# Acoustic coordinated reset therapy for tinnitus with perceptually relevant frequency spacing and levels

## Abstract

Acoustic coordinated reset (CR) therapy based on neuromodulation and neuroplasticity principles has been proposed for the treatment of tonal tinnitus. The original therapy involved periodic delivery of randomly ordered sequences of four low-level tones centered around the frequency of a tone that matched the tinnitus pitch, fT, with fixed ratios relative to fT and delivered several hours/day over several weeks. Here we transform the original CR tone selection method to a more perceptually-relevant equivalent rectangular bandwidth (ERB) frequency scale, the ERBN-number scale. Specifically, we provide a mathematical model that enables calculation of CR tones that accounts for fT- and hearing loss-related ERB widening and ERB overlaps and gaps of CR tone alignments. Further, the model ensures symmetric CR tone alignments based on modelling studies that indicate the effect is optimal if the CR stimuli are symmetrically spaced relative to the tinnitus-related population of abnormally synchronized cortical neurons to activate the adjacent sub-populations. We also present experimentally testable ERB-based CR tone alignment strategies and explain how to use the ERB-based model in experiments, clinical studies, other types of tinnitus sound treatment such as tailor-made notch music training and limitations of our approach.

## Introduction

Tinnitus is the perception of sound in the absence of a sound source external to the person1,2. Some forms of tinnitus, called “objective” or “secondary” tinnitus, can arise from an identifiable sound source in the body, for example arterial blood flow. However, most cases of tinnitus cannot be explained in terms of a physical sound source. “Subjective” or “primary” tinnitus is used to describe the perception of sound that does not have any obvious acoustic origin. Subjective tinnitus is often associated with damage to the peripheral auditory system, but it probably reflects more central nervous system changes that follow the peripheral damage, specifically deafferentiation3,4,5. One such change is abnormal neural synchrony; neurons that normally would show relatively independent patterns of firing may have abnormal responses that are highly synchronized6,7. A second possible change is tonotopic reorganization4; the remapping of acoustic frequency to neural position within an array of tonotopically organized neurons may be altered. However, there is no clear evidence for tonotopic map changes in the auditory cortex of humans with tinnitus8 or partial hearing loss9, although such reorganization may occur for severe or profound hearing loss of long duration, or when there are dead regions in the cochlea10,11.

The tinnitus percept is highly variable including self-reported differences in temporal characteristics (constant vs intermittent), perceptual locations (monaural, binaural and central), loudness (low to high) and subjective descriptors (humming, buzzing, ringing, banging, clicking, sea-like, machine-like, and squeaking and many other subjective descriptors)12. Regardless of these large ranges, the percept can be objectively and reliably characterized by careful psychoacoustic studies of pitch matching, loudness matching and masking using frequency-specific and broad band calibrated external stimuli. The most comprehensive and largest data set describing these characteristics12 reported a frequency-specific tinnitus in over 96.5% of cases with the tinnitus pitch matched to a tone (<1000 Hz, 27.6%; between 1001 and 3000 Hz, 10.3%; between 3001 and 6000 Hz, 17.3%; between 6001 and 8000 Hz, 10.3%; and >8000 Hz, 31%) and the small remainder requiring a non-frequency specific signal (white noise, 3.5%). Though it is possible for pitch matching to be affected by multiple pitch components of the percept, the primary pitch is likely the dominant factor associated with tonotopic locations in the auditory pathways.

The concept of coordinated reset (CR) neuromodulation was developed based on computational studies13. It aims at a desynchronization-induced synaptic modulation by delivery of specific spatio-temporal stimulus patterns, ultimately inducing sustained, long-lasting desynchronization14 via neuroplasticity principles and applies to several neurologic conditions. According to computational studies, the CR approach can be applied by means of both invasive and non-invasive stimulation15,16. Invasive electrical CR deep brain stimulation induced long-lasting motor effects in Parkinsonian monkeys17,18 and Parkinson’s patients19. Analogously, non-invasive acoustic CR stimulation induced long-lasting tinnitus loudness and annoyance reduction effects20 based on the concept that tonal tinnitus arises primarily from abnormal neural synchrony across a defined array of tonotopically organized neurons16.

The theoretical basis of CR therapy for tinnitus is to desynchronize the abnormal neural activity via the sequential presentation of sinusoidal tones with frequencies on either side of the frequency of a tone that is matched in pitch to the tinnitus16,20. A key aim of this approach is to reduce or eliminate the tinnitus percept for a long period after the therapy ceases. A proof of concept study of acoustic CR therapy in 63 patients with chronic tonal tinnitus showed that 12 weeks of acoustic CR treatment, delivered 4–6 h/day, caused significant therapeutic effects compared to baseline. The therapeutic effects persisted through a preplanned 4-week therapy pause and showed sustained long-term effects after 10 months of therapy, giving rise to 75% responders20. Therapeutic effects included a significant reduction of tinnitus questionnaire (TQ) scores21, and Visual Analog Scale (VAS) scores for loudness (VAS-L) and annoyance (VAS-A) compared to initial scores. In addition, acoustic CR therapy showed significant reductions in the frequency of the tinnitus percept and in tinnitus-related abnormal synchrony and effective neural connectivity as assessed by high density EEG recordings20,22,23. The clinical effects of acoustic CR treatment achieved in the prospective, randomized, single blind, placebo-controlled proof of concept study by Tass et al.20 were reproduced in two open label studies without sham control group24,25.

None of the computational studies published so far predicted optimal values of the overlaps of the spatial stimulation profiles (see Discussion). In particular, none of these existing neural network models have taken into account spatial stimulation profiles required to model basic tuning characteristics of the auditory cortex such as co-tuned excitation and inhibition as opposed to lateral inhibition26. By the same token, hearing loss-related spatially varying stimulation profiles at the CR tones have not been taken into account in computational studies. However, in all 1D, 2D and 3D neural network models used for the development of CR stimulation, the stimulation profiles of the different stimulation sites were symmetrical, i.e. overlaps of neighboring stimulation sites were identical, no matter whether the different spatial stimulation profiles were actually overlapping or not (see Discussion). Accordingly, we hypothesize that CR tones that enable symmetric stimulation profile overlaps and gaps of the stimulated cortical sub-populations of neurons will be favorable for desynchronization.

Filters play a key role in fundamental concepts of audiology and psychoacoustics27. For instance, the concept of critical bands introduced an “auditory filter” created by the cochlea in the inner ear27. A second tone delivered within the band of audio frequencies belonging to an auditory filter interferes with the perception of a first tone due to auditory masking27. The equivalent rectangular bandwidth (ERB) is another concept associated with the auditory filter28,29. The ERB uses a rectangular band-pass filter approximation, where the ERB passes the same amount of energy as its corresponding auditory filter, and provides the relationship between the auditory filter, frequency, and the critical bandwidth28,29.

The purpose of the current report is to develop a more advanced, auditory filter-based theoretical basis for the selection of the acoustic CR therapy tones that incorporates both a more perceptually relevant frequency scale than fixed ratios relative to the frequency matched to the tinnitus pitch, as well as known changes to the perceptually relevant frequency scale associated with stimulus level and sensorineural hearing loss. Based on this theoretical approach we analyze the selection of CR tones used so far and design novel CR therapy tone arrangements that can be tested experimentally.

## Results

### ERB framework

The hypothesis aims to select CR tones that stimulate different brain sites e.g. within the primary auditory cortex with symmetrically overlapping stimulation profiles. Because the spatial extent of the area with abnormal neuronal synchrony is unknown, in a first step, the CR tones are symmetrically placed around the frequency of the tinnitus, with two CR tones above and two below the tinnitus frequency20. The tonotopic organization that originates in the cochlea is preserved at all levels of the auditory nervous system up to the primary auditory cortex30. Hence, the location of the sub-population of cortical neurons excited by a given CR tone will depend on the frequency of the CR tone. The spatial extent of the sub-population of neurons will depend mainly on the frequency selectivity of the cochlea at the place tuned to the frequency of the tone. This selectivity can be quantified using the bandwidth of the auditory filter centered at that frequency29. This bandwidth increases with increasing sound level and with increasing sensorineural hearing loss at that frequency31,32,33. For optimal tinnitus reduction, we hypothesize that the relative overlap of the spatial stimulation profiles of neighboring CR tones be identical and symmetrically arranged relative to the profile of the frequency of a tone matched to the pitch of the tinnitus, designated here as fT13,16,34.

