Abstract
Most commercial microfluidic droplet generators rely on the planar flowfocusing configuration implemented in polymer or glass chips. The planar geometry, however, suffers from many limitations and drawbacks, such as the need of specific coatings or the use of dedicated surfactants, depending on the fluids in play. On the contrary, and thanks to their axisymmetric geometry, glass capillarybased droplet generators are a priori not fluiddependent. Nevertheless, they have never reached the market because their assembly requires fastidious and not scalable fabrication techniques. Here we present a new device, called Raydrop, based on the alignment of two capillaries immersed in a pressurized chamber containing the continuous phase. The dispersed phase exits one of the capillaries through a 3Dprinted nozzle placed in front of the extraction capillary for collecting the droplets. This nonembedded implementation of an axisymmetric flowfocusing is referred to nonembedded coflowfocusing configuration. Experimental results demonstrate the universality of the device in terms of the variety of fluids that can be emulsified, as well as the range of droplet radii that can be obtained, without neither the need of surfactant nor coating. Additionally, numerical computations of the NavierStokes equations based on the quasisteadiness assumption allow to provide an explanation to the underlying mechanism behind the drop formation and the mechanism of the dripping to jetting transition. Excellent predictions were also obtained for the droplet radius, as well as for the drippingjetting transition, when varying the geometrical and fluid parameters, showing the ability of this configuration to enventually enhance the dripping regime. The monodispersity ensured by the dripping regime, the robustness of the fabrication technique, the optimization capabilities from the numerical modelling and the universality of the configuration confer to the Raydrop technology a very high potential in the race towards highthroughput droplet generation processes.
Similar content being viewed by others
Introduction
In recent years, droplet microfluidics^{1,2,3,4,5} has become an important tool for many different applications, including fundamental studies on emulsification^{6}, crystallization^{7} and chemical reaction^{8}, temperaturecontrolled tensiometry^{9}, molecules encapsulation^{10}, particle synthesis^{11}, biochemical assays^{12}, immunoassays^{13}, digital PCR^{14}, directed evolution^{15}, drug discovery^{16}, singlecell analysis^{17} or cell and gene manipulations^{18,19}. Nowadays, all the commercially available and most of labmade droplet generators are based on a flowfocusing technology implemented in rectangular microchannels fabricated by lithography, and made of polydimethylsiloxane (PDMS), polymers or glass. However, this planar configuration has many limitations, mostly due to the contact between the walls of the microchannels and both phases at the junction, requiring laborious and often ephemeral wettability treatments of these walls. On the contrary, due to their axisymmetric configuration, glass capillary systems do not have this drawback since the dispersed phase is never in contact with the walls of the outer capillary^{20}. Yet their widespread use is limited by the difficulty to implement this technology in an easytouse device. And even if coflow configurations have been set up using commercially available components^{21} (see Fig. 1a,b), these devices produce large droplets (\(>100~\upmu\)m) at low throughput and are unable to generate small (\(< 100~\upmu\)m) and monodisperse droplets at high throughput (\(>1\) kHz), as realised in planar flow focusing configuration. Furthermore, the centering of the capillary into an outer flow capillary is challenging. Even though the flow focusing configuration has also been designed using glass capillaries by inserting two circular capillaries into a square outer flow capillary, which greatly simplifies centering of the capillaries^{22} (see Fig. 1c,d), three main limitations remain. Firstly, the restrictions necessary to obtain a focusing effect at the tip of the inner injection and extraction capillaries are obtained by pulling and breaking a heated glass capillary, a very artdependent technique limiting the use of this design in largescale microfluidic applications. Secondly, the confined space for flowing the continuous phase around the inner capillaries limits its flow rate, hence the throughput, in the droplet generation process, particularly with highly viscous fluids. And finally, the interfacing with the tubing carrying the fluids is typically realised by gluing hypodermic needles over the embedded capillaries previously glued on a glass slide. This fabrication process not only limits the reuse of the device because it can not be disassembled and properly cleaned, but also make it not suitable for largescale production. Alternative procedures have been developed to improve the usability of embedded capillary devices^{23,24}, but none can overcome all these limitations.
A configuration proposed by Evangelio et al.^{25} has revealed a promising alternative by placing the extraction tube without any surrounding confinement in front of the injection tube (see Fig. 1e). The continuous phase is therefore accelerated to the extraction tube, thus creating a pressure drop, similar to a Venturi tube, even though viscous forces are dominant in this case. The authors have exploited this pressure drop to create a stretched jet of the dispersed phase, that further destabilizes into microbubbles or droplets, a mechanism which is referred to as the “tip streaming”. However, their system exclusively works in the jetting regime, which does not intrinsically guarantee the droplet monodispersity associated to the dripping regime, as realised in embedded glass capillaries^{22}, even though the polydispersity of droplet sizes obtained with the tip streaming configuration could be lower than the one obtained with other jetting mode configurations. In this work, we consider a new configuration based on the configuration proposed by Evangelio et al.^{25} but in which the dripping regime is enforced.
Our device relies on the alignment of two glass capillaries inside a pressurised chamber, similarly to Evangelio et al.^{25}. However, while their configuration relies on an inner diameter of the injection capillary larger than the one of the extraction capillary, such as it can stably operate in the jetting mode only, our configuration can operate in the dripping mode also because the injection capillary has a smaller diameter than the extraction one. This configuration could only be obtained thanks to the proper combination of cuttingedge machining and 3D printing techniques to fabricate a micronozzle connected at the tip of the injection capillary (see Fig. 2d) and align it with the extraction capillary, enforcing the dripping of small droplets as in Utada et al.^{22}. This nonembedded design presents both the characteristics of a coflow (axisymetric geometry) and a flowfocusing (dramatic local accelerations of the continuous phase), and is thereby called nonembedded coflowfocusing. As illustrated in Fig. 1f, this configuration fills the gap in the design array of microfluidic droplet generators. The technological breakthrough of this design is twofolds. Firstly, it enables the generation of monodisperse droplets, intrinsic to the dripping regime, at high throughput and for a wide variety of fluids, thanks to the fact that the continuous phase is not confined before entering the extraction capillary, allowing for flushing very viscous continuous phases. Secondly, it takes benefit of specific fabrication techniques and materials compatible with a largescale production of the device, while ensuring a very precise and reproducible alignment of the two capillaries in the chamber. Additionally, the device is made plugandplay thanks to the standard connections and the possibility to easily assemble and disassemble all parts for cleaning.
Besides the axisymmetric configuration proposed by Evangelio et al.^{25} in the jetting mode, one should mention the recent experiment performed by CruzMazo et al.^{26} that exploits the axisymmetric flowfocusing in the dripping mode. To our knowledge, this is the closest configuration to the one of the Raydrop. Yet these authors used exclusively gas as continuous phase and found that the liquid drop size is independent of the flow rate of the dispersed phase, a feature that suggests a quasistatic behavior, as exploited in the present work.
