Abstract
In most applications of ultrashort pulse lasers, temporal compressors are used to achieve a desired pulse duration in a target or sample, and precise temporal characterization is important. The dispersionscan (dscan) pulse characterization technique usually involves using glass wedges to impart variable, welldefined amounts of dispersion to the pulses, while measuring the spectrum of a nonlinear signal produced by those pulses. This works very well for broadband fewcycle pulses, but longer, narrower bandwidth pulses are much more difficult to measure this way. Here we demonstrate the concept of selfcalibrating dscan, which extends the applicability of the dscan technique to pulses of arbitrary duration, enabling their complete measurement without prior knowledge of the introduced dispersion. In particular, we show that the pulse compressors already employed in chirped pulse amplification (CPA) systems can be used to simultaneously compress and measure the temporal profile of the output pulses ontarget in a simple way, without the need of additional diagnostics or calibrations, while at the same time calibrating the oftenunknown differential dispersion of the compressor itself. We demonstrate the technique through simulations and experiments under known conditions. Finally, we apply it to the measurement and compression of 27.5 fs pulses from a CPA laser.
Introduction
Ultrashort laser pulses are a key tool for many applications in science and technology, from materials processing to timeresolved spectroscopy and attosecond science, with wavelength ranges spanning from the THz to the extreme ultraviolet and beyond. Commonly used sources include modelocked laser oscillators, laser amplifiers, optical parametric oscillators/amplifiers and pulse postcompressors. These sources are usually coupled to a temporal pulse compressor, either to obtain a compressed pulse or other pulse duration (or chirp) on target. For example, achieving maximum compression (and hence maximum signal) at the focus of a nonlinear microscope requires precompensating the dispersion introduced by all the optics within the system (e.g. filters, polarizers, and objectives) between the laser source and the sample plane, which invariably requires an adequate pulse compressor. Temporal pulse compressors are also an integral part of a very important class of ultrafast sources, namely chirpedpulse amplification (CPA) laser systems and optical parametric chirped pulse amplification (OPCPA) systems.
Knowing the temporal profile of the pulses on target is very important for optimizing the compression and for establishing the pulse quality and duration, which, in turn, determines the attainable peak power and temporal resolution in timeresolved experiments. Since the advent of ultrashort laser pulses, the community of users has been working in the development of methods to measure such short events, most of them based on nonlinear effects involving the pulse itself or a comparable ultrashort pulse^{1}. Some of these techniques are wellestablished and present many variants.
Recently, the dispersionscan (dscan) technique was introduced^{2,3}, which originally consisted in the measurement of the secondharmonic generation (SHG) and/or sumfrequency generation (SFG) signal produced by a pulse in a nonlinear medium for different and wellknown values of dispersion introduced by a chirped mirror and glass wedge compressor around the maximum compression point. The resulting measurement, known as the dscan trace, encodes information about the spectral phase of the pulses. By coupling this measurement with a suitable mathematical model and optimization algorithm, one is able to fully retrieve the spectral phase of the pulses from the dscan trace without the need of approximations. The dscan technique has several advantages, including a robust inline experimental setup devoid of temporal delay lines, beam splitting or beam recombination, as well as relaxed SHG conversion bandwidth requirements compared to other techniques, as demonstrated in ref.^{3}. Furthermore, dscan is highly robust to noise and can correctly retrieve the spectral phase even when there is no SHG signal for a given frequency, as long as there is SFG signal resulting from the mixing of that frequency with other parts of the pulse spectrum. Additionally, a dscan setup also doubles as a pulse compressor, which together with its measurement capability allows one to simultaneously measure the pulse and to optimize its compression.
Until now, the variable dispersion has mostly been introduced with glass wedges of known material and angle. By controlling the steps of the wedge insertion, the amount of glass crossed by the pulse is easily determined and its dispersion can be obtained from the corresponding refractive index calculated, e.g., from Sellmeier equations. This information is used within the optimization algorithm that calculates the spectral phase of the pulse, which is expressed as a multiparameter unknown variable^{2}. This implementation and approach has enabled, e.g., the generation and measurement of subtwocycle^{4} and even nearsinglecycle^{5,6,7,8} amplified laser pulses postcompressed in a hollowcore fibre (HCF). On the other hand, there is also strong interest in measuring longer, multicycle pulses, as directly produced by Ti:Sapphire CPA laser systems and by OPCPAs, since this is determinant for many applications involving those pulses, such as materials processing, ultrafast spectroscopy, pulse postcompression, highharmonic generation and laser particle acceleration experiments, among others. However, the relatively narrower bandwidth of these pulses, compared to e.g., postcompressed pulses, makes it very difficult in practice to introduce enough dispersion with standard glass wedges to obtain an adequate dscan trace.