The frequencies of the four tones used in a proof of concept study20 and two open label studies24,25 were specified as fixed percentages relative to fT, with two placed below fT and two placed above fT. The frequencies of the tones were defined by:

$${{\rm{f}}}_{1}=0.766{{\rm{f}}}_{{\rm{T}}}$$
(1)
$${{\rm{f}}}_{2}=0.9{{\rm{f}}}_{{\rm{T}}}$$
(2)
$${{\rm{f}}}_{3}=1.1{{\rm{f}}}_{{\rm{T}}}$$
(3)
$${{\rm{f}}}_{4}=1.4{{\rm{f}}}_{{\rm{T}}}$$
(4)

However, it is not clear whether the selected frequencies were optimally spaced. The selection of frequencies based on a perceptually relevant frequency scale might be more appropriate. In the next section we consider the effect of selecting the frequencies of the tones based on such a scale.

We use a frequency scale based on the equivalent rectangular bandwidth (ERB) of the auditory filter, as determined from masking experiments using notched noise or spectrally rippled noise with human listeners28,29. The average value of the ERB for young listeners with normal audiometric thresholds measured at moderate sound levels is denoted ERBN1. Its value in Hz is given by:

$${{\rm{ERB}}}_{{\rm{N}}}=24.7(0.00437\,{\rm{f}}+1)$$
(5)

where f is the center frequency in Hz29. This equation gives a good prediction of ERB values estimated psychoacoustically using masking experiments for center frequencies spanning almost the entire range of human hearing from about 50 Hz35,36 to 15,000 Hz37.

### Estimation of ERB widths from audiometric thresholds

The tonotopic mapping in the cochlea is preserved at higher levels in the auditory system30, so constant steps on the ERB scale correspond to approximately constant spatial distances for neural sub-populations that are tonotopically organized, including the auditory cortex. Recent data suggest that tonotopic spatial maps do not change in the auditory cortex of humans with tinnitus8 or with partial sensorineural hearing loss9. In other words, the mapping of frequency to position in the auditory cortex is not markedly changed by changes in hearing sensitivity associated with sensorineural hearing loss. However, there is considerable evidence showing that the ERB widths of the auditory filters increase with increasing signal level and therefore with increasing sensorineural hearing loss. For a given degree of sensorineural hearing loss at a specific frequency there can be marked individual variations in the value of the ERB, but, on average, the ERB increases with increasing sensorineural hearing loss32,33,38. The spatial extent of the tonotopically organized neural sub-population in the auditory cortex activated by a tone of a given frequency can reasonably be assumed to be proportional to the ERB of the auditory filter centered at that frequency. On average, this ERB will increase with increasing sensorineural hearing loss at that frequency, hence increasing the spatial spread of the activated neural sub-population.

We estimated the ERB-width values at low sensation levels (SLs) for any frequency based on the hearing level at that frequency, using data presented by Moore et al.32. Moore et al.32 provided data on the value of the ERB for center frequencies of 2000, 4000, and 6000 Hz and audiometric thresholds between 0 and 80 dB HL. Based on their data, the dependence of the ERB on the hearing loss h for the range between 0 and 50 dB HL can be modelled with good accuracy by:

$${\rm{ERB}}({h})={{\rm{ERB}}}_{{\rm{N}}}\ast \{1+[{h}/(50{\rm{dB}}\,{\rm{HL}})]\}$$
(6)

For hearing loss greater 50 dB HL there was no such clear relationship32. For frequencies below 2000 and above 6000 Hz, the relationship between amount of hearing loss and auditory filter bandwidth is less well established and this approximation may be less accurate. However, more than 60% of cases of tonal tinnitus have values of fT that fall within the range 2000–6000 Hz12 and there is no reason to expect that the approximation will be substantially different for fT outside of this frequency range.

The effect of hearing threshold on ERB widths is illustrated in Fig. 1. The upper panel (Fig. 1A) shows the values of ERBN for normal hearing according to Eq. 5 (thin line with crosses) and for ERB (h) according to Eq. 6 (thicker line with diamonds) for a representative audiogram for a typical tinnitus patient (Fig. 1B) as a function of frequency. Based on Eq. 6, overlaps or gaps between ERBs of neighboring CR tones, ERB (fj) and ERB (fk), as well as between ERBs of CR tones ERB (fj) and the tinnitus ERB (fT) can be analyzed and quantified (Fig. 2, see Methods).

### Standard CR stimulation

Figure 3 illustrates the effect of fT and of hearing loss on the alignment and overlap of the ERB-wide bands centered around the frequencies of each of the CR tones and fT, as a function of fT. Figure 3A,C,E are for a case of normal hearing with h = 0 dB HL for all audiometric frequencies, and Fig. 3B,D,F are for a case of sensorineural hearing loss using the audiogram shown in Fig. 1B where the magnitude of the hearing loss changes with audiometric frequency. In Fig. 3A,B the abscissa shows frequency in Hz and the ordinate shows the frequencies of the ERB band edges relative to the ERB band edges of fT. The colored areas show the extent of the ERB-wide band around the frequency of each CR tone and around fT. Let us first consider Fig. 3A. Based on the ERBN scale, the mutual arrangement of the standard CR therapy tones, given by Eqs 1420, as well as their relation to the tinnitus ERB, is asymmetrical spacing.

#### Alignment of CR tones relative to the tinnitus ERB at fT (normal hearing)

The ERB-wide bands around f2 and f3 overlap only slightly with the ERB-wide band around fT for high values of fT, because ERBN is relatively small (about 11% of the center frequency) at high frequencies (Fig. 3A). However, for low frequencies, the value of ERBN relative to the center frequency increases (to about 20% of the center frequency at 250 Hz), so the overlap of the ERB-wide bands around f2 and f3 with the ERB-wide band around fT increases (Fig. 3A). The alignment of CR tones 2 and 3 relative to the tinnitus tone is asymmetrical, and this asymmetry depends on fT in that the relative overlap rdERB (3,fT) exceeds rdERB (2,fT) for all values of fT, and the overlap ratio rdERB (3,fT)/rdERB (2,fT) increases with increasing fT (Fig. 3C). Other than with CR tones 2 and 3, the ERB-wide bands around f1 and f4 do not overlap with the ERB-wide band around fT. The alignment of CR tones 1 and 4 relative to the tinnitus tone is asymmetrical in that rdERB (fT, 4) = (bT − a4)/(bT − aT), the gap between the ERB-wide band around f4 and the ERB-wide band around fT, exceeds rdERB (1, fT) = (b1 − aT)/(bT − aT), the gap between the ERB-wide band around f1 and the ERB-wide band around fT, by a factor of at least 2 (Fig. 3C). Furthermore, this gap decreases with increasing fT, so that the asymmetry of the alignment of CR tones 1 and 4 relative to fT depends on fT.

#### Alignment of CR tones relative to each other (normal hearing)

The mutual alignment of the CR tones is asymmetrical, and the asymmetry depends on fT (Fig. 3A,E). The ERB-wide band around f1 and f2 overlaps up to 560 Hz (Fig. 3A,E). For greater values of fT there is a gap between both ERBs. The ERB-wide bands of CR tones 2 and 3 have a minor overlap up to 270 Hz and a gap in between for greater values of fT (Fig. 3A,E). The ERB-wide bands of CR tones 3 and 4 do not overlap for all values of fT (Fig. 3A,E). The gap between CR tones 3 and 4 increases with increasing fT (Fig. 3A,E). Because only neighboring CR tones can overlap it is sufficient to calculate the relative overlap of neighboring ERBs as defined by Eqs 1517 and 1820.

To study the impact of hearing loss on the alignment of the ERB-wide bands of the CR tones and the ERB-wide band of the tone matched to the pitch of the tinnitus frequency, we consider a case of hearing loss using the audiogram shown in Fig. 1B.