The ultimate aim in droplet generation being to predict the size of droplets, initial theoretical works have relied on linear stability analysis of the coflow configuration and found that the dripping and jetting regimes exhibit the main features of absolutely and convectively unstable flows, respectively^{20,27}. However, Cordero et al.^{28} have shown that the frequency selection in the dripping regime is not ruled by the absolute frequency predicted by the stability analysis. Thus, recognizing the nonlinear behaviour of droplet formation in the dripping regime, many numerical studies have been performed in the coflow configuration using the NavierStokes equations together with diverse methods to describe the interfacial dynamics, as for instance the finite volume method combined with continuoussurfaceforce method^{29}, front tracking method^{30} or levelset method^{31}. For the flowfocusing configuration in a crossjunction microchannel, one can mention the diffuse interface or phasefield method^{32}, or the work by Wu et al.^{33} using threedimensional lattice Boltzmann simulations. Nevertheless, computationally intensive threedimensional simulations and the number of geometrical parameters involved in a planar flowfocusing configuration have prevented so far systematic and complete numerical analysis of the drop formation in such a configuration. Using Finite Element Method (FEM) with adaptive meshing in a diffuseinterface framework, Zhou et al.^{34} have simulated an axisymmetric flowfocusing configuration, yet involving a large number of geometrical parameters. The nonembedded coflowfocusing (see Fig. 1e,f) presented here reduces the configuration to the minimum number of independent parameters, as compared to planar flowfocusing or embedded coflowfocusing devices (see Fig. 1c,d), then making more accessible an exhaustive parametric analysis. Yet the task is still huge and some additional simplifications are needed.
Several authors have recognised the quasistatic behavior of the dripping mode. For instance, Garstecki et al.^{35} claimed that the quasistatic character of their data collapse forms the basis for controlled, highthroughput generation of monodisperse fluid dispersions. In the stepemulsification configuration, Li et al.^{36} solved the quasistatic shape of an elongated drop using the 1D equations of the HeleShaw cell. Inspired by these quasistatic behaviors, and thinking about the coflowfocusing configuration, which is intrinsically axisymmetric, the analogy with the pendant droplet becomes evident. In this context, the quasistatic assumption applies, provided the liquid is injected at a sufficiently low flow rate into the droplet in formation. Boucher et al.^{37} for instance have shown that the dripping of a pendant droplet coincides with a folding bifurcation, indicating the maximum drop volume above which surface tension cannot sustain the drop weight anymore. But gravity can obviously be replaced by other forces, such as surface forces^{38}, or electric forces^{39}. We claim in this paper that the quasistatic approach can be adopted analogously to the pendant drop method, provided the flow rate of the dispersed phase is small enough and that the role of gravity is essentially played by the hydrodynamic forces acting on both phases and originated by the imposed continuous flow rate. As mentioned above, this situation has been experimentally considered by CruzMazo et al.^{26}, but only in the case of air as continuous phase. We here consider all situations in the quasistatic approach, from zero viscosity ratio corresponding to inviscid droplets, to infinite viscosity ratio corresponding to highly viscous droplets.
In the context of simulating multicomponents flows, interface tracking methods, such as FEM with moving mesh algorithm, are very accurate for simulating the onset of droplet breakup but have difficulties in simulating through and past the transitions as it requires cutandconnect procedures for the mesh^{40}. Assuming that the volume of the dispersed phase preceding the pinchoff corresponds exactly to the volume of the droplet after breakup, there is in fact no need to simulate the dynamics beyond the transition if one only wants to determine the volume of the generated droplet. Following this approach, MartinezCalvo et al.^{41} have applied FEM to investigate the breakup of liquid threads and determine the volume of satellite droplets, with an impeccable accuracy. They have solved the NavierStokes equations using the finite element software Comsol, and more precisely, the weak form Partial Differential Equation (PDE) and the Arbitrary LagrangianEulerian (ALE) modules. Note that the jet breaking is intrinsically transient as it results from the RayleighPlateau instability but the droplet generation, as mentioned above, can be quasistatic in the limit of low dispersed flow rate. Van Brummelen et al.^{42} having shown that solving the transient NavierStokes equation to obtain the steady solution is often inefficient, RiveroRodriguez et al.^{43} have developed the Boundary Arbitrary LagrangianEulerian (BALE) method to facilitate the solution of steady NavierStokes equations with free surfaces. RiveroRodriguez and Scheid^{44,45} have then applied this method to study the dynamics of deformable and offcentered bubbles in microchannels, allowing for exhaustive parametric analysis, taking the advantage of continuation methods suitable for tracking stationary solutions. The same approaches are undertaken in the present paper in which the unprecedented quasistatic simplification allows for an extended parametric analysis in the context of flowfocusing.
After detailing the Raydrop configuration in section Materials and methods, we show in section Experiments the experimental measurements of droplet size and frequency for two different geometries, with the aim to identify experimentally the dripping to jetting transition when increasing both the dispersed and the continuous flow rates. Section Modelling is dedicated to the model description and validation in regards to the experimental results, and especially the relevance of the quasistatic approach for determining the droplet radius in the dripping regime in the limit of small dispersed flow rate. Section Parametric analysis focuses on the quasistatic approach, first for inviscid droplets, then for viscous ones, followed by a parametric analysis of the geometrical parameters, and a discussion on the role of inertia for properly predicting the dripping to jetting transition when increasing the continuous flow rate. General conclusions and perspectives are given in the last section, Discussions and Conclusions.
Materials and methods
The droplet generator Raydrop is made of a metallic chamber filled with the continuous phase, in which two inserts supporting the glass capillaries are introduced on the lateral sides in such a way that the capillaries are perfectly aligned and almost in contact at the centre of the chamber, as shown in Fig. 2.
Two glass windows on the top and bottom faces of the device close the chamber and allow the observation of the droplets. Leakage from the chamber is prevented by the use of Orings seals on the windows and inserts. A nozzle is printed in the photoresin IPL with a submicrometric resolution using a 3Dprinter Photonic Professional GT from Nanoscribe company and then glued onto the tip of the injection glass capillary. Capillaries (Postnova Analytics) are coated with a polyimide film on the external diameter, ensuring a very high mechanical resistance. They are held into the inserts in such a way that for all combinations, alignment is guaranteed and a fixed gap between the nozzle and the extraction capillary is maintained. As demonstrated later, one can then change in a few minutes the couple of nozzle and extraction capillary in order to have access to other ranges of droplet radii, which represents a great advantage as compared to most of other glass capillary devices. Connections to the fluids inlets and outlets are ensured by standard Upchurch fittings directly screwed on the inserts for the connection to the capillaries and on the chamber for the continuous phase supply and bleed.
As shown in Fig. 2d, the input capillary supporting the nozzle provides the dispersed phase while the output capillary collects droplets carried by the continuous phase. At the entrance of the output capillary, the continuous phase pressurised in the chamber encounters a dramatic acceleration due to the change of section and therefore squeezes the dispersed phase flowing out of the nozzle, resulting in the formation of droplets (see Fig. 2e).
Fluids are injected in the device using a pressure controller (MFCSEZ, Fluigent) and the flow rates are measured using flowmeters (Flow Unit, Fluigent), as shown in Fig. 2f. A highspeed camera (MotionPro Y3, IDT) operating up to 10000 frames per second is connected to a microscope with a 10x magnification to visualise the droplets. Recorded images are then processed with a Python script to detect the contour of the droplets and determine their size.
