The aim of the pilot study was to evaluate the effect of Vagus Nerve Stimulation (VNS) paired with sounds in chronic tinnitus patients. All participants were implanted and randomized to a paired VNS (n = 16) or control (n = 14) group. After 6 weeks of home therapy, all participants received paired VNS. The device was used on 96% of days with good compliance. After 6 weeks, the paired VNS group improved on the Tinnitus Handicap Inventory (THI) (p = 0.0012) compared to controls (p = 0.1561). The between-group difference was 10.3% (p = 0.3393). Fifty percent of the participants in the paired VNS group showed clinically meaningful improvements compared to 28% in controls. At one year, 50% of participants had a clinically meaningful response. The therapy had greater benefits for participants with tonal and non-blast induced tinnitus at the end of 6 (24.3% vs. 2%, p = 0.05) and 12 weeks (34% vs. 2%, p = 0.004) compared to controls with 80% and 70% responding at 6 months and 1 year, respectively. Adverse effects were mild and well-tolerated and the therapy had a similar safety profile to VNS for epilepsy. VNS paired with tones may be effective for a subgroup of tinnitus patients and provides impetus for a larger pivotal study.
Tinnitus is the perception of a chronic ringing or buzzing sound in the absence of external stimulation. The reactions to tinnitus have been categorized into four primary functions of daily living activities: emotions, hearing, sleep and concentration1,2,3,4. Several counseling and sound-based therapies have been developed to help some patients manage these reactions5,6,7,8,9,10. Those that incorporate cognitive behavior therapy principles have received particular attention11,12,13. However, many patients with chronic tinnitus continue to remain refractory to treatments.
The psychological model of tinnitus proposed by Dauman and Tyler14 distinguishes the physiologic mechanisms of tinnitus from the reactions to tinnitus15. Preece et al.16 categorized mechanisms into three broad categories: (1) deafferentation and central- induced changes, (2) increases in spontaneous activity, and (3) increases in cross-fiber correlation17. One of the proposed mechanisms of tinnitus is generally thought to arise from hyperactive neurons in the central auditory system18,19,20,21,22. Neurons deprived of auditory input respond to the same frequencies as neighboring neurons that receive input from undamaged parts of the cochlea22. This results in an increase in the number of neurons in the primary auditory cortex that respond to a narrow range of frequencies. Subsequently, spontaneous activity and synchrony set in, which can create the perception of tinnitus23,24.
Pairing vagus nerve stimulation (VNS) with tones can reverse this maladaptive neuroplasticity in an animal model25. VNS promotes neuroplasticity that is specific to the paired experience by triggering a timed burst of neurotransmitters25. The tones paired with VNS were specifically chosen to asynchronously activate auditory neurons surrounding the tinnitus frequency (excluding the tinnitus frequency), thereby reducing the synchronous activity of neurons in the hyperactive tinnitus region26,27.
The first feasibility study using this conceptual framework was an open-label, 10-patient study28. Stimulation settings were identical to those used in the animal experiments25. After 4 weeks of daily VNS-tone pairing, tinnitus symptoms improved in 40% of the participants and this improvement lasted for at least 2 months28.
The aim of the pilot trial was to further explore the safety and efficacy of paired VNS therapy in a double-blind, randomized, controlled study in a larger sample of participants (n = 30), for a longer period of time, and at four independent sites. This pilot study was not powered a priori to detect efficacy. The main intent was to provide both safety and efficacy information to support the design and implementation of a pivotal study for US market approval.
Sixty-two individuals were screened and thirty were implanted (see Methods). All participants followed the standard protocol visit schedule. No participants discontinued therapy during the randomized portion of the study and all participants elected to continue treatment during the long-term portion of the study. There were no significant protocol deviations that impacted the study results or conclusions. Baseline demographics of enrolled and implanted participants are provided in Table 1.
In general, adverse events were mild and well-tolerated. No new or unexpected events were observed. Moderate adverse events involved two participants who experienced iatrogenic vocal-cord paralyses (hoarseness or voice weakness) after implant that lasted longer than 12 weeks. In one of those participants, the affected side returned back to normal while in the other participant, hoarseness and vocal cord movement improved but the vocal cord paralyses did not recover completely. This participant was offered the possibility of gel injections and speech therapy but declined both. One of these events occurred with a surgeon new to VNS surgery. One lead fracture occurred which required replacement surgery after re-engineering of the lead. There were no other related serious adverse events.
All 30 implanted participants completed the randomized portion of the study (6 weeks) and all participants continued performing the therapy daily for the next 6 weeks (12 weeks total). Participants continued to do the therapy periodically afterwards and were evaluated at quarterly time-points for up to a year. For the first 6 weeks, the device was used on 96% of total possible days. Participants received an average of 273 (±38 SD) stimulations per day (approximately 2.5 (±0.1 SD) hrs/day) during the first 6 weeks.
The highest number of missed days for a single participant was 31% (13 of 42). The protocol required the device to be used for at least four (4) out of every seven (7) days (57%; or 18 or fewer missed days) with no more than three (3) unused days in a row; all participants met the compliance criteria. There were no compliance issues with the system, even though use required that the participant actively start treatment (by starting their computer and hitting the start button) and be near their laptop for 2.5 hours. The stimulation parameters were confirmed upon return clinic visits by establishing communication with the IPG.
