom R and Bioconductor40 scripts. Variants, including substitutions and small insertions and deletions, were called using VarScan241 with the following parameters: –min-coverage 10 –min-reads2 2 –min-avg-qual 15 –min-var-freq 0.1 –strand-filter 0. Finally, variants were annotated with a combination of annovar42 and custom scripts. bam-readcount was then used to compute additional quality parameters (quality of the bases supporting the reference and alternative alleles, position of the changes in the reads, number of mismatches present in reads supporting the alternative allele, etc.) that were combined to produce a set of variant quality indicators to guide the variant validation process. Final variant annotation included: basic variant quality parameters, gene and transcript annotations and effects, presence in variation databases (dbSNP and ClinVar), population frequencies (1000 G, ExAC and ESP6500) and in-silico prediction of effects in protein function (Polyphen2, SIFT, MutationAssessor, MutationTaster, PROVEAN, CADD). All information, including coverage metrics, variants, variant quality estimators and annotations, was stored in a PostgreSQL database. Variants were filtered using the regioneR bioconductor package43 according to the clinical indication prior to generating the variant lists. The preliminary algorithm for exon-level copy-number estimation includes different normalization steps on the exon mean depth of coverage (including mean sample coverage and mean exon coverage across samples), similar to Kang et al.44, and was run using a set of custom scripts. CLC Genomics Workbench v6 (Qiagen) was used to analyze data from I2HCP V1. Repeatability was measured by comparing the variants called for four samples prepared and sequenced twice under the same conditions. Reproducibility was calculated for twelve samples prepared once and sequenced twice in independent runs. In both cases, the concordance was computed as the number of common variants divided by the total number of variants identified. 95% confidence intervals were computed using the exact method from Hmisc R package. Uniformity was defined as the percentage of bases with a coverage within ± 20% of the mean coverage. C30 was defined as the percentage of ROI bases with coverage ≥30x.
How to cite this article: Castellanos, E. et al. A comprehensive custom panel design for routine hereditary cancer testing: preserving control, improving diagnostics and revealing a complex variation landscape. Sci. Rep. 7, 39348; doi: 10.1038/srep39348 (2017).
Publisher's note: Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Bosdet, I. E. et al. A clinically validated diagnostic second-generation sequencing assay for detection of hereditary BRCA1 and BRCA2 mutations. J Mol Diagn 15, 796–809, doi: 10.1016/j.jmoldx.2013.07.004 (2013).
De Leeneer, K. et al. Massive parallel amplicon sequencing of the breast cancer genes BRCA1 and BRCA2: opportunities, challenges, and limitations. Hum Mutat 32, 335–344, doi: 10.1002/humu.21428 (2011).
Feliubadalo, L. et al. Next-generation sequencing meets genetic diagnostics: development of a comprehensive workflow for the analysis of BRCA1 and BRCA2 genes. Eur J Hum Genet 21, 864–870, doi: 10.1038/ejhg.2012.270 (2013).
Michils, G. et al. Molecular analysis of the breast cancer genes BRCA1 and BRCA2 using amplicon-based massive parallel pyrosequencing. J Mol Diagn 14, 623–630, doi: 10.1016/j.jmoldx.2012.05.006 (2012).
Chong, H. K. et al. The validation and clinical implementation of BRCAplus: a comprehensive high-risk breast cancer diagnostic assay. PLoS One 9, e97408, doi: 10.1371/journal.pone.0097408 (2014).
Couch, F. J. et al. Inherited mutations in 17 breast cancer susceptibility genes among a large triple-negative breast cancer cohort unselected for family history of breast cancer. J Clin Oncol 33, 304–311, doi: 10.1200/JCO.2014.57.1414 (2015).
De Leeneer, K. et al. Flexible, scalable, and efficient targeted resequencing on a benchtop sequencer for variant detection in clinical practice. Hum Mutat 36, 379–387, doi: 10.1002/humu.22739 (2015).
Judkins, T. et al. Development and analytical validation of a 25-gene next generation sequencing panel that includes the BRCA1 and BRCA2 genes to assess hereditary cancer risk. BMC Cancer 15, 215, doi: 10.1186/s12885-015-1224-y (2015).
Tung, N. et al. Frequency of mutations in individuals with breast cancer referred for BRCA1 and BRCA2 testing using next-generation sequencing with a 25-gene panel. Cancer 121, 25–33, doi: 10.1002/cncr.29010 (2015).
Walsh, T. et al. Detection of inherited mutations for breast and ovarian cancer using genomic capture and massively parallel sequencing. Proc Natl Acad Sci USA 107, 12629–12633, doi: 10.1073/pnas.1007983107 (2010).
