Recent developments such as multi-harmonic Atomic Force Microscopy (AFM) techniques have enabled fast, quantitative mapping of nanomechanical properties of living cells. Due to their high spatiotemporal resolution, these methods provide new insights into changes of mechanical properties of subcellular structures due to disease or drug response. Here, we propose three new improvements to significantly improve the resolution, identification, and mechanical property quantification of sub-cellular and sub-nuclear structures using multi-harmonic AFM on living cells. First, microcantilever tips are streamlined using long-carbon tips to minimize long-range hydrodynamic interactions with the cell surface, to enhance the spatial resolution of nanomechanical maps and minimize hydrodynamic artifacts. Second, simultaneous Spinning Disk Confocal Microscopy (SDC) with live-cell fluorescent markers enables the unambiguous correlation between observed heterogeneities in nanomechanical maps with subcellular structures. Third, computational approaches are then used to estimate the mechanical properties of sub-nuclear structures. Results are demonstrated on living NIH 3T3 fibroblasts and breast cancer MDA-MB-231 cells, where properties of nucleoli, a deep intracellular structure, were assessed. The integrated approach opens the door to study the mechanobiology of sub-cellular structures during disease or drug response.
Changes in the mechanical properties of whole living cells are known to be closely related to a large variety of physiological and pathological processes1,2,3,4,5,6. However, much less is known about the corresponding changes in the mechanical properties of individual subcellular components including organelles. Yet, the changes in the properties of these components might precede the change in global cellular properties and serve as valuable prognostic markers. Molecular and structural alterations of subcellular components like the cytoskeleton7,8, nucleus, and nucleolus are known to be a marker of diseases9,10. One way to characterize subcellular organelles involves their isolation and separation from other components11,12. This, however, leads to complicated preparation steps and raises questions about the preservation of native properties of the isolated components.
Several new techniques have recently emerged to characterize the mechanical properties of the subcellular components directly inside living cells. One such method, optics-based noninvasive Brillouin microscopy, has been used to quantify the mechanical properties of plant extracellular matrices13 and mammalian cells14 at the submicrometer scale. Other methods involve particle tracking inside the cell to measure the local viscoelastic properties of the cytoplasm15.
AFM has been successfully used in recent years to characterize the mechanical properties of cells16,17,18. One of the limitations of this technique is that the cantilever interacts with the surface of the cell, and that the obtained data describe the topmost sublayers only. Thus, an information about subcellular structures that are localized deeper below the cell surface including cytoskeletal elements, nucleus, mitochondria, lysosomes, and Golgi apparatus generally remains unavailable.
To address this challenge, many efforts have been made recently to image sub-surface features through tip-generated stress or electric fields in AFM19,20,21, some of which might be applied to the living cells22,23,24,25. Nanomechanical holography, a method involving ultrasonic waves that are detected by an AFM probe in contact with the cell surface, has been applied to image the inner structures of cells and to detect the presence of nanoparticles inside the cells of exposed animals24. However, difficulties in interpreting the images in terms of the mechanical properties of the subsurface structures limit the broader application of this technique. Force-spectroscopy-based methods were also suggested, where each force curve is divided by sections based on the indentation depth. Each section is then processed by using a linear elastic23 or viscoelastic model22 to analyze in-depth properties of the sample. These techniques, however, have limited time-resolution due to the acquisition of many force curves. Subcellular features have often been observed with Peak Force Tapping26,27 or dAFM28,29 methods. These prior techniques are generally limited either in spatial, or temporal resolution while imaging live cells or are unable to quantify the properties of the sub-cellular object being imaged.
We have recently developed a multi-harmonic dynamic AFM (dAFM) method to map quantitatively the nanomechanical properties of soft biological samples in a liquid environment at kHz frequencies29,30,31,32, which is compatible with commercial AFM systems equipped with a direct-excitation setup. Here, we demonstrate that with some key additional advances this method provides unprecedented observations of the deep intracellular structures (nucleoli) and suggest a way to quantitatively extract the mechanical properties of these subcellular structures. Three new approaches were combined here: (1) microcantilever with long-carbon tips to minimize hydrodynamic artifacts and improve the spatial resolution; (2) simultaneous imaging with spinning disk confocal (SDC) microscopy to confirm the identity of intracellular structures, their size and location in the living cell; (3) computational approaches to estimate the mechanical properties of subcellular structures.
