Biofilm formation causes prolonged wound infections due to the dense biofilm structure, differential gene regulation to combat stress, and production of extracellular polymeric substances. Acinetobacter baumannii, Staphylococcus aureus, and Pseudomonas aeruginosa are three difficult-to-treat biofilm-forming bacteria frequently found in wound infections. This work describes a novel wound dressing in the form of an electrochemical scaffold (e-scaffold) that generates controlled, low concentrations of hypochlorous acid (HOCl) suitable for killing biofilm communities without substantially damaging host tissue. Production of HOCl near the e-scaffold surface was verified by measuring its concentration using needle-type microelectrodes. E-scaffolds producing 17, 10 and 7 mM HOCl completely eradicated S. aureus, A. baumannii, and P. aeruginosa biofilms after 3 hours, 2 hours, and 1 hour, respectively. Cytotoxicity and histopathological assessment showed no discernible harm to host tissues when e-scaffolds were applied to explant biofilms. The described strategy may provide a novel antibiotic-free strategy for treating persistent biofilm-associated infections, such as wound infections.
It is estimated that over 7.5 million people in the United States are affected by chronic wounds, with a projected financial burden on healthcare systems of $20 billion annually1. Individuals with chronic wounds often have an impaired inflammatory response and are unable to efficiently fight infectious microorganisms1,2,3 that acquire nutrients from injured tissue while resisting phagocytosis3,4. Moreover, 65% of all healthcare-associated infections have biofilms present in them according to a 2011 survey by the Centers for Disease Control and Prevention. Biofilms persist in chronic wounds and resist antibiotic treatments5,6,7.
Chronic wounds are associated with bacterial and/or fungal biofilm formation in the wound bed2. Biofilms are clustered microbial communities found on surfaces, embedded within a matrix known as extracellular polymeric substance (EPS) that protects the microbes from some antibiotics8,9. Staphylococcus aureus, Pseudomonas aeruginosa, and Acinetobacter baumannii are commonly associated with biofilm formation in wound infections4,5,10,11,12. Biofilms of S. aureus lead to a chronic infectious state in wounds by stimulating the production of pro-inflammatory cytokines and prolonging the inflammatory phase of the wound1,13. Public and healthcare concerns have arisen because of the increasing prevalence of bacteria resistant to antibiotics and antiseptics10,14. Thus, it is crucial to develop new treatment approaches to combat biofilm-infected wounds that do not involve antibiotics.
Electrical stimulation is broadly described as the act of passing an electric current across a wound bed via electrodes that are placed over the area of interest15,16. Electrical stimulation can increase tissue perfusion, decrease edema, promote angiogenesis, and direct cell migration in wound tissue, all of which can be important aspects of the wound repair process17,18,19. Moreover, such stimulation has demonstrated bactericidal effects against bacteria such as P. aeruginosa20,21 and Escherichia coli22.
The antibacterial mechanism of electrical stimulation is not well understood, which has in turn resulted in poor standardization of its applications16,21,23,24,25,26. A recent study demonstrated inhibition of P. aeruginosa by applying DC voltage (3.5 V); the authors ascribed this effect to electrochemical production of toxic compounds27, but this mechanism was not confirmed23,25,27. Moreover, electrical stimulation has varying effects that depend on how it is employed (e.g., DC versus AC, voltage and current settings, lengths of application time)16,28. For example, one study found that the application of electric current in a copper mesh electrode at a current density of 32 μA/cm2 for 2 hours, 3 times daily resulted in elimination of P. aeruginosa from infected skin ulcers21 while a similar study required the application of electrical current on the same material at a current density of 52 μA/cm2 for 72 hours to eradicate P. aeruginosa in an infected wound model20. Exploring electrochemical processes occurring on electrode surfaces should make it possible to standardize these applications.
Our previous publications28,29 showed that hydrogen peroxide (H2O2) can be produced at controlled concentrations near the surface of an electrochemical scaffold by keeping the scaffold at a suitable fixed potential. Our H2O2-producing electrochemical scaffold (e-scaffold) eliminated large (4-log) microbial communities in biofilms. Moreover, the H2O2-producing e-scaffold successfully eradicated A. baumanii biofilms when combined with a hyperosmotic agent, maltodextrin, after 24 hours of treatment. For the current work, we developed an e-scaffold that produces hypochlorous acid (HOCl) near the electrode surface. HOCl acts faster to eradicate pathogenic bacteria and has lower cytotoxicity towards mammalian cells than sodium hypochlorite (NaOCl) or H2O230. HOCl eradicates bacteria by inhibiting bacterial growth, cell division, and protein synthesis; oxidizing sulfhydryl enzymes and amino acids; decreasing adenosine triphosphate production; breaking DNA; and depressing DNA synthesis22,31,32. HOCl is already used as a wound cleansing agent33, and published work demonstrates that HOCl acts as an antimicrobial, anti-biofilm agent that also promotes wound healing30,34,35,36,37. Sakarya et al. showed that a stabilized HOCl solution decreased cell numbers in biofilms and had favorable effects on fibroblast and keratinocyte migration in vitro compared to povidone iodine34. Previous studies showed that HOCl solution is not painful to patients, eliminates wound odor38, and is effective in cleansing, debriding, and promoting wound healing39. HOCl is considered an excellent adjunct therapy for infection treatment as well as a safe and effective wound cleansing agent40.
The goal of this study was to develop a technology that would allow the sustained delivery of controlled concentrations of HOCl to a biofilm wound bed. To the authors’ knowledge, this is the first report of a potentiostatically controlled wound dressing delivering continuous controlled concentrations of HOCl to eradicate wound biofilms. The e-scaffold is a device that oxidizes chloride ions (Cl−) from phosphate-buffered saline (PBS) or physiological chloride ions (Cl−) in blood and wound fluid to chlorine (Cl2) (Equation 141); the Cl2 then reacts with water to produce HOCl (Equation 2). Low concentrations (below a CT50 of 286 μM30) of HOCl can be antimicrobial without causing cytotoxicity to host tissue40. Our hypothesis is that HOCl generated electrochemically by the e-scaffold can eliminate biofilms without damaging host tissue.
