Abstract
Digital Breast Tomosynthesis (DBT) is a modern 3D Computed Tomography Xray technique for the early detection of breast tumors, which is receiving growing interest in the medical and scientific community. Since DBT performs incomplete sampling of data, the image reconstruction approaches based on iterative methods are preferable to the classical analytic techniques, such as the Filtered Back Projection algorithm, providing fewer artifacts. In this work, we consider a ModelBased Iterative Reconstruction (MBIR) method well suited to describe the DBT data acquisition process and to include prior information on the reconstructed image. We propose a gradientbased solver named Scaled Gradient Projection (SGP) for the solution of the constrained optimization problem arising in the considered MBIR method. Even if the SGP algorithm exhibits fast convergence, the time required on a serial computer for the reconstruction of a real DBT data set is too long for the clinical needs. In this paper we propose a parallel SGP version designed to perform the most expensive computations of each iteration on Graphics Processing Unit (GPU). We apply the proposed parallel approach on three different GPU boards, with computational performance comparable with that of the boards usually installed in commercial DBT systems. The numerical results show that the proposed GPUbased MBIR method provides accurate reconstructions in a time suitable for clinical trials.
Introduction
Digital Breast Tomosynthesis (DBT) is a quite recent 3D Computed Tomography (CT) technique providing a volumetric breast reconstruction as a stack of 2D images, each representing a crosssectional slice of the breast itself^{1}. When compared with traditional 2D mammography, DBT has the advantage of separating the breast anatomical tissues, which can be overlapped in a mammography, and this generally reduces false negative diagnosis. At the same time, since the DBT source emits Xrays only from a small number of angles in an arc trajectory, DBT provides a low radiation dose comparable to the radiation dose used in a standard mammography. For this reason, DBT is appealing to the medical and scientific community as a breast screening routine^{2,3}.
Until a few years ago, the introduction of DBT in a clinical setting has been slowed down by the absence of efficient algorithms for image reconstruction from a limited number of projections. It is well known that the traditional Filtered Back Projection (FBP)^{4} analytic algorithm amplifies noise and artifacts in the case of lowsampled data^{3}. Iterative algorithms are known to perform better than FBP in the case of tomosynthesis data, but they require computational times incompatible with a clinical use^{3,5}, since in clinical DBT examinations the breast reconstruction must be provided in 40 to 60 seconds. Nevertheless, the recent advent of low cost parallel architectures, such as Graphics Processing Units (GPUs), has provided the chance for a remarkable reduction of the image reconstruction time, making iterative methods a realistic alternative to analytic algorithms.
Among the different iterative approaches in Xray CT (see^{5} for a detailed classification of iterative methods in CT), the so called ModelBased Iterative Reconstruction (MBIR) methods are now getting growing attention. In general, they try to model the acquisition process as accurately as possible, since they take into account system geometry, physical interactions of photons in the projections and prior information about the acquired volume. This approach produces better results than the traditional FBP in terms of quality of the reconstructed image and artifacts reduction, especially for lowdose or limited data Xray CT. The MBIR methods can be described in a unifying framework as constrained or unconstrained minimization problems^{5,6,7}, involving a fittodata function and a regularizing function, acting as a prior on the solution. A widely used regularization term, proposed in sparse tomography by^{8} and then used by many authors^{9,10,11,12,13,14}, is the Total Variation (TV) function, whose edge enhancing properties are very effective on medical images. Herein, we consider the general constrained minimization formulation:
where LS(x) is the least squares fittodata function, TV(x) denotes the Total Variation regularizer, λ > 0 is the regularization parameter and the request of nonnegativity on the solution is due to physical constraints.
In the CT framework, very popular algorithms for solving (1) are based on alternate minimization of the two terms of the objective function of (1). They process few views at a time with Ordered Subset strategies for the least squares term in (1) and use a fixed step size gradient descent approach on the TV function^{6,8,11,12,15}. However, the two steps strategy may have a rather slow convergence, while an algorithm that computes the new approximate solution in a single descent step, through the computation of the gradient of the whole objective function, generally has a faster convergence^{16}. In sparse tomography, these gradientbased minimization methods have been used in^{7,9,17}. In this work, we propose to solve (1) by an accelerated gradient scheme belonging to the class of Scaled Gradient Projection (SGP) methods^{18,19}. The SGP methods have been recently applied in lowsampled Xrays cone beam CT (CBCT) image reconstruction, with very good results in terms of image accuracy^{13,14,20}. In particular, in^{14} the authors proposed a SGP method for Xrays CT image reconstruction and applied it to a phantom simulation using a geometry different from DBT limited angles. Since SGP showed a very fast convergence in the first iterations, we choose a similar approach also for our DBT application.