Twenty-seven (27) of thirty (30) participants (90%) kept their device and equipment through at least 6-months. Two participants had their device explanted and one participant returned his equipment prior to 24 weeks. The two participants that were explanted did have an improvement on their tinnitus symptoms (THI decrease of 39.13% and 55.17% respectively after 12 weeks of treatment). The participant that returned the device prior to 24 weeks had only a 2.7% decrease on the THI after 12 weeks. An additional participant had their device removed after 10 months. Twenty of 30 (67%) participants routinely used the device through 12 months. An additional 5 participants still had the device implanted for sporadic use (83.3% used the device at least some during the long-term study). Only 5 participants (3 participants were explanted and 2 participants returned the equipment) stopped using the system within 1 year (17%).
The paired VNS group improved on the THI (−17.7%; 95% confidence interval, −28% to −7.3%, p = 0.0012) at the end of the randomized portion of the study (6 weeks) while the control group did not (−7.3%; 95% confidence interval, −27.5% to 12.7%, p = 0.1561). The between-group difference was 10.3% (95% confidence interval, −10.49% to 31.12%, p = 0.3393) (Table 2). The median (interquartile range) was −17.78% (−28.67, −2.08) for the paired VNS and −4.91% (−25.71, 3.03) for the control group at the end of 6 weeks. Figure 1 shows the individual participant data for the paired VNS and control group. There was wide variability in the THI response with some participants showing a decrease while others showed no change. A wide range of tinnitus severity was also observed at the baseline visit. THI scores ranged from 20 to 96 in the paired VNS group and 24 to 80 in the control group. Note that one participant in the control group displayed a very large decrease between baseline and 6 weeks and then an increase in the THI score between 6 and 12 weeks. The reason for this variability is unclear. The participant had a high THI score at baseline (80), indicating someone who was greatly distressed or anxious.
Responder rate for the THI was defined as percentage of participants who showed a clinically meaningful response on the THI29. At the end of 6 weeks, 50% (8/16) of the participants in the paired VNS group showed a >20% improvement on the THI compared to 28% (4/14) in controls (20% cut-off: χ2 = 1.43, p = 0.23).
After the 6 weeks of the blinded portion of the study, both groups were unblinded. After this time point, participants in the paired VNS group continued to receive the paired VNS while the control (unpaired) group crossed-over to receive paired VNS. Once unblinded, clinicians had the option to use different stimulation parameters (e.g., increasing or decreasing the stimulus intensity, changing the stimulation pulse width or interval between paired stimulations).
A repeated measures ANOVA was performed with condition (paired VNS vs. control) as the between-subjects’ variable and the THI at baseline, 6 weeks, and 12 weeks as the within-subjects’ variable. This analysis revealed a main effect for THI (F = 6.18, p = 0.006, η 2 = 0.31) demonstrating that in comparison to baseline (M = 53.03, SD = 20.45), a decrease was obtained after 6 weeks (M = 45.87, SD = 22.13, p = 0.023) and 12 weeks (M = 43.73, SD = 23.68, p = 0.004). We did not observe the main effect for condition (paired VNS vs. control) (F = 0.001, p = 0.97, η 2 < 0.001) or an interaction effect (F = 0.10, p = 0.91, η 2 = 0.007). The paired VNS group had a mean suppression of 17.71% (Md = 17.78%, SD = 19.45) on the THI, while the control group had a mean suppression of 7.39% (Md = 4.92%, SD = 34.97) after 6 weeks of treatment (t = 1.02, p = 0.32). Analysis for THQ, TFI, loudness severity, loudness match and MML are shown in Table 3.
Fifty percent (8/16) of the participants in the paired VNS group exhibited a clinically meaningful improvement (>20% improvement) on the THI at the end of 6 weeks and 56% responded after 12 weeks. In the control group, only 28% (4/14) of the participants responded at the end of 6 weeks. After these participants crossed over to receive paired VNS, the responder rate increased to 43% at 12 weeks. Participants were followed up to one year and outcome measures and responder rates from both groups were pooled at 6 months and 1 year since both groups were receiving paired VNS therapy for the long-term portion of the study (Supplementary Tables 4 and 5).
At 6 months, responder rates for participants in both groups that continued to receive paired VNS were 56% (n = 16), 35% (n = 17) and 52% (n = 17) for the THI, THQ and TFI respectively. At 1 year, responder rates were 50% (n = 16), 64% (n = 17) and 50% (n = 16) for the THI, THQ and TFI, respectively. It should be noted that of the 14 participants in the VNS group and control group that responded to the paired VNS therapy during their first 6 weeks of paired VNS (8/16 VNS, 6/14 cross-over controls), eight participants used the device until one year and 6/8 (75%) continued to be responders at one year (average decrease of 54% on the THI).
An earlier study suggested using a 7-point absolute cut-off for the THI as being clinically meaningful30. Using this cut-off, responder rates for the THI were 62% at 6 months and 50% at 1 year. Some authors have suggested using a 13-point cut-off for the TFI31. Using this cut-off, responder rates were 35% and 37% at 6 months and 1 year, respectively. A summary of all responder rates is provided in the Supplementary Materials. The changes on outcomes at 1 year from available participants are shown in Table 4.