To verify HOCl generation, custom-made microelectrodes were employed to measure HOCl concentrations near the surface of the e-scaffold. S. aureus, A. baumannii and P. aeruginosa biofilms were subjected to HOCl e-scaffold treatment for various lengths of time. The addition of exogenous HOCl for biofilm treatment was also compared to e-scaffold-generated HOCl. Finally, uninfected ex vivo porcine explants were subjected to HOCl- producing e-scaffolds for a preliminary assessment of cytotoxicity, and HOCl e-scaffold efficacy was tested against infected porcine explants. The e-scaffold produces a continuous supply of HOCl at a low concentration, a process that is very different from simply applying HOCl directly to a wound. By using a three-electrode system, it is possible to control reactions on the e-scaffold and quantify the amount of generated HOCl. This allows us to control the concentration of HOCl, which is a significant difference from previous studies.
Results and Discussion
HOCl is generated on the electrochemical scaffold surface
Application of a potential above +1.138 VAg/AgCl generated HOCl electrochemically at ~50 µm from the surface of the e-scaffold (Fig. 1A). Approximately 5 µM HOCl was produced at a potential of +1.5 VAg/AgCl (Fig. 1A). This value is below the reported toxic concentration of HOCl30, and we therefore used +1.5 VAg/AgCl for the remaining experiments. The production of HOCl did not cause variation in pH, as the pH measured near the surface of the e-scaffold was similar to the pH measured 2,000 µm above the surface (Fig. 1B). This is important because HOCl exists in equilibrium with OCl− and Cl2 gas. At pH 7.5 (pKa of HOCl), HOCl is present in equal amounts with OCl−, while Cl2 exists prominently at lower pH42. With the pH of the solution between 3.5 and 6.5, HOCl will be the predominant species relative to either OCl− or Cl2. Operating at pH below 6.5 also has beneficial impacts on wound healing. Chronic wounds have a pH range of 7.15–8.943, but delivery of oxygen from red blood cells is more efficient when the pH is <7.444. Moreover, reports show that lowering the wound pH using Manuka honey can lead to a decreased wound size45. Furthermore, studies show that low pH promotes epithelialization of wounds43,46,47,48 and can inhibit bacterial growth and reduce activities of proteases that are detrimental to wound healing49. Our e-scaffold does not change the pH in the wound bed and thus does not cause undesired effects.
In vitro biofilms were eradicated by the HOCl- generating electrochemical scaffold
All viable S. aureus cells were eradicated after 3 hours of treatment with the HOCl e-scaffold (Fig. 2A), all viable A. baumannii cells were eradicated after 2 hours of treatment (Fig. 2B), and all viable P. aeruginosa cells were eradicated after 1 hour of exposure (Fig. 2C).
E-scaffold-generated HOCl and exogenously added HOCl show similar efficacy
To determine the mechanism by which HOCl kills biofilm cells, we investigated two possibilities: (1) pH change to a highly alkaline state eradicates biofilms, and (2) HOCl is the dominant reactive chlorine species produced during the e-scaffold-mediated oxidation process. Changes in pH are unlikely, as shown in Fig. 1B (the pH stayed at ~6, which is within the range, 6.5–3.5, where HOCl predominates). To determine whether HOCl is the sole effector against biofilms, we compared the efficacy of the e-scaffold to that of HOCl added exogenously either as a single dose or continuously, which mimics the e-scaffold mode of operation. We estimated that a 3-hour exposure to the e-scaffold produced a total of 0.12 mmoles (=17 mM) of HOCl. When we treated S. aureus biofilms for 3 hours with the scaffold or by introducing 0.12 mmoles of HOCl exogenously (at one time or by continuously adding 150.43 mM HOCl solution at a rate of 5.25 μL/minute over a 3-hour period), the biofilms were eradicated (Fig. 2D), consistent with HOCl being the dominant compound killing the biofilm cells. Adding the total amount of HOCl exogenously at one time eradicated biofilm as well as the HOCl e-scaffold; the drawback of this approach is that the initial high concentration exceeds the reported cytotoxicity limit (286–477 μM)30. Continuously adding HOCl presents challenges including how to control the concentration near the biofilm, whereas such control is an inherent property of the e-scaffold.
To test whether electrochemically generated HOCl is the dominant mechanism for biofilm removal, we removed the Cl− from the medium, which prevents the production of HOCl when the e-scaffold is operated at suitable potentials. To do this we used custom-prepared phosphate buffer solution with 32 mM dipotassium phosphate (K2HPO4) and 18 mM monopotassium phosphate (KH2PO4). No KCl was added to the solution, and the pH was adjusted using 1 M NaOH or 1 M HNO3, when needed. When phosphate buffer solution was used, the e-scaffold killing mechanism was neutralized (Fig. 3A). This shows that the presence of Cl− ions is key in the electrochemistry of the HOCl e-scaffold. Although this does not mean that HOCl is the only mechanism involved, we can conclude that HOCl is the dominant species present in the solution and is mostly responsible for in vitro anti-biofilm activity. At pH ranging between 6.3 and 6.7, an average of 90% of Cl− oxidation product is HOCl, with the remainder OCl− (<10%)30, and it is likely that both species contribute to the antibacterial effects of the e-scaffold (Fig. 3B).
We should note that the data in Fig. 3A also show that the use of an Ag/AgCl electrode did not eradicate biofilms, because a polarized electrode was used when Cl− was removed. Moreover, we found that the e-scaffold and control showed nearly identical bacterial counts after 24 hours when the electrode was not polarized (Fig. SI 2). This control experiment demonstrates that none of the e-scaffold components (including the Ag/AgCl reference electrode) had biocidal effects.
Electrochemically produced HOCl does not damage host tissue and eradicates ex vivo biofilms
The efficacy of the HOCl e-scaffold was tested on biofilms grown on porcine dermal explants. After 3 hours of exposure to the HOCl e-scaffold, S. aureus biofilm cell counts dropped by ~6 logs (Fig. 4A). Moreover, the HOCl e-scaffold reduced viable A. baumannii and P. aeruginosa biofilm cells by ~5 log and ~3 log after 2 and 1 hours of treatment, respectively. The HOCl e-scaffold was more effective than a previously described H2O2 e-scaffold28 (~5 log-reduction of viable cells compared to ~3 log-reduction by the H2O2 e-scaffold) both in terms of log reduction and time required. Cytotoxicity studies showed that HOCl- exposed porcine dermal explants did not decline in viability with e-scaffold treatment (Fig. 4B). Moreover, blinded histopathological assessment did not show significant differences between the untreated and treated porcine dermal explants (Fig. 5). These results suggest that a 3-hour exposure to a HOCl e-scaffold does not damage host tissue.