Results and discussion
Cantilevers with long tips to improve nanomechanical contrast
Quantitative multi-harmonic imaging on living cells first and foremost requires microcantilevers that are directly excited, i.e. the excitation force is applied only to the microcantilever33,34,35,36. This includes magnetic, Lorenz force, piezoelectric, and photothermal excitation methods. The use of directly excited AFM microcantilevers or multi-harmonic AFM of living cells has been described previously30,31. Here, we will focus on the implementation that uses deflection feedback (0th harmonic feedback) on living cells. This approach results in an order of magnitude improvement in scanning speed when compared to the traditional feedback on the first harmonic amplitude28,29.
The main principles of application of dAFM (AM-AFM) technique with deflection feedback on very soft samples such as living cells (modulus < 1 MPa) were considered previously28,30,31. At the imaging set point for deflection (A0), the tip is not intermittently tapping the cell but rather permanently interacts with it over the entire oscillation cycle. Thus, higher harmonics (2nd and beyond) of microcantilever vibration are usually not observed while utilizing resonant microcantilevers for imaging mammalian cells; however higher harmonics are observed for stiffer bacterial cells and viruses31,32 where the tip intermittently taps the sample. The multi-harmonic observables (amplitudes A0, A1, and phase φ1) acquired on a living cell can be related to the local viscoelastic properties. The tip oscillation amplitude is small compared to the average indentation, which allows the use of contact mechanics models such as Hertz’s37, Sneddon’s38, or models with bottom effect correction39,40. From these mechanical models, the local storage Estorage and loss Eloss Young’s moduli can be obtained. However, as shown in30,31, it is important to consider the hydrodynamic coupling between the cantilever and the sample for correct data processing. The developed theory is described in the Supplementary Material.
When any dAFM technique is used in liquid, the squeeze-film hydrodynamic effect between the oscillating cantilever, the surrounding liquid, and a sample surface reduces the quality factor of the cantilever30,41,42,43 as well as its natural frequency. The oscillation amplitude decreases while the phase lag increases as the resonant cantilever approaches the surface. The extent of this effect depends on the oscillation frequency (through the unsteady Reynold’s number) and the gap (through the squeeze number)41,43. As a rule of thumb for AFM cantilevers oscillating near hard surfaces, the hydrodynamic effect generally starts to be noticeable when the gap becomes comparable to the cantilever half-width41,43, which is usually several 10’s of micrometers. Less studied but equally important is that the squeeze film effect also depends on the viscoelastic properties of the surface44 approached by the AFM microcantilever in liquid. In turn, the viscoelastic cell properties also depend on the height of the cell surface from the substrate, which changes from the nuclear region to the periphery of the cell40. This leads to two distinct artifacts that challenge quantitative dAFM methods on living cells:
The tip height for soft AFM cantilevers usually used for imaging living cells are small (of the order of 3–5 µm). Thus, when the AFM tip contacts a living cell in liquid, the main AFM cantilever body is well within the range to experience significant squeeze film damping. As a result, the harmonic observables Anear and φnear right before the tip-surface contact can be substantially different from the values Afar and φfar several hundred microns above the surface where cantilever tuning is usually performed. this difference cannot be attributed to local tip-sample interaction properties but rather to the long-range hydrodynamic effect.
The hydrodynamic coupling to the surface and the resulting resonant cantilever’s Anear and φnear values are different on cells as compared to when the cantilever is located at the same height above a rigid surface30. The coupling differs even when the cantilever is scanning at the same height above different regions of the same cell (nucleus region versus periphery).
Thus, the hydrodynamic squeeze film effect acts as a long-range interaction between the cantilever body and the surface which significantly reduces the method’s resolution. Moreover, the dependence of Anear and φnear values on the effective viscoelastic properties of the surface over which the cantilever is located can cause significant local mechanical property/topography-associated artifacts in the observables even before the tip contacts the surface.
Since most of this hydrodynamic artifact arises from the coupling of the cantilever beam and the surface, the tip height serves as an offset, which can control a magnitude of the effect. To reduce the squeeze-film hydrodynamic effect and boost the resolution of sub-cellular imaging, we grew 10–15 µm long high aspect ratio carbon tips on the 3–5 µm tip of BL-TR400PB cantilevers (Fig. 1A), thus increasing the tip height by 300–500%. The radius of carbon tips was 50–300 nm (150 nm typical), which is large enough to reduce the possibility of cell membrane penetration and damage. No cell damage was observed from comparing confocal imaging before and after the scanning (Fig. S1).