Next, we looked at the proportion of participants that dropped a category on the THI scale as they progressed through therapy. For the paired VNS group (n = 16), 43.75% of participants had moderate tinnitus at baseline, which decreased to 12.5% at 12 weeks. This was accompanied by large increase in the proportion of participants with mild tinnitus from 25% at baseline to 56% at 12 weeks. In the unpaired group (n = 14), a large proportion of participants had severe tinnitus prior to treatment (42.86%). This dropped to 7.14% at the end of 12 weeks (this period included the paired VNS therapy) (Fig. 2).
We evaluated the responses within this tinnitus population to determine if certain participants responded better to the therapy than others. VNS treatment appeared to have greater benefits for participants that did not have hissing tinnitus and/or blast-induced tinnitus. After excluding this subset of participants (n = 11 participants excluded), a 24.3% improvement was obtained on the THI in the paired VNS group, compared to only 2% in the control group at the end of 6 weeks (p = 0.05). After 12 weeks, the THI improved in the paired VNS group by 34% compared to 2% in controls (p = 0.004). Responder rate was 80% and 70% at 6 months and one year respectively. The average THI decrease for these responders was 44% and 40% respectively, in this subpopulation.
This pilot study was significantly different from the earlier open label feasibility study in several ways. First, this was a double-blind, randomized controlled trial. Second, participants in both groups were implanted and controls crossed over (after 6 weeks) to receive the paired VNS therapy. Third, participants continued to use the device over one year and were followed up with quarterly assessments. Finally, the earlier open label feasibility study used an implanted lead with an external stimulator. Participants had to be in the clinic each day to receive stimulation for 4 weeks (20 weekdays). The current study had the stimulator and lead both implanted such that therapy could be delivered at home.
There was a wide variability in the responses across participants. For example, two participants showed large decreases in the THI during the control portion of the study. It is possible that the improvement may have resulted from a placebo effect or that VNS may be modulating the emotional component of the tinnitus32,33,34,35,36,37,38. When considering the latter, it should be noted that the use of VNS in this study is distinctly different from the use of VNS for epilepsy and patients with major depression, where VNS is given 24 hours per day, seven days a week, and the stimulation is not paired with any specific sensory input. In contrast, in this study, substantially less VNS was administered compared to epilepsy32 and depression33 studies (2.5 minutes of stimulation/day for tinnitus vs. 144 minutes of stimulation/day for epilepsy/depression). Therefore, it is unclear whether the unpaired VNS in the control group may have contributed to the therapeutic benefits in some participants and/or was a placebo effect. Our preclinical studies had previously demonstrated that unpaired VNS did not reverse the tinnitus percept or plasticity in a rat animal model of tinnitus25. Moreover, no differences in depression scores (BDI) were obtained between the groups and no overall decrease in BDI was observed in the long term. Subsequent studies will explore the timing of paired VNS to have a better understanding of the mechanism of action.
Our results show that VNS was associated with reductions in tinnitus severity in a subgroup of tinnitus patients (tonal and non-blast induced tinnitus). In animal studies, tinnitus was evaluated in a gap-detection behavior model whereby rats underwent noise exposure and were subsequently tested for tonal tinnitus percept25. Therefore, the study model was a noise exposure model (non-blast induced) with tonal tinnitus. It is possible that the pathophysiology of blast-induced tinnitus is different from tinnitus due to prolonged noise exposure. It is also possible that a different pairing paradigm may be needed for hissing quality of tinnitus. Given the wide heterogeneity of tinnitus pathophysiology and symptoms, paired VNS therapy may be effective within certain subgroups.
We did not observe large consistent decreases for the MML or loudness match measures. Use of psychoacoustic measures such as the MML have been problematic in earlier studies39,40 and loudness matching is known to be subject to learning effects41. In this study, there was high variability in MML measurements across baselines with an average difference of 7.4 dB (range 6.2 to 29 dB), making interpretation of the results difficult.
VNS is known to be a safe and well-tolerated procedure that has been performed in approximately 100,000 patients worldwide for the treatment of severe depression and epilepsy42,43. Two participants reported hoarseness or vocal weakness lasting more than 12-weeks. As in our case, this has previously been associated with VNS surgery and is comparable to the rates seen in other early VNS studies for the treatment of epilepsy. Nevertheless, the events did not interfere with compliance. Future studies will assess vocal cord function prior to surgery and use experienced VNS surgeons with enhanced surgical training.
All participants completed the randomized phase of the study and side effects due to stimulation were minimal (i.e. hoarseness). The pairing of VNS with tones revealed no new types of adverse effects, suggesting that the safety profile is similar to VNS for epilepsy and depression.
Although the therapy was time consuming (2.5 hours a day, 7 days a week), compliance during the first 6 weeks of the study was very good. After the first 12 weeks, participants could use the device for differing amounts, although typically for 1.5 to 2.5 hours a day, for 3–7 days a week. Only 2 out of 30 participants were explanted after 6 months. Both participants had a decrease on their THI between 30–60% at 6 months suggesting that both participants believed the benefit had plateaued. One participant returned the computer after 6 months, due to a weak improvement on the THI. Most other participants elected to remain implanted and try new or different settings. It is possible that periodic use of the device may have accounted for minimal or late onset of improvement in some participants.
