As the review of the European particle physics strategy is underway and planned future particle accelerators are being discussed worldwide, new, improved silicon tracking detector concepts are currently being studied for use at the proposed new facilities. This Technical Review outlines the current state of the art in silicon tracking detectors for particle physics along with examples of some recent applications in other areas. The various technologies currently available in particle physics are listed and their relative merits for different uses and environments are discussed. The silicon detector research programmes to allow optimal scientific returns from future particle physics experiments are described along with their current status.
Particle physics experiments depend on the ability to accurately track the paths of particles produced in high-energy collisions.
Frontier experiments demand ever higher collision rates both to maximize sensitivity to rare processes and subtle effects, and to minimize the statistical errors on measurements.
The high-rate capability, tracking precision, sensor granularity, readout speed and radiation resistance of the current generation of silicon detectors has led to their widespread adoption, particularly in collider experiments (which typically require the most demanding specifications).
New technologies are emerging, both from within particle physics and through the design of specialist variants of commercial optical sensors, which offer significant improvements in terms of particle-track spatial resolution, fast-timing capability, detector radiation hardness, scattering material and cost per unit area.
Other disciplines that already use particle physics silicon tracking technologies can only benefit from adopting these developments as they become available.
Subscribe to Journal
Get full journal access for 1 year
only $8.67 per issue
All prices are NET prices.
VAT will be added later in the checkout.
Rent or Buy article
Get time limited or full article access on ReadCube.
All prices are NET prices.
Moore, G. Cramming more components onto integrated circuits, Electronics 38, 114–117 (1968).
Hartmann, F. Evolution of Silicon Sensor Technology in Particle Physics 2nd edn (Springer, 2017).
Moll, M. Displacement damage in silicon detectors for high energy physics. IEEE Trans. Nucl. Sci. 65, 1561–1582 (2018).
Spieler, H. Semiconductor Detector Systems (Oxford Univ. Press, 2005).
Garcia-Sciveres, M. & Wermes, N. A review of advances in pixel detectors for experiments with high rate and radiation. Rep. Prog. Phys. 81, 066101 (2018).
Lutz, G. Semiconductor Radiation Detectors (Springer-Verlag, 1999).
Dijkstra, J. L. H. Overview of silicon detectors. Nucl. Instrum. Methods A 494, 86–93 (2002).
Sze, S. Physics of Semiconductor Devices 2nd edn (Wiley, 1981).
Sze, S Semiconductor Devices, Physics and Technology (Wiley, 1985).
Horisberger, R. Pixel detectors for LHC. Nucl. Instrum. Methods A 284, 185–191 (1996).
Apollinari, G. et al. High-Luminosity Large Hadron Collider (HL-LHC) Technical Design Report CERN-2017-007-M (CERN, 2017).
The High Luminosity LHC (HL-LHC) Accelerator Project http://hilumilhc.web.cern.ch/.
The CMS Collaboration CMS Technical Design Report for the Pixel Detector Upgrade CERN-LHCC-2012-016 (CERN, 2012).
The CMS Collaboration CMS, Tracker Technical Design Report CERN-LHCC-98-06 (CERN, 1998).
Azzi-Bacchetta, P. et al. The CDF intermediate silicon layers detector. Nucl. Phys. Proc. Suppl. 78, 307–310 (1999).
The ATLAS Collaboration The Inner Detector Technical Design Report CERN-LHCC-97-16 and CERN-LHCC-97-17 (CERN, 1997).
The ATLAS Collaboration Technical Design Report for the ATLAS Inner Tracker Pixel Detector CERN-LHCC-2017-021 (CERN, 2017).
Abbott, B. Production and Integration of the ATLAS insertable B-layer. J. Instrum. 13, T05008 (2018).
The CMS Collaboration The Phase-2 Upgrade of The CMS Tracker CERN-LHCC-2017-009 (CERN, 2017).
The ATLAS Collaboration Technical Design Report for the ATLAS Inner Tracker Strip Detector CERN-LHCC-2017-005 and ATLAS-TDR-025, (CERN, 2017).
Moll, M. & Casse, G. RD50 Collaboration, CERN https://rd50.web.cern.ch/rd50/.
Unno, Y. et al. Development of n+-in-p large-area silicon microstrip sensors for very high radiation environments — ATLAS12 design and initial results. Nucl. Instrum. Methods A 765, 80–90 (2014).