As can be seen in Fig. 1C,D for these long-tipped AFM microcantilevers the squeeze film induced φfar—φnear (10 vs. 35 degrees for the standard microcantilevers) and Anear/Afar ratio (0.74 vs. 0.24 for the standard cantilever) are significantly reduced in PBS on a glass surface and also on hydrogel and cell (Fig. S2). The reduction in hydrodynamic effect also led to weaker topography dependence of φnear (Fig. 2C,D) and Anear. Specifically, when standard tips were used, the phase signal φ1, which is affected by φnear, showed a significant negative correlation with the cell height (p < 0.001, Pearson correlation test for the part of the curves over the cell on Fig. 2C). No such correlation (p = 0.5) was found for the long-tipped AFM microcantilevers (Fig. 2D), which confirms the artificial origin of the correlation observed for standard tips. Longer tip AFM cantilever also ensure that the amplitude and phase variables change during a scan solely due to local tip-sample interactions. This leads to higher spatial contrast in dAFM observables as well as more reliable nano-mechanical maps. In particular, the small actin cytoskeleton elements could be distinguished much better in the phase signal (Fig. 2), which consequently led to the better resolved nanomechanical maps (Figs. 2B,D, 4A). Moreover, the long tips with high aspect-ratio are generally helpful to observe tall objects like cells to avoid possible interactions between the tip sides or cantilever beam and the cell surface. Not only do cantilever body-cell interactions lead to image distortions, but small cantilever-cell gaps cause a change in the hydrodynamic loading. The latter fact is observed as an increase in the phase beyond the tall cell (with respect to the cantilever beam axis, Fig. 2A).
The modified cantilevers were characterized by laser Doppler vibrometry to estimate the spring constant from thermal vibrations in air using the equipartition theorem. The tip shape was characterized using SEM images (Fig. 1A), the tip radius was estimated by a fit with a parabolic function (Fig. 1A, inset). The indentation of polyacrylamide gels with known mechanical properties was performed to confirm the contact geometry. Generally, the obtained force curves fit well (Adjusted R squared > 0.95) with the contact model for the paraboloid (Hertz model) (Fig. 1B). The Young’s modulus of the polyacrylamide gel measured with the modified cantilevers was within 5% of the modulus measured with colloidal probes (5 µm diameter).
The idea of using long tips is not novel by itself and such cantilevers were used before, for example, to improve the output of the contact resonance45, Force Mapping46, PeakForce Tapping47,48, and high-speed AFM49. Extra-long cantilever tips (~ 60 µm or 100–300 µm) have been used to keep the cantilever beam from submerging into the liquid during the scanning, thus removing most of the hydrodynamic damping and preserving high Q-factor (100) and operational frequencies (hundreds of kHz)50,51. However, they require low (50–100 μm) and controllable thickness of the liquid layer above the sample surface. In the method described here, the cantilevers are fully immersed into the liquid and thus have low Q-factor and operating frequencies. Our goal with this design is to increase the resolution of sub-cellular imaging using multi-harmonic AFM. This approach simplifies the conversion of the observables to local mechanical properties and facilitates comparison with the other AFM methods where low frequencies are used as well. The viscoelastic behavior of living cells is known to vary widely depending on the frequency range52. Other methods, like Brillouin microscopy, operate at much higher frequencies (GHz range) at which origin of the contrast and relevance to the cell physiological behavior is less studied. Also, the spatial resolution of Brillouin images is currently not high enough to see the individual actin stress fibers14.
Observation of subcellular and subnuclear structures with improved contrast
The higher contrast achieved using the long tips allowed us to observe and identify different subcellular structures in the nanomechanical maps including stress fibers and nucleoli (Figs. 3, 4, 5) corresponding to a spatial resolution of ≈ 100–200 nm. Nucleoli are sites of ribosome biogenesis inside the nucleus. They could be easily detected in the phase and amplitude images, but not in the topography or deflection error images, thus confirming that the contrast is caused by mechanical properties of the buried object (Figs. 3A, 4A). We confirmed that these structures are indeed nucleoli by comparing AFM images with confocal fluorescence images of the same live cell stained for nucleic acids (Figs. 3B, 4B). Nucleoli can be detected as the brightest objects in such images. The nucleoli were seen previously with AFM subsurface imaging based on the analysis of deeper force curve segments22,23. The benefit of our setup is that the confocal z-stacks can be used to estimate the size and location of these organelles in the living cell. Vertical cross-sections over the nucleoli are presented in Fig. 3D with additional F-actin staining to visualize the cell borders. The depth of the nucleoli as measured from the z-stacks is up to several microns from the upper cell surface. Interestingly, the nucleus itself generally was not seen in the nanomechanical maps, indicating that it is not much stiffer than the surrounding cytoplasm.