On the other hand, the variable pulse compressors used in ultrafast CPA systems, which usually employ diffraction gratings, prisms, grisms or combinations of some of these elements, are capable of introducing much larger dispersion than glass wedges and are designed to compress the output pulses of the specific laser system they are part of, providing optimized dispersion and minimum pulse duration for a specific position of the compressor. However, the compressors in CPA laser systems are usually not calibrated, in the sense that neither their exact dispersion at a given position nor their (differential) dispersion change per step are known to arbitrary order, since this is not a necessary requirement for their operation.
In this work, we present a selfcalibrating dispersionscan technique capable of measuring and compressing ultrashort laser pulses over a broad range of pulse parameters, as produced by a broad range of ultrafast sources, and where prior knowledge of the amount of dispersion introduced for each position or step of the compressor is not required. The reconstruction algorithm is capable of retrieving not only the ultrashort pulse, but also of obtaining the differential dispersion introduced during the scan (hence the term selfcalibrating dispersionscan), which is an important property with great practical advantages with respect to previous works^{9,10,11}. As a consequence, dscan measurements can be performed with virtually any dispersion scanning system. For instance, the noncalibrated internal pulse compressors in CPA laser systems can be used to directly measure their output pulses, which effectively enables extending the dscan technique to longer, narrower bandwidth laser pulses, with durations from tens of femtoseconds up to several picoseconds and more, while maintaining the distinctive advantages of the technique.
Results and Discussion
Theoretical demonstration of selfcalibrating dscan with ~30fs pulses
The selfcalibrating dscan operation (detailed in the Methods Section) is based on the numerical retrieval of the spectral phase of the pulses using a nonlinear optimization algorithm, where the spectral phase is treated as a multiparameter unknown variable, and where the unknown dispersion of the dispersion scanning system or compressor is described by a theoretical model of its functional dependence on the compressor position or step. The fundamental spectrum of the pulse can be measured directly or retrieved from the measured dscan trace^{2,7}. During numerical optimization, the dscan trace is initially simulated for a random guess phase and then this phase is iteratively modified until the simulated dscan trace converges to the experimental one using a merit function based on the comparison between both traces. If the intensity calibrated fundamental spectrum is independently measured, there is no need to calibrate the intensity of the nonlinear signal (e.g., SHG) used for the dscan trace, since we can use the dispersion marginal of the trace or a generalized wavelengthdependent error function to obtain this calibration^{2,6}. In the present work, our optimization algorithm retrieves not only the spectral phase of the pulse but also the unknown phase introduced by the compressor during the scan. For the pulse, each point of its frequencydependent phase is treated as an independent variable, which enables measuring an arbitrary spectral phase. For the compressor dispersion, the phase is described by a Taylor series truncated to a given arbitrary order. The pulse and its dscan trace are not affected by constant and linear spectral phase terms, so these two terms are not considered. Actually, for many relevant situations, we can model the dispersion introduced per step as the combination of only two unknown parameters: the group delay dispersion (GDD) and the 3rdorder dispersion (TOD). For example, the prism compressor used in the final part of this work will be modeled with GDD and TOD, in agreement with previous works^{12,13}. Please note that these approximations refer to the variable part of the compressor dispersion, which tends to be wellbehaved (or even monotonic, as in the case of a chirped mirror and wedge compressor), and not to the spectral phase of the pulse itself, which can be arbitrarily complex. If necessary, more terms can be added (e.g., 4thorder dispersion). The compressor dispersion can be also described by a complex function of frequency, but this is usually not necessary in most practical realworld situations, as demonstrated further below.