Even if the SGP algorithm is expected to produce accurate reconstructions in few iterations, the high computational cost of each iteration still prevents the completion of the reconstruction process in a suitable time on a serial architecture. Exploiting the computational power of modern GPU boards, we aim to perform the expensive operations of each SGP iteration in a time consistent with the practical constraints imposed by the DBT application. Similar parallel approaches have been investigated in the case of 3D Xray CT image reconstruction^{9,10,11,15,21}, but none of these schemes is based on gradient methods accelerated by scaling strategies nor is optimized for the particular case of DBT data. To achieve our goal, we design a parallel SGP version in which the most time consuming task involved in each iteration, represented by the computation of the gradient of the objective function in (1), is distributed according to the hardware features of commonly available low cost GPU boards. The proposed implementation is evaluated in terms of time and image quality by reconstructing images from DBT simulated and real projections on three different GPU boards. The experiments show that the parallel SGP implementation performs a number of iterations suitable for reconstructing images with enhanced quality in about one minute (as usually required in clinical trials).
Results
Digital breast tomosynthesis imaging
Following the ongoing technological development, the medical imaging community is investing in innovation, looking for healthier and more reliable tools. To set an example, in traditional 2D mammography, cancerous masses are often camouflaged by the superposition of dense breast tissue on the final image, hence 3D breast reconstructions are getting increasing interest. On the other hand, the radiation risk from Xray based medical imaging is a matter of fact, especially in classical 3D CT where hundreds of Xray scans are performed from different points over a circular trajectory around the patient, in order to get a complete data set producing an accurate reconstructed image. As a consequence, classical 3D CT is not suitable for screening tests. The socalled tomosynthesis (a quite recent tomographic routine) tries to overcome this issue: it is characterized by the acquisition of a small number of projections captured from a limited angular range around the patient thus resulting in a faster and safer examination than traditional 3D CBCT^{1}. As a matter of fact, DBT has been included in the diagnostic and screening settings in adjunct to digital mammography in some European countries, such as Italy^{22,23}.
Figure 1(a) illustrates a draft of the tomosynthesis device for the specific case of the breast imaging: the Xray source emits a cone beam at each angle and its trajectory is restricted from the classical circular one to a Cshape path. In Fig. 1(b) we report a more technical draw of the DBT geometry mashed on the YZplane in order to better visualize the device setting, and in Fig. 1(c) we show the Xray cone beam projection for one recording unit on the detector. Due to the lack of data, the resolution of the reconstructed images along the indepth Zdirection is coarser than in the XYplane. Moreover, the incomplete data sampling leads to wellstudied artifacts. In particular, DBT images present two different types of artifacts: inplane artifacts, consisting in a dark path in the direction of the source motion around the dense objects, and indepth artifacts, appearing as N_{θ} shadows in the slices preceding and following the object along the Zdirection^{24}.
More specifically, the quality of the reconstructed images depends on the number N_{θ} of acquired projections, typically from 10 to 25, and on the angular range, usually from 15 to 50 degrees.
Materials
We have tested the proposed SGP method on two real data sets, fitting the European DBT Protocol^{22}, obtained by the digital system Giotto Class of the IMS Giotto Spa company and on a synthetic data set obtained by simulating the projections of a digital phantom with the Giotto geometry. In Giotto apparatus, the reference system is set as in Fig. 1(b). In the central position, the Xray source is approximately 69 cm above the detector and it shoots N_{θ} = 11 projections with a low dose protocol, in a very limited angular range from −15 to 15 degrees. The flat detector has 3580 × 2812 square recording units with edges equal to 0.085 mm and it is large enough to contain the widest compressed breast projections. As usual habit, in order to speed up the reconstruction phase, in a preprocessing step the projections are cut to eliminate those portions of the data which do not contain useful information for the breast reconstruction. If the resulting crop consists of n_{x} × n_{y} pixels for each projection, then the available data set has size N_{d} = n_{x} × n_{y} × N_{θ}. The system uses a polychromatic ray with energies in a narrow range around 20 keV to avoid the photon scattering. As often happens in CT reconstruction algorithms, we approximate the polychromatic ray with a monochromatic one.
The first data set (D1) considered has size 3000 × 1500 × 11 and is obtained by scanning the Tomophan Phantom TSP004, by The Phanto Laboratory^{25}. The phantom contains aluminum Test Objects (TOs) embedded into a clear urethane material and it has a semicircular shape with a thickness of 42 mm. Due to its structure, this Quality Control (QC) accreditation phantom is used for different tests (see the phantom data sheet^{25}). Among them, we concentrate on evaluating the homogeneity of the inplane aluminum objects, the contrast between the luminescent aluminum objects and the dark background, and the measure of the noise in the background. In particular, in the reconstructed image we will focus on three 0.500 mm aluminum beads spaced 10 mm along the Zdirection and hence positioned on three different layers.
The second data set tested (D2) has size 3000 × 1500 × 11 and it is obtained by scanning the BR3D Breast Imaging Phantom^{26} (model 020), produced by the CIRS company, which is widely used to assess the detectability of lesions of different sizes. The objects of interests, constituted by microcalcifications, fibers and masses, are within a heterogeneous background composed by epoxy resin to simulate pure adipose and pure glandular tissues, mixed together to mimic a real breast. In particular, we will analyze the reconstruction of a cluster of CaCO3 small beads, simulating microcalcifications of 0.290 mm in diameter.