Nucleoli were observed both in NIH 3T3 and MDA-MB-231 cells, in the latter cell type in smaller numbers but with a larger size (Figs. 3, 4). Detection of nucleoli from nanomechanical maps was hampered by the presence of stiff apical actin fibers in the NIH 3T3 cells (Fig. 5). From the nanomechanical maps, the effective storage Young’s modulus above the nucleoli was ~ 2 times higher than the modulus of the surrounding area for both types of cells. The primary function of the nucleolus is ribosome biogenesis, which is closely related to the metabolic state of a cell10,53,54. The nucleolus is a non-membranous cellular body and observed high stiffness could be caused by a high concentration of proteins and RNAs in a constrained space, which serves presumably to improve reaction and regulation efficiency. The size of the nucleolus positively correlates with ribosomal RNA synthesis, which in turn is governed by cell growth and metabolism. The larger size of nucleoli is characteristic of actively growing cancer cells55, which was observed here in MDA-MB-231 cells. Recent studies have also uncovered the role of the nucleolus in aging53 and establish the fact that nucleoli act as sensors of stress, regulating the cellular response to pathological insults56. While such studies mostly covered molecular alterations in the nucleoli, much less is known about their structural and mechanical properties. Some insights on structure-mechanics connection were obtained by studying isolated nucleoli11, but these findings remain to be confirmed in living cells.
Apical stress fibers were detected on nanomechanical maps of NIH 3T3 cells (Figs. 3, 5). They represent a special subtype of actin stress fibers called perinuclear actin cap and characterized by apical over-the-nucleus location with large adhesions at the basal cell surface (dome-like shape), connection with the nucleus via LINC complexes, and participation in mechanotransduction57,58. The perinuclear actin cap is presented in different cell types (mesenchymal stem cells, fibroblasts, endothelial cells) but is usually lost in cancerous cells. Due to its superficial apical location and high effective stiffness, the stress fibers from the perinuclear cap are often observed on the AFM images of cells taken in contact mode, both in topography and deflection error signal. They can also be distinguished in the phase images and on the nanomechanical maps (Fig. 3A).
3D nanomechanical characterization of sub-surface organelles
Next, we estimated the mechanical properties of nucleoli using the data available from both the AFM and SDCM experiments on the same cell (MDA-MB-231). A representative FE model was created based on confocal imaging, which accounted for the cell thickness, location and size of the selected nucleolus (Fig. 6A,B). An axisymmetric model with the indentation right above the nucleolus was created for simplicity, and only data for Estorage (storage modulus) and keff (local effective stiffness) were used. keff corresponds to the conservative force gradient (see supplementary materials for details) and, among other parameters, depends on the difference between the properties of nucleolus and the surrounding matrix. The goal was to estimate the modulus of the nucleolus based on the data from AFM nanomechanical maps (Fts, Estorage, and keff values) acquired directly above this nucleolus and in its vicinity. The value of the modulus of the matrix around the nucleoli was prescribed based on the values from the nanomechanical maps in the vicinity of the nucleoli (with bottom-effect correction and excluded data above the stress fibers). During the FE simulation, the modulus of the nucleoli Enuc was continuously increased relative to the modulus of the surrounding nucleoplasm Em, and the force-indentation curves were recorded (Fig. 6C). Then, for each curve, the value of keff from the simulation was calculated at the experimental force Fts (0.8 nN) and compared with the experimental value of keff (0.016 nN/nm) (Fig. 6D). From this analysis, we find that the simulated keff approaches the experimental one at high values of the nucleolus modulus (Enuc/Em > 10). This means that nucleoli are much stiffer than the nucleoplasm with the lowest bound of an estimate of Estorage ≈ 100 kPa. The upper bound of the estimate, however, cannot be currently determined given the low stiffness of the selected cantilever (as selected for live-cell imaging) and its resulting inability to deform the nucleolus enough for quantitatively ascertaining the upper bound of the modulus measurement. Similar results were obtained for NIH 3T3 fibroblasts (data not shown). Therefore, we might conclude that the nucleolus is a very stiff organelle, probably due to high molecular packing density11. Nanomechanical estimations of buried objects in living cells might enable further studies in the area of cell physiology and toxicology.
Streamlining AFM cantilever’s with long-carbon tips can significantly minimize their long-range hydrodynamic interactions with the cell surface, thus significantly enhancing the spatial resolution of nanomechanical maps and minimizing topography-related artifacts. Combining multi-harmonic AFM with simultaneous Spinning Disk Confocal Microscopy (SDC) and live-cell fluorescent markers enables the unambiguous correlation between observed heterogeneities in nanomechanical maps with specific subcellular structures. By integrating organelle size and depth data from SDC with heterogenous nanomechanical stiffness maps from multi-harmonic AFM and computational approaches, it is possible to quantitatively estimate the mechanical properties of sub-cellular structures such as nucleoli in living cells for the first time. This integration of dAFM, SDC, and computational approaches opens the door to mechanobiology of sub-cellular structures such as nucleoli, mitochondria, Golgi apparatus, and ribosomes, and relating changes in sub-cellular organelle properties with key cell metamorphosis or disease.