We start by presenting the selfcalibrating dscan method through numerical simulations assuming 28 fs pulses, from which we retrieve both the compressor dispersion (Fig. 1) and the pulse measurement (Fig. 2). Such pulse durations are usually beyond the range of a chirped mirror and wedge compressor, so the scan must rely on another type of compressor, such as a grating or a prism compressor. Then, we demonstrate the technique through experiments with broadband pulses using glass wedges of known dispersion and ~7 fs pulses and compare it to the standard dscan technique. Finally, we apply it to the measurement of pulses with approximately 28 fs from a CPA laser system.
As a first test set (30 simulations) we calculated dscan traces for simulated ultrashort pulses using the same spectral resolution for the SHG signal as in the CPA laser pulse measurements presented further below, and used a measured fundamental spectrum from the same system. Also, we used 51 values of prism insertion per scan, as in those experiments. The spectral phase imparted on the simulated initial pulse was a combination of GDD = 200 fs^{2}, TOD = +5000 fs^{3}, plus an oscillatory term given by \(0.25\,\cos \,[100(\omega {\omega }_{0})+\pi /10]\), where \(\omega \) is the frequency and \({\omega }_{0}\) is the central frequency of the laser pulse. These parameters result in a theoretical trace [Fig. 2(a)] that is very representative of actual systems.
In the simulations, for the compressor dispersion we used the following values of GDD/L and TOD/L (referring to the dispersion of the grating or prism compressor per unit insertion length): GDD/L = 150, 200, 250, 300, 350, and 400 fs^{2}/mm; TOD/L = −1000, −500, 0, 500, and 1000 fs^{3}/mm. Using always the same simulated pulse described before, and for each combination of GDD/L and TOD/L, we calculated the corresponding dscan trace, which produced the set of 30 simulations. We then used the selfcalibrating retrieval, which gave us the spectral phase as well as the GDD/L and TOD/L for each case. In all the retrievals, the initial guess pulse phase was flat (Fouriertransform limited pulse) and the guess GDD/L and TOD/L were 250 fs^{2}/mm and 0 fs^{3}/mm, respectively. In Fig. 1, we show the comparison between the values used in the simulation and the values obtained with the selfcalibrating dscan algorithm. The achieved agreement demonstrates that it is possible to apply this scheme to compressors with unknown dispersion at least up to the 3^{rd} order. As an example, we show in Fig. 2(a) the theoretical dscan trace created to test the numerical retrieval for simulation number 15 (GDD/L = 250 fs^{2}/mm and TOD/L = 1000 fs^{3}/mm). The retrieved selfcalibrating dscan trace is plotted in Fig. 2(b) to illustrate the achieved convergence and agreement.
We then analyzed the capability of our proposed scheme to retrieve the unknown pulse. The calculated spectral phase matches the theoretical one, as shown in Fig. 2(c). The same stands for the comparison between the theoretical and the retrieved intensity and phase in the temporal domain [Fig. 2(d)], which indicates that the pulse is accurately retrieved. Also, the obtained pulse duration of 28.0 ± 0.1 fs (FWHM) is in excellent agreement with the theoretical pulse duration of 28.0 fs. The statistical analysis was done over the 30 retrievals obtained for the different differential dispersions shown in Fig. 1, with the error calculated as the standard deviations of the retrieved functions (spectral phase and temporal intensity and phase) for each value of the independent variable (wavelength and time, respectively). In the plots, this error is magnified 10 times in order to be distinguishable. The low values of these errors are indicative of the precision of the retrieval in these simulations.
Experimental demonstration and comparison with standard dscan: retrieval of the pulse and of the compressor dispersion
In a first experimental test, we used a Ti:Sapphire fewcycle ultrafast laser oscillator with a central wavelength of around 800 nm and a Fourier limit of approximately 7 fs, including chirped mirrors and glass wedges to compress the pulse and perform the dscan, as done previously^{2,14}. The dscan trace was measured by recording the SHG spectra generated by the output beam focused on a BBO crystal (10 µm thick, cut for type I SHG) using a fibrecoupled spectrometer (HR4000, Ocean Optics Inc.) as a function of the dispersion introduced by fixed insertion steps of BK7 glass wedges (8° angle), which was calculated from Sellmeier equations. This allows comparing the results from the standard dscan algorithm^{2} with the present selfcalibrating dscan, where we have modeled the dispersion of BK7 by fitting to orders 2–3 (GDD and TOD) and orders 2–4 (up to 4thorder dispersion) of the corresponding Taylor expansion.