Finally, we have tested the SGP algorithm on a simulated data set (D3), obtained from a digital phantom we designed. The D3 data set has 3200 × 1100 × 11 elements, computed as projections of the digital phantom under the geometric setting of the Giotto Class device onto a detector with 0.100 mm element pitch. We added Gaussian noise with SNR = 50 to the computed projections (where the SNR value is computed as the logarithm of the ratio between the noisy projections and the noise, multiplied by a factor of 20), to better simulate a real Xray acquisition. Our phantom reproduces the main features of the BR3D Breast Imaging Phantom: it contains several test objects reproducing microcalcifications, fibers and masses, immersed into a uniform adiposelike background. The attenuation coefficients of the test objects and background are taken from a reconstruction of the BR3D Breast Imaging Phantom performed on a commercial system. The analysis on the reconstructed images carried out in the next subsection will concentrate on the detection of a cluster containing 0.300 mm diameter beads, simulating microcalcifications of medium size.
The data sets analysed during the current study are available at the website: https://cloud.hipert.unimore.it/s/ptqjtMdwXA7sc5N.
Reconstructions
In this paragraph we evaluate the quality of the reconstructions obtained with the proposed SGP method on the data sets presented above. We are interested in monitoring how the algorithm performs for an increasing number of iterations, hence we focus on the reconstructions achieved in 4, 12 and 30 iterations.
The volume to be reconstructed is discretized into N_{v} = N_{x} × N_{y} × N_{z} volumetric elements (voxels), where N_{x}, N_{y}, N_{z} are the number of elements along the three Cartesian directions. We observe that in DBT imaging N_{v} > N_{d}. The reconstructions of real phantoms have inplane voxel size of 0.090 mm, whereas the indepth one is 1.000 mm. In all the tests, the regularization parameter λ has been tuned by a trial and error procedure. The images presented in this section show slices of the reconstructed volume parallel to the XYplane. We remind that the source moves along the Cshaped path on the Y direction. For each phantom, the intensities of the reconstructions are shown in the same gray scale (as arbitrary unit).
Tomophan phantom results
We report in Fig. 2 some results obtained on the D1 data set. Figure 2(a–c) show the reconstructions after 4, 12, 30 iterations inside a region of 160 × 320 pixels (corresponding to 14.4 mm × 28.8 mm) on the layer 19, where the lower metallic sphere is on focus. The yellow rectangle in figures (a–c) defines the region where we computed the mean and the standard deviation (StdDev) values reported under each corresponding figure in order to analyze the background uniformity and the noise. Above the bead, we can also see the expected metallic artifacts produced by the two specks located on layers 39 and 29. The artifacts corresponding to the highest speck are in the form of 11 (i.e., the number of projections views) small circles, while the artifacts produced by the central sphere are in the form of a light strip. The intensity of these artifacts can be analysed on the profile identified by the orange line of Fig. 2(d); the reconstructions of the profile at different iterations are compared in Fig. 2(e).
As concerns the in focus bead, we study the orange profile in Fig. 2(f) compared among the three considered reconstructions in Fig. 2(g). The blue line shows that after 4 iterations the recovered values inside the sphere are quite low and nonuniform, whereas the bead detection improves getting higher values of about 0.092 and 0.115 in the reconstructions at 12 and 30 iterations, respectively.
BR3D breast phantom results
Now we analyse the results obtained from the data set D2, characterized by breastlike background and objects of interest. They are perfectly on focus on the selected layer. In Fig. 3(a–c) we report the reconstruction crops corresponding to a 145 × 145 pixel region (13.05 mm × 13.05 mm) containing a cluster of microcalcifications. In Fig. 3(d) we plot the corresponding inplane profiles on the central speck along the Ydirection.
Digital phantom results
In order to better analyse the accuracy of the MBIR approach in the challenging task of a microcalcification detection, we present here the results obtained on the simulated D3 data set. The digital phantom to be reconstructed from the D3 data set has size 3000 × 1000 × 50 with voxel size of 0.100 mm along the X and Y axes and 1.000 mm along the Z direction. Figure 4(a–c) show the results obtained from D3 inside the 173 × 173 pixel ROI (corresponding to 17.3 mm × 17.3 mm) of the central layer of the synthetic phantom, containing small microcalcifications. We have zoomed over one microcalcification in Fig. 4(e–g) where we can appreciate how the 3voxel wide object is already detected after only 4 iterations. Fig. 4(d) shows the behaviour of the objective function minimized in (1) with respect to the iteration number. In Fig. 4(i) we plotted the profile of the yellow line of Fig. 4(h) reconstructed after 4, 12 and 30 iterations and, with black dots, the solution obtained when the convergence criterion is met (i.e., after 64 iterations). The green dotted line corresponds to the exact profile.
Parallel executions
The reconstructing software has been implemented in C code on a commercially available high end computer, equipped with Intel i7 7700K CPU at 4.2 GHz, 32 GB of RAM and 1 TB of Solid State Disk (SSD), and its parallel implementation has been performed on NVIDIA GPUs by means of the CUDA SDK^{27}. The program has been tested on different GPUs, with different memories, number of CUDA cores and, obviously, price point.