Kurian, A. W. et al. Clinical evaluation of a multiple-gene sequencing panel for hereditary cancer risk assessment. J Clin Oncol 32, 2001–2009, doi: 10.1200/JCO.2013.53.6607 (2014).
LaDuca, H. et al. Utilization of multigene panels in hereditary cancer predisposition testing: analysis of more than 2,000 patients. Genet Med 16, 830–837, doi: 10.1038/gim.2014.40 (2014).
Rehm, H. L. Disease-targeted sequencing: a cornerstone in the clinic. Nat Rev Genet 14, 295–300, doi: 10.1038/nrg3463 (2013).
Feliubadaló, L. et al. Benchmarking of Whole Exome Sequencing and Ad Hoc Designed Panels for Genetic Testing of Hereditary Cancer. Sci. Rep. 6, 37984, doi: 10.1038/srep37984 (2016).
Deans, Z., Watson, C. M., Charlton, R. et al. Practice guidelines for targeted next generation sequencing analysis and interpretation. (Date of acces: 23/04/2016) http://www.www.acgs.uk.com/media/774807/bpg_for_targeted_next_generation_sequencing_may_2014_final.pdf. (2014).
Ellard, S., Charlton, R. . Lindsay, H. et al. Practice guidelines for targeted next generation sequencing analysis and interpretation (Date of acces: 23/04/2016) http://www.cmgs.org/BPGs/best_practice_guidelines.htm (2012).
Gargis, A. S. et al. Assuring the quality of next-generation sequencing in clinical laboratory practice. Nat Biotechnol 30, 1033–1036, doi: 10.1038/nbt.2403 (2012).
Matthijs, G. et al. Guidelines for diagnostic next-generation sequencing. Eur J Hum Genet 24, 2–5, doi: 10.1038/ejhg.2015.226 (2016).
Rehm, H. L. et al. ACMG clinical laboratory standards for next-generation sequencing. Genet Med 15, 733–747, doi: 10.1038/gim.2013.92 (2013).
Robson, M. E. et al. American Society of Clinical Oncology Policy Statement Update: Genetic and Genomic Testing for Cancer Susceptibility. J Clin Oncol 33, 3660–3667, doi: 10.1200/JCO.2015.63.0996 (2015).
Robson, M. E., Storm, C. D., Weitzel, J., Wollins, D. S. & Offit, K. American Society of Clinical Oncology policy statement update: genetic and genomic testing for cancer susceptibility. J Clin Oncol 28, 893–901, doi: 10.1200/JCO.2009.27.0660 (2010).
Weiss, M. M. et al. Best practice guidelines for the use of next-generation sequencing applications in genome diagnostics: a national collaborative study of Dutch genome diagnostic laboratories. Hum Mutat 34, 1313–1321, doi: 10.1002/humu.22368 (2013).
Mattocks, C. J. et al. A standardized framework for the validation and verification of clinical molecular genetic tests. Eur J Hum Genet 18, 1276–1288, doi: 10.1038/ejhg.2010.101 (2010).
Hastings, R. et al. The changing landscape of genetic testing and its impact on clinical and laboratory services and research in Europe. Eur J Hum Genet 20, 911–916, doi: 10.1038/ejhg.2012.56 (2012).
Desmond, A. et al. Clinical Actionability of Multigene Panel Testing for Hereditary Breast and Ovarian Cancer Risk Assessment. JAMA Oncol 1, 943–951, doi: 10.1001/jamaoncol.2015.2690 (2015).
Lincoln, S. E. et al. A Systematic Comparison of Traditional and Multigene Panel Testing for Hereditary Breast and Ovarian Cancer Genes in More Than 1000 Patients. J Mol Diagn 17, 533–544, doi: 10.1016/j.jmoldx.2015.04.009 (2015).
Rahman, N. Mainstreaming genetic testing of cancer predisposition genes. Clin Med (Lond) 14, 436–439, doi: 10.7861/clinmedicine.14-4-436 (2014).
Messiaen, L. M. & Wimmer, K. In Neurofibromatoses Vol. 16 Monogr Hum Gene t (ed Kaufmann, D. ) 63–77 (Karger, 2008).
Vrijenhoek, T. et al. Next-generation sequencing-based genome diagnostics across clinical genetics centers: implementation choices and their effects. Eur J Hum Genet 23, 1142–1150, doi: 10.1038/ejhg.2014.279 (2015).
Bowdin, S., Ray, P. N., Cohn, R. D. & Meyn, M. S. The genome clinic: a multidisciplinary approach to assessing the opportunities and challenges of integrating genomic analysis into clinical care. Hum Mutat 35, 513–519, doi: 10.1002/humu.22536 (2014).