Experiments
To demonstrate the operation of the device, we used two nozzles of tip radii \(R_n = 15\) or \(45~\upmu\)m, coupled with an extraction capillary of internal radius \(R_e= 75\) or \(225~\upmu\)m, respectively. The distance H between the nozzle tip and the extraction capillary is 50 or 75 \(\upmu\)m, respectively, the thickness e of the walls of the extraction capillary is 75 \(\upmu\)m and the external angle of the nozzle \(\alpha\) taken perpendicularly from the main flow axis is 50\(^\circ\). We used ultrapure water, obtained from a Sartorius water filtration system, as dispersed phase and light mineral oil, purchased from SigmaAldrich, as continuous phase. Both phases are filtered with PTFE 0.45 \(\upmu\)m syringe filters. The interfacial tension of the bare interface between the two fluids was measured to be 49.5 mN/m using the pendant drop method (tensiometer Kruss DSA100). Capillarybased devices do not necessitate the use of a surfactant to generate droplets, in both water in oil (W/O) and oil in water (O/W) cases. Even though surfactants can be used to stabilise the emulsion after the droplets are collected and stored, the literature mentions that its presence does not affect the droplet formation (see for instance Erb et al.^{46}). Indeed, adsorption times at the interface of the liquids are on a typical scale of tens of milliseconds whereas the droplet formation occurs in a few milliseconds. To validate this assumption in the case of our device, droplets generated with and without surfactant (Span 80, 2% w/w in the continuous phase) have been compared, with no difference. On the contrary, surfactants however do have an influence on the droplet formation in planar configuration, as they modify the contact angle at the triple line between the continuous phase, the disperse phase and the wall. This triple line could be of major nuisance in droplet formation. Fluid properties and geometrical parameters for two different couples of nozzle and extraction capillary mentioned above are listed in Table 1. Subscripts d and c refer to the dispersed and continuous phases, respectively, \(\mu\) being the dynamic viscosity, \(\gamma\) the interfacial tension and \(\rho\) the density of the fluids.
Considering the first couple of nozzle and extraction capillary (see dimensions in Table 1), for each value of the oil flow rate \(Q_c\) between 20 and 350 \(\upmu\)L/min, the flow rate of the dispersed water \(Q_d\) is increased until it reaches the drippingjetting transition. The results are shown in Fig. 3a where the size of the bullet is proportional to the size of the droplet and the color scale indicates the frequency of droplets generation.We observe with the geometry of couple 1 that droplets are generated in a range of radius R from 25 to 60 \(\upmu\)m with a high monodispersity (coefficient of variation: CV \(< 1\)%). For a given \(Q_d\), the droplet radius decreases by increasing \(Q_c\), while for a given \(Q_c\), it remains very stable as \(Q_d\) is increased, with a maximum radius variation of 15% at \(Q_c = 350 ~\upmu\)L/min. Additionally, as the system operates for \(Q_c \gtrsim 200\) \(\upmu\)L/min, the drippingjetting transition (dashed line in Fig. 3a) reaches a plateau indicating that the drippingjetting transition is only determined by the geometry in this region. Yet the droplet generation frequency increases to reach a maximum of 5500 Hz in this case. We have also tested the sensitivity of the device to small geometrical modifications such as a small misalignment of both capillaries, namely of about a fraction of the nozzle tip radius, i.e. \(<R_n/4\). The influence of the distance H between the nozzle tip and the extraction capillary has also been tested in the range 0 to 2.5 \(R_n\). In both tests, droplet radii have been found comparable to these obtained with the nominal values of the parameters.
With the geometry of couple 2 (see dimensions in Table 1), droplets can be generated in a range of radius from 60 to 300 \(\upmu\)m (see Fig. 3b). While only the dripping regime could be observed for couple 1, because of the limitation of the maximum pressure drop attainable, higher flow rates of the continuous phase \(Q_c\) could be reached with couple 2 corresponding to a less resistive geometry. Consequently, two drippingjetting transitions could be observed : as for couple 1, a drippingjetting transition at increasing \(Q_d\) (black dashed line in Fig. 3b) and another drippingjetting transition when \(Q_c\) is increased above a threshold value (red dashed line in Fig. 3b). As shown in Fig. 4, this second transition at high \(Q_c\) separates a monodisperse regime with CV \(< 1\)% from a polydisperse regime with CV \(> 4\)%, a value still much smaller than typical CV values observed in the jetting regime at sufficiently large \(Q_d\), usually larger than 10%. Indeed, in this jetting mode, the jet is uniform in radius and acts like a noise amplifier, therefore being more sensitive to external noise. But in the jetting regime at sufficiently large \(Q_c\), the jet is nonuniform as it is stretched in the flow direction, which has been shown by Gordillo et al.^{47} to have a stabilizing effect on the RayleighPlateau instability. This plays in favor of a mode selection at the tip of the jet, resulting in a lower polydispersity visible on the images of Fig. 4a where no polydisperse regime seems to be reached whatever the value of \(Q_c\), reminiscent to tip streaming regime^{25}. Actually, a statistical analysis of the diameter of a large number of droplets (\(>200\)) allows to highlight the steep transition in CV that occurs for \(Q_c\) at around \(5200\pm 500\) \(\upmu\)L/min and plotted in Fig. 4b. This transition is in good agreement with the prediction of the numerical model described below.
Modelling
With the goal to predict the droplet radius generated in the dripping regime using the Raydrop, we propose in this section to model the nonembedded coflowfocusing configuration in transient and then using the quasistatic approach. After a validation with the experimental results, we show how to determine the droplet size in the dripping regime, as well as the dripping to jetting transition, when varying the continuous flow rate alone.
Transient
Geometry of the nonembedded coflowfocusing configuration is considered in an axisymmetric coordinate system (r, z), with r and z the radial and axial coordinates, respectively, as sketched in Fig. 5.The properties and parameters for the two phases will be given with the generic subscript i referring to d and c for the dispersed and the continuous phases, respectively. Each phase has thus a density \(\rho _i\), a dynamic viscosity \(\mu _i\) and a flow rate \(Q_i\). The timedependent volumes of the two phases are denoted by \(\mathscr {V}_i\) and are separated by the surface \(\Sigma _m\), characterized by an interfacial tension \(\gamma\). The surface of the nozzle and the extraction capillary are denoted by \(\Sigma _n\) and \(\Sigma _e\), respectively. The domain considered is truncated at \(\Sigma _{\infty }\), \(\Sigma _{\mathrm{{in}}}\) and \(\Sigma _{\mathrm{{out}}}\), which are sufficiently far from the droplet generation zone to not affect the results. The dispersed flow rate \(Q_d\) is injected in the form of a fully developed Poiseuille flow through the capillary of the nozzle at \(\Sigma _{\mathrm{{in}}}\), and both fluids leave the device through the crosssection of the extraction capillary \(\Sigma _{\mathrm{{out}}}\) at a flow rate \(Q_c+Q_d\). In addition to the geometrical parameters already defined in Table 1, one can define \(L\) as the distance taken on the symmetry axis from the nozzle tip to the tip of the meniscus, as sketched in Fig. 5.
The motion of the fluids is governed by the continuity and the Navier–Stokes equations,
where \({{\mathbf{v}}}_i\) is the velocity vector and \(\mathbf {\tau }_i=p_i {\mathscr {I}}+ \mu _i \left[ \mathbf {\nabla }{{\mathbf{v}}}_i + \left( \mathbf {\nabla }{{\mathbf{v}}}_i \right) ^T \right]\) is the stress tensor of the phase i with \(p_i\) being the pressure field and \({\mathscr {I}}\) the identity matrix. Defining \({\mathbf{n}}_i\) as the outer normal to the domain \(\mathscr {V}_i\) at any of its boundaries, as sketched in Fig. 5, the nostress boundary condition at \(\Sigma _{\infty }\) writes
The flow being considered as fully developed (i.e. parallel flow), upstream of the nozzle as well as downstream of the extraction capillary, we have
where the pressures, \(P_d\) and \(P_c\), correspond to the ones imposing the given flow rates
with t the time. Noslip boundary conditions at the nozzle and at the extraction capillary are applied
The continuity of velocity and stress balance at the interface, which is pinned at the nozzle tip, writes
where \({\mathscr {I}}_S= {\mathscr {I}} {\mathbf{n}}_i {\mathbf{n}}_i\) is the surface identity tensor. The kinematic condition writes
where \(\mathbf {x}\in \Sigma _{m}\) and \(\partial _t \mathbf {x}\cdot {\mathbf{n}}_d\) represents the normal velocity of the interface.