In this study, the stimulation settings were identical to those used in the animal and the feasibility studies25,28. However, it is not known whether the stimulation parameters used to activate the human vagus nerve are optimal. Recent animal research demonstrated that moderate intensity of VNS (0.4 and 0.8 mA) drive greater plasticity compared to VNS at higher intensities (1.2 and 1.6 mA)44. However, it is possible that different frequencies, different pulse widths or amplitudes might reveal better results45.
Individuals with tinnitus may certainly benefit from counseling and sound therapy. However, it should be noted that the individuals enrolled in this study had tinnitus for 18 years on average and had tried several other therapies, making them a refractory population. The data here showed an improvement on all three questionnaires as well as loudness severity rating after one year of treatment. For responders, the average decrease in THI was 47% (in terms of absolute decrease, these responders had an average THI decrease of 20 points from baseline) at one year. Although not all individuals participated in the long-term follow-up, these results are encouraging and suggest a long-lasting therapeutic benefit.
A limitation of the study was the small number of participants and sites due to the pilot nature of the study. Potential clinical effects of promising new tinnitus interventions should be first tested in smaller trials, which can give important information on the effect size of the intervention and may help to identify subgroups of patients being more likely to respond to the tested intervention46. This information is necessary to design large prospective placebo-controlled clinical trials, which are costly and time consuming46. Based on the results from this study, a 11% difference between paired VNS and control group with a common standard deviation of 20 would indicate 53 participants per group (106 total participants) and would have 80% power. This pilot study serves as primary support for the design and implementation of a larger pivotal study (120 participants) for US market approval.
Given the heterogeneity of chronic tinnitus, it is unlikely that any single treatment will be effective for all patients with tinnitus47. The observation of a robust response in a subgroup of patients that corresponded most closely to our preclinical data suggests that paired VNS is associated with reductions in tinnitus severity for certain types of tinnitus. The results of this study demonstrate that VNS may be promising for a subgroup of patients with chronic tinnitus. The 120-subject pivotal study has been approved by the FDA to start enrollment.
The study was a two-arm, double-blind, randomized, and controlled study in 30 participants with chronic sensorineural tinnitus. Participants in both groups were implanted with the VNS device and randomized to either a paired VNS (n = 16) or control (n = 14) group (Fig. 3). Participants performed the treatment at home for approximately 2.5 hours/day, 7 days/week, for 6 weeks. During the first 6 weeks, the VNS group received VNS paired with tones (discussed below). The control group also received VNS but was unpaired from tones. After 6 weeks, participants in the paired VNS group continued to receive paired VNS while participants in the control group crossed over to receive paired VNS. Both groups were followed up to a year.
Participants were enrolled at 4 centers in the United States (Supplementary Table 1). The trial was performed under an approved FDA Investigational Device Exemption (IDE, #G130140) and registered on clinicaltrials.gov (NCT01962558) on 9th October 2013. Informed consent was obtained in compliance with the requirements set forth in U.S. Food and Drug Administration, Code of Federal Regulations Title 21.
With respect to safety, a sample of 30 participants allowed adequate power to detect the incidence of rare safety and device events. A sample of 30 participants yielded 95% probability that the study will reveal at least one occurrence of all events or complications that occur in participants at a rate of 9.5% or greater. In addition, implantation and follow-up of 30 participants for 6 weeks will yield 1,260 days of exposure. In this case, the threshold for detection decreases to a very unlikely event, one that occurs in only 0.2375% of days. In other words, if an event has a probability of 0.002375 or 1 event in 422 days of exposure, we have a 95% probability of observing this event in a sample of 30 participants with 1,260 days of exposure. As participants continue into the long-term portion of the study, the ability to detect rare events increases as the exposure increases.
Participants and Enrollment
Sixty-two participants were screened across the study sites. Of the 62 participants, 32 did not proceed due to (1) participants decision, typically regarding the participants or a family member’s concern about surgery or time commitment for the study (15), discontinuation by the site after the study implant limit was reached (5), failure to meet ongoing tinnitus levels (4), medication contraindication (3), other medical issues (3) and excessive hearing loss (2). Thirty participants were enrolled and implanted which included 25 males and 5 females.
Inclusion and Exclusion Criteria
Key inclusion criteria included individuals with sensorineural tinnitus who (a) were 22 to 65 years-of-age (b) had primarily a tonal quality to their tinnitus (c) had either unilateral or bilateral tinnitus (d) had experienced tinnitus for at least one year in duration (e) had engaged in at least one tinnitus therapy program and found it unhelpful. Key exclusion criteria included (a) acute or intermittent tinnitus (b) Meniere’s disease, retro-cochlear disease or evidence of active middle-ear disease (c) any active implanted device such as a pacemaker or other neurostimulator or any other investigational device or drug (d) Beck Depression Inventory (BDI) of 30 or greater (e) Any drug known to mimic, increase, or decrease release or removal of a diffuse neuromodulator, such as norepinephrine, dopamine, serotonin, benzodiazepines, acetylcholine, psychoactive medications or medication known to cause or increase tinnitus.
