The use of a structure with a remote fixed point around which a mechanism can rotate is called remote centre of motion (RCM). The technique is widely used in minimally invasive surgery to avoid excess force on the incision site during the robot’s motion. Here we describe the design, fabrication and characterization of an origami-inspired miniature RCM manipulator for teleoperated microsurgery (the mini-RCM has mass 2.4 g and size 50 mm × 70 mm × 50 mm), which is actuated by three independently controlled linear actuators with concomitant sensing (each mini-LA has mass 0.41 g and size 28 mm × 7 mm × 3.6 mm). The mini-RCM has a payload capacity of approximately 27 mN and attains a positional precision of 26.4 μm. We demonstrate its potential utility as a precise tool for teleoperated microsurgery by performing 0.5-mm-square tracing and micro-cannulation teleoperated microsurgical procedures under a microscope. Teleoperation using the mini-RCM reduced the deviation from the desired trajectory by 68% compared to manual operation. In addition, the mini-RCM allows gravity compensation and back drivability for safety. Its compact, simple structure facilitates manufacture.
Subscribe to Journal
Get full journal access for 1 year
only $8.25 per issue
All prices are NET prices.
VAT will be added later in the checkout.
Tax calculation will be finalised during checkout.
Rent or Buy article
Get time limited or full article access on ReadCube.
All prices are NET prices.
The source data for the figures presented in this paper can be found in the Supplementary information. Source data are provided with this paper.
Motion control code is available from the corresponding authors upon reasonable request.
Taylor, R. H. A perspective on medical robotics. Proc. IEEE 94, 1652–1664 (2006).
Jung, M., Morel, P., Buehler, L., Buchs, N. C. & Hagen, M. E. Robotic general surgery: current practice, evidence, and perspective. Langenbeck’s Arch. Surg. 400, 283–292 (2015).
Menaker, S. A. et al. Current applications and future perspectives of robotics in cerebrovascular and endovascular neurosurgery. J. Neurointerv. Surg. 10, 78–82 (2018).
Das, H., Zak, H., Johnson, J., Crouch, J. & Frambach, D. Evaluation of a telerobotic system to assist surgeons in microsurgery. Comput. Aided Surg. 4, 15–25 (1999).
Maddahi, Y. et al. Treatment of glioma using neuroarm surgical system. BioMed Res. Int. 2016, 9734512 (2016).
van Mulken, T. J. M. et al. First-in-human robotic supermicrosurgery using a dedicated microsurgical robot for treating breast cancer-related lymphedema: a randomized pilot trial. Nat. Commun. 11, 757 (2020).
Edwards, T. L. et al. First-in-human study of the safety and viability of intraocular robotic surgery. Nat. Biomed. Eng. 2, 649–656 (2018).
Gijbels, A. et al. In-human robot-assisted retinal vein cannulation, a world first. Ann. Biomed. Eng. 46, 1676–1685 (2018).
Aksungur, S. Remote center of motion (RCM) mechanisms for surgical operations. Int. J. Appl. Math. Electron. Comput. 3, 119 (2015).
Vander Poorten, E. et al. Robotic retinal surgery. In Handbook of Robotic and Image-Guided Surgery (ed. Abedin-Nasab, M. H.) 627–672 (Elsevier, 2020).
Kuo, C.-H., Dai, J. & Dasgupta, P. Kinematic design considerations for minimally invasive surgical robots: an overview. Int. J. Med. Robot. 8, 127–145 (2012).
Kim, U. et al. S-surge: novel portable surgical robot with multiaxis force-sensing capability for minimally invasive surgery. IEEE/ASME Trans. Mechatron. 22, 1717–1727 (2017).
Kim, C. et al. Three-degrees-of-freedom passive gravity compensation mechanism applicable to robotic arm with remote center of motion for minimally invasive surgery. IEEE Robot. Autom. Lett. 4, 3473–3480 (2019).
Felton, S. Origami for the everyday. Nat. Mach. Intell. 1, 555–556 (2019).