#### Alignment of CR tones relative to the tinnitus ERB at fT (hearing loss)

The ERBs of the CR tones as well as the tinnitus ERB are broadened at frequencies fT that have hearing loss. Accordingly, the overlap of the ERB-wide bands around f2 and f3 with the ERB-wide band around fT is greater than in the case of normal hearing and ranges between 40% and 60%, depending on fT (Fig. 3B,D). Consequently, the tinnitus ERB is covered by the ERB-wide bands around f2 and f3 except for fT between 450 Hz and 1500 Hz. The gap between the ERB of CR tone 1 and the tinnitus ERB is smaller, especially for values of fT with greater hearing loss (Fig. 3B,D). The gap between the ERB of tone 1 and the tinnitus ERB is smaller than in the case of normal hearing, in particular at 4000 Hz where the hearing loss is the greatest (Fig. 3B,D).

#### Alignment of CR tones relative to each other (hearing loss)

The hearing loss-induced ERB-broadening causes a tighter alignment of the ERBs of the CR tones. The ERB-wide bands of f1 and f2 overlap for all values of fT. The ERBs of CR tones 2 and 3 overlap at lower values of fT as well as for values of fT with greater hearing loss (Fig. 3B,D). The gap between the ERBs of CR tones 3 and 4 is markedly reduced compared to the normal hearing case, and both ERB bands nearly coincide at 4000 Hz, corresponding to most hearing loss (Fig. 3B,D).

In summary, due to the hearing loss given by the audiogram in Fig. 1B, the ERBs of CR tones 1 and 2 as well as the ERBs of CR tones 2 and 3 overlap for all values of fT. The tinnitus ERB is covered by the ERBs of CR tones 2 and 3 over a wide range of values of fT. Unlike in the normal hearing case, only the ERB of CR tone 4 does not overlap with the ERB of its neighboring CR tone. The mutual overlaps and gaps depend on the amount of hearing loss and, hence, on fT.

### Design of hearing threshold adapted CR tone alignment

In principle, with the methods from Section 3 a variety of different types of CR tone alignments can be designed. To constrain possible options, based on previous studies (see Section 2) we hypothesize that CR tone alignments with identical relative overlap between adjacent CR tone ERBs may be favorable for desynchronization. The procedure for the calculation of the CR tones requires the following input:

• The audiogram in dB HL (including standardized frequencies greater than the typical 8000 Hz limit in most audiograms). We perform a linear interpolation on a logarithmic frequency axis.

• The tinnitus frequency fT.

• The actual value of rdERB, the relative ERB overlap, as defined by Eqs 1517 or 1820, where rdERB (1,2) = rdERB (2,3) = rdERB (3,4).

• The number of CR tones. Standard CR therapy employs 4 tones20. However, translating findings from computational studies in the field of deep brain stimulation (DBS)39, different numbers of tones might be optimal and should, in particular, be adapted to the spatial dimension of the synchronized neuronal population in relation to the ERB-related area of neural tissue activated at the different simulation sites.

### Tinnitus-centered CR tone alignment

In contrast to the tone alignment of standard acoustic CR stimulation, for this type of alignment one CR tone equals the tinnitus tone fT. In the case of a symmetric alignment of e.g. five CR tones, the center CR tone (f3) equals the tinnitus tone fT, as illustrated in Fig. 4A for coinciding ERB edges (i.e. aj+1 = bj and hence rdERB (j, j + 1) = 0). The two lower tones (f1, f2) and the two higher tones (f4, f5) are grouped around fT such that their ERB band edges coincide (Fig. 4A).

### Tinnitus-centered CR tone alignment with coinciding ERB bands

For normal hearing (Fig. 5A) as well as the hearing loss shown in Fig. 1B (Fig. 5B) we vary the tinnitus frequency fT between 250 Hz and 10,000 Hz and determine the CR tones f1, …, f5 with coinciding neighboring ERB edges. Only for normal hearing the ERB width decreases, and the frequency ratios fj/fT tend to 1 from above (f4, f5) and below (f1, f2) as fT increases (Fig. 5A). In contrast, this pattern no longer holds in the hearing loss case, where the ERB width attains maxima at 3000 Hz and 4000 Hz for CR tones 2, 3, 4, and 5 (Fig. 5B).

### Tinnitus-centered CR tone alignment with ERB overlaps

For normal hearing (Fig. 5C) as well as for the same audiogram from Fig. 1B (Fig. 5D) and tinnitus frequency fT between 250 Hz and 10,000 Hz we now calculate the CR tones f1, …, f5 with neighboring ERBs symmetically overlapping by 25%, i.e. rdERB (1,2) = rdERB (2,3) = rdERB (3,4) = 0.25, with rdERB (j, k) from Eqs 1517. Due to the 25% ERB overlap the CR frequency ratios fj/fT are, in general, closer to 1 (for j = 1, 2, 4, 5). The overall picture with narrowing ERB width is preserved for the normal hearing case (Fig. 5C). By the same token, in the hearing loss case maximal ERB widths as well as maximal frequency ratios fj/fT (for j = 4, 5) and minimal frequency ratios fj/fT (for j = 1, 2) are observed at 3000 Hz and 4000 Hz (Fig. 5D).

### Tinnitus-centered CR tone alignment with ERB gaps

To illustrate identical relative ERB gaps, we consider the normal hearing case (Fig. 5E) and use the audiogram from Fig. 1B (Fig. 5F), vary the tinnitus frequency fT between 250 Hz and 10,000 Hz and determine the CR tones f1, …, f5 such that the gaps between neighboring ERBs are identical and equal to 25%, i.e. rdERB (1, 2) = rdERB (2, 3) = rdERB (3, 4) = −0.25, with rdERB (j, k) from Eqs 1517. As a consequence of the 25% ERB gap the CR frequency ratios fj/fT are, in general, more spread, (for j = 1, 2, 4, 5). However, for the normal hearing case ERB widths narrow with increasing tinnitus frequency fT (Fig. 5E). As before, in the hearing loss case maximal ERB widths together with maximal frequency ratios fj/fT (for j = 4, 5) and minimal frequency ratios fj/fT (for j = 1, 2) are observed at 3000 Hz and 4000 Hz (Fig. 5F).

### Tinnitus-enclosing CR tone alignment

Alternatively, CR tones also can be symmetrically aligned on both sides of the tinnitus tone fT, where none of the CR tones coincides with fT. For instance, we may align the 4 CR tones relative to fT so that the coinciding ERB edges of CR tones 2 and 3 coincide with the ERB edges of the tinnitus tone fT, i.e. a3 = b2 = fT (Fig. 4B). Alternatively, we can design an ERB-balanced alignment of CR tones 2 and 3 relative to the tinnitus tone fT that works whether the ERBs of CR tones 2 and 3 overlap, share an edge or are separated by a gap. To this end, we require the tinnitus ERB to be located in the middle of the entire interval formed by the ERBs of CR tones 2 and 3, so that fT is in the middle, which means that fT = (a2 + b3)/2 has to be fulfilled. This condition holds irrespective of whether ERB band edges coincide (Fig. 6A), overlap (Fig. 6B) or are separated by a gap (Fig. 6C).

### Tinnitus-enclosing CR tone alignment with coinciding ERB edges and fT coinciding with ERB edges

For the case of normal hearing (Fig. 7A) and for the audiogram shown in Fig. 1B (Fig. 7B) and tinnitus frequency fT between 250 Hz and 10,000 Hz we determine the CR tones f1, …, f4 so that the ERB edges of CR tones 2 and 3 coincide with the ERB edges of the tinnitus tone fT (Fig. 4B). In addition, ERB edges of neighboring CR tones also coincide (Fig. 4B). For normal hearing the ERB width decreases and the frequency ratios fj/fT continuously tend to 1 from above (f3, f4) and below (f1, f2) as fT increases (Fig. 7A). For the hearing loss case this pattern no longer holds, and the ERB width attains maxima, e.g., at 3000 Hz and 4000 Hz for CR tones 1–4 (Fig. 7B).