The instantaneous equivalent radius of the meniscus \(R_{m}(t)\) is defined as the radius of a sphere of a volume equivalent to the one of the dispersed phase beyond the nozzle and bounded by the meniscus surface \(\Sigma _m\), namely
It should be noted that at the pinchoff time \(t_*\), the volume preceding the neck is the one corresponding to the droplet \(\frac{4}{3} \pi R^3\). In Fig. 6, the volumes of revolution reconstructed from the shaded areas are related to \(R_{m}\) and R. The meniscus shape and volume for an arbitrary time and for the pinchoff time are sketched in Fig. 6a,b, respectively. The instantaneous equivalent radius of the meniscus \(R_{m}(t)\) is timedependent and shadowed at both times, whereas the radius of the drop R is only defined at the pinchoff time \(t_*\), and is therefore not timedependent.
The equations are nondimensionalised using \(R_n\) for the length scale, \(\gamma / \mu _c\) for the velocity scale and \(\gamma /R_n\) for the pressure scale, leading to the following dimensionless flow parameters
with \({\mathrm{Ca}}_c\) the capillary number for the continuous phase, \(\bar{R}_e\) the radius ratio, \({\bar{Q}}\) the flow rate ratio, \(\lambda\) the viscosity ratio, \(\mathrm{{La}}\) the Laplace number and \(\phi\) the density ratio. Finally, the other geometrical parameters are all made dimensionless relative to the diameter of the nozzle tip:
The system of Eqs. (1)–(7) governs the droplet formation using the Raydrop. The domain variables are \(p_i\) and \({{\mathbf{v}}}_i\), the surface variable is \(\mathbf {x}\), and the global variables are \(P_i\). This system of equations is solved using the Finite Element Method (FEM) with the help of the software Comsol and quadratic Lagrangian elements, with exception of the pressure for which linear elements have been used. For the deformable domain, the moving mesh application mode has been used. It implements the Arbitrary LagrangianEulerian (ALE) method combined with the Boundary Arbitrary LagrangianEulerian (BALE) method previously developed in Ref.^{44}. In every simulation, the initial shape of the meniscus is half a sphere out of the nozzle. Transient simulations have been carried out using a firstorder backwardEuler time discretization until the pinchoff time. The criteria for the pinchoff is when the neck radius has reached 2% of the nozzle tip radius.
Finally, we note that most of the results presented below have been obtained with the Stokes equations, i.e. with the lefthand side of the momentum balance in Eq. (1) set to \(\mathbf{0}\). We have indeed checked numerically (not shown) that for the maximum Reynolds number corresponding to our experimental conditions (see Table 1), or equivalently for a maximum Laplace number \(\mathrm{{La}}\) of about 100, inertia has no influence on these results. Therefore, all results presented below are for \(\mathrm{{La}}=0\), unless specified otherwise.
Quasistatic approach
In general, the system behavior is transient but considering the quasistatic (QS) limit is of great interest. This limit corresponds to the situation of \(Q_d \rightarrow 0\), meaning that the flow of the dispersed phase is negligible, and thus set to zero, i.e. \(Q_d=0\). Consequently, the system of Eqs. (1)–(7) can be solved for stationary solutions, cancelling the timederivatives in Eqs. (1) and (7), and parametrizing the volume of the liquid meniscus with its quasistatic equivalent radius \(\bar{R}_{m}\), instead of using time t in Eq. (4b). This QS approach thus facilitates the parametrical analysis and allows to provide an explanation to the underlying mechanism behind the drop formation. By convenience, quasistatic simulations have been carried out using parametric continuation in \(\bar{L}\) instead of \({\mathrm{Ca}}_c\), this latter being considered as a simulation output. As for the transient simulation, the initial shape for the start of the continuation procedure is assumed to be half a sphere. But afterwards, at each step of the procedure the initial shape is taken by continuation on the parameter L¯ from the previous solution. Hence the quasistatic solution does not depend on the initial guess for the droplet shape.
For comparison purposes, we show in Table 2 the correspondence between the equivalent radii obtained from the transient and the quasistatic approaches.
Model validation
The model in the Stokes limit computed first in transient is validated by comparison with experimental data, as shown in Fig. 7. Figure 7a displays the superposition of the droplet shape with a circle of radius R (green line) obtained from the equivalent radius of the meniscus \(R_m\) predicted by the model for couple 1. In Fig. 7b, the interfacial shape of a drop in formation is shown just before the pinchoff of the meniscus for couple 2. In both cases, an excellent agreement is observed. In Fig. 7c,d, the experimental and numerical drop sizes are compared for a wide range of \({\mathrm{Ca}}_c\), showing again an excellent agreement for the two geometries (couples 1 and 2). It can be observed that as the continuous flow rate increases, the generated drop becomes smaller. The shaded area in Fig. 7d corresponds to the drippingjetting transition identified experimentally in Fig. 4 in the range \(0.23 \lesssim {\mathrm{Ca}}_d \lesssim 0.28\). Finally, note that \(Q_d\) has been kept small enough to ensure \({\bar{Q}} \ll 1\), such as it should have no influence on the droplet size, then suggesting that the QS limit is applicable.
In order to verify the range of validity of the QS limit, Fig. 8a shows the predictions of the droplet size in transient for various values of \({\mathrm{Ca}}_d\), as well as in the QS limit as represented by the black solid line, denoted as \(\bar{R}^{\mathrm{QS}}\) and effectively obtained for \({\mathrm{Ca}}_d = 4\times 10^{6}\pi ^{1}\).This limit lies slightly below the numerical curve plotted in Fig. 7c, as represented by the orange area in Fig. 8a for \(\log _{10}(\pi {\mathrm{Ca}}_d) \approx 0.67\). Figure 8a shows that for increasing values of \({\mathrm{Ca}}_d\), the system leads to a monotonous (yet logarithmic) increase of the drop size, as compared to the QS limit prediction.
In order to understand how the flow rate of the continuous phase affects the size of the drop as a function of the geometry and of the properties of the given pair of fluids, the comparison between the transient and the quasistatic solutions is further investigated. In Fig. 8b, the dynamics of the meniscus shape predicted from a quasistatic simulation (\({\bar{Q}} = 0\)), as well as from transient simulations of several values of \({\bar{Q}}\) is presented for a given value of \({\mathrm{Ca}}_c\).
Parameters \(\bar{L}\) and \(\bar{R}_{m}\) were chosen as representative quantities of the meniscus shape and are plotted as trajectory lines following the time during the formation process. One notices that the lines diverge as \({\bar{Q}} \rightarrow 0\). Remarkably, solving the model using the quasistatic simulation for exactly \({\bar{Q}} = 0\) depicts the existence of a turningpoint corresponding to a folding bifurcation. This is analogous to the detachment of a pendant drop as discussed in the introduction and further developed below. We note \(\bar{R}_{m}^D\) the value of \(\bar{R}_{m}\) at the turningpoint, as labeled in Table 2.
In the transient simulations with finite values of \({\bar{Q}}\), it can be observed (i) that the meniscus is less elongated for larger \(\bar{Q}\) and given \(\bar{R}_{m}\), especially close to the turningpoint, which naturally leads to larger droplets as shown in Fig. 8a, and (ii) that larger menisci, i.e. \(\bar{R}_{m}>\bar{R}_{m}^D\), can be attained for finite dispersed flow rates. At that stage, \(\bar{L}\) exhibits a large increase for a small increase of \(\bar{R}_{m}\), due to the elongation concentrated in the neck region shown in Fig. 7b, i.e. the menisci evolve towards the pinchoff, leading to the formation of a drop of size \(\bar{R}\).