In details, we have considered the following GPU boards:
GTX 1060: 6 GB of RAM, 1280 CUDA cores, launch price 250$;
GTX 1080: 8 GB of RAM, 2560 CUDA cores, launch price 700$;
Titan V: 12 GB of RAM, 5120 CUDA cores, launch price 3000$.
We tested two different parallel implementations on the Titan V board: the first one (denoted as Titan V_1) has the same approach considered for the GTX 1060 and GTX 1080 boards, where the data cannot be fully stored in RAM and many data transfers between the CPU and the GPU are required during each SGP iteration, while the second (denoted as Titan V_2) exploits the larger GPU RAM to store all the needed data. For more details on the parallel implementations, please refer to the corresponding section.
The results shown in this paragraph are obtained on the D3 synthetic data set, whose volume is made of N_{v} = 1.5·10^{8} voxels. We have identified four main tasks in the SGP algorithm: the Forward and Backward projections, the TV evaluation and all the remaining operations (Other), mainly consisting of scalar products and vector sums. We will analyse the parallel algorithm performance through these four tasks.
By profiling a serial execution of the program, we can draw the pie chart in Fig. 5(a) of the computational time for the different tasks composing a single SGP iteration (for a comprehensive description of the algorithm, please see the Methods sections): 88% of the time is spent for the Forward and Backward projections, 4% for the evaluation of the TV function and of its gradient and 8% for all remaining SGP operations.
Figure 5(b) shows the execution times in milliseconds per iteration for each of the 4 computational kernels; the last column reports the global speedup. The time required for a Forward or Backward projection decreases from about 4 minutes in the serial execution to 1–3 seconds on the GTX boards and to less than 300 milliseconds for the Titan V_1 case. The parallelization of the TV function reduces the time from 23 seconds to 1 second, while the execution time for Other decreases from about 40 to 7 seconds: these times are almost constant on all the three boards since they are mainly due to data communications between CPU and GPU, which is independent from the considered GPU board. Different performances are provided by the Titan V_2 case which achieves a speedup of almost 470. We remark that the reported times correspond to an SGP iteration with only one execution of the inner backtracking loop; each further backtracking step requires to evaluate only the TV component of the objective function, but it does not involve Forward or Backward projections.
Figure 6 shows the time per iteration on the three considered GPUs for data sets simulated by projecting the digital phantom used in D3 with a varying number of angles N_{θ} ∈ {11, 21, 31, 41}.
Discussion
Looking at Fig. 2 and recalling the characterizations of the Tomophan Phantom, we can make the following remarks. It is evident from Fig. 2(a–c) that for increasing iterations the contrast between the lower aluminum sphere and the background enhances. Moreover, when the algorithm approaches the solution of (1), the sphere gets more and more homogeneous inside, as can be viewed by the plots in Fig. 2(g). The mean and standard deviation reported under Fig. 2(a–c) confirm that the noise keeps very low along the iterations. This proves that the TV regularization term efficiently smooths the noise and that a suitable value for the regularization parameter has been chosen. There are visible artifacts in the upper part of the images which are due to the small number of projection angles. However, from the plots in Fig. 2(e) we observe that their amplitude is very small and their contrast from the background decreases with ongoing iterations. We also notice that all the artifacts are spread along the Y axis direction according to the Xray source motion. From all the above considerations, we can conclude that increasing the iterations from 4 through 30 improves the quality of the reconstructed image for this test phantom.
The images in Fig. 3(a–c) show how the small beads of the microcalcification cluster are detected by the solver after very few iterations, but with low contrast from the background. After 30 iterations, the contrast is far enhanced. Figure 3(d) provides a double piece of information on solution changes during the iterations: it confirms that the contrast improves with the increasing number of iterations and that the microcalcification width gets closer to the expected one of 290 micrometers (corresponding to about 3 voxels wide).
By analysing the results of the D3 data set we compare the reconstructed images with a ground truth. The regularizing effect of the TV prior is visible in Fig. 4(e–g), where the object spread is progressively reduced in favour of a higher contrast with the background. Figure 4(i) confirms that values obtained for the microcalcification get closer to the ground truth by increasing the number of iterations, whereas the exact background is recreated from the first iterations. Since the black dots, representing the solution when the convergence criterion is met, overlap the 30 iteration profile, we can conclude that the reconstruction after 30 iterations is the best possible to obtain. We can also point out that the plot in Fig. 4(d) shows a remarkable decrease of the objective function in the first iterations. As expected, there is a discrepancy between the computed solution and the exact one which is mainly due to the subsampling of data in the DBT acquisition process.
From the performed tests we can conclude that the proposed modelbased method, considering the L2TV mathematical model and using the SGP optimization algorithm, provides accurate DBT image reconstructions in few tens of iterations. Concerning the comparison with other iterative optimization algorithms, the SGP algorithm exploits a scaling acceleration strategy for improving the performance of the standard Gradient Projection BarzilaiBorwein (GPBB) method^{14} and the GPBB method has been proved to outperform widely used alternating minimization methods in the work by Park et al.^{9}.