The RD53 Collaboration RD53A Integrated Circuit Specifications CERN-RD53-PUB-15-001 (CERN, 2015).
Parker, S., Kenny, C. & Segal, J. 3D — a proposed new architecture for solid-state radiation detectors. Nucl. Instrum. Methods A 395, 328–343 (1997).
Da Via, C. et al. 3D silicon detectors status and applications. Nucl. Instrum. Methods A 549, 122–125 (2005).
Pellegrini, G. et al. First double-sided 3-D detectors fabricated at CNM-IMB. Nucl. Instrum. Methods 592, 38–43 (2008).
Sadrozinski, H. et al. Ultra-fast silicon detectors (UFSD). Nucl. Instrum. Methods A 831, 18–23 (2016).
Cartiglia, N. et al. The 4D pixel challenge. J. Instrum. 11, C12016 (2016).
The LHCb Collaboration LHCb Vertex Locator Technical Design Report CERN-LHCC-2001-011, (CERN, 2001).
Collins, P. The LHCb VELO (VErtex LOcator) and the LHCb VELO upgrade. Nucl. Instrum. Methods A 699, 160–165 (2013).
Sadrozinski, H. et al. 4D tracking with ultra-fast detectors. Rep. Prog. Phys. 81, 026101 (2018).
Cartiglia, N. et al. Beam test results of a 16 ps timing system based on ultra-fast silicon detectors. Nucl. Instrum. Methods A 850, 83–88 (2017).
The FCC Collaboration The future circular collider study https://fcc.web.cern.ch/Pages/default.aspx.
Faccio, F. et al. Radiation-induced short channel (RISCE) and narrow channel (RINCE) effects in 65 and 130 nm MOSFETs. IEEE Trans. Nucl. Sci. 62, 2933–2940 (2015).
Menouini, M. et al. 1 GRad total dose evaluation of 65 nm CMOS technology for the HL-LHC upgrades. J. Instrum. 10, C05009 (2015).
Peric, I. A novel monolithic pixelated particle detector implemented in high-voltage CMOS technology. Nucl. Instrum. Methods A 582, 876–885 (2007).
Wermes, M. From hybrid to CMOS pixels… a possibility for LHC’s pixel future? J. Instrum. 10, C12023 (2015).
Pernegger, H. et al. First tests of a novel radiation hard CMOS sensor process for depleted monolithic active pixel sensors. J. Instrum. 12, P06008 (2017).
Besson, A. et al. From vertex detectors to inner trackers with CMOS pixel sensors. Nucl. Instrum. Methods A 845, 33–37 (2017).
Price, T. MAPS technology for vertexing, tracking, and calorimetry. Phys. Proc. 37, 932–939 (2012).
The Linear Collider Collaboration http://www.linearcollider.org/.
Boland, M. et al. Updated Baseline for A Staged Compact Linear Collider CERN–2016–004 (CERN, 2017).
Goldstein, J. Development of detector technologies for ILC vertexing. Proc. Sci. https://doi.org/10.22323/1.287.0050 (2017).
Abelev, B. et al. Technical design report for the upgrade of the ALICE inner tracking system. J. Phys. G Nucl. Part. Phys. 41, 087002 (2014).
Lutz, G. et al. DEPFET-detectors: new developments. Nucl. Instrum. Methods A 572, 311–315 (2007).
Luetticke, F. et al. The ultralight DEPFET pixel detector of the Belle II experiment. Nucl. Instrum. Methods A 845, 118–121 (2017).
Turchetta, R. et al. A monolithic active pixel sensor for charged particle tracking and imaging using standard VLSI CMOS technology. Nucl. Instrum. Methods A 458, 677–689 (2001).
Valin, I. et al. A reticle size CMOS pixel sensor dedicated to the STAR HFT. J. Instrum. 7, C01102 (2012).
Baudot, J. et al. Development of single- and double-sided ladders for the ILD vertex detectors. Preprint at https://arxiv.org/abs/1203.3689 (2012).
Brau, J. et al. Monolithic CMOS pixel detector for international linear collider vertex detection. Pramana 69, 1009–1013 (2007).
Hanranek, M. et al. Readout chip for column parallel CCD, CPR2A. Nucl. Instrum. Methods A 607, 640–647 (2009).