### Tinnitus-enclosing CR tone alignment with coinciding ERB edges and with ERB-based balancing around fT

To illustrate the effect of an ERB width-related symmetrical spacing of the CR tones 2 and 3 in relation to the tinnitus ERB (Fig. 6A), for normal hearing (Fig. 7C) and for the audiogram from Fig. 1B (Fig. 7D) and tinnitus frequency fT between 250 Hz and 10,000 Hz, we determine the CR tones f1, …, f4 so that the ERB symmetrical spacing condition fT = (a2 + b3)/2 is fulfilled for each fT (Fig. 6A). Due to the ERB symmetrical spacing around fT, the latter does not coincide with the ERB edges of CR tones 2 and 3, and the entire CR tone and ERB alignment is slightly shifted towards smaller frequency ratios (Fig. 7C,D). Apart from that shift, the overall ERB and CR tone alignment is very similar to the case without ERB symmetrical spacing around fT (compare Fig. 7A,B with 7 C,D).

### Tinnitus-enclosing CR tone alignment with ERB overlaps and with ERB-based symmetry around fT

Here we design CR tones with overlapping ERBs for normal hearing (Fig. 7E) and for the audiogram from Fig. 1B (Fig. 7F) and tinnitus frequency fT between 250 Hz and 10,000 Hz. We determine the CR tones f1, …, f4 so that all relative overlaps of neighboring ERBs are 25%, i.e. rdERB (1,2) = rdERB (2,3) = rdERB (3,4) = 0.25, with rdERB (j,k) from Eqs 1517 (Fig. 7E,F). For the placement of the CR tones relative to the tinnitus frequency fT we use the ERB symmetry condition fT = (a2 + b3)/2 for each fT (Fig. 6B). Except for the ERB overlaps, the overall pattern of CR tone and ERB arrangement (Fig. 7E,F) is similar to the case without gaps (Fig. 7C,D).

### Tinnitus-enclosing CR tone alignment with ERB gaps and with ERB-based symmetry around fT

In contrast to coinciding (Fig. 7C,D) or overlapping (Fig. 7E,F) CR-tone ERBs, here we calculate CR tones f1, …, f4 with symmetric ERB gaps for normal hearing (Fig. 7G) and for the hearing loss case from Fig. 1B (Fig. 7H) and tinnitus frequency fT between 250 Hz and 10,000 Hz. To this end, relative gaps between neighboring CR tone ERBs are chosen to be identical at 25%, i.e. rdERB (1,2) = rdERB (2,3) = rdERB (3,4) = −0.25, where rdERB (j,k) is given by Eqs 1517 (Fig. 7G,H). As in Fig. 7C–F we employ the ERB symmetry condition fT = (a2 + b3)/2 for each fT (Fig. 6C). Similarly as in Fig. 7C–F the ERB balancing around fT slightly shifts the entire CR tone and ERB arrangement towards smaller frequency ratios. Apart from the gaps the overall CR tone and ERB arrangement (Fig. 7G,H) resembles that with coinciding (Fig. 7C,D) or overlapping (Fig. 7E,F) ERB edges.

## Discussion

Tass et al.20 proposed an acoustic CR neuromodulation therapy for tinnitus based on neuroscience principles and computational modelling13,14,16. The computational studies published so far do not provide predictions about optimal values of the overlaps of the spatial stimulation profiles in spatial brain coordinates (see below). However, previous studies indicate that favorable desynchronizing effects can be obtained with spatially symmetrically arranged stimulation profiles, where two cases were considered: (i) The overlap of the stimulation profiles of all pairs of neighboring stimulation sites in the brain was either identical, or (ii) the stimulation profiles of all pairs of neighboring stimulation sites were non-overlapping and equidistant (see below). Accordingly, we hypothesized that because of the strict tonotopic organization of the auditory system from the sensory end organ to the auditory cortex, favorable alignments of the CR tones on the frequency axis should enable an identical overlap of neighboring spatial stimulation profiles in brain spatial coordinates.

To this end, we propose here a more perceptually relevant transformation of the frequency scale for determination of optimal frequency spacing of the CR tones, the ERBN-number scale29. We assumed that the mapping of frequency to position in the cortical tonotopic map is not affected by mild to moderate sensorineural hearing loss, but that the widths of the auditory filters are affected by sensorineural hearing loss, and that these widths influence the degree of overlap of the ERB band around the tinnitus frequency fT and the ERB bands around each of the CR tones, denoted dERB number (j, fT), where j denotes the number of the CR tone. We also considered the change in ERB bandwidth associated with the frequency dependent individual hearing levels of the patient. The theoretical model proposed here advances the original theory16,20 by incorporating a more relevant perceptual frequency scale, the ERB, as well as changes in ERB values associated with stimulus level and magnitude of sensorineural hearing loss up to 50 dB HL. With this ERB-based approach we analyzed the alignment of the original four CR tones, specified as fixed percentages relative to fT (f1 = 0.766fT, f2 = 0.9fT, f3 = 1.1fT, f4 = 1.4fT), employed in a proof of concept study20 and two subsequent studies24,25.

We used the hearing threshold adapted ERB(h) as defined in Eq. 6 instead of a hearing-threshold adapted ERB number scale because for frequency dependent hearing threshold h(f), in general, there is no closed-form formula for ERB number (f). Furthermore, we are focusing on ERB relationships in the vicinity of the tinnitus frequency and, hence, do not need to count the number of ERBs subjacent to fT.

The mutual alignment of the CR tones as well their alignment relative to the tinnitus ERB are highly asymmetric with respect to mutual distance, overlap with tinnitus ERB, and overlap of neighboring CR tones, in the case of both normal hearing and hearing loss (see Section 4.1). In particular, irrespective of the hearing loss, the ERB overlaps of neighboring CR tones are typically not identical (Fig. 3E,F). Furthermore, the alignment pattern of tinnitus ERB and CR tone ERBs strongly vary with tinnitus frequency fT (Fig. 3C,D). Because the majority of patients undergo a significant reduction of fT in the course of a 12-week acoustic CR therapy20,24,25, the ERB overlaps will vary in the course of the treatment, too. If the efficacy of the CR therapy depends on the ERB alignment of the tones and, especially, on the ERB overlaps, treatment efficacy will vary during the course of the treatment, too. The pronounced and, in addition, time-varying asymmetry of the CR tone alignment might be a reason why effects of the original acoustic CR therapy needed weeks to build up20,24,25 and may even explain variability of treatment effects across patients.

Other factors to be considered for enhancing therapeutic effects of acoustic CR neuromodulation include overall therapeutic stimulation equivalent to therapeutic “dose”. Longer stimulation per day that could be implemented into hearing aids being worn by the patient up to 18 hours/day rather than use of a custom device that requires removal of an existing hearing aid already worn by many tinnitus patients that itself provides some tinnitus benefit2. Use of a modified hearing aid to provide the tinnitus therapy tones could also objectively record the total therapy duration using “data logging”, a common feature of contemporary hearing aids. Development of algorithms implemented by contemporary hearing aids also can optimize CR tone selection that include improvements in automated pitch matching and automated incorporation of hearing levels. By the same token, a clinically established dosage regimen, e.g. using ERB-based principles of CR tones alignment might further increase the efficacy of acoustic CR stimulation as illustrated in a computational study40. However, all of these potential improvements still require solid clinical testing.

Because we do not know the exact spatial dimension and location of tinnitus-related abnormally synchronized populations in the individual patient’s brain and their exact relationship to the tinnitus ERB, the optimal number of CR tones as well the most appropriate alignment strategy, sparing the tinnitus frequency fT (Fig. 7) as opposed to including a CR tone at fT (Fig. 5), remain to be determined. In fact, the number of CR tones as well as their alignment relative to fT might have to be personalized and recalibrated during the course of the treatment to achieve optimal stimulation effects. Establishing such methods requires controlled comparative clinical studies, where either different patient groups are treated with different ERB-based CR tone alignments and/or number of CR tones or where one or more patient groups receive different ERB-based CR tone alignments and/or number of CR tones in a cross-over protocol. To reduce related efforts, duration and costs, it might be favorable to first select optimal candidate parameters by means of acute studies with EEG recordings, i.e. by studying electrophysiological effects and after-effects of short stimulation epochs of, say 15 min as, for instance, in a study comparing two different types of acoustic CR stimulation protocols and a sham (i.e. detuned CR) stimulation protocol41. Electrophysiological after-effects, in particular, the post-stimulus time required for resynchronization of e.g. delta band oscillations in the auditory cortex, turned out to distinguish between effective and less effective versions of acoustic CR stimulation41. The most promising parameters revealed in such an acute study should ultimately be tested in a proof of concept study, e.g. providing a comparison to standard acoustic CR stimulation.