As we stated in the Introduction, the drawback of modelbased methods is the long computational time required. From the results presented in Fig. 5 we can notice that the computational times are dramatically reduced in the parallel algorithm execution, with speedups ranging from 35× to 57× thanks to increasing computational power for GTX1060, GTX1080 and Titan V boards. The execution of the Titan V_2 version further breaks down the time, getting a speedup of almost 500, due to the particular implementation that avoids most of the data transfer between the CPU and the GPU memory. Figure 6 proves that our parallel implementation linearly scales with the size of the problem, providing different slopes for the different boards. In particular the Titan V_1 has an incremental slope of 50 with respect to the GTX 1060 which is about 600, showing an enhanced scalability. We can state that the proposed parallel implementation drastically cuts down the computational time, making the execution times for the reconstruction of real volumes compatible with clinical requirements. This is possible thanks to the ability of the SGP algorithm to provide suitable reconstructions in a few tens of iterations, which can be performed in less than one minute by the most recent boards, like the Titan V. Taking into account that the market of GPUs rapidly evolves towards more and more powerful and less costly boards, the proposed MBIR approach represents a useful tool for achieving the goal of getting large volumetric images of superior quality at affordable costs in the near future.
Methods
Scaled gradient projection method
In this paragraph we briefly describe the serial version of the SGP algorithm which has been implemented to solve the minimization problem (1) for DBT reconstruction. The SGP method is a firstorder descent method for the solution of a general minimization problem of the form: \({\rm{\arg }}\,{{\rm{\min }}}_{x\ge 0}f(x)\), where \(x\in {{\mathbb{R}}}^{n}\) and \(f:{{\mathbb{R}}}^{n}\to {\mathbb{R}}\) is a convex differentiable function. In order to accelerate the classical Gradient Projection method, SGP introduces at each iteration k a scaling procedure^{18}, by multiplying the descent direction \(\nabla f({x}^{(k)})\) by a diagonal matrix D_{k} with entries in the interval \([\frac{1}{{\rho }_{k}},{\rho }_{k}],{\rho }_{k} > 1\), (we call \({ {\cal{D}} }_{{\rho }_{k}}\) the set of these matrices). Moreover, SGP exploits BarzilaiBorwein^{28} type rules for the choice of the steplength to ensure a fast decrease of the objective function.
In details, after initializing \({x}^{(0)}\ge 0,\gamma ,\sigma \in (0,1),0 < {\alpha }_{{\min }}\le {\alpha }_{{\max }},{\alpha }_{0}\in [{\alpha }_{{\min }},{\alpha }_{{\max }}]\), \({\rho }_{0} > 0\), \({D}_{0}\in { {\cal{D}} }_{{\rho }_{0}}\), the following steps are repeated for k = 0, 1… until a stopping criterion is satisfied.
 1.
Compute the scaled descent projected direction d^{(k)} as \({d}^{(k)}={P}_{+}({x}^{(k)}{\alpha }_{k}{D}_{k}\nabla f({x}^{(k)})){x}^{(k)}\), where P+(z) is the Euclidean projection of the vector \(z\in {{\mathbb{R}}}^{n}\) onto the non negative orthant.
 2.
Perform a backtracking on the computed direction d^{(k)} starting with η = 1:
$$\begin{array}{c}while\,f({x}^{(k)}+\eta {d}^{(k)}) > f({x}^{(k)})+\sigma \eta \nabla f{({x}^{(k)})}^{T}{d}^{(k)}\\ \eta =\gamma \,\eta ;\end{array}$$  3.
Compute the new iterate: \({x}^{(k+1)}={x}^{(k)}+\eta {d}^{(k)}\).
 4.
Update \({\rho }_{k+1}=\sqrt{1+{10}^{15}/{(k+1)}^{2.1}}\) and the diagonal scaling matrix \({D}_{k+1}\in { {\cal{D}} }_{{\rho }_{k+1}}\).
 5.
Update the steplength \({\alpha }_{k+1}\in [{\alpha }_{min},{\alpha }_{max}]\).
The update of the scaling matrix D_{k+1} is performed through a splitting of the gradient of the objective function in its positive and negative parts as in^{14} and the definition of ρ_{k+1} is aimed at avoiding restrictive bounds on the diagonal entries of D_{k+1} in the initial phase of the iterative process and satisfying the SGP convergence conditions^{19} by asymptotically forcing D_{k+1} towards the identity matrix. The update of the steplength α_{k+1} is obtained by using an alternate BarzilaiBorwein strategy; in particular, we use adaptive alternations of the two classical BarzilaiBorwein rules proposed in^{29} and applied also in^{14}.