Easton, D. F. et al. Gene-panel sequencing and the prediction of breast-cancer risk. N Engl J Med 372, 2243–2257, doi: 10.1056/NEJMsr1501341 (2015).
Stadler, Z. K., Schrader, K. A., Vijai, J., Robson, M. E. & Offit, K. Cancer genomics and inherited risk. J Clin Oncol 32, 687–698, doi: 10.1200/JCO.2013.49.7271 (2014).
Pagon, R. A., Adam, M. P., Ardinger, H. H. et al. editors. GeneReviews (University of Washington, 1993–2016 Available from: http://www.ncbi.nlm.nih.gov/books/NBK1116/) (Date of acces: 23/04/2016).
Neiman, M. et al. Library preparation and multiplex capture for massive parallel sequencing applications made efficient and easy. PLoS One 7, e48616, doi: 10.1371/journal.pone.0048616 (2012).
De Leeneer, K. et al. Practical tools to implement massive parallel pyrosequencing of PCR products in next generation molecular diagnostics. PLoS One 6, e25531, doi: 10.1371/journal.pone.0025531 (2011).
Yates, A. et al. Ensembl 2016. Nucleic Acids Res 44, D710–716, doi: 10.1093/nar/gkv1157 (2016).
Li, H. Aligning sequence reads, clone sequences and assembly contigs with BWA-MEM. arXiv.org arXiv:1303.3997v2 [q-bio.GN] (Date of acces: 23/04/2016) (2013).
Li, H. et al. The Sequence Alignment/Map format and SAMtools. Bioinformatics 25, 2078–2079, doi: 10.1093/bioinformatics/btp352 (2009).
Quinlan, A. R. & Hall, I. M. BEDTools: a flexible suite of utilities for comparing genomic features. Bioinformatics 26, 841–842, doi: 10.1093/bioinformatics/btq033 (2010).
Huber, W. et al. Orchestrating high-throughput genomic analysis with Bioconductor. Nat Methods 12, 115–121, doi: 10.1038/nmeth.3252 (2015).
Koboldt, D. C., Larson, D. E. & Wilson, R. K. Using VarScan 2 for Germline Variant Calling and Somatic Mutation Detection. Curr Protoc Bioinformatics 44, 15 14 11-15 14 17, doi: 10.1002/0471250953.bi1504s44 (2013).
Wang, K., Li, M. & Hakonarson, H. ANNOVAR: functional annotation of genetic variants from high-throughput sequencing data. Nucleic Acids Res 38, e164, doi: 10.1093/nar/gkq603 (2010).
Gel, B. et al. regioneR: an R/Bioconductor package for the association analysis of genomic regions based on permutation tests. Bioinformatics 32, 289–291, doi: 10.1093/bioinformatics/btv562 (2016).
Kang, H. P. et al. Design and validation of a next generation sequencing assay for hereditary BRCA1 and BRCA2 mutation testing. PeerJ 4, e2162, doi: 10.7717/peerj.2162 (2016).
We thank the IGTP Genomics Core Facility (Pilar Armengol, Anna Oliveira) and the IGTP HPC Core Facility (Iñaki Martinez de Ilarduya). We also thank Ana Beatriz Sanchez and Patricia Barrero for their collaboration. This work has been supported by: the Spanish Association Against Cancer (AECC); the IMPPC; the Spanish Ministry of Science and Innovation, Carlos III Health Institute (ISCIII) (PI11/1609; PI14/00577)(RTICC RD12/0036/008) Plan Estatal de I + D + I 2013–2016, and co-financed by the FEDER program; and the Government of Catalonia (2014 SGR 338). CL, ES and IB acknowledge the support of the Spanish (Asociación de Afectados de Neurofibromatosis) and Catalan (ACNefi) Neurofibromatosis Patient Associations.
The authors declare no competing financial interests.
About this article
Cite this article
Castellanos, E., Gel, B., Rosas, I. et al. A comprehensive custom panel design for routine hereditary cancer testing: preserving control, improving diagnostics and revealing a complex variation landscape. Sci Rep 7, 39348 (2017). https://doi.org/10.1038/srep39348
BARD1 Pathogenic Variants Are Associated with Triple-Negative Breast Cancer in a Spanish Hereditary Breast and Ovarian Cancer Cohort
Mutational spectrum by phenotype: panel‐based NGS testing of patients with clinical suspicion of RASopathy and children with multiple café‐au‐lait macules
Clinical Genetics (2020)
A comprehensive custom panel evaluation for routine hereditary cancer testing: improving the yield of germline mutation detection
Journal of Translational Medicine (2020)
Diagnostics of Mutations in MMR/EPCAM Genes and Their Role in the Treatment and Care of Patients with Lynch Syndrome
The Journal of Molecular Diagnostics (2020)