Standard acoustic CR stimulation with adequately pitch matched tinnitus tone causes acute clinical effects after 15 min20,41. Clinical improvement can readily and quickly be assessed by means of VAS scores for tinnitus loudness and tinnitus annoyance20,41. An additional, technical more sophisticated means for calibration for acoustic CR stimulation is to measure sound stimulation-induced reduction of EEG delta band power (over the auditory cortex or of reconstructed auditory cortex currents underlying EEG) and prolonged delta resynchronization41. These EEG effects evolve on the same time scale as the clinical effects, i.e. 15 min20,41. In clinical studies it still remains to be shown whether the EEG-based calibration is clinically superior to the simple VAS-based calibration, in particular given that an EEG-based calibration is more time consuming. However, already the simple VAS-based calibration could provide an adequate means to compare different stimulation parameter settings. For instance, by means of a therapeutic sound stimulation app the patient might test different settings and select the favorable one which provides strongest tinnitus relief. For example, one could start with an approach that is reasonably close to the standard acoustic CR20 and aims at improving the treatment outcome step-by-step by comparisons: One could start with 4 CR tones with ERB-based tinnitus-enclosing alignment with coinciding ERB bands of neighboring CR tones, sparing the tinnitus frequency fT (Fig. 7C,D). After 15 min sound stimulation the patient turns stimulation off for 10 min. After the 10 min pause the patient resumes sound stimulation for 15 min, where only one parameter is changed, for instance, by setting the relative ERB overlap to 25% (Fig. 7E,F). During the 15 min stimulation epochs as well as during the 10 min pauses tinnitus loudness and tinnitus annoyance are evaluated by VAS scores (as it was done in an EEG study41). Based on clinically relevant sound stimulation-induced VAS reduction42 the more efficient setting will be chosen, say, the coinciding ERB bands of neighboring CR tones might be better. VAS-based pairwise selections of stimulation settings differing by just one parameter value, e.g. the relative ERB overlap, may be repeatedly be performed and ultimately provide psychophysically optimized stimulation parameters. This approach can equally be applied to optimize other stimulation parameters, for instance, the number of CR tones (e.g. by comparing VAS reduction achieved with 4 as opposed to 6 CR tones) or different tone alignment schemes.

To put our manuscript and subsequent, secondary efforts (see below) into perspective, we note that hearing threshold adapted CR stimulation was invented at Stanford by GRP and PAT43. Their patent application43 comprises the model for hearing loss-induced ERB widening (Eq. 6), ERB overlap and gap calculations (Eqs 1120) using ERB edges (Eqs 710) and an ERB overlap-based CR tone alignment strategy43. Only one publication has been reported that did not use the original CR tones, a study in 25 patients where CR tone frequencies and levels were adapted to hearing thesholds44. According to their Fig. 1 in this study44 the ratios between the CR tone frequencies and the tinnitus frequency fj/fT were (i) adapted to stimulation intensities of up to 80 dB HL, (ii) did not depend on the tinnitus frequency, and (iii) were based on the assumption that tuning curves of CR tones should optimally overlap by 25–30%44. The exact procedure used by44 was not fully disclosed. However, our model for the hearing threshold-related ERB widening (Eq. 6) presented here (i) is valid only up to hearing loss of 50 dB HL32 (see below), (ii) is based on the concept that the ratios between the CR tone frequencies and the tinnitus frequency fj/fT significantly depend on the tinnitus frequency, because the ERB width depends on both tinnitus frequency fT and hearing loss (Eqs 5 and 6)32 and (iii) notes that neither computational nor pre-clinical data suggest that an overlap of the tuning curves of CR tones of 25–30% could be considered as optimal (see below). The differences between our hypothesis presented here and what was used in this clinical study44 means that a meaningful comparison is not possible. In addition, the lack of results obtained with standard CR stimulation20 make it difficult to infer any superiority of the results reported in this clinical study44.

There are limitations to our approach. Eq. 6 is key, because it describes the dependence of the ERB on the hearing loss h for the range between 0 and 50 dB HL. For hearing loss exceeding 50 dB HL existing experimental data do not support this model32. Accordingly, the approach presented here should only be applied for cases up to 50 dB HL. By the same token, Eq. 6 is based on experimental data on the value of the ERB for center frequencies of 2000, 4000, and 6000 Hz32. For frequencies outside the range 2000–6000 Hz, the relationship between amount of hearing loss and auditory filter bandwidth still has to be established. Accordingly, for tinnitus frequencies that are outside the range 2000–6000 Hz Eq. 6 might not be valid and, hence, our approach has to be used with great care. However, because more than 60% of cases of tonal tinnitus fall within the range 2000–6000 Hz12, our approach is justified for a relevant portion of tonal tinnitus patients.

Based on our ERB analysis, the arrangement of the original CR tones used for standard acoustic CR neuromodulation20 is highly asymmetric. Let us consider this aspect from a computational point of view. CR stimulation was computationally developed by means of a number of computational studies in qualitatively different 1D model neural networks34,39,45,46, 1D neural networks with periodic boundary conditions (i.e. rings)15,16, 2D13,14,47,48,49,50,51 and 3D neural networks52,53. In these computational-based studies equally spaced linear39,45,53, circular15,16,54,55 or rectangular13,14,40,47,48,49,50,51 arrangements of the stimulation sites were used. The sub-populations, stimulated by the different stimulation sites, either did not overlap at all13,45, or their overlap depended on stimulation amplitude, spatial arrangement of stimulation sites and spatial stimulation profile13,14,15,16,39,40,46,47,48,49,50,51,52,53,54,55,56,57,58. To this end, different stimulation profiles were used to model current spread in brain tissue caused by electrical stimulation59,60.

The stimulation amplitude turned out to be an important stimulation parameter. For instance, in accordance with theoretical predictions34, invasive electrical CR deep brain stimulation at weak intensities (i.e. with pulse amplitudes of only a third of that used for standard deep brain stimulation) delivered to the subthalamic nucleus in Parkinsonian monkeys caused a long-lasting relief of motor symptoms, persisting for several weeks after cessation of stimulation17,18. In contrast, the after-effect of electrical CR deep brain stimulation at high intensities (i.e. with pulse amplitudes identical to those used for standard deep brain stimulation) was limited to only the first five days after cessation of stimulation17. Stimulation amplitude matters for CR stimulation, irrespective of the stimulation modality. In general, at higher stimulation amplitudes the dynamics of the single stimulated neuron may qualitatively change, e.g. firing rates may change significantly, and model neurons may be blocked34,56. In addition, the stimulation amplitude affects the spatial overlap of the different stimulated sub-populations. In a neural network model without spike-timing-dependent plasticity (STDP), the stimulation amplitude and the spatial decay rate of the stimulus profile were varied, while the equidistantly positioned stimulation sites were fixed34. The spatial decay rate was an approximation of an electric field of a line charge of finite length, relevant for electrical stimulation of brain tissue60. For fixed stimulation amplitude, larger spatial overlaps of the stimulation profiles caused a more pronounced desynchronization (in terms of a uniform distribution of the neurons’ phases), whereas smaller spatial overlaps of the stimulation profiles led to cluster states (neurons do not fire coincidentally, but are arranged in different sub-populations firing at equally distributed times)34. For fixed stimulation profiles, the actual spatial overlap of the sub-populations stimulated at different stimulation sites increases with an increase of the stimulation amplitude34. Hence, for a wide range of the overlap of the stimulation profiles as well as the stimulation intensity and, hence, the actual overlap of the stimulated sub-populations, CR stimulation effectively caused a desynchronization34. However, the presence of spike timing dependent plasticity (STDP) may fundamentally change the spontaneous (i.e. stimulation-free) as well as stimulation-induced dynamics of neural networks56. For instance, in the presence of STDP, CR stimulation may be effective without causing phase resets of sub-populations56. So far, in neural networks with STDP the impact of the spatial decay rate of the stimulus profile on the desynchronizing impact of CR stimulation has not been investigated systematically. Also, in34 CR stimulation with fixed CR sequences was used, whereas acoustic CR stimulation employs CR sequences randomly varying from one cycle to another20. CR variants that differ with respect to their sequence variation have different stimulation effects56.