The algorithm is stopped when:
where maxiter is the maximum number of iterations allowed. For more implementation details and for the convergence properties of SGP refer to^{18,19}. Even if the theoretical convergence rate \({\mathscr{O}}(1/k)\) on the objective function values is lower than the rate \({\mathscr{O}}(1/{k}^{2})\) of some optimal firstorder methods^{17,30}, the practical performance of SGP method is very well comparable with the convergence rate of the optimal algorithms^{19}.
Discrete mathematical model
The MBIR methods for CT image reconstruction are built on the discretization of the LambertBeer law, relating the acquired and emitted intensities of Xrays^{31} in the tomographic process. The resulting model is a linear system of the form Ax = b where b collects all the \({N}_{d}={N}_{\theta }\times {n}_{x}\times {n}_{y}\) tomographic noisy projections, acquired during \({N}_{\theta }\) Xray scans as \({n}_{x}\times {n}_{y}\) projection images, lexicographically reordered into a vector shape; the unknown x is a N_{v}dimensional vector resulting from the 3D discretization of the volume. The operator projecting a volume \(x\) onto the detector according to the geometry of the CT device is called Forward Projection (FwP). This is mathematically modeled by computing an \({N}_{d}\times {N}_{v}\) coefficient matrix A, where each element represents the geometrical contribution of one voxel to one pixel of the detector for each projection angle, and multiplying the matrix by a N_{v}dimensional vector of the object space. The operator acting from the projection space to the object space is called Backward Projection (BwP) and it is implemented via multiplication of A^{T} by a N_{d}dimensional vector of the projection space.
In the particular case of DBT, due to the subsampling of the limited angle geometry, \({N}_{d} < {N}_{v}\), hence the linear system \(Ax=b\) is underdetermined and admits infinite solutions. In order to force the uniqueness of the solution and enhance the quality of the reconstruction, the problem is formulated as a minimization task as in (1), where the objective function is defined as:
Here, \(\parallel \nabla {x}_{j}{\parallel }_{2}\) is computed with respect to the three Cartesian directions with forward differences, while β > 0 is a small constant introduced to overcome the nondifferentiability of the discrete TV function whenever \(\parallel \nabla {x}_{j}{\parallel }_{2}\) is zero for some j.
The function f(x) defined in (3) represents the objective function of the constrained minimization problem that we solve with the SGP method, whose gradient is: \(\nabla f(x)={A}^{T}Ax{A}^{T}b+\lambda \nabla TV(x)\). We remark that in, the serial execution, the main computational cost at each SGP iteration is given by one FwP and one BwP involving the matrixvector product by A and \({A}^{T}\) in the gradient formula (as shown in Fig. 5(a)). Moreover, the matrix is too large to be stored in central memory and it must be reevaluated at each matrixvector operation (further details in “Parallel implementation”). The computation of the matrix A plays a key role in the accuracy of the CT modeling. Among the several approaches available in literature to define the tomographic matrix, we considered the distancedriven (DD) one, proposed for the 3D geometry in 2004^{32}: it fits very well the CBCT geometry and is suitable to be split among many processors. The draft in Fig. 1(c) schematically represents the DD Forward Projection process for one pixel i, in a 2D scheme of the coronal view. In particular, fixed an angle \({\theta }_{k}\) and a pixel i, the corresponding row of A contains the weights representing the contribution of all the voxels to the projection onto the pixel i. To compute the weights, the DD algorithm first identifies which voxels are crossed by the Xrays reaching the \(ith\) pixel (the voxels with bold edges in Fig. 1(c)); then it quantifies the contribution of all and only the identified voxels as the volume of the intersection between each voxel and the Xray beam. We observe that A is a very sparse matrix; in our tests, it always has a density factor of the order of 10^{−7}. To compute the \(\nabla TV(x)\) function in each voxel, we used forward differences involving the voxel values^{14}.
Parallel implementation
The proposed SGP method is suitable for parallel implementation on GPU as reported in^{33,34}. We describe in this section the parallel implementation of the SGP method for DBT image reconstruction. We exploit the massive parallel architecture of the GPUs by distributing the work to hundreds of small cores. A high level of parallelism is achieved by exploiting hundreds of computing pipelines, usually grouped into processing clusters, which can work both in single and double precision. Within each of these processing clusters, one hardware scheduler dispatches small groups of threads to the pipelines. To manage the communications between the CPU and the GPU, the division of the work and the partition of thread in blocks, we use CUDA^{27}, a wellknown Application Programming Interface developed by NVIDIA.
As already observed, the typical size of the data is so large that storing the whole matrix A in RAM is unfeasible. For example, in the data set D3 this would amount to (3000 × 1000 × 50) × (3200 × 1100 × 11) = 5.8·10^{15} elements (about 46 Petabytes in double precision data), which is much larger than the capacity of the most recent RAM. Furthermore, the number of nonzero elements (see Fig. 6(b)) makes the storage of A in a sparse matrix data structure inconvenient, since it would involve too large data transfer from the CPU memory to the GPU one, representing a crucial bottleneck for the GPU performance (see Fig. 7(a)). In D3 data set, a sparse matrix data structure representation requires at least 110 GB and the considered GPUs take about ten seconds for the memory transfer (almost an order of magnitude more than our approach). As a consequence, we organize the parallelization of the FwP and BwP operations by computing the nonzero entries of A every time we need them.