In Fig. 9, the flow field and meniscus shape are represented at two different times. While in Fig. 9a the volume exhibits a quasisteady meniscus since \(\bar{R}_{m}(\bar{t}_1)<\bar{R}_{m}^D\), in Fig. 9b the volume is no longer in equilibrium since \(\bar{R}_{m}(\bar{t}_2) > \bar{R}_{m}^D\), leading to the formation of a neck. On the one hand, it can be observed that, for a quasistatic meniscus, streamlines are tangent to the meniscus revealing the quasistatic character of the flow, i.e. the righthand side term of Eq. (7) is negligible and thus set to zero, i.e. \(Q_d=0\). On the other hand, for larger volumes, the quasistatic meniscus no longer exists and it exhibits a neck which eventually breaks for later times. In this situation, the meniscus evolves dynamically as revealed by the streamlines which are no longer tangent to the meniscus, and the righthand side of Eq. (7) is not negligible anymore, even in the limit of \({\bar{Q}} \rightarrow 0\). Finally, the transient solution shown in Fig. 9c for a larger \({\mathrm{Ca}}_c\) depicts the shape of a jet with streamlines again parallel to the interface, thus suggesting a quasistatic solution like in Fig. 9a. As it will be shown below, this solution corresponds to the jetting regime in the quasistatic limit, occurring at a \({\mathrm{Ca}}_c\) larger than the transition value, denoted by \({\mathrm{Ca}}_c^*\).
Parametric analysis
Results obtained so far support the quasistatic approach for two reasons: (i) theoretically, the occurrence in Fig. 8b of a folding bifurcation in this limit allows for transition tracking between dripping and jetting modes; (ii) experimentally, the size of the droplets shown in Fig. 3 barely varies with \(Q_d\) at constant \(Q_c\), suggesting that size predictions in the limit of \(Q_d \rightarrow 0\) should also apply for finite \(Q_d\), which then only influences the frequency of droplet generation, at least to some extent. In this section, quasistatic simulations are therefore used as a predictive tool to evaluate the influence of fluid properties and geometric parameters on the drop formation.
Inviscid droplets
Starting with the inviscid droplet limit, i.e. \(\lambda =0\), we plot in Fig. 10a the quasistatic meniscus length \(\bar{L}\) as a function of \({\mathrm{Ca}}_c\) for constant equivalent radii of the meniscus \(\bar{R}_{m}\) (blue lines), the arrows indicating the decrease of \(\bar{R}_{m}\). It can be observed that in some range of \({\mathrm{Ca}}_c\) corresponding to large values of \(\bar{R}_{m}\), the curves of constant \(\bar{R}_{m}\) exhibit a turningpoint. The loci of the turningpoints are then depicted by the red curve, as determined by \(\partial {\mathrm{Ca}}_c / \partial \bar{L}_{\bar{R}_{m}} = 0\). The shadowed region above this curve is therefore the unstable region. Consequently, a dripping occurs at a fixed \({\mathrm{Ca}}_c\) as the volume of the meniscus is quasistatically increased up to the red curve corresponding to \(\bar{R}_{m}=\bar{R}_{m}^D\). Quasistatic solution no longer exists for higher volumes, as already shown in Fig. 8b for \({\mathrm{Ca}}_c=0.1\) and \(\lambda =1/23\).
Notably, the locus of turning points stops at \({\mathrm{Ca}}_c^*\) (red cross in Fig. 10a), thus representing the maximum \({\mathrm{Ca}}_c\) for which the dripping regime in the QS limit occurs. Indeed, as a matter of fact, this maximum no longer exists on the right of the locus of the inflectionpoint (green curve), determined by \(\partial ^2 {\mathrm{Ca}}_c / \partial \bar{L}^2 _{\bar{R}_{m}} = 0\). Quasistatic and transient simulations show that for \({\mathrm{Ca}}_c>{\mathrm{Ca}}_c^*\), a jet is established as was shown in Fig. 9. In Fig. 10a, and for a fixed \({\mathrm{Ca}}_c > {\mathrm{Ca}}_c^*\), as the volume of the meniscus increases, the tip of the jet advances and the jet is formed behind. Remarkably, the form of the jet does not vary behind the tip, which recalls the steady shape of the fluid cone in the tip streaming process^{25,48}.
In Fig. 10b, the folding bifurcation locus is shown in the (\(\bar{R}_{m}^D\),\({\mathrm{Ca}}_c\))plane. It can be observed that the equivalent radius is smaller for larger values of the continuous flow rate, as the viscous forces exerted by the continuous phase on the meniscus is larger, inducing the detachment of a smaller droplet. This recalls the analogy with the pendant droplet, provided the hydrodynamic forces are substituted to the gravity forces. We also plot in blue dashed line the equivalent radius \(\bar{R}^{\mathrm{QS}}\) corresponding to the volume of revolution preceding the neck, as obtained in transient in the quasistatic limit, i.e. for \({\mathrm{Ca}}_c \rightarrow 0\) (see Table 2). As \({\mathrm{Ca}}_c\) increases, the position of the neck moves from the nozzle at \({\mathrm{Ca}}_c \rightarrow 0\) to the tip of the meniscus at \({\mathrm{Ca}}_c^*\), for which the neck disappears. This explains the dramatic decrease of \(\bar{R}^{\mathrm{QS}}\) when approaching the jetting transition, even though it only occurs for inviscid droplets and is therefore not really physical. Indeed, a finite viscosity of the dispersed phase is found to regularise this singularity (see Fig. 8a).
Now, for lower \({\mathrm{Ca}}_c\), \(\bar{R}^{\mathrm{QS}}\) is shown to follow the same behavior than \(\bar{R}_{m}^D\), even though it remains smaller because of the volume that remains attached to the nozzle behind the neck. Consequently, it confirms that except near the drippingjetting transition, the value \(\bar{R}_{m}^D\) is well representative of the drop radius determined in transient in the quasistatic limit, but without relying on the pinchoff formation.
Finally, it is worth mentioning that the equivalent radius of the meniscus for dripping in Fig. 10b can be correlated by
The prefactor and mostly the exponent are similar to those found by Lan et al.^{31} for dripping in a coflow configuration, namely \(\bar{R}=1.07 \, {\mathrm{Ca}}^{0.34}Re_d^{0.06}\) with \(Re_d=2\rho _dQ_d/(\pi \mu _d R_n)\) the Reynolds number of the dispersed flow. Apart from the slight influence of inertia, i.e. \(Re_d^{0.06}\approx 1.2\), not included in Eq. (11) since obtained for a Stokes flow and \(\lambda =0\), it shows that the drop size predictions in the dripping mode are comparable in coflow and coflowfocusing.
Viscous droplets
Next, the influence of the viscosity ratio \(\lambda\) on the meniscus equivalent radius is considered. Each blue line in Fig. 11 corresponds to a constant meniscus equivalent radius \(\bar{R}_{m}^D\) in the (\(\lambda\),\({\mathrm{Ca}}_c\))plane, while the black arrow indicates the decrease of the meniscus volume. We observe that, as \(\lambda\) increases, smaller values of \({\mathrm{Ca}}_c\) are needed to generate a drop of the same size. This can be explained by the fact that for inviscid droplet, only the normal component of the viscous stresses exerted at the interface contributes to the net force responsible for the droplet detachment, while, as \(\lambda\) increases, the tangential (or shear) component is added to the force, provoking the detachment of smaller droplets. For the same reason, the drippingjetting transition occurs for larger values of \({\mathrm{Ca}}_c^*\) as \(\lambda\) increases, as shown by the red curve in Fig. 11, which starts from the value of \({\mathrm{Ca}}_c^*=0.188\) in the limit of inviscid droplets, to large and possibly asymptotically infinite value for viscous droplets corresponding to \(\lambda \gtrsim 5\times 10^{2}\). It thus appears that beyond this value, the viscous shear stress exerted on the droplet is always large enough to ensure the dripping mode.