Biofilms contain extracellular polymeric substance (EPS), which restricts the diffusion of molecules and can bind to some antimicrobials. For example, the negatively charged EPS protects cells by preventing positively charged aminoglycoside antibiotics from permeating into the biofilm, possibly through electrostatic interactions50,51. Davenport et al.52 demonstrated that A. baumannii is protected against tobramycin regardless of which species of pathogen produces the EPS. Nevertheless, previous literature showed that the reduction of antibiotic penetration varies depending on the bacteria and antibiotic used53. Biofilms also restrict penetration by biocides other than antibiotics, such as H2O254,55 and HOCl56. It is our contention that by continuously generating HOCl at low concentrations, this restriction can be mitigated. In addition, HOCl is 80–100 times more potent as a germicide than NaOCl because HOCl is neutral and penetrates bacteria more readily than the charged OCl− 57. From the cellular perspective, Gram-positive bacteria have a thicker peptidoglycan layer than Gram-negative bacteria; peptidoglycan is a covalently linked polymer matrix that is composed of peptide-linked β-(1–4)-N-acetyl hexosamine58. This added thickness makes Gram-positive bacteria more resistant to mechanical and chemical stress59, which may explain the longer time required to kill S. aureus biofilm cells (Fig. 2).
The current study used a three-electrode system, whereas earlier work by Sandvik et al. (2013) employed a current-controlled, two-electrode system with which the presence of free chlorine was demonstrated26. The authors concluded that electric current killed S. epidermidis. We used a three-electrode system to control the potential of the working electrode to generate HOCl, which allowed the development of a system that is independent of electrode size and system geometry60.
HOCl is a naturally occurring compound that is one of several oxidative species (including O−2 and H2O2) that are produced by neutrophils and macrophages61,62. Inside bacterial cells, HOCl decreases ATP production in E. coli, as well as in Pseudomonas and Staphylococcus species63. HOCl eradicates cells by inhibiting bacterial growth, cell division, and protein synthesis; oxidizing sulfhydryl enzymes and amino acids; decreasing adenosine triphosphate production; breaking DNA; and depressing DNA synthesis22,31,32. Figure 6 illustrates the multi-mechanism action of HOCl.
Some current literature supports the claim that electrical stimulation accelerates wound healing through galvanotaxis64,65 and provides antimicrobial effects20,21,66. Methods of application include dressings, electrode placement, and practitioner-assisted control65,67,68,69. Bioelectric dressings such as Procellera® and Posifect RD® DC are emerging as a useful method for delivering electrical stimulation. Both dressings are focused on accelerated wound healing67,70,71. Electrical stimulation studies vary with respect to which modality works best against infection; they employ various ranges of DC voltages, current settings, polarity of the electrodes placed on wounds, lengths of application and other variables16,28. Generally, researchers hypothesize that electrical current and electrical fields are responsible for antimicrobial effects20,21,66,72 but acknowledge that the production of toxic compounds may also play a role. Our work demonstrates a specific mode of action in which HOCl is produced near the surface of a specifically tuned e-scaffold and eradicates bacterial biofilms.
The presented work establishes a foundation for an alternative antibiotic-free strategy for treating persistent biofilm infections on wound surfaces through the controlled local generation of HOCl near the infected wound surface.
Electrochemical scaffold assembly
An e-scaffold with a 2.86-mm diameter was used for in vitro experiments. Carbon fabric (Zoltek Companies Inc., St. Louis, MO, Panex 30 PW-06) was used as the electrode material because of its flexibility and electrochemical performance. The working electrode was cut into a circular shape (diameter of 2.86 cm) with a 1.3 × 1.5 cm rectangular tail for attaching the wire (Fig. 7A top). The counter electrode was cut to a short-half-circle shape with a diameter of 2.6 cm with a same-sized rectangular tail (Fig. 7A). This was done because we wanted to use the extra space on the ventral side of the working electrode as a pseudo-reference electrode (for future research projects). A thin layer of silicone rubber was placed between the working and counter electrodes to prevent electrical contact and to improve e-scaffold rigidity. Before assembly, the carbon fabric was treated in 1 M HCl overnight and washed with deionized water. Nylon sew snaps (Dritz, Spartanburg, SC, item #85) were used to connect titanium wires (TEMCo, Amazon.com, catalog #RW0524) to the fabric by looping the wire into the snap before attaching it to the fabric. Silicone adhesive was used to affix the working and counter electrodes, which were kept out of direct contact with one another (Fig. 7B). A custom-made Ag/AgCl reference electrode was constructed according to published protocols73. We confirmed that the reference electrode by itself did not eradicate biofilms. New e-scaffolds were used for each experiment. For quality control, the ohmic resistance between the titanium wire and the carbon fabric was measured using a Fluke 87 V multimeter (Fluke, Everr, WA). E-scaffolds with a resistance higher than 5 Ω were discarded. Ti wire was used to connect carbon fabric to the potentiostat; ~0.5 cm of the Ti wire was exposed to the solution. We used Ti wires with a 150-μm diameter, and the total surface area of the Ti wires (~5 × 10−4 cm2) was significantly less (by a factor of >10,000) than the total surface area of the working electrode (~10.9 cm2). The resistance between the Ti wire and the carbon fabric was measured before and after the experiments, and we detected no change. For our e-scaffolds, the working electrode is equivalent to an anode and the counter electrode is equivalent to a cathode.