In summary, symmetric stimulation site arrangements have been used to study and optimize CR stimulation. However, stimulation site arrangements with pronounced asymmetries might, in principle, be feasible, but have hardly been studied computationally so far. Hence, computational studies into that direction might generate relevant predictions that can be tested experimentally.

The acoustic CR therapy requires a frequency-specific tinnitus with one or a few dominant tones. In a study in 1440 patients a frequency-specific tinnitus was reported in over 96.5% of cases12. In the presence of several tones of similar loudness and/or annoyance, the highest tone could serve as first target tinnitus tone fT, since in the majority of patients the tinnitus tone decreases in the course of standard acoustic CR therapy20,25. However, a clinical study is required to investigate whether acoustic CR with ERB-based tone spacing may also cause pronounced changes of tinnitus pitch.

Our new approach allows well defined studies to be conducted to assess the effectiveness of stimulus settings. It is expected that optimization of stimulus protocols using the proposed modifications that consider a perceptually relevant frequency scale and individual hearing thresholds will improve the effectiveness of the original CR therapy by reducing the amount of therapy needed either per day, or number of weeks, and possibly increasing the permanency of the intervention.

Our approach may also be applied to other sound treatments. For instance, tailor-made notch music training (TMNMT) aims to reduce tinnitus-related brain activity by means of lateral inhibition by activating neurons in the neighborhood of the tinnitus frequency fT, but sparing activation of neurons at fT61. To this end, patients listened to notch-filtered music where a frequency band of 1/2-octave around the individual tinnitus frequency fT was removed from the power spectrum of the music, whereas the frequencies at the edge of the notch were amplified61. In a double-blind randomized controlled trial of 3 months duration in 100 patients suffering from chronic tonal tinnitus TMNMT was compared with a placebo (moving frequency notch) condition62. No effect was revealed for the primary outcome measures62. The width of the notch filter and the width of the amplified edge enclosing the notch were neither adapted to the tinnitus frequency fT nor to the hearing threshold. Hence, one might hypothesize that the efficacy of TMNMT may increase by adapting its filter characteristics to tinnitus frequency fT as well as hearing threshold.

## Methods

### Estimation of ERB overlap

The degree of overlap between the ERB-wide frequency band centered on the frequency of a given CR tone and the ERB-wide frequency band centered at fT is determined by two factors: the separation of the ERB edges of the two bands and the ERB widths of the two bands. For a normal-hearing ear and for sounds presented at a moderate level, for example, an ERB band centered at 934 Hz would have band edges at 872 Hz and 997 Hz, while an ERB band centered at 1066 Hz would have band edges at 996 Hz and 1136 Hz. However, for an ear with sensorineural hearing loss, the width of each band is increased, while the mapping of the center frequency of the ERB remains approximately the same. Hence the ERB-wide bands around the two frequencies will overlap by an amount depending on the two frequencies and the magnitude of the hearing loss.

We next consider the notation illustrated in Fig. 2 that shows a representative fT, (black font) and the four CR tones (blue, pink, green and red font) determined by Eqs 14 on a linear frequency scale. Centered on each of these frequencies is the ERB bandwidth using Eq. 5 for each tone with lines extended to the frequency axis that define the center frequency and the frequencies at the edges of the bands. The symbols aj and bj denote the lower and upper edge frequencies of the ERB for the j-th CR tone (j = 1–4) at center frequency fj. The symbols aT and bT denote the lower and upper edge frequencies of the ERB-wide band centered at fT. The frequencies at the upper and lower edges of the ERB band centered around CR tone j, aj and bj, were calculated as:

$${{\rm{a}}}_{{\rm{j}}}={{\rm{f}}}_{{\rm{j}}}-0.5\ast {\rm{ERB}}({\rm{h}}\,{\rm{at}}\,{{\rm{f}}}_{{\rm{j}}})$$
(7)
$${{\rm{b}}}_{{\rm{j}}}={{\rm{f}}}_{{\rm{j}}}+0.5\ast {\rm{ERB}}({\rm{h}}\,{\rm{at}}\,{{\rm{f}}}_{{\rm{j}}})$$
(8)

Similarly, the frequencies at the upper and lower edges of the ERB band centered around fT, aT and bT, were calculated as:

$${{\rm{a}}}_{{\rm{T}}}={{\rm{f}}}_{{\rm{T}}}-0.5\ast {\rm{ERB}}({\rm{h}}\,{\rm{at}}\,{{\rm{f}}}_{{\rm{T}}})$$
(9)
$${{\rm{b}}}_{{\rm{T}}}={{\rm{f}}}_{{\rm{T}}}+0.5\ast {\rm{ERB}}({\rm{h}}\,{\rm{at}}\,{{\rm{f}}}_{{\rm{T}}})$$
(10)

We calculate the degree of overlap of the ERB band centered at the frequency of a given CR tone (f1, f2, f3, and f4) with the ERB band centered at fT. For example, the overlap of the upper edge of the ERB band centered at f2 with the ERB band centered at fT is b2 − aT. If this quantity is negative, then there is a gap and no overlap. If this quantity is zero, the band edges coincide. Positive values indicate the amount of overlap.

Because ERBN increases with frequency f according to Eq. 5, already in the normal hearing condition there is a fundamental asymmetry of the spatial “stimulation profiles” and, hence, the corresponding overlaps have not yet been taken into account in computational studies (see Sec. 2). Other ERB arrangements and corresponding ERB overlaps are conceivable and can be used for computational and experimental studies.

### Tinnitus-centered ERB overlap calculation

Considering the tinnitus ERB to be the target of the CR intervention, we calculate the relative overlap of the ERB-wide band around CR tone j and the ERB-wide band around fT by

$${\rm{rdERB}}\,(1,\,{{\rm{f}}}_{{\rm{T}}})=({{\rm{b}}}_{1}-{{\rm{a}}}_{{\rm{T}}})/({{\rm{b}}}_{{\rm{T}}}-{{\rm{a}}}_{{\rm{T}}})$$
(11)
$${\rm{rdERB}}\,(2,\,{{\rm{f}}}_{{\rm{T}}})=({{\rm{b}}}_{2}-{{\rm{a}}}_{{\rm{T}}})/({{\rm{b}}}_{{\rm{T}}}-{{\rm{a}}}_{{\rm{T}}})$$
(12)
$${\rm{rdERB}}\,({{\rm{f}}}_{{\rm{T}}},\,3)=({{\rm{b}}}_{{\rm{T}}}-{{\rm{a}}}_{3})/({{\rm{b}}}_{{\rm{T}}}-{{\rm{a}}}_{{\rm{T}}})$$
(13)
$${\rm{rdERB}}\,({{\rm{f}}}_{{\rm{T}}},\,4)=({{\rm{b}}}_{{\rm{T}}}-{{\rm{a}}}_{4})/({{\rm{b}}}_{{\rm{T}}}-{{\rm{a}}}_{{\rm{T}}}).$$
(14)

### Therapy tone-centered ERB overlap calculation

Aiming at symmetric ERB overlaps of neighboring CR tones, we determine the relative overlap of the ERB-wide band around CR tone j and the ERB-wide band around tone k by

$${\rm{rdERB}}\,(1,\,2)=({{\rm{b}}}_{1}-{{\rm{a}}}_{2})/\,{\rm{\min }}[({{\rm{b}}}_{1}-{{\rm{a}}}_{1}),\,({{\rm{b}}}_{2}-{{\rm{a}}}_{2})]$$
(15)
$${\rm{rdERB}}\,(2,\,3)=({{\rm{b}}}_{2}-{{\rm{a}}}_{3})/\,{\rm{\min }}[({{\rm{b}}}_{2}-{{\rm{a}}}_{2}),\,({{\rm{b}}}_{3}-{{\rm{a}}}_{3})]$$
(16)
$${\rm{rdERB}}\,(3,\,4)=({{\rm{b}}}_{3}-{{\rm{a}}}_{4})/\,{\rm{\min }}[({{\rm{b}}}_{3}-{{\rm{a}}}_{3}),\,({{\rm{b}}}_{4}-{{\rm{a}}}_{4})]$$
(17)