Randomization and blinding
The contract research organization (CRO) performing the data analysis supplied the randomization for each participant to the site via a phone call with unblinded site personnel who performed the device programming and set the device to the proper group device settings. The participant initiated each therapy session but was unaware of the group allocation (there was a “start” button, but no information on the specific settings was shown to the participant). This allowed the blinded condition to be maintained, since participants were often unable to perceive VNS or only sporadically perceive VNS either via neck tingling or voice hoarseness. The audiologist performing the assessments on a specific participant was not allowed to operate the device settings for that participant so as to remain blinded to that participant’s settings. Participants were randomized using SAS (PROC PLAN) and stratified by site. Each site had its own randomization sequence, with an initial block of two followed by blocks of four. For each participant, the site called the CRO to receive the treatment assignment, taken sequentially from the randomization list for that site.
VNS Implantation and Stimulation Parameters
Device implantation was typically performed under general anesthesia by an otolaryngologist. One participant, however, was implanted using local anesthesia, at their request. Details of the device implantation have been described previously28. After surgical implantation and approximately one week of recovery, participants were randomized to either the paired VNS group (n = 16) or to the active control therapy group (n = 14).
Participants were admitted to the surgical center/hospital on the morning of the device implantation and were discharged within 24 hours. The implantation involved placement of the lead’s stimulation electrodes on the left vagus nerve in the carotid sheath. The lead connector was then tunneled subcutaneously to a pocket created in the left ancillary or pectoral region where it was attached to the implantable pulse generator.
The device consisted of an implantable pulse generator (Model 1000 Serenity®), an implantable lead and electrode (Model 3000), and an external controller system (Fig. 4). The external controller system included a laptop computer (Dell Inspiron) with high quality circumaural headphones (Sennheiser, HD280-PRO), running the Tinnitus Application Programming Software (TAPS Model 4000) and an external controller. The external controller (Model 2000, connected to the laptop via USB) communicated wirelessly with the IPG stimulator. The software enabled the audiologist to program the stimulation parameters (amplitude (mA), frequency (Hz), pulse width (µs), duration (ms)), review captured participants’ programming history, and check lead impedance and battery status. The software also captured participants’ programming history.
Participants started the therapy after approximately one week of recovery from surgery. Stimulation was delivered to the left vagus nerve since this is the most common practice in VNS for epilepsy and depression. However, since the upstream targets are bilateral, stimulation likely affects both sides of the cerebral hemispheres. Each VNS stimulation consisted of fifteen 0.8 mA, constant current, charge balanced pulses (100 µs pulse width, at 30 Hz). The duration of the VNS pulse train was 0.5 seconds (Fig. 5a). Each pulse train was delivered approximately every 30 seconds for 2.5 hours. In no instance were settings outside those used for VNS in epilepsy or depression (output currents were ≤3.5 mA, frequencies were ≤30 Hz, pulse widths were ≤1000 µs and duty cycles (ON time / OFF times) ≤50%.
In the paired VNS group, each 0.5 s VNS pulse was presented simultaneously with a 0.5 s tone every 30 s for 2.5 hrs. Therapy tones excluded one or more of the participant’s tinnitus frequencies. The tones paired with VNS were at least ½ octave away from the most prominent tinnitus frequency for each individual participant. The frequencies ranged from 170 to 16000 Hz. The sounds were played at an intensity based on the participant’s comfort level and adjusted for any hearing loss at different frequencies and intensities were limited to 80 dB SPL. Each of the tone frequencies was made to appear to arise from various 3D locations (programmed using a KEMAR head model) in order to avoid a bias of presenting a tone (paired with VNS) from a single spatial location. The frequency and intensity (dB SPL) of each tone were randomly selected each time a VNS pulse was delivered.
In the control (unpaired) group, VNS was not paired with tones (10 minutes of tones only, 5 minutes of silence and no VNS; 2 hours of VNS only; 5 minutes of silence and no VNS, and 10 minutes of tones only) during the 2.5-hour period (Fig. 5b).
Delivery of Home-based Therapy
Both groups received therapy for 6 weeks (randomized portion of the study). The control group then crossed over to the paired VNS group settings after 6 weeks while the VNS group continued the original paired VNS therapy.
After the audiologist programmed the system settings, participants trained in proper use of the device to initiate daily stimulation at home. Although the stimulation was self-delivered by the participant at their home, a test therapy session was administered at the site under supervision of the investigator. VNS was administered to determine whether the participant could tolerate the standard settings and initiate therapy appropriately. If the participant was unable to tolerate the standard settings, output current was reduced from 0.8 mA in 0.1 mA steps until a tolerable level was reached. After a tolerable level of output current was obtained, the site operator verified that the participant could hear the tones and that the tones were coming from various locations as perceived through the headphones (site personnel would listen to the tones through the headphones and then have the participant listen and verify the tones could be heard).
Participants were contacted at least once during the first seven days of therapy so that the site could check that the participant was delivering therapy appropriately for the full 2.5-hour session. Participant compliance with performing at-home therapy was verified at sites by reviewing the data log files (log files had date/time/stimulation information), such that each participants daily use could be verified.