Fixed \({N}_{thread}\) as the number of threads per block, we created a grid of \(\lceil {N}_{v}/{N}_{thread}\rceil \) blocks. Within this configuration, each thread i independently computes the \(ith\) row of the system matrix A according to the DD strategy.
In the FwP evaluation, each thread i computes the product between the \(ith\) line of A and the column vector and saves the result in the \(ith\) position of the resulting vector. Due to the independence of all these computations, we do not need any data access synchronization. Inside the BwP step we call exactly the same DD function to compute the matrix A, hence we perform the product involving \({A}^{T}\) in a vector manner. Each thread i still computes one row of A (i.e., column of \({A}^{T}\)), it multiplies the contribution of each \(jth\) voxel to the \(ith\) element of input vector and atomically sums the result to the \(jth\) element of output vector.
For the parallelization of the TV and its gradient, we only need to operate on the values of the voxels. However, the data access pattern is not coalesced and this limits the speedup of its implementation. All remaining computations in the SGP steps (previously labelled by Other) are vector operations and they are mainly executed in parallel with CUBLAS library functions provided within the CUDA environment.
All the computations are performed in double precision arithmetic, since the results obtained with single precision numbers were not satisfying. As a result, we need more than 11 GB of GPU memory to process the D3 data set. Since not all the commercial GPUs are equipped with such a large amount of memory, each parallel task in the Fwp, BwP, TV and Others is separately executed on the GPU on portions of data (called chunks) and the final results are collected on the CPU memory. Hence, the vectors of size \({N}_{v}\) or \({N}_{d}\) involved in the parallel computations are divided into chunks of size \({{N}}_{x}\times {{N}}_{y}\times {{N}}_{chunk}\) and \({n}_{x}\times {n}_{y}\times {n}_{chunk}\), respectively, where both \({{N}}_{chunk}\) and \({n}_{chunk}\) are integers depending on the memory available on the device. An example is depicted in Fig. 7(b). Since the amount of data transferred between the CPU and the GPU is constant, the number of chunks does not have a significant impact on overall performance.
References
Males, M., Mileta, D. & Grgic, M. Digital breast tomosynthesis: A technological review. In Proceedings ELMAR2011, 41–45 (IEEE, 2011).
Andersson, I. et al. Breast tomosynthesis and digital mammography: a comparison of breast cancer visibility and birads classification in a population of cancers with subtle mammographic findings. European radiology 18, 2817–2825 (2008).
Das, M., Gifford, H. C., O’Connor, J. M. & Glick, S. J. Penalized maximum likelihood reconstruction for improved microcalcification detection in breast tomosynthesis. IEEE Transactions on Medical Imaging 30, 904–914 (2010).
Feldkamp, L. A., Davis, L. & Kress, J. W. Practical conebeam algorithm. Josa a 1, 612–619 (1984).
Beister, M., Kolditz, D. & Kalender, W. A. Iterative reconstruction methods in xray ct. Physica medica 28, 94–108 (2012).
Sidky, E. Y. et al. Enhanced imaging of microcalcifications in digital breast tomosynthesis through improved imagereconstruction algorithms. Medical physics 36, 4920–4932 (2009).
Sidky, E. Y., Jørgensen, J. H. & Pan, X. Convex optimization problem prototyping for image reconstruction in computed tomography with the chambolle–pock algorithm. Physics in Medicine & Biology 57, 3065 (2012).
Sidky, E. Y., Kao, C.M. & Pan, X. Accurate image reconstruction from fewviews and limitedangle data in divergentbeam ct. Journal of Xray Science and Technology 14, 119–139 (2006).
Park, J. C. et al. Fast compressed sensingbased cbct reconstruction using barzilaiborwein formulation for application to online igrt. Medical physics 39, 1207–1217 (2012).
Jia, X., Lou, Y., Li, R., Song, W. Y. & Jiang, S. B. Gpubased fast cone beam ct reconstruction from undersampled and noisy projection data via total variation. Medical physics 37, 1757–1760 (2010).
Matenine, D., Goussard, Y. & Després, P. Gpuaccelerated regularized iterative reconstruction for fewview cone beam ct. Medical physics 42, 1505–1517 (2015).
Sidky, E. Y. & Pan, X. Image reconstruction in circular conebeam computed tomography by constrained, totalvariation minimization. Physics in Medicine & Biology 53, 4777 (2008).
Loli Piccolomini, E. & Morotti, E. A fast total variationbased iterative algorithm for digital breast tomosynthesis image reconstruction. Journal of Algorithms & Computational Technology 10, 277–289 (2016).
Loli Piccolomini, E., Coli, V., Morotti, E. & Zanni, L. Reconstruction of 3d xray ct images from reduced sampling by a scaled gradient projection algorithm. Computational Optimization and Applications 71, 171–191 (2018).