The results for the pulse retrieval are shown in Fig. 3, where the experimental trace in Fig. 3(a) is compared with the selfcalibrating trace in Fig. 3(b). The pulse is retrieved for the material insertion corresponding to the optimum compression (the shortest output pulse). For the three cases shown (known dispersion; fit to orders 2–3; fit to orders 2–4), the calculated spectral phases show a very good agreement [Fig. 3(c)], as do the pulse temporal intensity and phase [Fig. 3(d)]. The retrieved pulse durations are, respectively, 7.4, 7.3 and 7.3 fs (FWHM). To complete this first experimental demonstration of the validity of the selfcalibrating method, we compared the calibration of the dispersion obtained by the algorithm with the values for BK7 calculated from Sellmeier equations. Since the linear phase term is irrelevant here, we calculated the variation of the group delay, \({\tau }_{g}\), per mm, referred to the central wavelength of 800 nm. These results are given in Fig. 3(e), where a very good agreement is observed.
Application of selfcalibrating dscan to a real CPA laser system
As stated previously, the selfcalibrating dscan technique can be applied to any CPA or OPCPA laser system, by using its internal (and in most cases uncalibrated) compressor as the dispersion scanning module. Therefore, the laser system can be used to perform a selfdiagnostic, where both the output pulses and the compressor calibration can be simultaneously obtained. To demonstrate this, we applied the technique to a CPA Ti:Sapphire laser system, comprising a doublepass compressor with two Brewsterangled prism pairs, delivering pulses with a Fouriertransform limit of 22 fs (FWHM) centered at 785 nm (Femtolasers Femtopower Compact Pro CEP).
The dispersion is scanned by varying the prism insertion in the beam path (Fig. 4). The Brewsterangled prisms are made of LaKN16 glass, have an apex angle of approximately 60°, are separated by approximately 2 m and are aligned for minimum deviation. After the prism compressor, the laser output was focused by a goldcoated offaxis parabola (OAP) onto a nonlinear crystal (BBO, 10 µm thick, cut for type I SHG). The SHG signal was filtered with highpass optical filters (Schott BG filters) and acquired with a fibrecoupled spectrometer (S2000, Ocean Optics Inc.). Additionally, we recorded the spectrum of the fundamental pulse with a second spectrometer (PC2000, Ocean Optics Inc.). For the dscan measurement, we varied the prism insertion along a range of 20 mm with a linear stage in steps of 0.4 mm; less (more) prism insertion means more (less) negative chirp. This way, we produced a complete dscan trace from 51 values of prism insertion, in which we tracked the SHG signal from negatively to positively chirped pulses, going through optimum compression. We also ensured there were no nonlinear effects inside the prism compressor (see the Methods Section). Under these conditions, we took five experimental dscan traces for which the spectral phase, GDD/L and TOD/L were all retrieved in the manner explained previously. The experimental and retrieved dscan traces are shown in Figs. 5(a) and (b), respectively, for one of the experimental measurements.
The statistics of the compressor dispersion calculated from the optimization algorithm are GDD/L = 277.5 ± 2.5 fs^{2}/mm and TOD/L = 175 ± 15 fs^{3}/mm at 785 nm, so both parameters are obtained with reasonable precision. The absolute errors are calculated from the standard deviation of the 5 independent measurements and retrievals, which is the same data set used to calculate the errors in the pulse retrieval in the spectral and temporal domains (the standard deviation is calculated for the 5 retrieved values of the corresponding functions at each wavelength and time, respectively).
In order to ascertain the accuracy of the retrieved differential GDD and TOD of our compressor, we have estimated these parameters using analytical expressions for the phase introduced by a fourprism compressor as a function of the interprism distances and insertion^{15}, also taking into account the propagation in the prisms. For our prism material (LaK16 glass) and geometry (design wavelength of 800 nm; distance between prism pairs of 205 cm) we obtained values of GDD/L = 273 fs^{2}/mm and TOD/L = 187 fs^{3}/mm at 785 nm, in very good agreement with the dispersion retrieved by the selfcalibrating d scan algorithm.