HOCl concentration and pH measurements
The HOCl microelectrode was constructed similarly to an H2O2 microelectrode following previously described protocols73. However, to make the HOCl microelectrodes work correctly in the electrical field, we built an integrated reference electrode. This allowed us to place the tip and reference electrodes very close to each other and move them together without changing the distance between them. The construction processes of H2O2 and HOCl microelectrodes are identical; the only difference is the polarization potential used for detecting HOCl. Details of the construction process are published elsewhere74. The microelectrodes had tip diameters of <20 µm. HOCl microelectrodes were polarized to +0.35 VAg/AgCl and calibrated using a sodium hypochlorite (NaOCl) solution diluted in PBS (initial pH = 6.3) with a concentration ranging from 0 to 600 µM. The pH of the calibration solutions did not vary significantly with NaOCl concentration (~0.1 pH unit). An example calibration curve is shown in Fig. SI 2. The microelectrode response was linear from 0 to 157 µM, with a detection limit of 0.25 µM (S/N = 2) and a response time of less than three seconds. Custom-made Ag/AgCl reference electrodes were used for HOCl measurements73. Two independent Gamry© Interface 1000ETM potentiostats (Gamry© Instruments, Warminster, PA) were used to operate the HOCl microelectrode and the e-scaffold. The microelectrode was initially placed on top of the working electrode side of the e-scaffold, which was located using a Zeiss Stemi 2000 stereomicroscope (Carl Zeiss© Microscopy). The e-scaffold was placed in a 6-well plate containing 7 mL of PBS with an initial pH of 6.3. A stationary profile of the HOCl concentration near the surface (~50 µm) of the e-scaffold was measured. A stationary profile was obtained by placing the microelectrode tip ~50 µm above the e-scaffold surface (to measure HOCl concentration) while changing the potential of the working electrode from 0 VAg/AgCl to 2 VAg/AgCl with a scan rate of 0.010 V/s. During the stationary profile measurement, the microelectrode tip was not moved but the electrode potential changed. This allowed us to determine the onset potential of HOCl generation (Fig. 1A). The electrical current response from the HOCl microelectrode during the scan was recorded. This response was later used to calculate the concentration of HOCl. A computer-controlled stepper motor (Physik Instrumente©, PI M-230.10 S, part no. M23010SX) linked to custom LabVIEW software (National Instruments, Austin, TX, USA) was used to manipulate microelectrode movements and control the location of the tip. The vertical range of the motorized actuator was 8 mm with a 40-nm resolution. After the microprofiling system was set up, the HOCl microelectrode was placed within the micromanipulator arm. A stereomicroscope (Zeiss Stemi 2000) was used to locate the tip of the microelectrode held by the micromanipulator. Using the coarse knobs on the micromanipulator, the microelectrode tip was moved above the scaffold, lowered, and positioned to near the e-scaffold surface. The stepper motor was then used to move the microelectrode with a step size of 5 μm until the tip touched the e-scaffold surface, which is used as a reference point for distance. The microelectrode was then moved to ~50 µm above the e-scaffold surface to measure the stationary HOCl profiles.
The pH microelectrode was constructed following previously described protocols73. The microelectrodes had tip diameters of <20 µm and were constructed with liquid ion exchange (LIX, Sigma-Aldrich, 95297) membrane tips73. Custom-made Ag/AgCl reference electrodes with an agar salt bridge at the tip were used for pH measurements. The pH microelectrodes were calibrated in buffer solutions (pH = 4, 7, and 10, Cole-Palmer 910104, 910107, and 910110). The pH measurements were performed while the e-scaffold was polarized at +1.5 VAg/AgCl. The pH profiles were recorded when the current of the polarized e-scaffold reached a steady state (~22 minutes of polarization). The pH microelectrode was moved using the stepper motor in the z-direction at a displacement rate of 100 µm every 10 seconds. The pH sensor output was recorded using a Kiethley© 6537 Electrometer. The pH measurements were performed in an experimental setup and under conditions identical to those of the HOCl microelectrode measurements. The bulk pH before and after the measurements varied between 6.3 and 6.4.
Bacterial strains, media and culture preparation
Pure cultures of S. aureus (ATCC BAA-1747), A. baumannii (ATCC #BAA-1605), and P. aeruginosa PAO1 were used in this study. All organisms were plated on trypticase soy agar (TSA) (Fisher Scientific, DF0369-17-6). Trypticase soy broth (TSB) (Fisher Scientific, DF0370-17-3) was used to culture S. aureus and A. baumannii, and Lennox L Broth (LB) (ThermoFisher, 12780–052) was used to culture P. aeruginosa. To prepare the inoculum, 5 mL of medium and 1 colony of bacteria were added to a 50-mL polypropylene tube and incubated at 37 °C on a shaker table (75 rpm, overnight growth, stationary incubator).
Overnight cultures (18 hours of growth in a 37 °C incubator, shaking at ~90 rpm) of S. aureus, A. baumannii, and P. aeruginosa were diluted to an OD600 of 0.5. One milliliter of a diluted culture was dispensed to each well in a 6-well plate (VWR Intnl Inc, 29442–042) and left for 1 hour for cell attachment. The wells were then washed twice with 1 mL of sterile TSB. TSB (2 mL) was added, and the plates were incubated for 24 hours at room temperature. The medium was replaced, and incubation was continued for 48 hours total. LB was used for P. aeruginosa biofilm growth instead of TSB.
Electrochemical scaffold treatment of in vitro biofilms
After 48 hours of biofilm growth, spent medium on the biofilms was replaced with 7 mL of sterilized PBS solution (32 mM dipotassium phosphate (K2HPO4), 18 mM monopotassium phosphate (KH2PO4), and 0.9% w/v sodium chloride (NaCl), pH 6.3-6.35). The PBS was titrated with 1 M HNO3 or 1 M NaOH in the event that the pH was above or below the desired pH. Next, the e-scaffold was placed directly on top of the biofilm and then polarized at +1.5 VAg/AgCl for the selected treatment time. While we were unable to measure the exact distance between the e-scaffold and the biofilm, this e-scaffold is made of flexible fabric electrodes that conform to the surface below them. The potential at the working electrode was ~ −1.6 VAg/AgCl, which is well below the compliance voltage of the potentiostat, indicating that the counter electrode did not limit the current. The surface areas of the working and counter electrodes exposed to the PBS were ~6.42 cm2 and ~2.5 cm2, respectively. The e-scaffold was conditioned twice (in PBS only and in PBS on biofilm) prior to being polarized by running cyclic voltammetry (0.100 V/s and 0.010 V/s scan rates, 3 cycles, scan from 0 VAg/AgCl to 2 VAg/AgCl). To test whether conditioning affects biofilms prior to polarization, a control experiment with e-scaffold (but no polarization) was performed. When we conditioned the e-scaffold in PBS, there was no detectable killing of the biofilm (Fig. SI 1). Untreated biofilms were used as a control (this control did not include an e-scaffold or a reference electrode). A control experiment without the presence of chloride ions was also performed to determine whether the dominant species produced with the e-scaffold surface is HOCl. The e-scaffold and Ag/AgCl reference electrode were positioned similarly to their positioning in the setup shown in Fig. 8A and polarized for 3 hours (3 hours was chosen for this control because it is the longest treatment duration used in our experiments). This control also determined whether the Ag/AgCl reference electrode itself has any killing effect on the biofilm. After treatment, biofilms were scraped from the surface of the 6-well plate, vortexed to homogenize the biofilms, and washed with 1 mL of PBS three times (with centrifugation in between). The biofilms were harvested after the duration of the treatment times (3 hours, 2 hours, 1 hour, and 30 mins). Viable cell counts were determined using the drop plate method75. Figure 8 shows the experimental setup for testing the e-scaffold on in vitro biofilms and the scheme of electrochemical reactions.