Materials and methods
NIH 3T3 and MDA-MB-231 cell lines were maintained in Dulbecco’s modified Eagle’s medium (DMEM) containing 10% FBS and 1% antibiotic/antimycotic solution (Invitrogen, Carlsbad, CA) in a humidified 5% CO2 atmosphere at 37 °C. Prior to AFM experiments, cells were re-plated onto 50 mm glass-bottom cell culture dishes (FluoroDish, World Precision Instruments, Sarasota, FL) coated with 30 μg/mL fibronectin for 30 min (Sigma-Aldrich, St. Louis, MO) and grown for an additional period of 1–2 days to final confluency of ~ 60–70%. SiR-actin staining59 was performed according to the manufacturer’s protocol (Cytoskeleton, Inc., Denver, CO). Briefly, the cells were incubated with 200 nM SiR-dye and 10 µM verapamil (broad-spectrum efflux pump inhibitor improving the staining efficiency) diluted in regular growth media overnight. The cells were stained with CellMask Green plasma membrane staining solution (final concentration 5 ug/ml; Life Technology) and SYTO 85 Orange nucleic acid stain (5 µM; Life Technology) for 15 min right before AFM experiments, which were conducted after several washing steps with PBS in normal cell growth medium with 20 mM HEPES.
Polyacrylamide gel was prepared from stock solutions of acrylamide (40%, Sigma-Aldrich, USA) and bis-acrylamide (1%, Sigma-Aldrich, USA) in PBS and polymerized by addition of the photoinitiator Irgacure 2959 (0.5% w/v, (2-Hydroxy-4′-(2-hydroxyethoxy)-2-methylpropiophenone, Sigma-Aldrich, USA). Final concentrations of acrylamide (4%) and bis-acrylamide (0.2%, w/v) were chosen to prepare a PAAm gel with approximately 1–2 kPa Young’s modulus, which is close to the cell stiffness. Polymerization was done between two coverslips under 10 min of UV-exposure, prepared PAAm gels had a ~ 100 μm thickness. AFM measurements were performed in PBS containing 0.1% Triton X-100 detergent (Sigma-Aldrich, USA) to decrease probe-gel adhesion60. The Young’s modulus of the gel was calculated from AFM indentation experiments conducted with the CSC38 cantilevers (rectangular, Micromash Inc., Estonia) modified with 5 μm diameter silicon dioxide bead (Microspheres-Nanospheres, Corpuscular, NY, USA). Force curves were obtained at 4 µm/s piezo speed.
Dynamic AFM material property mapping
We used the MFP-3D-Bio AFM (Asylum Research, an Oxford Instruments Company, Santa Barbara, CA) mounted on an IX-71 inverted optical microscope (Olympus, Tokyo, Japan) and integrated with an SDC microscope Andor Revolution XD (Andor Technology, South Windsor, CT). BL-TR400PB cantilevers (Olympus/Asylum Research) with a nominal spring constant of 0.09 N/m, a nominal tip radius of 42 nm (± 12 nm), and a half-opening angle α = 35° were used in our previous works. The modified version of these cantilevers with a 15-µm tall tip made of diamond-like carbon with 50–300 nm radius and 11° tilt compensation (nanotools USA LLC) was employed here. The value of cantilever spring constant was determined with laser Doppler vibrometry (Polytec MSA-400 Micro System Analyzer from Polytec GmbH, Waldbronn, Germany) from thermal vibrations in air using the equipartition theorem61. The cantilever was mounted on the iDrive Lorentz force excitation holder that enables direct excitation of the cantilever. Before and after measurements, the relationship between the photodiode signal and cantilever deflection (sensitivity factor S) was calibrated from the thermal tune spectra acquired far from the sample surface62. The AFM images were acquired with the feedback on the mean deflection to improve acquisition speed (1–2 Hz)29 with setpoint ~ 5–15 nm (force ~ 0.5–1.5 nN, corresponds to indentation of 50–200 nm). The acquisition time was 5–10 min per image. The amplitude far from the sample was selected to be approximately 10–20 nm. The conversion of AFM observables to the maps of mechanical properties was done according to29, the theory is reproduced briefly in Supplementary Material. The Hertz’s contact model for paraboloid indenter geometry was used; correction for the finite sample thickness was applied using the correction coefficients from63.