Practically, the influence of \(\lambda\) on \({\mathrm{Ca}}_c\) for the dripping mode can be described by a correlation involving the limits for \(\lambda \rightarrow 0\) and \(\lambda \rightarrow \infty\),
where \({\mathrm{Ca}}_{c,0}\) and \({\mathrm{Ca}}_{c,\infty }\) are the \({\mathrm{Ca}}_c\) numbers that produce menisci of size \(\bar{R}_{m}^D\) in both limits, respectively, whereas the transition is centered at \(\lambda _*\). These functions have been obtained numerically (see Supplementary Information), the correlations of which are \({\mathrm{Ca}}_{c,0}\approx 0.5952 (\bar{R}_{m}^D)^{2.936}\) (as inverted from Eq. (11)), \({\mathrm{Ca}}_{c,\infty }\approx 0.1666 (\bar{R}_{m}^D)^{2.496}\) and \(\lambda _* \approx 0.09497(\bar{R}_{m}^D)^{1.609}\), applicable in the range \(1.5 \lesssim \bar{R}_{m}^D \lesssim 3.5\). Using these correlations in Eq. (12) allows to estimate the dimensionless droplet size \(\bar{R}_{m}^D\) for a given viscosity ratio \(\lambda\) and a given capillary number of the continuous phase \({\mathrm{Ca}}_c\), provided the geometry verifies the dimensionless parameters corresponding to couple 1, namely \({\bar{R}}_e=5\), \({\bar{H}}=3.33\), \({\bar{e}}=5\) and \(\alpha = 50^\circ\). The sensitivity of the droplet size on these parameters is analysed in the next section.
Figure 11 also shows that droplets generated in the highly viscous limit (\(\lambda =10^2\)) are always smaller than those generated in the almost inviscid limit (\(\lambda =10^{3}\)), for a fixed \({\mathrm{Ca}}_c\), because of the additional shear stresses contributing to the force involved in the dripping mechanism. This argument should also be completed by the fact that if the viscosity of the dispersed phase increases, the flow resistance at the neck of the droplet increases, reducing the flow towards the droplet, hence contributing to produce smaller droplet size too^{31}.
Influence of geometrical parameters
The quasistatic approach is now used to study the influence of the geometrical parameters \(\bar{R}_{e}\), \(\bar{H}\) and \(\alpha\) (see Fig. 5) as compared to the nominal values corresponding to the geometry of couple 1 (see Table 1), with \(\lambda =1/23\).
It can be observed in Fig. 12a that a smaller ratio of the extraction capillary diameter with the nozzle diameter leads to smaller drops for the same \({\mathrm{Ca}}_c\), since the viscous stresses exerted by the continuous phase on the meniscus obviously increases with the confinement, i.e. as \({\bar{R}}_e\) decreases. Accordingly, we name \({\bar{R}}_e\) the unconfinement parameter. On the contrary, the drippingjetting transition at \({\mathrm{Ca}}_c^*\) (crosses) is almost insensitive to the radius of the extraction capillary, except for \({\bar{R}}_e < 3\), indicating that strong confinement can displace the dripping to jetting transition towards larger value of \(Ca_c\) and eventually enhance the dripping regime.
Figure 12b shows that the meniscus equivalent radius is quite insensitive to \(\bar{H}\) for \(\bar{H}<2\), above which a larger value of \(\bar{H}\) leads to a larger meniscus. It can be explained by the flow field coming from an infinite medium inside a tube in absence of the dispersed phase and the nozzle, where the velocity is uniform inside the extraction capillary but decays in the axis of symmetry as moving away from the entrance. The length scale of this effect is of the order of the radius of the capillary. For larger \(\bar{H}\), the flow rate should be higher to keep the same stresses around the meniscus, and thus the same droplet size. Interestingly, the drippingjetting transition (crosses) occurs for larger \({\mathrm{Ca}}_c^*\) as \({\bar{H}}\) is increased. Reversely, this transition becomes independent on \({\bar{H}}\) for \({\bar{H}} \le 0\) (see Coefficients of Variation (CV) in Supplementary Information), i.e. when the injection capillary is embedded into the extraction capillary, as for a simple coflow configuration. Accordingly, we name \(\bar{H}\) the nonembedment parameter. Now, increasing \({\bar{H}}\) toward positive values seems to displace the dripping to jetting transition towards larger value of \(Ca_c\), thus enhancing the dripping regime. However, we found experimentally that the dripping regime is not stable for \({\bar{H}} >4\) and rather gives place to tipmultibreaking regime, as reported by Zhu et al.^{49} (see Supplementary Information).
Finally, no influence of \(\alpha\) is found for the meniscus equivalent radius but small and nonmonotonic influence on \({\mathrm{Ca}}_c^*\) is revealed, with a maximum around \(50^\circ\), as shown in Fig. 12c. Note also that no significant influence of the thickness of the extraction tube wall, \({\bar{e}}\), could be identified, such as this parameter has been disregarded from our parametric analysis. Contrarily, we show in Supplementary Information that opening the inner diameter of the extraction capillary with a given angle can displace the quasistatic dripping to jetting transition to larger values of \(Ca_c\), as already reported in planar flowfocusing^{50}, thus also enhancing the dripping regime.
Influence of inertia
As mentioned in section Modelling, all results for the drop size have been obtained with the Stokes equation, i.e. with \(\mathrm{{La}}=0\), since no influence of inertia were found for \(\mathrm{{La}}\) up to 100 on the drop size predictions. Nevertheless, we found that inertia can still significantly influence the occurrence of the quasistatic drippingjetting transition. Figure 13 shows that inertia effects enhance the drippingjetting transition, i.e. \({\mathrm{Ca}}_c^*\) decreases with increasing \(\mathrm{{La}}\) for a fixed \(\lambda\), and that the enhancement is more pronounced as \(\lambda\) increases. On the contrary, inertia plays almost no role in the inviscid limit (\(\lambda \rightarrow 0\)).
Experimentally, this transition could only be identified with the geometry of couple 2 and the fluid properties given in Table 1. The corresponding point is shown in Fig. 13 for \(\lambda =1/23\approx 4\times 10^{2}\) and \(\mathrm{{La}}=3.6\), with the error bar covering the range in \({\mathrm{Ca}}_c\) over which the transition was identified (see Fig. 4). Despite the large error bar, the agreement with the quasistatic approach including inertia (see orange line for \(\mathrm{{La}}=4\) in Fig. 13) is satisfactory and confirms the important role of inertia for viscous droplets. Indeed, for the case of less viscous droplets, another pair of fluids corresponding to \(\lambda \approx 6\times 10^{3}\) and \(\mathrm{{La}}=0.1\) has also been tested experimentally, showing again an excellent agreement with the modelling, and demonstrating a much lower influence of inertia effects for low viscosity ratios.