Exogenous addition of HOCl
S. aureus was chosen for this test because its cell membrane is more difficult to penetrate than those of the other bacteria used in this study. S. aureus biofilms were prepared in a 6-well plate using the same protocol described for in vitro biofilm treatment. Exogenous HOCl was added to the biofilm well to reach a final concentration equivalent to the HOCl concentration produced by the e-scaffold. To calculate the theoretical total amount of HOCl produced on the e-scaffold after 3 hours, we integrated the current passed over the treatment period to calculate the charge (C) and divided by the Faraday constant (96,487 C/mol). This gives the moles of electrons exchanged with the solution. We assumed that the oxidation reaction of Cl− ions dominated the measured anodic current according to Equation 1 (assuming 100% Faradaic efficiency). We also assumed that the Cl2 gas generated in the medium was completely converted to free reactive chlorine compounds according to Equation 2. The final concentration of HOCl produced was calculated from the HOCl equivalent of Cl− oxidation divided by the solution volume. Overall, the equivalent of 0.12 mmoles (17 mM) of HOCl in the well was generated by the e-scaffold after 3 hours. This translates to a ~6 mM increase/hour for the 3-hour treatment. Thus, the volumetric flow rate, 314.76 µL/hour, was used for the continuous addition of HOCl to S. aureus biofilms. From the stock solution of NaOCl (10–15% w/v available chlorine, 1.4 M–2.1 M NaOCl, Sigma Aldrich, 424044), a concentration of 150.43 mM was prepared and aliquoted using a 3-mL syringe (BD, Fisher Scientific, 14-823-436). The spent medium on the S. aureus biofilm was replaced with 7 mL of PBS. Then, using a Legato 270 P syringe pump (kd Scientific, 78–8272), HOCl solution was pumped into the biofilm at a volumetric flow rate of 314.76 µL/hour.
From the remaining solution of 150.43 mM, a concentration of 17 mM was prepared in an 8-mL total volume of PBS. About 7 mL of PBS was added to the control biofilm, while 7 mL of 17 mM HOCl was added to S. aureus biofilms and incubated for 3 hours. After the exogenous HOCl treatments, biofilms were harvested by scraping the 6-well plate surface. The biofilms were homogenized and washed 3 times with 1 mL of PBS (with centrifugation in between to collect cell pellets). Cell counts were performed using the drop plate method75.
Explant preparation, cytotoxicity determination, and histopathology
We followed previously established protocols76,77. Ear samples were obtained from a local abattoir (C&L Lockers, Moscow, ID) at the time that private client healthy pigs were butchered. Fresh ears were immediately cooled to 4 °C during transport and kept for less than an hour at this temperature before being processed in the laboratory. The tissues were cleaned with 70% ethanol, and the hair was removed using an electric clipper. The epidermis layer was excised with a Padgett’s dermatome and discarded. Then the dermis layer was excised and placed in a Petri dish containing 20 mL of serum-free Dulbecco’s modified Eagle’s medium (DMEM) (Thermo Scientific, 11995040) with no supplement. The excised dermis layer was sectioned at a thickness of approximately 500 μm, using a Padgett’s dermatome, and punched into 12-mm-diameter discs, excluding layers with visible skin abnormalities (scratches, erosion or scars). Next, the dermis layer punches were used with the dermal side down to seed polycarbonate transwell inserts (Greiner Bio-One North America, Inc., 657641) with a 0.4-μm pore size membrane separating each explant from the outer well prefilled with 2 mL of cell nutrient medium. The nutrient medium consisted of DMEM supplemented with ampicillin (50 μg/mL) and amphotericin B (0.4 μg/mL).
For cytotoxicity tests, uninfected explants (after 24-hour incubation at 37 °C, 95% humidity, and 5% CO2) were exposed to the HOCl e-scaffold for 3 hours, and so were 2 controls: (1) explants processed on the day of the pig ear extraction (day 1 fresh explant) and (2) explants that were not exposed to the e-scaffold (day 2 control explant). After application of the e-scaffold (12 mm diameter working electrode), porcine explant cell viability was quantified using the PrestoBlue cell viability reagent (Thermo Fisher, A-13262) per standard protocols (Life Technologies). Briefly, explants were incubated in 300 μL of 10% PrestoBlue (in DMEM) for 3 hours at 37 °C. Absorbance of the medium was then measured at 570 nm and 600 nm. The percent reduction of PrestoBlue was calculated from this absorbance and the molar extinction coefficient of oxidized and reduced PrestoBlue. A normalized viability score of 100% was assigned to the explant showing the highest percent reduction of PrestoBlue for this control.
Explants exposed to HOCl e-scaffold and PBS only (control) were fixed in 10% formalin for histopathological analysis. For this experiment, the explants subjected to histopathological analysis were independent from the explants used for cytotoxicity testing. The resulting light micrographs were subjected to treatment-blind evaluation by a board-certified anatomic and clinical pathologist. For each case, we used independent explants (3 biological replicates for each treatment). We imaged each sample from each replicate from several locations on the slide (total of 4–6 images), and all the observed images were consistent (only select images are shown here).
Explants were processed as described above. Overnight cultures of S. aureus, A. baumannii, and P. aeruginosa were diluted to an OD600 of 0.5. Explants were infected with 5 μL of diluted bacteria (~1 × 109 CF/mL). After 4 days of biofilm growth (inside an incubator at 37 °C, 95% humidity, and 5% CO2), explants were subjected to e-scaffold treatment as described above for in vitro biofilms (the e-scaffold was overlaid on the explant, which was placed on a transwell insert). Four milliliters of PBS (pH = 6.3) were added into the transwell insert as electrolyte. Treatment times were 3 hours for S. aureus, 2 hours for A. baumannii, and 1 hour for P. aeruginosa. Biofilms from the explants were collected by scraping and vortexing the explant in PBS. Then the biofilm cells were washed 3 times in 1 mL of PBS. Finally, viable cells were evaluated using the drop plate method75.