McGaffin, M. G. & Fessler, J. A. Alternating dual updates algorithm for xray ct reconstruction on the gpu. IEEE Transactions on computational imaging 1, 186–199 (2015).
Graff, C. G. & Sidky, E. Y. Compressive sensing in medical imaging. Applied optics 54, C23–C44 (2015).
Jensen, T. L., Jørgensen, J. H., Hansen, P. C. & Jensen, S. H. Implementation of an optimal firstorder method for strongly convex total variation regularization. BIT Numerical Mathematics 52, 329–356 (2012).
Bonettini, S., Zanella, R. & Zanni, L. A scaled gradient projection method for constrained image deblurring. Inverse Problems 25, 015002 (2008).
Bonettini, S. & Prato, M. New convergence results for the scaled gradient projection method. Inverse Problems 31, 095008 (2015).
Coli, V. L., Ruggiero, V. & Zanni, L. Scaled firstorder methods for a class of largescale constrained least square problems. In AIP Conference Proceedings, vol. 1776, 040002 (AIP Publishing, 2016).
Flores, L. A., Vidal, V., Mayo, P., Rodenas, F. & Verdú, G. Parallel ct image reconstruction based on gpus. Radiation Physics and Chemistry 95, 247–250 (2014).
European Reference Organisation for Quality Assured Breast Screening and Diagnostic Services. www.euref.org. Protocol for the Quality Control of the Physical and Technical Aspects of Digital Breast Tomosynthesis Systems.
Bernardi, D. et al. Digital breast tomosynthesis (dbt): recommendations from the italian college of breast radiologists (icbr) by the italian society of medical radiology (sirm) and the italian group for mammography screening (gisma). Radiol. Med. 122, 723–730 (2017).
Hu, Y.H., Zhao, B. & Zhao, W. Image artifacts in digital breast tomosynthesis: Investigation of the effects of system geometry and reconstruction parameters using a linear system approach. Medical physics 35, 5242–5252 (2008).
Tomophan Image Quality Phantom. https://www.phantomlab.com/tomophanphantom. Tomophan Phantom  The Phantom Laboratory.
Computerized Imaging Reference Systems. https://www.cirsinc.com/products/a11/51/br3dbreastimagingphantom/. BR3D Breast Imaging Phantom, Model 020.
Nvidia. Cuda programming guide version 10.0. Nvidia Corporation (2018).
Barzilai, J. & Borwein, J. M. Twopoint step size gradient methods. IMA journal of numerical analysis 8, 141–148 (1988).
Frassoldati, G., Zanni, L. & Zanghirati, G. New adaptive stepsize selections in gradient methods. Journal of Industrial and Management Optimization 4, 299–312 (2008).
Nesterov, Y. Introductory lectures on convex optimization (Kluwer Academic, Dordrecht, 2004).
Epstein, C. Introduction to the mathematics of medical imaging (SIAM, Philadelphia, 2007).
De Man, B. & Basu, S. Distancedriven projection and backprojection in three dimensions. Physics in Medicine & Biology 49, 2463 (2004).
Zanella, R. et al. Towards realtime image deconvolution: application to confocal and sted microscopy. Scientific reports 3, 2523 (2013).
Ruggiero, V., Serafini, T., Zanella, R. & Zanni, L. Iterative regularization algorithms for constrained image deblurring on graphics processors. Journal of Global Optimization 48, 145–157 (2010).
Acknowledgements
The authors thank the anonymous reviewers for their valuable comments and suggestions. This research has been partially supported by GNCSINDAM Italy (Research Projects 2018).
Author information
Authors and Affiliations
Contributions
E.M. and E.L.P., designed the experiments, R.C., J.C.H., E.M. and L.Z., designed the algorithm and executed the experiments, R.C., E.L.P., E.M. and L.Z. analysed the results and wrote the paper. All the authors reviewed and approved the manuscript.
Corresponding author
Ethics declarations
Competing interests
The authors declare no competing interests.
Additional information
Publisher’s note Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The images or other third party material in this article are included in the article’s Creative Commons license, unless indicated otherwise in a credit line to the material. If material is not included in the article’s Creative Commons license and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this license, visit http://creativecommons.org/licenses/by/4.0/.
About this article
Cite this article
Cavicchioli, R., Hu, J.C., Loli Piccolomini, E. et al. GPU acceleration of a modelbased iterative method for Digital Breast Tomosynthesis. Sci Rep 10, 43 (2020). https://doi.org/10.1038/s4159801956920y
Received:
Accepted:
Published:
DOI: https://doi.org/10.1038/s4159801956920y
Further reading

Backward Projection Imaging of ThroughWall Radar Based on Airspace Nonuniform Sampling
Journal of Russian Laser Research (2022)

Technical evaluation of image quality in synthetic mammograms obtained from 15° and 40° digital breast tomosynthesis in a commercial system: a quantitative comparison
Physical and Engineering Sciences in Medicine (2021)
Comments
By submitting a comment you agree to abide by our Terms and Community Guidelines. If you find something abusive or that does not comply with our terms or guidelines please flag it as inappropriate.