Participants were instructed to do therapy in a quiet room and to either read a magazine or book while sitting in a comfortable chair. They were also instructed not to sleep, watch TV (with audio), or have extended conversations during therapy. However, they were allowed to work on a muted computer. The intent was to allow the participant to hear the tones while still being able to perform some other tasks. Although participants were instructed to deliver therapy in their home at approximately the same time every day, they were allowed to deliver therapy at different times due to scheduling conflicts. If participants needed a bathroom break or if other social interruptions (i.e., telephone call) occurred, they were given a 15-minute break. Furthermore, if a participant missed more than an average of three therapy sessions a week, or more than 3 sessions in a row, that participant was eliminated from the primary analysis, but was included in any intent-to-treat analysis. Delivery of therapy was verified through review of the records on the participant’s computer at each study visit.
Baseline Screening and Outcome Measures
Baseline screening included audiometry (through 12 kHz with pure tones for both air and bone conduction stimuli including monosyllabic word recognition testing) and tinnitus pitch matching. Hearing thresholds were measured at 0.5 kHz, 1 kHz, 2 kHz, 3 kHz, 4 kHz, 6 kHz, 8 kHz and 12.5 kHz (Fig. 6). Assessments included Tinnitus Handicap Inventory (THI), Tinnitus Handicap Questionnaire (THQ), Tinnitus Functional Index (TFI), loudness severity (participants were asked to rate the loudness of their tinnitus on a scale from 0–100; 0 meaning no tinnitus and 100 indicating loudest tinnitus that they can imagine), as well as psychoacoustic outcome measures including Minimum Masking Level (MML) and loudness matching48,49. Beck Depression Inventory (BDI)50, State-Trait Anxiety Inventory (STAI)51, and SF-1252 were also obtained. Since this was a pilot study, no primary outcome measure was designated. Participants had three baseline visits prior to starting therapy. The first visit was performed soon after consent was obtained to confirm eligibility. A second was performed at least two weeks later. Device implantation was then scheduled and participants were randomized to the paired VNS group or the control group (described below). A third baseline assessment was performed after surgery but prior to therapy initiation in order to determine whether surgery or anesthesia had any significant impact on assessments. All participants had their outcome measures administered in the clinic.
Baseline characteristics were displayed for each group by mean and standard deviation for continuous variables and numbers per category for categorical variables. Baseline characteristics were compared between groups using a t-test or a chi-squared test. Change from baseline was calculated for each group as percent or absolute change from baseline to 6 weeks. The within-group change in outcome (THI) from baseline to week 6 was analyzed with a one-sample t-test (H0: change = 0) and presented as a mean difference (with 95% confidence interval) for each group. Between group comparisons were made with two-sample t-test. For all comparisons, alpha was set at 0.05. No adjustments were made for multiplicity and each analysis was tested at the 0.05 level.
A repeated measures ANOVA was performed with condition (paired VNS vs. Control) as the between-subjects’ variable and the THI (at baseline, 6 weeks and 12 weeks) as the within-subject’s variable. A similar analysis was conducted for the THQ, TFI, loudness severity, MML and Loudness match. To test if the variance between the two groups (paired vs. control) was equal, we applied a Mauchly sphericity test. This test did not show statistically significant differences between the variances (Mauchly W = 0.84, χ2 = 4.79, p = 0.091).
A responder analysis was also performed to determine clinically meaningful effects. For the THI, a 20% reduction or an absolute 7-point reduction was considered clinically meaningful29,30,52. A 10-point reduction on the THQ and a 13- or 7-point decrease for the TFI were considered clinically meaningful31,53. To compare the responders and non-responders for both the paired VNS and control group, a χ2-coefficient was calculated. Statistical analysis was done using MATLAB (Mathworks, Inc.), SPSS and SAS v 9.4.
Bhatt, J. M., Lin, H. W. & Bhattacharyya, N. Prevalence, Severity, Exposures, and Treatment Patterns of Tinnitus in the United States. JAMA Otolaryngol. Neck Surg. 92697, 1–7 (2016).
Langguth, B. & Landgrebe, M. Tinnitus and depression. Textb. Tinnitus 12, 493–498 (2011).
Pattyn, T. et al. Tinnitus and anxiety disorders: A review. Hear. Res. 1–11 (2015).
Tyler, R. et al. Development and validation of the tinnitus primary function questionnaire. Am. J. Audiol. 23, 260–272 (2014).
Hoare, D. J., Kowalkowski, V. L., Kang, S. & Hall, D. A. Systematic review and meta-analyses of randomized controlled trials examining tinnitus management. Laryngoscope 121, 1555–1564 (2011).
Jastreboff, P. J. & Hazell, J. W. A neurophysiological approach to tinnitus: clinical implications. Br. J. Audiol. 27, 7–17 (1993).
Mohr, A. Your Life and Tinnitus in The Consumer Handbook on Tinnitus (ed. Tyler, R.) (Auricle Inc, 2008).
Searchfield, G. Hearing Aids and Tinnitus in Tinnitus Treatment Clinical Protocols (ed. Tyler, R.) 161–75 (Thieme, 2005).