All experiments were performed at least in triplicate. In vitro and ex vivo CFU data were analyzed using a t-test to identify statistically significant differences. A one-way ANOVA was used with a Tukey test to identify any significant differences in the explant viability test. We considered P ≤ 0.05 to be the threshold for statistical significance. Calculations and statistical analyses were performed using Sigma Plot© (versions 12.0 and 12.5).
The data supporting the findings in this work are available from the corresponding authors upon request.
Chronic Wound Care: The Essentials. (HMP Communications, LLC, 2014).
James, G. A. et al. Biofilms in chronic wounds. Wound Repair Regen 16, 37–44, https://doi.org/10.1111/j.1524-475X.2007.00321.x (2008).
Zhao, G. et al. Biofilms and Inflammation in Chronic Wounds. Adv Wound Care (New Rochelle) 2, 389–399, https://doi.org/10.1089/wound.2012.0381 (2013).
de la Fuente-Núñez, C., Reffuveille, F., Fernández, L. & Hancock, R. E. W. Bacterial biofilm development as a multicellular adaptation: antibiotic resistance and new therapeutic strategies. Current Opinion in Microbiology 16, 580–589, https://doi.org/10.1016/j.mib.2013.06.013 (2013).
Otto, M. Staphylococcal biofilms. Staphylococcal biofilms 322, 207–228 (2008).
Høiby, N., Bjarnsholt, T., Givskov, M., Molin, S. & Ciofu, O. Antibiotic resistance of bacterial biofilms. International Journal of Antimicrobial Agents 35, 322–332, https://doi.org/10.1016/j.ijantimicag.2009.12.011 (2010).
Stewart, P. S. & Costerton, J. W. Antibiotic resistance of bacteria in biofilms. Lancet 358, 135–138 (2001).
Beyenal, H., Lewandowski, Z. & Harkin, G. Quantifying biofilm structure: Facts and fiction. Biofouling 20, 1–23, https://doi.org/10.1080/0892701042000191628 (2004).
Bjarnsholt, T. The role of bacterial biofilms in chronic infections. APMIS 121, 1–51, https://doi.org/10.1111/apm.12099 (2013).
Costerton, J. W., Stewart, P. S. & Greenberg, E. P. Bacterial biofilms: A common cause of persistent infections. Science 284, 1318–1322 (1999).
Surucuoglu, S., Gazi, H., Kurutepe, S., Ozkutuk, N. & Ozbakkaloglu, B. Bacteriology of surgical wound infections in a tertiary care hospital in Turkey. East African medical journal 82, 331–336 (2005).
Percival, S. L. et al. The antimicrobial efficacy of a silver alginate dressing against a broad spectrum of clinically relevant wound isolates. International Wound Journal 8, 237–243, https://doi.org/10.1111/j.1742-481X.2011.00774.x (2011).
Thurlow, L. R. et al. Staphylococcus aureus biofilms prevent macrophage phagocytosis and attenuate inflammation in vivo. Jun 186, 6585–6596 (2011).
Smith, T. L. et al. Emergence of vancomycin resistance in Staphylococcus aureus. The New England Journal of Medicine 340, 493–501 (1999).
Kloth, L. C. Electrical stimulation technologies for wound healing. Advances in wound care 3, 81–90 (2014).
Isseroff, R. R. & Dahle, S. E. Electrical stimulation therapy and wound healing: Where are we now? Advances in Wound Care 1, 238–243, https://doi.org/10.1089/wound.2011.0351 (2012).
Ennis, W. J., Lee, C., Plummer, M. & Meneses, P. Current status of the use of modalities in wound care: electrical stimulation and ultrasound therapy. Plastic and reconstructive surgery 127(1), 93S, https://doi.org/10.1097/PRS.0b013e3181fbe2fd (2011).
Tofail, S. A. M. & Bauer, J. Electrically polarized biomaterials. Advanced Materials 28, 5470–5484, https://doi.org/10.1002/adma.201505403 (2016).
Feldman, D. S., Andino, R. V. & Jennings, J. A. 037 Clinical evaluation of an electrical stimulation bandage (Posifect dressing). Wound Repair and Regeneration 13, A4–A27, https://doi.org/10.1111/j.1067-1927.2005.130215ak.x (2005).
Rowley, B. A., McKenna, J. M., Chase, G. R. & Wolcott, L. E. The influence of electrical current on an infecting microorganism in wounds. Annals of the New York Academy of Sciences 238, 543–551, https://doi.org/10.1111/j.1749-6632.1974.tb26820.x (1974).
Wolcott, L. E., Wheeler, P. C., Hardwicke, H. M. & Rowley, B. A. Accelerated healing of skin ulcers by electro therapy preliminary clinical results. Southern Medical Journal 62, 795–801 (1969).
McKenna, S. M. & Davies, K. J. The inhibition of bacterial growth by hypochlorous acid. Possible role in the bactericidal activity of phagocytes. The Biochemical journal 254, 685 (1988).
Assimacopoulos, D. Low intensity negative electric current in the treatment of ulcers of the leg due to chronic venous insufficiency. Preliminary report of three cases. American journal of surgery 115, 683–687, https://doi.org/10.1016/0002-9610(68)90101-3 (1968).
Ojingwa, J. C. & Isseroff, R. R. Electrical stimulation of wound healing. Journal of Investigative Dermatology 121, 1–12 (2003).
Carley, P. J. & Wainapel, S. F. Electrotherapy for acceleration of wound-healing-low-intensity direct-current. Archives of Physical Medicine and Rehabilitation 66, 443–446 (1985).
Sandvik, E. L., McLeod, B. R., Parker, A. E. & Stewart, P. S. Direct electric current treatment under physiologic saline conditions kills Staphylococcus epidermidis biofilms via electrolytic generation of hypochlorous acid. PLoS One 8, 14, https://doi.org/10.1371/journal.pone.0055118 (2013).
Maadi, H. et al. Effect of alternating and direct currents on Pseudomonas aeruginosa growth in vitro. African Journal of Biotechnology 9, 6373–6379 (2010).