Equations 1517 employ a normalization by the smaller ERB. This will be favorable if larger overlaps relative to one of the ERBs have to be avoided. Conversely, if large ERB overlaps are desirable, one can use

$${\rm{rdERB}}\,(1,\,2)=({{\rm{b}}}_{1}-{{\rm{a}}}_{2})/\,{\rm{\max }}[({{\rm{b}}}_{1}-{{\rm{a}}}_{1}),\,({{\rm{b}}}_{2}-{{\rm{a}}}_{2})]$$
(18)
$${\rm{rdERB}}\,(2,\,3)=({{\rm{b}}}_{2}-{{\rm{a}}}_{3})/\,{\rm{\max }}[({{\rm{b}}}_{2}-{{\rm{a}}}_{2}),\,({{\rm{b}}}_{3}-{{\rm{a}}}_{3})]$$
(19)
$${\rm{rdERB}}\,(3,\,4)=({{\rm{b}}}_{3}-{{\rm{a}}}_{4})/\,{\rm{\max }}[({{\rm{b}}}_{3}-{{\rm{a}}}_{3}),\,({{\rm{b}}}_{4}-{{\rm{a}}}_{4})]$$
(20)

In principle, one can also use the ERB of the lower (or higher) frequency as reference ERB for normalization. In that case, however, according to Eq. 6 the hearing impairment determines the reference ERB that may render the interpretation of the results more difficult.

## References

1. 1.

Moore, B. C. J. In Tinnitus (eds Eggermont, J. J., Zeng, F.-G., Fay, R. R. & Popper, A. N.) 187–253 (Springer, 2012).

2. 2.

Tunkel, D. E. et al. Clinical practice guideline: tinnitus. Otolaryngology–head and neck surgery: official journal of American Academy of Otolaryngology-Head and Neck Surgery 151, S1–s40, https://doi.org/10.1177/0194599814545325 (2014).

3. 3.

Eggermont, J. J. On the pathophysiology of tinnitus; a review and a peripheral model. Hearing Research 48, 111–123 (1990).

4. 4.

Eggermont, J. J. Cortical tonotopic map reorganization and its implications for treatment of tinnitus. Acta Otolaryngologica Suppl. 556, 9–12 (2006).

5. 5.

Roberts, L. E. et al. Ringing ears: The neuroscience of tinnitus. Journal of Neuroscience 30, 14972–14979 (2010).

6. 6.

Weisz, N. et al. The neural code of auditory phantom perception. Journal of Neuroscience 27, 1479–1484 (2007).

7. 7.

Noreña, A. J. & Farley, B. J. Tinnitus-related neural activity: Theories of generation, propagation, and centralization. Hearing Research 295, 161–171 (2013).

8. 8.

Langers, D. R., de Kleine, E. & van Dijk, P. Tinnitus does not require macroscopic tonotopic map reorganization. Frontiers in Systems. Neuroscience 6, 2 (2012).

9. 9.

Eggermont, J. J. Acquired hearing loss and brain plasticity. Hearing Research 343, 176–190 (2017).

10. 10.

Moore, B. C. J. Dead regions in the cochlea: Diagnosis, perceptual consequences, and implications for the fitting of hearing aids. Trends in Amplification 5, 1–34 (2001).

11. 11.

Irvine, D. R. & Wright, B. A. Plasticity of spectral processing. International Review of Neurobiology 70, 435–472 (2005).

12. 12.

Savastano, M. Characteristics of tinnitus: investigation of over 1400 patients. The Journal of otolaryngology 33, 248–253 (2004).

13. 13.

Tass, P. A. A model of desynchronizing deep brain stimulation with a demand-controlled coordinated reset of neural subpopulations. Biological Cybernetics 89, 81–88 (2003).

14. 14.

Tass, P. A. & Majtanik, M. Long-term anti-kindling effects of desynchronizing brain stimulation: a theoretical study. Biological Cybernetics 94, 58–66, https://doi.org/10.1007/s00422-005-0028-6 (2006).

15. 15.

Popovych, O. V. & Tass, P. A. Desynchronizing electrical and sensory coordinated reset neuromodulation. Frontiers in human neuroscience 6, 58, https://doi.org/10.3389/fnhum.2012.00058 (2012).

16. 16.

Tass, P. A. & Popovych, O. V. Unlearning tinnitus-related cerebral synchrony with acoustic coordinated reset stimulation: theoretical concept and modelling. Biological Cybernetics 106, 27–36 (2012).

17. 17.

Tass, P. A. et al. Coordinated reset has sustained aftereffects in Parkinsonian monkeys. Annals of neurology 72, 816–820, https://doi.org/10.1002/ana.23663 (2012).

18. 18.

Wang, J. et al. Coordinated Reset Deep Brain Stimulation of Subthalamic Nucleus Produces Long-Lasting, Dose-Dependent Motor Improvements in the 1-Methyl-4-phenyl-1,2,3,6-tetrahydropyridine Non-Human Primate Model of Parkinsonism. Brain stimulation 9, 609–617, https://doi.org/10.1016/j.brs.2016.03.014 (2016).

19. 19.

Adamchic, I. et al. Coordinated reset neuromodulation for Parkinson’s disease: proof-of-concept study. Movement disorders: official journal of the Movement Disorder Society 29, 1679–1684, https://doi.org/10.1002/mds.25923 (2014).

20. 20.

Tass, P. A., Adamchic, I., Freund, H. J., von Stackelberg, T. & Hauptmann, C. Counteracting tinnitus by acoustic coordinated reset neuromodulation. Restorative Neurolology and Neuroscience 30, 137–159 (2012).

21. 21.

Goebel, G. & Hiller, W. The tinnitus questionnaire. A standard instrument for grading the degree of tinnitus. Results of a multicenter study with the tinnitus questionnaire. HNO 42, 166–172 (1994).

22. 22.

Silchenko, A. N., Adamchic, I., Hauptmann, C. & Tass, P. A. Impact of acoustic coordinated reset neuromodulation on effective connectivity in a neural network of phantom sound. NeuroImage 77, 133–147, https://doi.org/10.1016/j.neuroimage.2013.03.013 (2013).

23. 23.

Adamchic, I., Toth, T., Hauptmann, C. & Tass, P. A. Reversing pathologically increased EEG power by acoustic coordinated reset neuromodulation. Human brain mapping 35, 2099–2118, https://doi.org/10.1002/hbm.22314 (2014).

24. 24.

Williams, M., Hauptmann, C., Patel, N. & Acoustic, C. R. Neuromodulation Therapy for Subjective Tonal Tinnitus: A Review of Clinical Outcomes in an Independent Audiology Practice Setting. Frontiers in neurology 6, 54, https://doi.org/10.3389/fneur.2015.00054 (2015).

25. 25.

Hauptmann, C. et al. Acoustic Coordinated Reset Neuromodulation in a Real Life Patient Population with Chronic Tonal Tinnitus. BioMed research international 2015, 569052, https://doi.org/10.1155/2015/569052 (2015).

26. 26.

Oswald, A. M., Schiff, M. L. & Reyes, A. D. Synaptic mechanisms underlying auditory processing. Current opinion in neurobiology 16, 371–376, https://doi.org/10.1016/j.conb.2006.06.015 (2006).

27. 27.

Fletcher, H. Auditory patterns. Reviews of Modern Physics 12 ( 1 ), 47–65, doi:10.1103%2FRevModPhys.12.47 (1940).

28. 28.

Moore, B. C. J. & Glasberg, B. R. Suggested formulae for calculating auditory-filter bandwidths and excitation patterns. Journal of the Acoustical Society of America 74, 750–753 (1983).

29. 29.

Glasberg, B. R. & Moore, B. C. J. Derivation of auditory filter shapes from notched-noise data. Hearing Research 47, 103–138 (1990).

30. 30.

Palmer, A. R. In Hearing (ed. B. C. J. Moore) 75–121 (Academic Press, 1995).

31. 31.

Moore, B. C. J. & Glasberg, B. R. Formulae describing frequency selectivity as a function of frequency and level and their use in calculating excitation patterns. Hearing Research 28, 209–225 (1987).

32. 32.

Moore, B. C. J., Vickers, D. A., Plack, C. J. & Oxenham, A. J. Inter-relationship between different psychoacoustic measures assumed to be related to the cochlear active mechanism. Journal of the Acoustical Society of America 106, 2761–2778 (1999).