Calancha Paredes, C. et al. Progress in the development of the vertex detector with fine pixel CCD at the ILC. Proc. Sci https://doi.org/10.22323/1.198.0022 (2014).
Valerio, P. et al. A prototype hybrid pixel detector ASIC for the CLIC experiment. J. Instrum. 9, C01012 (2014).
Cortina Gil, E. & Soung-Yee, L. SOIPIX Programme and applications. J. Instrum. 10, C08018 (2015).
The circular electron positron collider http://cepc.ihep.ac.cn/.
Casse, G. Recent developments on silicon detectors. Nucl. Instrum. Methods A 732, 16–20 (2013).
Adloff, C. et al. Calorimetry for lepton collider experiments — CALICE results and activities. Preprint at https://arxiv.org/abs/1212.5127 (2012).
The CALICE Collaboration, Welcome to CALICE https://twiki.cern.ch/twiki/bin/view/CALICE/WebHome.
Magnan, A.-M. HGCAL: a high-granularity calorimeter for the endcaps of CMS at HL-LHC. J. Instrum. 12, C01042 (2017).
Kluge, A. et al. The TDCpix readout ASIC: a 75 ps resolution timing front-end for the NA62 Gigatracker hybrid pixel detector. Nucl. Instrum. Methods A 732, 511–514 (2013).
Takakhashi, J. et al. Silicon drift detectors for the STAR/SVT experiment at RHIC. Nucl. Instrum. Methods A 439, 497–506 (2000).
Nouais, D. et al. The ALICE silicon drift detector system. Nucl. Instrum. Methods A 501, 119–125 (2003).
Mager, M. ALPIDE, the monolithic active pixel sensor for the ALICE ITS upgrade. Nucl. Instrum. Methods A 824, 434–438 (2016).
The Electron Ion Collider, A machine that will unlock the secrets of the strongest force in Nature https://www.bnl.gov/eic/.
Abelleira Fernandez, J. et al. A large hadron electron collider at CERN: report on the physics and design concepts for machine and detector. J. Phys. G. Nucl. Part. Phys. 39, 075001 (2012).
The AMS Collaboration, The alpha magnetic spectrometer https://home.cern/about/experiments/ams.
The Fermi Gamma-ray space telescope, NASA https://fermi.gsfc.nasa.gov/.
Sadrozinski, H. et al. Operation of the preclinical head scanner for proton CT. Nucl. Instrum. Methods A 831, 394–399 (2016).
Taylor, J. et al. Proton tracking for medical imaging and dosimetry. J. Instrum. 10, C02015 (2015).
Esposito, M. et al. PRaVDA: the first solid-state system for proton computed tomography. Eur. J. Med. Phys. 55, 149–154 (2018).
Campbell, M. The medipix collaboration https://medipix.web.cern.ch/.
Schmitt, B. The SLS detectors group, The Paul Scherrer Institut https://www.psi.ch/detectors/detectors-group.
Matsumura, H. et al. Improving charge-collection efficiency of SOI pixel sensors for X-ray astronomy. Nucl. Instrum. Methods A 794, 255–259 (2015).
Butler, M. Sensor market set to soar. Photonics Media https://www.photonics.com/a62311/Sensor_Market_Set_to_Soar (2017).
Guerrini, N. et al. A high frame rate, 16 million pixels, radiation hard CMOS sensor. J. Instrum. 6, C03003 (2011).
Anelli, A. et al. Radiation tolerant VLSI circuits in standard deep submicron CMOS technologies for the LHC experiments: practical design aspects. IEEE Trans. Nucl. Sci. 46, 1690–1696 (1999).
Faruqi, G. M. A. Direct imaging for electron microscopy. Nucl. Instrum. Methods A 878, 180–190 (2018).
Rogmagnoli, G. et al. Silicon micro-fluidic cooling for NA62 GTK pixel detectors. Microelectron. Eng. 145, 133–137 (2015).
Sedgwick, I. et al. LASSENA: A 6.7 megapixel, 3-sides buttable wafer-scale CMOS sensor using a novel grid-addressing architecture. International Image Sensor Society http://www.imagesensors.org/Past%20Workshops/2013%20Workshop/2013%20Papers/08-2_030_Sedgwick_paper.pdf (2013).