Spinning disk confocal fluorescence imaging
The fluorescent images were acquired by a spinning-disk confocal microscope Andor Revolution XD with 100 × N.A. 1.4 oil immersion or 60 × N.A. water immersion objective on the Olympus IX-71 inverted optical microscope. Generally, the z-stacks were acquired before and after the acquisition of an AFM image of the same cell. The spacing between the optical sections was chosen based on the Nyquist criterion (0.21 μm for the used setup). The imaging parameters (laser intensity and exposition time) were adjusted in preliminary experiments to decrease the acquisition time and still preserve high signal-to-noise ratio and low phototoxicity. The fluorescent images were analyzed by Fiji (NIH, Bethesda, MD).
Finite element simulations
Finite element (FE) simulations were conducted in Abaqus CAE (version 14, Simulia Corp., Providence, RI). The axisymmetric system was created based on the real dimension of the cantilever tip and the cell. The latter one was measured from SDC images. The material models used for both the nucleolus and surrounding matrix was a linear elastic material (isotropic, homogeneous, Poisson’s ratio = 0.49). The bottom surface of the sample was constrained; the contact between the indenter and the sample was considered to be frictionless. The mesh was refined in the contact region, and a mesh sensitivity study was conducted in advance.
The datasets generated during and/or analyzed during the current study are available from the corresponding author on reasonable request.
Kuznetsova, T. G., Starodubtseva, M. N., Yegorenkov, N. I., Chizhik, S. A. & Zhdanov, R. I. Atomic force microscopy probing of cell elasticity. Micron 38, 824–833 (2007).
Lekka, M. Discrimination between normal and cancerous cells using AFM. Bionanoscience 6, 65–80 (2016).
Dumont, S. & Prakash, M. Emergent mechanics of biological structures. Mol. Biol. Cell 25, 3461–3465 (2014).
Özkaya, N., Nordin, M., Goldsheyder, D. & Leger, D. Mechanical properties of biological tissues. Fundam. Biomech. 86, 221–236 (2012).
Vining, K. H. & Mooney, D. J. Mechanical forces direct stem cell behaviour in development and regeneration. Nat. Rev. Mol. Cell Biol. 18, 728–742 (2017).
Lee, G. Y. H. & Lim, C. T. Biomechanics approaches to studying human diseases. Trends Biotechnol. 25, 111–118 (2007).
Olson, M. F. & Sahai, E. The actin cytoskeleton in cancer cell motility. Clin. Exp. Metastasis 26, 273–287 (2009).
Khatau, S. B., Kim, D.-H., Hale, C. M., Bloom, R. J. & Wirtz, D. The perinuclear actin cap in health and disease. Nucleus 1, 337–342 (2010).
Friedl, P., Wolf, K. & Lammerding, J. Nuclear mechanics during cell migration. Curr. Opin. Cell Biol. 23, 55–64 (2011).
Lam, Y. W. & Trinkle-Mulcahy, L. New insights into nucleolar structure and function. F1000Prime Rep. 7, 48 (2015).
Louvet, E., Yoshida, A., Kumeta, M. & Takeyasu, K. Probing the stiffness of isolated nucleoli by atomic force microscopy. Histochem. Cell Biol. 141, 365–381 (2014).
Lombardi, M. L. & Lammerding, J. Altered mechanical properties of the nucleus in disease. Methods Cell Biol. 98, 121–141 (2010).
Elsayad, K. et al. Mapping the subcellular mechanical properties of live cells in tissues with fluorescence emission-Brillouin imaging. Sci. Signal. 9, rs5 (2016).
Antonacci, G. & Braakman, S. Biomechanics of subcellular structures by non-invasive Brillouin microscopy. Sci. Rep. 6, 1–7 (2016).
Wirtz, D. Particle-tracking microrheology of living cells: Principles and applications. Annu. Rev. Biophys. 38, 301–326 (2009).
Efremov, Y. M., Bagrov, D. V., Dubrovin, E. V., Shaitan, K. V. & Yaminskii, I. V. Atomic force microscopy of animal cells: Advances and prospects. Biophysics 56, 257–267 (2011).
Kasas, S., Stupar, P. & Dietler, G. AFM contribution to unveil pro- and eukaryotic cell mechanical properties. Semin. Cell Dev. Biol. 73, 177–187 (2018).
Okajima, T. atomic force microscopy for the examination of single cell rheology. Curr. Pharm. Biotechnol. 13, 2623–2631 (2015).
Soliman, M., Ding, Y. & Tetard, L. Nanoscale subsurface imaging. J. Phys. Condens. Matter 29, 173001 (2017).
Cadena, M. J., Chen, Y., Reifenberger, R. G. & Raman, A. Sub-surface AFM imaging using tip generated stress and electric fields. Appl. Phys. Lett. 110, 123108 (2017).
Cadena, M. J., Reifenberger, R. G. & Raman, A. High resolution subsurface imaging using resonance-enhanced detection in 2nd-harmonic KPFM. Nanotechnology 29, 405702 (2018).