Sultana, S. T. et al. Electrochemical scaffold generates localized, low concentration of hydrogen peroxide that inhibits bacterial pathogens and biofilms. Scientific Reports 5, https://doi.org/10.1038/srep14908 (2015).
Sultana, S. T., Call, D. R. & Beyenal, H. Maltodextrin enhances biofilm elimination by electrochemical scaffold. Scientific Reports 6, https://doi.org/10.1038/srep36003 (2016).
Wang, L. et al. Hypochlorous acid as a potential wound care agent: part I. Stabilized hypochlorous acid: a component of the inorganic armamentarium of innate immunity. Journal of burns and wounds 6, e5 (2007).
Knox, W. E., Stumpf, P. K., Green, D. E. & Auerbach, V. H. The inhibition of sulfhydryl enzymes as the basis of the bactericidal action of chlorine. The Journal of Bacteriology 55, 451 (1948).
Barrette, W. C. Jr., Albrich, J. M. & Hurst, J. K. Hypochlorous acid-promoted loss of metabolic energy in Escherichia coli. Infection and Immunity 55, 2518 (1987).
Rani, S. A. et al. The in vitro antimicrobial activity of wound and skin cleansers at nontoxic concentrations. Adv. Skin Wound Care 27, 65–69, https://doi.org/10.1097/01.ASW.0000443255.73875.a3 (2014).
Sakarya, S., Gunay, N., Karakulak, M., Ozturk, B. & Ertugrul, B. Hypochlorous acid: An ideal wound care agent with powerful microbicidal, antibiofilm, and wound healing potency. Wounds-Compend. Clin. Res. Pract. 26, 342–350 (2014).
Selkon, J. B., Babbt, J. R. & Morris, R. Evaluation of the antimicrobial activity of a new super-oxidized water, Sterilox®, for the disinfection of endoscopes. Journal of Hospital Infection 41, 59–70, https://doi.org/10.1016/S0195-6701 (1999).
Robson, M. C. et al. Hypochlorous acid as a potential wound care agent part ii. Stabilized hypochlorous acid: Its role in decreasing tissue bacterial bioburden and overcoming the inhibition of infection on wound healing. Journal of Burns and Wounds 6 (2007).
Ramey, D. W. & Kinde, H. Commercial and homemade extremely dilute hypochlorous acid solutions are bactericidal against Staphylococcus aureus and Escherichia coli in vitro. J. Equine Vet. Sci. 35, 161–164, https://doi.org/10.1016/j.jevs.2014.12.004 (2015).
Selkon, J. B., Cherry, G. W., Wilson, J. M. & Hughes, M. A. Evaluation of hypochlorous acid washes in the treatment of chronic venous leg ulcers. Journal of Wound Care 15, 33–37, https://doi.org/10.12968/jowc.2006.15.1.26861 (2006).
Liden, B. A. Hypochlorous acid: Its multiple uses for wound care. Ostomy Wound Manag. 59, 8–10 (2013).
Armstrong, D. G. et al. Expert recommendations for the use of hypochlorous solution: Science and clinical application. Wounds-Compend. Clin. Res. Pract., 2–19 (2015).
CRC Handbook of Chemistry and Physics. Chemical Rubber Company handbook of chemistry and physics (1978).
Morris, J. C. The acid ionization constant of HOCl from 5 to 35°. The Journal of Physical Chemistry 70, 3798–3805, https://doi.org/10.1021/j100884a007 (1966).
Percival, S. L., McCarty, S., Hunt, J. A. & Woods, E. J. The effects of pH on wound healing, biofilms, and antimicrobial efficacy. Wound Repair and Regeneration 22, 174–186, https://doi.org/10.1111/wrr.12125 (2014).
Tymoczko, J. L., Stryer, L. & Berg, J. M. Biochemistry. 5th ed. edn., (New York: W.H. Freeman, 2002).
Gethin, G. T., Cowman, S. & Conroy, R. M. The impact of Manuka honey dressings on the surface pH of chronic wounds. International Wound Journal 5, 185–194, https://doi.org/10.1111/j.1742-481X.2007.00424.x (2008).
Leveen, H. H. et al. Chemical acidification of wounds. An adjuvant to healing and the unfavorable action of alkalinity and ammonia. Annals of surgery 178, 745–753, https://doi.org/10.1097/00000658-197312000-00011 (1973).
Nagoba, B. S., Suryawanshi, N. M., Wadher, B. & Selkar, S. Acidic Environment and Wound Healing: A Review. Wounds-a Compendium of Clinical Research and Practice 27, 5–11 (2015).
Schneider, L. A., Korber, A., Grabbe, S. & Dissemond, J. Influence of pH on wound-healing: a new perspective for wound-therapy? Archives of dermatological research 298, 413–420, https://doi.org/10.1007/s00403-006-0713-x (2007).
Nagoba, B. S., Suryawanshi, N., Wadher, B. & Selkar, S. Wounds-Compend. Clin. Res. Pract. 27, 5–11 (2015).
Ishida, H. et al. In vitro and in vivo Activities of Levofloxacin against Biofilm-Producing Pseudomonas aeruginosa. Antimicrobial Agents and Chemotherapy 42, 1641 (1998).
Shigeta, M. et al. Permeation of antimicrobial agents through Pseudomonas aeruginosa biofilms: A simple method. Chemotherapy 43, 340–345 (1997).
Davenport, E. K., Call, D. R. & Beyenal, H. Differential Protection from Tobramycin by Extracellular Polymeric Substances from Acinetobacter baumannii and Staphylococcus aureus Biofilms. Antimicrobial Agents and Chemotherapy 58, 4755–4761, https://doi.org/10.1128/aac.03071-14 (2014).
Singh, R., Ray, P., Das, A. & Sharma, M. Penetration of antibiotics through Staphylococcus aureus and Staphylococcus epidermidis biofilms. Journal of Antimicrobial Chemotherapy 65, 1955–1958, https://doi.org/10.1093/jac/dkq257 (2010).
Elkins, J. G., Hassett, D. J., Stewart, P. S., Schweizer, H. P. & McDermott, T. R. Protective role of catalase in Pseudomonas aeruginosa biofilm resistance to hydrogen peroxide. Applied and Environmental Microbiology 65, 4594–4600 (1999).