In Fig. 5(c), we show the measured spectrum and the retrieved spectral phase of the pulse, including its standard deviation. Like in the simulations presented before, the deviation is small and only becomes significant at the spectral edges, as expected, so it does not affect the temporal pulse intensity. The spectral phase of the pulse is given for the prism insertion that corresponds to the shortest pulse duration and presents a remaining negative cubic phase term, which produces prepulses in the temporal domain, as seen in Fig. 5(d). This residual phase is quite common in prism compressors (which are known to overcompensate the TOD of materials) and in principle can be minimized, for a given compressor position, by adjusting the TODcompensating mirrors within the laser system^{16}. The optimum pulse duration is 27.5 ± 0.2 fs (FWHM).
To conclude, we presented and demonstrated a selfcalibrating dscan technique, which allows measuring ultrashort pulses without prior knowledge of the dispersion introduced in the scan. Numerical simulations and experimental measurements of a fewcycle laser oscillator using a chirped mirror and glass wedge compressor showed that the retrieval algorithm calculates both the spectral phase of the pulse and the differential dispersion of the compressor. The technique was then applied to pulses produced by a CPA laser, where the dispersion scan was introducing by varying the internal pulse compressor of the system. Together with the pulse characterization, the selfcalibrating dscan method simultaneously calibrated the amount of dispersion introduced per step of the compressor and optimized the compression to find the shortest pulse duration generated from the system, in this case 27.5 fs. The present work demonstrates that the dscan technique can be extended to any ultrashort pulse duration range, and applied to any CPA or OPCPA system, as well as to any laser system coupled to an arbitrary compressor. Such systems can therefore be selfdiagnosed with the help of only a nonlinear medium and a spectrometer. The selfcalibrating dscan also allows performing ontarget diagnostics by placing the nonlinear medium at the target location and measuring the dscan trace at that point, without the need of additional calibrations or diagnostics, hence providing complete temporal information about the pulse effectively delivered by the laser system to the experimental target.
Methods
The essential information required to understand this work has already been presented along the manuscript, in particular regarding the experimental implementation of the dscan technique and the corresponding retrieval algorithm used to analyze the data, since it constitutes the core of the work. Here, we provide further details on the experiments and the algorithm employed for the selfcalibrating dscan retrieval.
Experimental details
All the experiments were performed using different outputs of the same Ti:Sapphire laser system from Femtolasers Produktions GmbH. The ultrafast oscillator (Femtosource Rainbow CEP) output is carrierenvelope phase (CEP) stabilized and delivers pulses with a Fouriertransform limit of around 7 fs at a repetition rate of 80 MHz, with a central wavelength of around 800 nm and an energy per pulse of up to 2.5 nJ. The CPA amplified output (FemtoPower Compact PRO CEP) provides 1 kHz, ~1 mJ pulses with a Fouriertransform limit of ~22 fs.
In the experiments, the SHG signal was optimized using the criterion of maximum signal and we verified that it was spatially homogeneous by laterally translating the collection fibre of the spectrometer.
In the particular case of a prism compressor, it is important to avoid nonlinear effects due to propagation, e.g. in the bulk of the prisms, which may vary along the dscan measurement, since the pulse intensity and path length within the bulk depend on the prism insertion. This is a wellknown issue with prism compressors, where maximum compression can occur too close to, or even inside, the last prism thus giving rise to unwanted selfphase modulation effects^{17}. Normally, this will not happen when using grating compressors, where the corresponding Bintegral is much lower. To avoid this problem, the output energy of the amplifier was reduced by decreasing the energy of the CPA pump laser together with further attenuation using beamsplitters placed after the amplification and before the prism compressor. This way we obtained a pulse energy of 20 μJ at the output. We checked that the dscan trace did not vary when reducing the energy by a factor of two (i.e. 10 μJ) and that the fundamental spectrum was also unmodified during the scan, which is indicative of the absence of nonlinear effects. Note that this energy reduction is not necessary when using lower energy laser systems or those equipped with a diffraction grating compressor, where the nonlinear phase accumulated during propagation in the compressor is in principle very low.