Tyler, R., Stocking, C., Secor, C. & Slattery, W. H. Amplitude modulated S-Tones can be superior to noise for tinnitus reduction. Am. J. Audiol. 23, 303–308 (2014).
Tyler, R. S., Noble, W., Coelho, C. B. & Ji, H. Tinnitus Retraining Therapy: Mixing Point and Total Masking Are Equally Effective. Ear Hear. 33, 588–594 (2012).
Andersson, G. & Kaldo, V. Internet-Based Cognitive Behavioral Therapy for Tinnitus. Online 60, 171–178 (2004).
Sweetow, R. W. Cognitive aspects of tinnitus patient management. Ear Hear. 7, 390–396 (1986).
Tyler, R. S., Gogel, S. A. & Gehringer, A. K. Tinnitus activities treatment. Prog. Brain Res. 166, 425–434 (2007).
Dauman, R. & Tyler, R. In Tinnitus 91: Proceedings of the Fourth International Tinnitus Seminar, Bordeaux, France. August 27–30, 1991 225–229 (1992).
Tyler, R. S. N. Models, P. Models, and Treatments for Tinnitus. Tinnitus Treat. Clin. Protoc. 1–22 (2006).
Preece, J. P., Tyler, R. S. & Noble, W. The management of tinnitus. Geriatrics and Aging 6, 22–28 (2003).
Eggermont, J. J. & Roberts, L. E. The Neuroscience of Tinnitus: Understanding Abnormal and Normal Auditory Perception. Front. Syst. Neurosci. 6, 1–4 (2012).
Bauer, C. A., Turner, J. G., Caspary, D. M., Myers, K. S. & Brozoski, T. J. Tinnitus and inferior colliculus activity in chinchillas related to three distinct patterns of cochlear trauma. J. Neurosci. Res. 86, 2564–2578 (2008).
Eggermont, J. J. Central tinnitus. Auris. Nasus. Larynx 30(Suppl), S7–12 (2003).
Holt, A. G., Bissig, D., Mirza, N., Rajah, G. & Berkowitz, B. Evidence of key tinnitus-related brain regions documented by a unique combination of manganese-enhanced MRI and acoustic startle reflex testing. PLoS One 5, e14260 (2010).
Kaltenbach, J. A. The dorsal cochlear nucleus as a contributor to tinnitus: mechanisms underlying the induction of hyperactivity. Prog. Brain Res. 166, 89–106 (2007).
Noreña, A. J. & Eggermont, J. J. Changes in spontaneous neural activity immediately after an acoustic trauma: Implications for neural correlates of tinnitus. Hear. Res. 183, 137–153 (2003).
Eggermont, J. J. Tinnitus and neural plasticity (Tonndorf lecture at XIth International Tinnitus Seminar, Berlin, 2014). Hear. Res. 319, 1–11 (2015).
Eggermont, J. J. & Roberts, L. E. Tinnitus: animal models and findings in humans. Cell Tissue Res. 361, 311–36 (2015).
Engineer, N. D. et al. Reversing pathological neural activity using targeted plasticity Nature 470, 101–4 (2011).
Engineer, N. D., Møller, A. R. & Kilgard, M. P. Directing neural plasticity to understand and treat tinnitus. Hear. Res. 295, 58–66 (2013).
Hays, S. A., Rennaker, R. L. & Kilgard, M. P. Targeting plasticity with vagus nerve stimulation to treat neurological disease. Prog. Brain Res. 207, 275–299 (2013).
De Ridder, D., Vanneste, S., Engineer, N. D. & Kilgard, M. P. Safety and efficacy of vagus nerve stimulation paired with tones for the treatment of tinnitus: A case series. Neuromodulation 17, 170–179 (2014).
Khedr, E. M., Rothwell, J. C., Ahmed, M. A. & El-Atar, A. Effect of daily repetitive transcranial magnetic stimulation for treatment of tinnitus: comparison of different stimulus frequencies. J. Neurol. Neurosurg. Psychiatry 79, 212–215 (2008).
Zeman, F. et al. Tinnitus handicap inventory for evaluating treatment effects: which changes are clinically relevant? Otolaryngol.–Head Neck Surg. 145, 282–287 (2011).
Meikle, M. B. et al. The tinnitus functional index: development of a new clinical measure for chronic, intrusive tinnitus. [Erratum appears in Ear Hear. 2012 May;33(3):443]. Ear Hear. 33, 153–176 (2012).
Chen, S.-P. et al. Vagus Nerve Stimulation Inhibits Cortical Spreading Depression. Pain (2015).
Oldani, L., Dell’Osso, B. & Altamura, A. C. Long-term effects of vagus nerve stimulation in treatment-resistant depression: A 5-year follow up case series. Brain Stimul. 8, 1229–1230 (2015).
Conway, C. R. et al. Cerebral blood flow changes during vagus nerve stimulation for depression. Psychiatry Res. - Neuroimaging 146, 179–184 (2006).
Golm, D., Schmidt-Samoa, C., Dechent, P. & Kröner-Herwig, B. Neural correlates of tinnitus related distress: An fMRI-study. Hear. Res. 295, 87–99 (2013).
Joos, K., van neste, S. & de Ridder, D. Disentangling depression and distress networks in the tinnitus brain. PLoS One 7, e40544 (2012).