Barber, G. et al. Operation of a silicon vertex detector in the NA14 photoproduction experiment. Nucl. Instrm Methods A 253, 530–536 (1987).
Contin, G. MAPS-based vertex detectors: operational experience in STAR and future applications. Proc. Sci. https://doi.org/10.22323/1.309.0019 (2018).
The author thanks his many colleagues in the international experimental particle physics community and especially those whose excellent presentations, articles, reviews and books are cited here. This Review could not have been completed without the encouragement of the Birmingham Particle Physics Group (whose recent research has relied to a significant degree on the UK’s Science and Technology Facilities Council 2015–2019 consolidated grant: ST/N000463/1) or without the wider support of the School of Physics and Astronomy at the University of Birmingham, UK. The author is also indebted to colleagues in the medical physics community and to the European Union’s Horizon 2020 Research and Innovation programme (under Grant Agreement no. 654168).
The author declares no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
- Bunch crossing
In a collider, particles are accelerated in a large number of bunches and they counter rotate in the accelerator and keep passing through each other to produce interactions.
Only data from bunch crossings containing a collision of interest are kept. Very fast processing of a subset of the data provides the ‘trigger‘ to readout that data from temporary on-detector data storage.
In a collider, the produced particle tracks spread out from the point where the collision occurred, the ‘primary vertex’. If one or more of these particles subsequently decay to produce more tracks, there are also ‘secondary vertices’ from which the additional tracks originate.
- Beam pipe
Particle collisions take place in the vacuum pipe ‘beam pipe’ in which the particle bunches circulate and collide.
Extrapolating tracks back to where they meet (that is, to the vertex they originate from).
The segmentation of the signal-collecting nodes defines the detector granularity and hence ability to cope with many simultaneous hits per unit area.
A detector that measures charged or neutral particle energy through measuring the particle showers they produce when they interact in a dense absorber.
- Bunch train
In some accelerators, bunches come grouped together in time (bunch trains) with relatively long periods of inactivity in between them.
- Radiation length
The thickness of the material (X0) required to reduce the mean energy of a traversing electron by 1/e \(\left( \sim 1/2.718\right)\).
- Sensing matrix
The nodes within the sensor measuring the signal arranged in a matrix of pixels.
- Zero suppression
Even after selection using the trigger, the number of electronics channels to readout is often too large, but many channels are empty. Zero suppression means only reading out channels with signals above a pre-set threshold.
- Space points
With strip detectors, there is no information where the particle causing a ‘hit’ occurred along the strip direction. Adding orthogonal strips still only gives two projections, which leaves ambiguities if many particles pass through the detector. Pixels measure directly the hit coordinates (space point with third coordinate constrained by the detector plane thickness).
- Molière radius
Lateral spread of the showers produced by electrons or photons (‘electromagnetic showers’) at depth through the material d = X0.
- Hadronic interaction length
Distance (d = λINT) over which the probability of a particle not having interacted falls by 1/e for hadrons (particles that mostly interact via the strong nuclear force).
- Pixelated fast-timing
Pixel detectors give true ‘space points’ (3D information), but if the timing of the hit is also recorded accurately (few tens of picoseconds) one can speak of 4D fast-timing pixels.
To marry a pixel sensor with a readout chip requires a high-density interconnect (typically conducting ‘bumps’) patterned onto one surface and then the two connected face-to-face.
- Point spread function
In astronomy, the response of an optical system to a source that itself approximates an infinitely narrow point source.
- Relative stopping power
Different materials slow down protons to a different degree, giving differences in where protons stop for a given incident energy.
- Multiplication layer
Detectors can be designed to create a high-field region just below the charge collecting node in which the collected charge is accelerated to sufficient energy to induce further electron–hole pair production ‘charge multiplication’.
A pixel is a 2D area on a detector in which the third coordinate is usually only determined by the thickness of the silicon plane, which is typically much larger than the pixel dimensions. By making the sensing thickness the same dimensions as the pixel gives a volume ‘voxel’ within which the hit occurred.
About this article
Cite this article
Allport, P. Applications of silicon strip and pixel-based particle tracking detectors. Nat Rev Phys 1, 567–576 (2019). https://doi.org/10.1038/s42254-019-0081-z
Impact of the innovations in semiconductor advanced technology on the tracking concepts in fundamental research
Journal of Instrumentation (2020)