Whitney, J. P., Sreetharan, P. S., Ma, K. Y. & Wood, R. J. Pop-up book MEMS. J. Micromech. Microeng. 21, 115021 (2011).
Wood, R. J., Avadhanula, S., Sahai, R., Steltz, E. & Fearing, R. S. Microrobot design using fiber reinforced composites. J. Mech. Des. 130, 52304–52311 (2008).
McClintock, H., Temel, F. Z., Doshi, N., Koh, J.-s & Wood, R. J. The milliDelta: a high-bandwidth, high-precision, millimeter-scale delta robot. Sci. Robot. 3, eaar3018 (2018).
York, P. A., Jafferis, N. T. & Wood, R. J. Mesoscale flextensional piezoelectric actuators. Smart Mater. Struct. 27, 15008 (2017).
York, P. A. & Wood, R. J. Nitinol living hinges for millimeter-sized robots and medical devices. In 2019 Int. Conf. on Robotics and Automation (ICRA) 889–893 (IEEE, 2019).
Gafford, J. B., Wood, R. J. & Walsh, C. J. Self-assembling, low-cost, and modular mm-scale force sensor. IEEE Sens. J. 16, 69–76 (2016).
Zhang, Z. G., Ueno, T. & Higuchi, T. Magnetostrictive actuating device utilizing impact forces coupled with friction forces. In 2010 IEEE Int. Symp. on Industrial Electronics 464–469 (IEEE, 2010).
Henderson, D. A. Simple ceramic motor . . . inspiring smaller products. In 10th Int. Conf. on New Actuators Vol. 50 (ICNA, 2006).
Hemsel, T. & Wallaschek, J. Survey of the present state of the art of piezoelectric linear motors. Ultrasonics 38, 37–40 (2000).
Morita, T. Miniature piezoelectric motors. Sens. Actuat. A 103, 291–300 (2003).
Mohammadi, F., Kholkin, A. L., Jadidian, B. & Safari, A. High-displacement spiral piezoelectric actuators. Appl. Phys. Lett. 75, 2488–2490 (1999).
Conway, N. J., Traina, Z. J. & Kim, S.-G. A strain amplifying piezoelectric MEMS actuator. J. Micromech. Microeng. 17, 781–787 (2007).
Poikselkä, K. et al. Novel genetically optimised high-displacement piezoelectric actuator with efficient use of active material. Smart Mater. Struct. 26, 95022 (2017).
Breguet, J. & Clavel, R. Stick and slip actuators: design, control, performances and applications. In Proc. 1998 Int. Symp. on Micromechatronics and Human Science 89–95 (IEEE, 1998).
Okamoto, Y. & Yoshida, R., S., M. The Development of a Smooth Impact Drive Mechanism (SIDM) Using a Piezoelectric Element: Technology Report 23–26 (Konica Minolta, 2004).
Nishimura, T., Hosaka, H. & Morita, T. Resonant-type smooth impact drive mechanism (SIDM) actuator using a bolt-clamped Langevin transducer. Ultrasonics 52, 75–80 (2012).
Park, J., Keller, S., Carman, G. P. & Hahn, H. T. Development of a compact displacement accumulation actuator device for both large force and large displacement. Sens. Actuat. A 90, 191–202 (2001).
Zhou, M. et al. Design and experimental research of a novel stick-slip type piezoelectric actuator. Micromachines 8, 150 (2017).
Gijbels, A., Poorten, E. B. V., Stalmans, P., Brussel, H. V. & Reynaerts, D. Design of a teleoperated robotic system for retinal surgery. In Int. Conf. Robot. Autom. 2357–2363 (IEEE, 2014).
Lum, M. J. H., Rosen, J., Sinanan, M. N. & Hannaford, B. Optimization of a spherical mechanism for a minimally invasive surgical robot: theoretical and experimental approaches. IEEE Trans. Biomed. Eng. 53, 1440–1445 (2006).
Hannaford, B. et al. Raven-II: an open platform for surgical robotics research. IEEE Trans. Biomed. Eng. 60, 954–959 (2013).