Guerrero, C. R., Garcia, P. D. & Garcia, R. Subsurface imaging of cell organelles by force microscopy. ACS Nano 13, 9629–9637 (2019).
Stühn, L., Fritschen, A., Choy, J., Dehnert, M. & Dietz, C. Nanomechanical sub-surface mapping of living biological cells by force microscopy. Nanoscale 11, 13089–13097 (2019).
Tetard, L. et al. Imaging nanoparticles in cells by nanomechanical holography. Nat. Nanotechnol. 3, 501–505 (2008).
Roduit, C. et al. Stiffness tomography by atomic force microscopy. Biophys. J. 97, 674–677 (2009).
Calzado-Martín, A., Encinar, M., Tamayo, J., Calleja, M. & San Paulo, A. Effect of actin organization on the stiffness of living breast cancer cells revealed by peak-force modulation atomic force microscopy. ACS Nano 10, 3365–3374 (2016).
Eghiaian, F., Rigato, A. & Scheuring, S. Structural, mechanical, and dynamical variability of the actin cortex in living cells. Biophys. J. 108, 1330–1340 (2015).
Efremov, Y. M., Cartagena-Rivera, A. X., Athamneh, A. I. M., Suter, D. M. & Raman, A. Mapping heterogeneity of cellular mechanics by multi-harmonic atomic force microscopy. Nat. Protoc. 13, 2200–2216 (2018).
Cartagena-Rivera, A. X., Wang, W.-H., Geahlen, R. L. & Raman, A. Fast, multi-frequency, and quantitative nanomechanical mapping of live cells using the atomic force microscope. Sci. Rep. 5, 11692 (2015).
Cartagena, A. & Raman, A. Local viscoelastic properties of live cells investigated using dynamic and quasi-static atomic force microscopy methods. Biophys. J. 106, 1033–1043 (2014).
Raman, A. et al. Mapping nanomechanical properties of live cells using multi-harmonic atomic force microscopy. Nat. Nanotechnol. 6, 809–814 (2011).
Cartagena, A., Hernando-Pérez, M., Carrascosa, J. L., de Pablo, P. J. & Raman, A. Mapping in vitro local material properties of intact and disrupted virions at high resolution using multi-harmonic atomic force microscopy. Nanoscale 5, 4729–4736 (2013).
Xu, X. & Raman, A. Comparative dynamics of magnetically, acoustically, and Brownian motion driven microcantilevers in liquids. J. Appl. Phys. 102, 034303 (2007).
Enders, O., Korte, F. & Kolb, H.-A. Lorentz-force-induced excitation of cantilevers for oscillation-mode scanning probe microscopy. Surf. Interface Anal. 36, 119–123 (2004).
Kiracofe, D., Kobayashi, K., Labuda, A., Raman, A. & Yamada, H. High efficiency laser photothermal excitation of microcantilever vibrations in air and liquids. Rev. Sci. Instrum. 82, 013702 (2011).
Labuda, A. et al. Comparison of photothermal and piezoacoustic excitation methods for frequency and phase modulation atomic force microscopy in liquid environments. AIP Adv. 1, 022136 (2011).
Hertz, H. Über die Berührung Fester Elastischer Körper. J. für die Reine u. Angew. Math. 92, 156–171 (1881).
Sneddon, I. N. The relation between load and penetration in the axisymmetric Boussinesq problem for a punch of arbitrary profile. Int. J. Eng. Sci. 3, 47–57 (1965).
Dimitriadis, E. K., Horkay, F., Maresca, J., Kachar, B. & Chadwick, R. S. Determination of elastic moduli of thin layers of soft material using the atomic force microscope. Biophys. J. 82, 2798–2810 (2002).
Gavara, N. & Chadwick, R. S. Determination of the elastic moduli of thin samples and adherent cells using conical atomic force microscope tips. Nat. Nanotechnol. 7, 733–736 (2012).
Tung, R. C., Jana, A. & Raman, A. Hydrodynamic loading of microcantilevers oscillating near rigid walls. J. Appl. Phys. 104, 114905 (2008).
Rankl, C., Pastushenko, V., Kienberger, F., Stroh, C. M. & Hinterdorfer, P. Hydrodynamic damping of a magnetically oscillated cantilever close to a surface. Ultramicroscopy 100, 301–308 (2004).
Sader, J. E. Frequency response of cantilever beams immersed in viscous fluids with applications to the atomic force microscope. J. Appl. Phys. 84, 64–76 (1998).
Gavara, N. & Chadwick, R. S. Noncontact microrheology at acoustic frequencies using frequency-modulated atomic force microscopy. Nat. Methods 7, 650–654 (2010).