Discussions and conclusions
In this paper, we have presented the Raydrop device allowing for the generation of droplets in an axisymmetric flowfocusing, that we have renamed nonembedded coflowfocusing. This configuration, together with the quasistatic approach, has enabled a systematic parametric analysis, considering the influence of all relevant parameters. In the quasistatic dripping regime, droplet size has been shown (i) to decrease with increasing the flow rate of the continuous phase, (ii) to decrease with increasing the viscosity ratio between the dispersed phase and the continuous phase, (iii) to increase with increasing the ratio of the radii between the extraction capillary and the nozzle, namely with decreasing confinement, (iv) to increase with the increasing interdistance between the nozzle and the extraction capillary, namely with decreasing embedment; all these dependencies being related to the viscous forces associated to the droplet detachment, like the gravity force in pendant droplets. On the contrary, the droplet size has been found to be rather insensitive to inertia, to the inclination angle of the nozzle and the inclination angle of the inner diameter of the extraction capillary, even though these parameters have been shown to affect the quasistatic dripping to jetting transition. Remarkably, decreasing confinement or embedment has been numerically shown to displace the transition towards larger values of \(Ca_c\) and enhance the dripping regime. This would however have to be confirmed with future experiments.
Besides the excellent agreement obtained between experiments and simulations, not only in the quasistatic limit but also in transient, we have experimentally identified the dripping to jetting transition when increasing the dispersed flow rate (\(Q_d\)) for a given continuous flow rate (\(Q_c\)). This transition has however not been explored in detail using our simulations as it requires full transient parametric analysis, hence much more computing resources than for simulating steady flows. Another justification for having disregarded a full parametric analysis in transient for the dripping regime is that the experimental droplet size has been shown to vary only slightly with \(Q_d\) at a fixed \(Q_c\). This characteristic is remarkable as it demonstrates the dominant role of the viscous forces induced by the continuous phase on the droplet formation and it therefore supports the pertinence of the quasistatic approach beyond its validity domain. Consequently, since increasing \(Q_d\) has little effect on the droplet size, it essentially modulates the droplet frequency (see Fig. 3).
While this study has been performed for water in oil, we illustrate in Fig. 14 the generation of bubbles and droplets involving a wide variety of pairs of fluids.
This figure demonstrates the universality of the Raydrop to operate properly independently of the wetting properties of the materials in contact with the fluids, and independently of the physicochemical properties of these fluids (interfacial tension, viscosity, density, miscibility). It additionally indicates that tuning the diameters of the nozzle tip and/or the extraction capillary enables to cover a wide range of droplet diameters, theoretically from 20 to 400\(\,\upmu\)m, with any given couple of fluids, a feature which is not achievable today with any other single device available.
In the context of a growing demand of controlled droplets in many areas, the Raydrop emerges therefore as a very robust and versatile solution easily implementable in laboratories with little experience and facilities in microfluidics.
References
Gunther, A. & Jensen, K. F. Multiphase microfluidics: from flow characteristics to chemical and materials synthesis. Lab Chip 6, 1487–1503 (2006).
Teh, S.Y., Lin, R., Hung, L.H. & Lee, A. P. Droplet microfluidics. Lab Chip 8, 198–220 (2008).
Tran, T., Lan, F., Thompson, C. & Abate, A. From tubes to drops: dropletbased microfluidics for ultrahighthroughput biology. J. Phys. D Appl. Phys. 46, 114004 (2013).
Anna, S. L. Droplets and bubbles in microfluidic devices. Annu. Rev. Fluid Mech. 48, 285–309 (2016).
Shang, L., Cheng, Y. & Zhao, Y. Emerging droplet microfluidics. Chem. Rev. 117, 7964–8040 (2017).
Bremond, N. & Bibette, J. Exploring emulsion science with microfluidics. Soft Matter 8, 10549–10559 (2012).
Candoni, N., Grossier, R., Lagaize, M. & Veesler, S. Advances in the use of microfluidics to study crystallization fundamentals. Ann. Rev. Chem. Biomol. Eng. 10, 59–83 (2019).
Song, H., Chen, D. L. & Ismagilov, R. F. Reactions in droplets in microfluidic channels. Angew. Chem. Int. Ed. 45, 7336–7356 (2006).
Lee, D., Fang, C., Ravan, A. S., Fuller, G. G. & Shen, A. Q. Temperature controlled tensiometry using droplet microfluidics. Lab Chip 17, 717–726 (2017).
Conchouso, D., Castro, D., Khan, S. A. & Foulds, I. G. Threedimensional parallelization of microfluidic droplet generators for a litre per hour volume production of single emulsions. Lab Chip 14, 3011–3020 (2014).
Nunes, J. K., Tsai, S. S. H., Wan, J. & Stone, H. A. Dripping and jetting in microfluidic multiphase flows applied to particle and fibre synthesis. J. Phys. D Appl. Phys. 46, 114002 (2013).
Baccouche, A. et al. Massively parallel and multiparameter titration of biochemical assays with droplet microfluidics. Nat. Protoc. 12, 1912–1932 (2017).
AliCherif, A., Begolo, S., Descroix, S., Viovy, J.L. & Malaquin, L. Programmable magnetic tweezers and droplet microfluidic device for highthroughput nanoliter multistep assays. Angew. Chem. Int. Ed. 51, 10765–10769 (2012).
Zhang, Y. & Jiang, H.R. A review on continuousflow microfluidic pcr in droplets: advances, challenges and future. Anal. Chim. Acta 914, 7–16 (2016).
Agresti, J. J. et al. Ultrahighthroughput screening in dropbased microfluidics for directed evolution. Proc. Nat. Acad. Sci. 107, 4004–4009 (2010).
Damiati, S., Kompella, U. B., Damiati, S. A. & Kodzius, R. Microfluidic devices for drug delivery systems and drug screening. Genes 9, 103 (2016).
Klein, A. et al. Droplet barcoding for single cell transcriptomics applied to embryonic stem cells. Cell 161, 1187–1201 (2015).
Brouzes, E. et al. Droplet microfluidic technology for singlecell highthroughput screening. Proc. Nat. Acad. Sci. 106, 14195–14200 (2009).
Ryckelynck, M. et al. Using dropletbased microfluidics to improve the catalytic properties of rna under multipleturnover conditions. RNA 21, 458–469 (2015).
Guillot, P., Colin, A., Utada, A. S. & Ajdari, A. Stability of a jet in confined pressuredriven biphasic flows at low Reynolds numbers. Phys. Rev. Lett. 99, 104502 (2007).
Serra, C., Berton, N., Bouquey, M., Prat, L. & Hadziioannou, G. A predictive approach of the influence of the operating parameters on the size of polymer particles synthesized in a simplified microfluidic system. Langmuir 23, 7745–7750 (2007).
Utada, A. S. et al. Monodisperse double emulsions generated from a microcapillary device. Science 308, 537–541 (2005).
Benson, B. R., Stone, H. A. & Prudhomme, R. K. An offtheshelf capillary microfluidic device that enables tuning of the droplet breakup regime at constant flow rates. Lab Chip 13, 4507–4511 (2013).
Bandulasena, M. V., Vladisavljević, G. T. & Benyahia, B. Versatile reconfigurable glass capillary microfluidic devices with lego®inspired blocks for drop generation and micromixing. J. Colloid Interface Sci. 542, 23–32 (2019).
Evangelio, A., CampoCorté, F. & Gordillo, J. M. Simple and double microemulsions via the capillary breakup of highly stretched liquid jets. J. Fluid Mech. 804, 550–577 (2016).
CruzMazo, F., Montanero, J. & GañánCalvo, A. Monosized dripping mode of axisymmetric flow focusing. Phys. Rev. E 94, 053122 (2016).