Selfcalibrating dscan and retrieval algorithm
The selfcalibrating dscan is used to measure the temporal (and spectral) amplitude and phase of the laser pulses. The electric field of the pulse is defined in the spectral domain as \(E(\omega )=A(\omega )\exp \{i\phi (\omega )\}\), where \(\omega \) is the frequency, and \(A(\omega )\), \(\phi (\omega )\) are the spectral amplitude and phase, respectively. The amplitude is directly obtained from the power spectral density defined as \(S(\omega )={A(\omega )}^{2}\), which is calculated from the signal measured with the spectrometer \(S(\lambda )\), including the correction scaling factor of \({\lambda }^{2}\) that ensures the conservation of the integral of the spectral distribution. During the dispersion scan, the compressor adds dispersion to the pulse, while the SHG signal is measured. The uncalibrated compressor (e.g. based on prisms or gratings), or the bulk material (e.g. wedges), introduces a phase \(\varphi (\omega )\) that depends on the compressor position (e.g. the prism or grating translation), or the inserted thickness of material (e.g. the wedge insertion), represented by \(z\). This phase is modeled by a truncated Taylor series as given by
where \({\omega }_{0}\) is the central frequency. As mentioned previously, the first two terms of the series do not affect the dscan trace and are disregarded, while the coefficients GDD, TOD, etc., are calibrated from the algorithm.
The measured SHG dscan trace as a function of the frequency and the compressor z position can be expressed as
where \( {\mathcal F} \) denotes the Fourier transform. The phase of the pulse, \(\phi (\omega )\), is sampled over a number of points (typically 32–48 points) that are evaluated as unknown variables. Then, the phase is interpolated to the frequency axis so that it is used within the algorithm to retrieve the pulse in the temporal domain. The unknown parameters of the compressor (GDD, TOD, etc.) and the sampled phase of the pulse therefore form a set of multiple unknown variables.
The calculation of those variables is done with a multivariable optimization algorithm (e.g., NelderMead Simplex^{2} or, in this case, the LevenbergMarquardt algorithm), in which the simulated dscan trace, \({S}_{simul}(\omega ,z)\), is computed and compared to the measured trace, \({S}_{meas}(\omega ,z)\). In this iterative algorithm, the difference between the traces is minimized (using a merit function^{2}) considering all the unknowns as a whole (sampled spectral phase and the dispersion of the compressor), therefore retrieving the pulse and calibrating the compressor at the same time. The pulse is calculated for every position of the compressor, z, so we can obtain the pulse measurement for any particular position, including for optimum compression (position that gives the shortest pulse duration and highest peak power).
We have tested our algorithm with experimental measurements, and we have systematically found the same solution, within the tolerance error, even when using initial conditions (guesses) for the spectral phase of the pulse and the compressor dispersion that deviate significantly from the solution. Also, there is no need to calibrate the amplitude of the measured SHG signal in the dscan trace, since this is done by the algorithm using a frequencydependent error function, defined as^{2}
which corresponds to the merit function (overall error) given by
The iteration is stopped when the convergence between measured and retrieved traces is reached, which is defined as the merit function not improving for a number of consecutive iterations (for example, when the relative variation of \(G\) is \( < {10}^{5}\) during 10 consecutive iterations).
Data availability
The datasets generated and/or analyzed during the current study are available from the corresponding author on reasonable request.
References
Walmsley, A. & Dorrer, C. Characterization of ultrashort electromagnetic pulses. Adv. Opt. Photon. 1, 308–437 (2009).
Miranda, M., Fordell, T., Arnold, C., L’Huillier, A. & Crespo, H. Simultaneous compression and characterization of ultrashort laser pulses using chirped mirrors and glass wedges. Opt. Express 20, 688–697 (2012).
Miranda, M. et al. Characterization of broadband fewcycle laser pulses with the dscan technique. Opt. Express 20, 18732–18743 (2012).
Alonso, B. et al. Characterization of subtwocycle pulses from a hollowcore fiber compressor in the spatiotemporal and spatiospectral domains. Appl. Phys. B 112, 105–114 (2013).
Silva, F. et al. Simultaneous compression, characterization and phase stabilization of GWlevel 1.4 cycle VISNIR femtosecond pulses using a single dispersionscan setup. Opt. Express 22, 10181–10191 (2014).