Tung, R. C., Killgore, J. P. & Hurley, D. C. Hydrodynamic corrections to contact resonance atomic force microscopy measurements of viscoelastic loss tangent. Rev. Sci. Instrum. 84, 073703 (2013).
Wang, A. & Butte, M. J. Customized atomic force microscopy probe by focused-ion-beam-assisted tip transfer. Appl. Phys. Lett. 105, 053101 (2014).
Schillers, H., Medalsy, I., Hu, S., Slade, A. L. & Shaw, J. E. PeakForce Tapping resolves individual microvilli on living cells. J. Mol. Recognit. 29, 95–101 (2016).
Efremov, Y. M., Shpichka, A. I., Kotova, S. L. & Timashev, P. S. Viscoelastic mapping of cells based on fast force volume and PeakForce Tapping. Soft Matter 15, 5455–5463 (2019).
Shibata, M., Uchihashi, T., Ando, T. & Yasuda, R. Long-tip high-speed atomic force microscopy for nanometer-scale imaging in live cells. Sci. Rep. 5, 1–7 (2015).
Guan, D., Barraud, C., Charlaix, E. & Tong, P. Noncontact viscoelastic measurement of polymer thin films in a liquid medium using long-needle atomic force microscopy. Langmuir 33, 1385–1390 (2017).
Guan, D., Charlaix, E., Qi, R. Z. & Tong, P. Noncontact viscoelastic imaging of living cells using a long-needle atomic force microscope with dual-frequency modulation. Phys. Rev. Appl. 8, 044010 (2017).
Rigato, A., Miyagi, A., Scheuring, S. & Rico, F. High-frequency microrheology reveals cytoskeleton dynamics in living cells. Nat. Phys. 13, 771–775 (2017).
Tiku, V. & Antebi, A. Nucleolar function in lifespan regulation. Trends Cell Biol. 28, 662–672 (2018).
Olson, M. O. & Dundr, M. Nucleolus: Structure and function. In eLS 1–9 (Wiley, 2015).
Ruggero, D. Revisiting the nucleolus: From marker to dynamic integrator of cancer signaling. Sci. Signal. 5, 5–8 (2012).
Avitabile, D. et al. Nucleolar stress is an early response to myocardial damage involving nucleolar proteins nucleostemin and nucleophosmin. Proc. Natl. Acad. Sci. USA 108, 6145–6150 (2011).
Khatau, S. B. et al. A perinuclear actin cap regulates nuclear shape. Proc. Natl. Acad. Sci. U. S. A. 106, 19017–19022 (2009).
Efremov, Y. M. et al. Anisotropy vs isotropy in living cell indentation with AFM. Sci. Rep. 9, 5757 (2019).
Lukinavičius, G. et al. Fluorogenic probes for live-cell imaging of the cytoskeleton. Nat. Methods 11, 731–733 (2014).
Efremov, Y. M., Wang, W.-H., Hardy, S. D., Geahlen, R. L. & Raman, A. Measuring nanoscale viscoelastic parameters of cells directly from AFM force-displacement curves. Sci. Rep. 7, 1541 (2017).
Lozano, J. R., Kiracofe, D., Melcher, J., Garcia, R. & Raman, A. Calibration of higher eigenmode spring constants of atomic force microscope cantilevers. Nanotechnology 21, 465502 (2010).
Schillers, H. et al. Standardized nanomechanical atomic force microscopy procedure (SNAP) for measuring soft and biological samples. Sci. Rep. 7, 5117 (2017).
Garcia, P. D. & Garcia, R. Determination of the elastic moduli of a single cell cultured on a rigid support by force microscopy. Biophys. J. 114, 2923–2932 (2018).
This work was financed in part by the National Science Foundation through the CMMI GOALI grant 1726274 to A.R. and the National Science Foundation grant 1146944-IOS to D.M.S. Y.M.E. and P.S.T. have been supported by the Ministry of Science and Higher Education of the Russian Federation within the framework of state support for the creation and development of World-Class Research Centers “Digital Biodesign and Personalized Healthcare” 075-15-2020-926.
The authors declare no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Efremov, Y.M., Suter, D.M., Timashev, P.S. et al. 3D nanomechanical mapping of subcellular and sub-nuclear structures of living cells by multi-harmonic AFM with long-tip microcantilevers. Sci Rep 12, 529 (2022). https://doi.org/10.1038/s41598-021-04443-w
This article is cited by
Investigation of TiO2 nanoparticle interactions in the fibroblast NIH-3T3 cells via liquid-mode atomic force microscope
Archives of Toxicology (2023)
Nonlinear Dynamics (2022)