Ramrath, L., Hofmann, U. G. & Schweikard, A. Spherical assistant for stereotactic surgery. In Int. Conf. Intell. Robot. Syst. 859–864 (IEEE/RSJ, 2007).
Sakai, T. et al. Design and development of miniature parallel robot for eye surgery. In 2014 36th Ann. Int. Conf. IEEE Engineering in Medicine and Biology Society 371–374 (IEEE, 2014).
Wong, T. Y., Klein, R., Klein, B. E. K., Meuer, S. M. & Hubbard, L. D. Retinal vessel diameters and their associations with age and blood pressure. Invest. Ophthalm. Vis. Sci. 44, 4644–4650 (2003).
Goldenberg, D., Shahar, J., Loewenstein, A. & Goldstein, M. Diameters of retinal blood vessels in a healthy cohort as measured by spectral domain optical coherence tomography. Retina 33, 1888–1894 (2013).
Bado, P., Clark, W. & Said, A. Ultrafast Laser Micromachining Handbook (Clark-MXR, 1999).
Baba, S. et al. Development of an advanced micro-neurosurgical robotic system for the deep surgical field. In First IEEE/RAS-EMBS Int. Conf. on Biomedical Robotics and Biomechatronics (BioRob 2006) 437–442 (IEEE, 2006).
Masamune, K. et al. Development of an MRI-compatible needle insertion manipulator for stereotactic neurosurgery. J. Image Guid. Surg. 14, 242–248 (1995).
Jun, C. et al. MR safe robot assisted needle access of the brain: preclinical study. J. Med. Robot. Res. 03, 1–11 (2018).
Li, G. et al. Robotic system for MRI-guided stereotactic neurosurgery. IEEE Trans. Biomed. Eng. 62, 1077–1088 (2015).
Monfaredi, R., Cleary, K. & Sharma, K. MRI robots for needle-based interventions: systems and technology. Ann. Biomed. Eng. 46, 1479–1497 (2018).
Uneri, A. et al. New steady-hand eye robot with micro-force sensing for vitreoretinal surgery. Proc. IEEE/RAS-EMBS Int. Conf. on Biomedical Robotics and Biomechatronics 2010, 814–819 (2010).
Wilson, J. et al. Intraocular robotic interventional surgical system (IRISS): mechanical design, evaluation, and master-slave manipulation. Int. J. Med. Robot. 14, e1842 (2017).
Gijbels, A. et al. Experimental validation of a robotic comanipulation and telemanipulation system for retinal surgery. In 5th IEEE RAS/EMBS Int. Conf. on Biomedical Robotics and Biomechatronics 144–150 (IEEE, 2014).
Meenink, H. C. M. et al. Robot-assisted vitreoretinal surgery. In Medical Robotics: Minimally Invasive Surgery (ed. Gomes, P.) 185–209 (Woodhead Publishing, 2012).
Nasseri, M. A. et al. The introduction of a new robot for assistance in ophthalmic surgery. In 35th Ann. Int. Conf. IEEE Engineering in Medicine and Biology Society (EMBC) 5682–5685 (IEEE, 2013).
We acknowledge the advice and discussions about fabrication techniques from G. Freeburn, P. York, D. Lee and all members of the Harvard Microrobotics Laboratory for their help and assistance.
The authors declare no competing interests.
Publisher’s note Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
a, The rail-unit. b, The runner-unit.
The signal from the function generator can be modified through the multiplier IC according to the output signal from the D/A. We can adjust the speed and motion direction of the runner-unit using the output signal of the D/A. The signal, generated by a proportional controller, is amplified and transmitted to the mini-LA. The displacement is measured and sampled by the mini-LA and a reference laser displacement sensor simultaneously.
Experimental determination of the mini-RCM’s motion range.
Supplementary Tables 1–5
Pop-up book assembly
Fail-safe Power outage
Source files of the collected data (.m .csv files).
About this article
Cite this article
Suzuki, H., Wood, R.J. Origami-inspired miniature manipulator for teleoperated microsurgery. Nat Mach Intell 2, 437–446 (2020). https://doi.org/10.1038/s42256-020-0203-4
npj Systems Biology and Applications (2021)