Guillot, P., Colin, A. & Ajdari, A. Stability of a jet in confined pressuredriven biphasic flows at low reynolds number in various geometries. Phys. Rev. E 78, 016307 (2008).
Cordero, M. L., Gallaire, F. & Baroud, C. N. Quantitative analysis of the dripping and jetting regimes in coflowing capillary jets. Phys. Fluids 23, 094111 (2011).
Richards, J. R., Beris, A. N. & Lenhoff, A. M. Drop formation in liquidliquid systems before and after jetting. Phys. Fluids 7, 2617–2630 (1995).
Hua, J., Zhang, B. & Lou, J. Numerical simulation of microdroplet formation in coflowing immiscible liquids. AIChE J. 53, 2534–2548 (2007).
Lan, W., Li, S. & Luo, G. Numerical and experimental investigation of dripping and jetting flow in a coaxial microchannel. Chem. Eng. Sci. 134, 76–85 (2015).
Bai, F., He, X., Yang, X., Zhou, R. & Wang, C. Three dimensional phasefield investigation of droplet formation in microfluidic flow focusing devices with experimental validation. Int. J. Multiph. Flow 93, 130–141 (2017).
Wu, L., Tsutahara, M., Kim, L. S. & Ha, M. Threedimensional lattice boltzmann simulations of droplet formation in a crossjunction microchannel. Int. J. Multiph. Flow 34, 852–864 (2008).
Zhou, C., Yue, P. & Feng, J. J. Formation of simple and compound drops in microfluidic devices. Phys. Fluids 18, 092105 (2006).
Garstecki, P., Stone, H. A. & Whitesides, G. M. Mechanism for flowrate controlled breakup in confined geometries: a route to monodisperse emulsions. Phys. Rev. Lett. 94, 164501 (2005).
Li, Z., Leshansky, A., Pismen, L. & Tabeling, P. Stepemulsification in a microfluidic device. Lab Chip 15, 1023–1031 (2015).
Boucher, E. A., Evans, M. J. B. & Frank, F. C. Pendent drop profiles and related capillary phenomena. Proc. R. Soc. Lond. A Math. Phys. Sci. 346, 349–374 (1975).
Valet, M., Pontani, L.L., Prevost, A. & Wandersman, E. Quasistatic microdroplet production in a capillary trap. Phys. Rev. Appl. 9, 014002 (2018).
Basaran, O. A. & Scriven, L. Axisymmetric shapes and stability of pendant and sessile drops in an electric field. J. Colloid Interface Sci. 140, 10–30 (1990).
Cristini, V. & Tan, Y.C. Theory and numerical simulation of droplet dynamics in complex flows—a review. Lab Chip 4, 257–264 (2004).
MartínezCalvo, A., RiveroRodriguez, J., Scheid, B. & Sevilla, A. Natural breakup and satellite formation regimes of surfactantladen liquid threads. J. Fluid Mech. 883, A35 (2020).
Van Brummelen, E., Raven, H. & Koren, B. Efficient numerical solution of steady freesurface navierstokes flow. J. Comput. Phys. 174, 120–137 (2001).
RiveroRodriguez, J., PerezSaborid, M. & Scheid, B. Pdes on deformable domains: Boundary arbitrary lagrangianeulerian (bale) and deformable boundary perturbation (DBP) methods. arXiv preprintarXiv:1810.10001 (2018).
RiveroRodriguez, J. & Scheid, B. Bubble dynamics in microchannels: inertial and capillary migration forces. J. Fluid Mech. 842, 215–247 (2018).
RiveroRodriguez, J. & Scheid, B. Mass transfer around bubbles flowing in cylindrical microchannels. J. Fluid Mech. 869, 110–142 (2019).
Erb, R. M., Obrist, D., Chen, P. W., Studer, J. & Studart, A. R. Predicting sizes of droplets made by microfluidic flowinduced dripping. Soft Matter 7, 8757 (2011).
Gordillo, J. M., Sevilla, A. & CampoCortès, F. Global stability of stretched jets: conditions for the generation of monodisperse microemulsions using coflows. J. Fluid Mech. 738, 335–357 (2014).
Suryo, R. & Basaran, O. A. Tip streaming from a liquid drop forming from a tube in a coflowing outer fluid. Phys. Fluids 18, 082102 (2006).
Zhu, P. et al. Tipmultibreaking in capillary microfluidic devices. Sci. Rep. 5, 11102 (2015).
Tan, Y.C., Cristini, V. & Lee, A. P. Monodispersed microfluidic droplet generation by shear focusing microfluidic device. Sensors Actuators B Chem. 114, 350–356 (2006).
Scheid, B., Dewandre, A. & Vitry, Y. Droplet and/or bubble generator (2019). PCT/EP2018/067960.
Acknowledgements
This work was financially supported by the FirstSpinOff program of the Walloon region as well as by the Fonds de la Recherche ScientifiqueF.N.R.S. (Postdoctoral Researcher position of B. Sobac and Senior Research Associate position of B. Scheid). We thank Lionel Matthys for having suggested the terminologies coflowfocusing, as well as Raydrop, in memory of Raymond Brandt who assisted us in the design.
Author information
Authors and Affiliations
Contributions
A.D., Y.V and B.Sc. conceived the idea of the presented device. A.D. and Y.V. developed it and A.D. carried out the experiments with the help of Y.V. and B.So. J.R. and B.Sc. developed the numerical method and J.R. performed the computations. A.D., J.R., B.So, B.Sc. analyzed and interpreted the experimental and computational data. A.D., J.R. and B.So. prepared the figures and wrote the first version of the manuscript. B.Sc. supervised this work, contributed to the writing and put the manuscript to its final version. All authors reviewed, commented and approved the manuscript.
Corresponding author
Ethics declarations
Competing interests
The results presented in this paper are covered by a patent application of the Université libre de Bruxelles^{51}. The Raydrop device is manufactured and marketed by Secoya Technologies, a spinoff company from the Université libre de Bruxelles in which A. Dewandre, Y. Vitry and B. Scheid hold shares. B. Sobac and J. RiveroRodriguez have no competing interests.
Additional information
Publisher's note
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Supplementary information
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons licence, and indicate if changes were made. The images or other third party material in this article are included in the article's Creative Commons licence, unless indicated otherwise in a credit line to the material. If material is not included in the article's Creative Commons licence and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this licence, visit http://creativecommons.org/licenses/by/4.0/.
About this article
Cite this article
Dewandre, A., RiveroRodriguez, J., Vitry, Y. et al. Microfluidic droplet generation based on nonembedded coflowfocusing using 3D printed nozzle. Sci Rep 10, 21616 (2020). https://doi.org/10.1038/s4159802077836y
Received:
Accepted:
Published:
DOI: https://doi.org/10.1038/s4159802077836y
This article is cited by

A onedimensional mathematical model for shearinduced droplet formation in coflowing fluids
Theoretical and Computational Fluid Dynamics (2024)

A Perspective Review of Droplets and Bubbles Formation in Microfluidics
Microgravity Science and Technology (2024)

Planesymmetric coflow capillary for the generation of monodisperse droplets
Microfluidics and Nanofluidics (2024)

Machine learningbased bead enumeration in microfluidics droplets enhances the reliability of monitoring bead encapsulation toward singlecell sorting applications
Microfluidics and Nanofluidics (2024)

High gravity material extrusion system and extruded polylactic acid performance enhancement
Scientific Reports (2023)
Comments
By submitting a comment you agree to abide by our Terms and Community Guidelines. If you find something abusive or that does not comply with our terms or guidelines please flag it as inappropriate.