Kosel, M., Brockmann, H., Frick, C., Zobel, A. & Schlaepfer, T. E. Chronic vagus nerve stimulation for treatment-resistant depression increases regional cerebral blood flow in the dorsolateral prefrontal cortex. Psychiatry Res. 191, 153–159 (2011).
Vanneste, S. et al. The neural correlates of tinnitus-related distress. Neuroimage 52, 470–480 (2010).
Staecker, H. et al. Selecting appropriate dose regimens for AM-101 in the intratympanic treatment of acute inner ear tinnitus. Audiol. Neurotol. 20, 172–182 (2015).
van de Heyning, P. et al. Efficacy and Safety of AM-101 in the Treatment of Acute Inner Ear Tinnitus V A Double-Blind, Randomized, Placebo-Controlled Phase II Study. Otol. 101 (2014).
Hoare, D. J., Edmondson-Jones, M., Gander, P. E. & Hall, D. A. Agreement and reliability of tinnitus loudness matching and pitch likeness rating. PLoS One 9, e114553 (2014).
Grimonprez, A., Raedt, R., Baeken, C., Boon, P. & Vonck, K. Neuroscience and Biobehavioral Reviews The antidepressant mechanism of action of vagus nerve stimulation: Evidence from preclinical studies. Neurosci. Biobehav. Rev. 56, 26–34 (2015).
Vonck, K. et al. Neuroscience and Biobehavioral Reviews Vagus nerve stimulation. 25 years later! What do we know about the effects on cognition? Neurosci. Biobehav. Rev. 45, 63–71 (2014).
Borland, M. et al. Cortical map plasticity as a function of vagus nerve stimulation intensity. Brain Stimul. 9, 117–23 (2016).
Usami, K., Kawai, K., Sonoo, M. & Saito, N. Scalp-recorded evoked potentials as a marker for afferent nerve impulse in clinical vagus nerve stimulation. Brain Stimul. 6, 615–623 (2013).
Dobie, R. Clinical Trials and Drug Therapy for tinnitus in Tinnitus: Theory and Management (ed. Snow, J) 266–77 (2004).
Tyler, R. et al. Identifying tinnitus subgroups with cluster analysis. Am. J. Audiol. 17, (2008).
Tyler, R. The psychoacoustical measurement of tinnitus in Tinnitus Handbook (ed. Tyler, R.) 149–79 (Singular Publishing Group, 2000).
Tyler, R. Psychoacoustical measurement of tinnitus for treatment evaluations in New Dimensions in Otorhinolaryngology Head and Neck Surgery (ed. Myers) 455–58 (Elsevier Publishing, 1985).
Beck, A., Steer, R., Ball, R. & Ranieri, W. Comparison of Beck Depression Inventories -IA and -II in psychiatric outpatients. J. Pers. Assess. 67, 588–597 (1996).
Spielberger C., Gorsuch R., Lushene R., Vagg P. & Jacobs G. Manual for the state-trait anxiety scale. Consulting Psychologists. SF-12 (1983).
Ware, J., Kosinski, M. & Keller, S. A 12-Item Short-Form Health Survey: construction of scales and preliminary tests of reliability and validity. Med Care. 34, 220–33 (1996).
Folmer, R. L. et al. Repetitive Transcranial Magnetic Stimulation Treatment for Chronic Tinnitus: A Randomized Clinical Trial. JAMA Otolaryngol. Head Neck Surg. 141, 716–22 (2015).
We would like to thank the participants who invested their time to participate in this study. We would like to thank Ernesto Diaz-Ordaz (Buffalo), Michael Seidman (Detroit), Marlan Hansen (Iowa), Bruce Gantz (Iowa), Larry Myers (Dallas) and Teresa Chan (Dallas) for performing the surgeries and Peter Roland for assistance in setting up the study. We would like to thank Michael Galvan (Research Point, Houston, TX) and Lingjiao Qi (Research Point, Houston, TX) for help with statistical analysis. We would like to thank Sarah Tillman and Amy Arnold for performing the audiological evaluations. This work was supported by grants from the National Institute on Deafness and other Communication Disorders (U44 DC010084-05).
Brent Tarver, Navzer Engineer, Chester Buress and David Pierce are employees of MicroTransponder, Inc. Michael Kilgard is a paid consultant for MicroTransponder, Inc.
Publisher's note: Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Electronic supplementary material
About this article
Cite this article
Tyler, R., Cacace, A., Stocking, C. et al. Vagus Nerve Stimulation Paired with Tones for the Treatment of Tinnitus: A Prospective Randomized Double-blind Controlled Pilot Study in Humans. Sci Rep 7, 11960 (2017). https://doi.org/10.1038/s41598-017-12178-w
Journal of Neural Engineering (2020)
Otolaryngologic Clinics of North America (2020)
Neuromodulation: Technology at the Neural Interface (2020)
Vagus nerve stimulation (VNS)-induced layer-specific modulation of evoked responses in the sensory cortex of rats
Scientific Reports (2020)
Restoration of Somatosensory Function by Pairing Vagus Nerve Stimulation with Tactile Rehabilitation
Annals of Neurology (2020)