Hassett, D. J. et al. Quorum sensing in Pseudomonas aeruginosa controls expression of catalase and superoxide dismutase genes and mediates biofilm susceptibility to hydrogen peroxide. Molecular Microbiology 34, 1082–1093 (1999).
Stewart, P. S., Rayner, J., Roe, F. & Rees, W. M. Biofilm penetration and disinfection efficacy of alkaline hypochlorite and chlorosulfamates. Journal of Applied Microbiology 91, 525–532, https://doi.org/10.1046/j.1365-2672.2001.01413.x (2001).
Hui, Y. H. Handbook of food science, technology, and engineering. (Boca Raton: Taylor & Francis, 2006).
Bugg, T. D. & Walsh, C. T. Intracellular steps of bacterial cell wall peptidoglycan biosynthesis: enzymology, antibiotics, and antibiotic resistance. Nat Prod Rep 9, 199–215 (1992).
Wada, A. et al. Rapid discrimination of Gram-positive and Gram-negative bacteria in liquid samples by using naoh-sodium dodecyl sulfate solution and flow cytometry (rapid gram discrimination with naoh-sds and fcm). 7, e47093, https://doi.org/10.1371/journal.pone.0047093 (2012).
Sultana, S. T., Babauta, J. T. & Beyenal, H. Electrochemical biofilm control: a review. Biofouling 31, 745–758, https://doi.org/10.1080/08927014.2015.1105222 (2015).
Weiss, S. J., Lampert, M. B. & Test, S. T. Long-lived oxidants generated by human neutrophils: Characterization and bioactivity. Science 222, 625–628 (1983).
Robinson, J. Reactive oxygen species in phagocytic leukocytes. Histochemistry and Cell Biology 130, 281–297, https://doi.org/10.1007/s00418-008-0461-4 (2008).
Barrette, W. C., Hannum, D. M., Wheeler, W. D. & Hurst, J. K. General mechanism for the bacterial toxicity of hypochlorous acid: abolition of ATP production. Biochemistry 28, 9172, https://doi.org/10.1021/bi00449a032 (1989).
Thakral, G. et al. Electrical stimulation to accelerate wound healing. Diabetic Foot & Ankle 4, 10.3402/dfa.v3404i3400.22081, https://doi.org/10.3402/dfa.v4i0.22081 (2013).
Ud-Din, S. & Bayat, A. Electrical stimulation and cutaneous wound healing: A review of clinical evidence. Healthcare 2, 445–467, https://doi.org/10.3390/healthcare2040445 (2014).
Canty, M., Luke-Marshall, N., Campagnari, A. & Ehrensberger, M. Cathodic voltage-controlled electrical stimulation of titanium for prevention of methicillin-resistant Staphylococcus aureus and Acinetobacter baumannii biofilm infections. Acta Biomaterialia 48, 451–460, https://doi.org/10.1016/j.actbio.2016.11.056 (2017).
Blount, L. A., Foster, A. S., Rapp, A. D. & Wilcox, A. R. The use of bioelectric dressings in skin graft harvest sites: A prospective case series. Journal of Burn Care & Research 33, 354–357, https://doi.org/10.1097/BCR.0b013e31823356e4 (2012).
Atalay, C. & Yilmaz, K. The effect of transcutaneous electrical nerve stimulation on postmastectomy skin flap necrosis. Breast Cancer Research and Treatment 117, 611–614, https://doi.org/10.1007/s10549-009-0335-z (2009).
Wirsing, P. G., Habrom, A. D., Zehnder, T. M., Friedli, S. & Blatti, M. Wireless micro current stimulation – an innovative electrical stimulation method for the treatment of patients with leg and diabetic foot ulcers. International Wound Journal 12, 693–698, https://doi.org/10.1111/iwj.12204 (2015).
Hampton, S. & Collins, F. Treating a pressure ulcer with bio-electric stimulation therapy. British journal of nursing (Mark Allen Publishing) 15, S14 (2006).
Hampton, S. & King, L. Healing an intractable wound using bio-electrical stimulation therapy. British journal of nursing (Mark Allen Publishing) 14, S30 (2005).
Niepa, T. H. R., Wang, H., Gilbert, J. L. & Ren, D. Eradication of Pseudomonas aeruginosa cells by cathodic electrochemical currents delivered with graphite electrodes. Acta Biomaterialia 50, 344–352, https://doi.org/10.1016/j.actbio.2016.12.053 (2017).
Lewandowski, Z. & Beyenal, H. Fundamentals of Biofilm Research. Second Edition edn, (CRC Press, 2014).
Atci, E., Babauta, J. T. & Beyenal, H. A hydrogen peroxide microelectrode to use in bioelectrochemical systems. Sensors & Actuators: B. Chemical 226, 429–435, https://doi.org/10.1016/j.snb.2015.12.004 (2016).
Chen, C.-Y., Nace, G. W. & Irwin, P. L. A 6 × 6 drop plate method for simultaneous colony counting and MPN enumeration of Campylobacter jejuni, Listeria monocytogenes, and Escherichia coli. Journal of Microbiological Methods 55, 475–479, https://doi.org/10.1016/S0167-7012 (2003).
Lone, A. G. et al. Staphylococcus aureus induces hypoxia and cellular damage in porcine dermal explants. Infection and Immunity 83, 2531–2541, https://doi.org/10.1128/iai.03075-14 (2015).
Lone, A. G. et al. Colonization of epidermal tissue by Staphylococcus aureus produces localized hypoxia and stimulates secretion of antioxidant and caspase-14 proteins. Infection and Immunity 83, 3026–3034, https://doi.org/10.1128/iai.00175-15 (2015).
This research was supported by NSF-CAREER award #0954186. The authors would like to acknowledge Dr. Audrey Schuetz (Mayo Clinic) for her help in reviewing the histopathology slides. Research reported in this publication was supported in part by the National Institutes of Health under award number R01 AI91594. The content is solely the responsibility of the authors and does not necessarily represent the official views of the National Institutes of Health.
The authors declare no competing interests.
Publisher’s note: Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Kiamco, M.M., Zmuda, H.M., Mohamed, A. et al. Hypochlorous-Acid-Generating Electrochemical Scaffold for Treatment of Wound Biofilms. Sci Rep 9, 2683 (2019). https://doi.org/10.1038/s41598-019-38968-y