Canhota, M., Silva, F., Weigand, R. & Crespo, H. Inline selfdiffraction dispersionscan of over octavespanning pulses in the singlecycle regime. Opt. Lett. 42, 3048–3051 (2017).
Miranda, M. et al. Fast iterative retrieval algorithm for ultrashort pulse characterization using dispersion scans. J. Opt. Soc. Am. B 34, 190–197 (2017).
Silva, F. et al. Strategies for achieving intense singlecycle pulses with inline postcompression setups. Opt. Lett. 43, 337–340 (2018).
Loriot, V., Gitzinger, G. & Forget, N. Selfreferenced characterization of femtosecond laser pulses by chirp scan. Opt. Express 21, 24879–24893 (2013).
Wnuk, P. et al. Discrete dispersion scanning as a simple method for broadband femtosecond pulse characterization. Opt. Express 24, 18551–18558 (2016).
Miranda, M., Silva, F., L’Huillier, A. & Arnold, C. Ultrashort Pulse Characterization from Dispersion Scans with a Grating Compressor. Conference proceeding at https://doi.org/10.1364/CLEO_AT.2016.JTu5A.67 (2016).
Fork, R.L., Martinez, O.E. & Gordon, J.P. Negative dispersion using pairs of prisms. Opt. Lett. 9, 150–152 (1984).
Martinez, O.E., Gordon, J.P. & Fork, R.L. Negative groupvelocity dispersion using refraction. J. Opt. Soc. Am. A 1, 1003–1006 (1984).
Alonso, B., Miranda, M., Sola, I.J. & Crespo, H. Spatiotemporal characterization of fewcycle laser pulses. Opt. Express 20, 17880–17893 (2012).
Cheng, Z. & Krausz, F. & Spielmann, Ch. Compression of 2 mJ kilohertz laser pulses to 17.5 fs by pairing doubleprism compressor: analysis and performance. Opt. Commun. 201, 145–155 (2002).
Sartania, S. et al. Generation of 0.1TW 5fs optical pulses at a 1kHz repetition rate. Opt. Lett. 22, 1562–1564 (1997).
Cavalieri, A.L. et al. Intense 1.5cycle near infrared laser waveforms and their use for the generation of ultrabroadband softxray harmonic continua. New J. Phys. 9, 242 (2007).
Acknowledgements
We thank Francisco Silva for very fruitful suggestions and discussions about the method. We also acknowledge funding from Junta de Castilla y León (SA116U13, SA046U16) and Spanish MINECO (FIS201344174P, FIS201571933REDT, FIS201787970R), and Fundação para a Ciência e Tecnologia, Portugal, via grants SFRH/BPD/88424/2012, SFRH/BSAB/105974/2015, UID/NAN/50024/2013, NORTE010145FEDER022096, and the Network of Extreme Conditions Laboratories (NECL), cofunded by COMPETE and FEDER.
Author information
Authors and Affiliations
Contributions
B.A. proposed the selfcalibrating algorithm. B.A. and H.C. conducted the experiments. I.J.S. and H.C. supervised the project. B.A. performed the simulations and analyzed the experimental results. All the authors were involved in the interpretation of the results and contributed to the manuscript preparation.
Corresponding author
Ethics declarations
Competing Interests
The authors declare no competing interests.
Additional information
Publisher's note: Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The images or other third party material in this article are included in the article’s Creative Commons license, unless indicated otherwise in a credit line to the material. If material is not included in the article’s Creative Commons license and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this license, visit http://creativecommons.org/licenses/by/4.0/.
About this article
Cite this article
Alonso, B., Sola, Í.J. & Crespo, H. Selfcalibrating dscan: measuring ultrashort laser pulses ontarget using an arbitrary pulse compressor. Sci Rep 8, 3264 (2018). https://doi.org/10.1038/s41598018217016
Received:
Accepted:
Published:
DOI: https://doi.org/10.1038/s41598018217016
This article is cited by

Detection and elimination of pulse train instabilities in broadband fibre lasers using dispersion scan
Scientific Reports (2020)

Robustness and capabilities of ultrashort laser pulses characterization with amplitude swing
Scientific Reports (2020)
Comments
By submitting a comment you agree to abide by our Terms and Community Guidelines. If you find something abusive or that does not comply with our terms or guidelines please flag it as inappropriate.