33. 33.

Moore, B. C. J. Cochlear Hearing Loss: Physiological, Psychological and Technical Issues, 2nd Ed., (Wiley, 2007).

34. 34.

Lysyansky, B., Popovych, O. V. & Tass, P. A. Desynchronizing anti-resonance effect of m: n ON-OFF coordinated reset stimulation. Journal of neural engineering 8, 036019, https://doi.org/10.1088/1741-2560/8/3/036019 (2011).

35. 35.

Jurado, C. & Moore, B. C. J. Frequency selectivity for frequencies below 100 Hz: comparisons with mid-frequencies. Journal of the Acoustical Society of America 128, 3585–3596 (2010).

36. 36.

Jurado, C., Pedersen, C. S. & Moore, B. C. J. Psychophysical tuning curves for frequencies below 100 Hz. Journal of the Acoustical Society of America 129, 3166–3180 (2011).

37. 37.

Zhou, B. Auditory filter shapes at high frequencies. Journal of the Acoustical Society of America 98, 1935–1942 (1995).

38. 38.

Pick, G., Evans, E. F. & Wilson, J. P. In Psychophysics and Physiology of Hearing (eds Evans, E. F. & Wilson, J. P.) 273–281 (Academic Press, 1977).

39. 39.

Lysyansky, B., Popovych, O. V. & Tass, P. A. Optimal number of stimulation contacts for coordinated reset neuromodulation. Frontiers in neuroengineering 6, 5, https://doi.org/10.3389/fneng.2013.00005 (2013).

40. 40.

Popovych, O. V., Xenakis, M. N. & Tass, P. A. The spacing principle for unlearning abnormal neuronal synchrony. PloS one 10, e0117205, https://doi.org/10.1371/journal.pone.0117205 (2015).

41. 41.

Adamchic, I. et al. Acute effects and after-effects of acoustic coordinated reset neuromodulation in patients with chronic subjective tinnitus. NeuroImage. Clinical 15, 541–558, https://doi.org/10.1016/j.nicl.2017.05.017 (2017).

42. 42.

Adamchic, I., Langguth, B., Hauptmann, C. & Tass, P. A. Psychometric evaluation of visual analog scale for the assessment of chronic tinnitus. Am J Audiol 21, 215–225, https://doi.org/10.1044/1059-0889(2012/12-0010) (2012).

43. 43.

Popelka, G. R. & Tass, P. A. Device and method for hearing threshold adapted acoustic stimulation. WTO 10(2016), 009 874.0 (2016).

44. 44.

Wurzer, H., Hauptmann, C. & Adapted Acoustic, C. R. Neuromodulation in Patients With Chronic Tonal Tinnitus and Hearing Loss. Frontiers in medicine 5, 288, https://doi.org/10.3389/fmed.2018.00288 (2018).

45. 45.

Lucken, L., Yanchuk, S., Popovych, O. V. & Tass, P. A. Desynchronization boost by non-uniform coordinated reset stimulation in ensembles of pulse-coupled neurons. Front Comput Neurosci 7, 63, https://doi.org/10.3389/fncom.2013.00063 (2013).

46. 46.

Tyulmankov, D., Tass, P. A. & Bokil, H. Periodic flashing coordinated reset stimulation paradigm reduces sensitivity to ON and OFF period durations. PloS one 13, e0203782, https://doi.org/10.1371/journal.pone.0203782 (2018).

47. 47.

Hauptmann, C. & Tass, P. A. Therapeutic rewiring by means of desynchronizing brain stimulation. Bio Systems 89, 173–181, https://doi.org/10.1016/j.biosystems.2006.04.015 (2007).

48. 48.

Tass, P. A. & Hauptmann, C. Therapeutic modulation of synaptic connectivity with desynchronizing brain stimulation. International journal of psychophysiology: official journal of the International Organization of Psychophysiology 64, 53–61, https://doi.org/10.1016/j.ijpsycho.2006.07.013 (2007).

49. 49.

Hauptmann, C. & Tass, P. A. Cumulative and after-effects of short and weak coordinated reset stimulation: a modeling study. J Neural Eng 6, 016004, https://doi.org/10.1088/1741-2560/6/1/016004 (2009).

50. 50.

Tass, P. A. & Hauptmann, C. Anti-kindling achieved by stimulation targeting slow synaptic dynamics. Restorative neurology and neuroscience 27, 589–609, https://doi.org/10.3233/rnn-2009-0484 (2009).

51. 51.

Tass, P. A. & Hauptmann, C. Long-term anti-kindling effects induced by short-term, weak desynchronizing stimulation. Nonl. Phen. Compl. Syst. 9, 298–312 (2006).

52. 52.

Hauptmann, C. & Tass, P. A. Restoration of segregated, physiological neuronal connectivity by desynchronizing stimulation. J Neural Eng 7, 056008, https://doi.org/10.1088/1741-2560/7/5/056008 (2010).

53. 53.

Ebert, M., Hauptmann, C. & Tass, P. A. Coordinated reset stimulation in a large-scale model of the STN-GPe circuit. Front Comput Neurosci 8, 154, https://doi.org/10.3389/fncom.2014.00154 (2014).

54. 54.

Zeitler, M. & Tass, P. A. Augmented brain function by coordinated reset stimulation with slowly varying sequences. Front Syst Neurosci 9, 49, https://doi.org/10.3389/fnsys.2015.00049 (2015).

55. 55.

Zeitler, M. & Tass, P. A. Anti-kindling Induced by Two-Stage Coordinated Reset Stimulation with Weak Onset Intensity. Front Comput Neurosci 10, 44, https://doi.org/10.3389/fncom.2016.00044 (2016).

56. 56.

Manos, T., Zeitler, M. & Tass, P. A. How stimulation frequency and intensity impact on the long-lasting effects of coordinated reset stimulation. PLoS computational biology 14, e1006113, https://doi.org/10.1371/journal.pcbi.1006113 (2018).

57. 57.

Manos, T., Zeitler, M. & Tass, P. A. Short-Term Dosage Regimen for Stimulation-Induced Long-Lasting Desynchronization. Frontiers in physiology 9, 376, https://doi.org/10.3389/fphys.2018.00376 (2018).

58. 58.

Zeitler, M. & Tass, P. A. Computationally Developed Sham Stimulation Protocol for Multichannel Desynchronizing Stimulation. Frontiers in physiology 9, 512, https://doi.org/10.3389/fphys.2018.00512 (2018).

59. 59.

Ranck, J. B. Jr. Which elements are excited in electrical stimulation of mammalian central nervous system: a review. Brain research 98, 417–440 (1975).

60. 60.

Richardson, K. A. et al. In vivo modulation of hippocampal epileptiform activity with radial electric fields. Epilepsia 44, 768–777 (2003).

61. 61.

Okamoto, H., Stracke, H., Stoll, W. & Pantev, C. Listening to tailor-made notched music reduces tinnitus loudness and tinnitus-related auditory cortex activity. Proceedings of the National Academy of Sciences of the United States of America 107, 1207–1210, https://doi.org/10.1073/pnas.0911268107 (2010).

62. 62.

Stein, A. et al. Clinical trial on tonal tinnitus with tailor-made notched music training. BMC neurology 16, 38, https://doi.org/10.1186/s12883-016-0558-7 (2016).

## Acknowledgements

We gratefully acknowledge support by the Binns family foundation.

## Author information

P.A.T. and G.P.R. conceived ERB-based CR neuromodulation. P.A.T. designed the different CR tone alignements. A.N.S. and P.A.T. calculated ERB overlaps and gaps and prepared figures. P.A.T. and G.R.P. wrote manuscript, A.N.S. approved manuscript.

Correspondence to Peter A. Tass.

## Ethics declarations

### Competing Interests

P.A.T. and G.R.P. are (employee) inventors of the underlying patent WTO 10 2016 009 874.0.

Publisher’s note Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.

## Rights and permissions

Reprints and Permissions

Tass, P.A., Silchenko, A.N. & Popelka, G.R. Acoustic coordinated reset therapy for tinnitus with perceptually relevant frequency spacing and levels. Sci Rep 9, 13607 (2019). https://doi.org/10.1038/s41598-019-49945-w