High positive margin rates in oncologic breast-conserving surgery are a pressing clinical problem. Volumetric X-ray scanning is emerging as a powerful ex vivo specimen imaging technique for analyzing resection margins, but X-rays lack contrast between non-malignant and malignant fibrous tissues. In this study, combined micro-CT and wide-field optical image radiomics were developed to classify malignancy of breast cancer tissues, demonstrating that X-ray/optical radiomics improve malignancy classification. Ninety-two standardized features were extracted from co-registered micro-CT and optical spatial frequency domain imaging samples extracted from 54 breast tumors exhibiting seven tissue subtypes confirmed by microscopic histological analysis. Multimodal feature sets improved classification performance versus micro-CT alone when adipose samples were included (AUC = 0.88 vs. 0.90; p-value = 3.65e−11) and excluded, focusing the classification task on exclusively non-malignant fibrous versus malignant tissues (AUC = 0.78 vs. 0.85; p-value = 9.33e−14). Extending the radiomics approach to high-dimensional optical data—termed “optomics” in this study—offers a promising optical image analysis technique for cancer detection. Radiomic feature data and classification source code are publicly available.
Breast cancer is the second-leading cause of cancer death among women in the United States1. At the time of diagnosis, the disease is most frequently early-stage, localized cancer, which is predominately treated with surgical resection followed by radiation therapy. The goal of the surgical procedure, termed a lumpectomy or breast-conserving surgery (BCS), is to remove the malignancy with a surrounding layer or margin of non-cancerous tissue. An ideal BCS procedure results in cancer-free tissue on the margins (i.e., negative margins), an outcome known to provide the best prognosis2,3,4,5. Determining the final margin status of a BCS resection takes a day or more to complete. Depending on clinical context, one or more positive margins confirmed by histopathology may necessitate a re-excision procedure to remove residual disease, given the known association of positive margins with ipsilateral breast tumor recurrence2,6,7,8,9. Today, ~ 20% of BCS procedures require a follow-up re-excision due to positive margins10,11,12,13. Several studies report the negative health and financial impacts of BCS re-excision procedures to the patient and healthcare system14,15,16,17,18. For these reasons, improving intraoperative margin assessment in BCS is a pressing clinical need. In this study, volumetric imaging by X-rays and surface imaging by optical light were combined in a classification pipeline following conventions described in radiomics. The purpose of the study was to demonstrate how this approach might improve diagnostic assessment of malignancy in breast lumpectomy tissues, and in so doing, provide the foundational work needed to extend the approach to margin assessment in the future.
Standard of care margin assessment techniques include gross tissue inspection by the surgeon19,20, projection X-ray or radiographic specimen imaging21, and at some medical centers, frozen section pathology22,23, imprint cytology24,25, and post-excision cavity shaving21. Gross tissue inspection is useful for detecting palpable invasive cancers but is not as effective at detecting primary lesions intertwined with dense fibroglandular tissue or DCIS, which frequently presents as a cluster of macrocalcifications without a localized mass26,27. Projection (i.e., two-dimensional) X-ray imaging coupled with intraoperative reading by a radiologist is a mainstay for analyzing margins, and X-rays offer excellent contrast between adipose and fibrous tissue and sensitivity to microcalcifications28. Furthermore, recent advances in volumetric X-ray imaging (i.e., computed tomography and tomosynthesis) have demonstrated sensing of all six anatomical margins of a specimen in a clinically relevant timeframe with high spatial resolution29,30,31. For this reason, volumetric X-ray imaging is emerging as a powerful ex vivo specimen scanning technology. However, a key limitation of X-ray imaging is its inability to differentiate normal, abnormal benign, and malignant fibrous tissues that may be relevant to diagnosis, thereby posing the risk of frequent false positives28,29. Intraoperative frozen section pathology and imprint cytology reduce positive margin rates, but the approaches are not widely adopted, because they are resource-intensive (i.e., requiring pathology staff in the operating room or surgical suite during the procedure) and suffer from slow turnaround times23,32,33. Post-excision cavity shaving is also effective at reducing positive margin rates, requiring ≤ 10 min in the operating room with minimal impact to patient cosmesis34,35. However, positive margin rates after cavity shaving are variable (6–24%)34,36,37,38,39,40,41, suggesting alternative or complementary margin assessment techniques are still needed.
Volumetric X-ray imaging coupled with a second imaging modality could be an attractive solution for rapidly analyzing an entire BCS specimen with increased contrast to fibrous tissues on the margin32,42. Pradipta et al. reviewed an array of margin assessment techniques32, ranging from intraoperative ultrasonography, bioimpedance spectroscopy, and ex vivo magnetic resonance imaging to a host of optical techniques, including optical coherence tomography, ultraviolet-photoacoustic microscopy, and fluorescence probes. Each technique offers advantages and disadvantages. DiCorpo et al. reported that the average surface area of a BCS specimen is ~ 45 cm2 30. A limitation of many proposed techniques is their relatively small field of view and/or long scan time, such that analyzing the entire surface of a specimen would be too time-consuming for clinical translation. Other limitations include: the need for exogenous contrast agents that must preferentially accumulate in target tissues and be proven safe for human use; or lack of evidence demonstrating efficacy when analyzing realistic BCS specimens, which can be amorphous, variable in size, and heterogeneous with a combination of adipose, fibroglandular, and potentially malignant tissues on the margin42.
In this study, breast tumor tissues were imaged with spatially co-registered micro-computed tomography (micro-CT) scanning and multi-wavelength spatial frequency domain imaging (SFDI). SFDI is a wide-field, noncontact, and rapid optical imaging modality first introduced by Cuccia et al.43. The technique involves the projection of one-dimensional, sinusoidal patterns of light (each at a discrete spatial frequency) onto the tissue surface and acquisition of the reflected light. At each spatial frequency, three phase-shifted patterns are projected and imaged. Each set of three phase-shifted images creates a demodulated reflectance map corresponding to the wavelength(s) and spatial frequency of illumination. Additional background information related to the optical imaging modality is available (Supplementary Material Appendix 1), and the interested reader is directed to a recent review of the modality44.
Previous studies have coupled machine learning models with SFDI data to classify or predict different breast tissues. One study used SFDI-derived optical scattering properties of breast tissue and an explicit mathematical model to predict the epithelial, stromal, and adipose fractions of breast tumor samples45. Another study applied high spatial frequency, monochromatic SFDI and a limited set of textural features to classify pairs of breast tumor subtypes using a support vector machine classifier with accuracies ranging from 55 to 95%46. Deep learning methods have also determined optical properties from raw SFDI data47,48,49. The use of a limited set of image features may not probe the image data sufficiently to extract the most useful signatures for differentiating tissues. On the other hand, the complexity of deep learning methods hinders model interpretation and thus limits potential for clinical translation. To address these issues, this study used a supervised machine learning pipeline based on a large number of Image Biomarker Standardization Initiative (IBSI)-compliant radiomic features50. The pipeline tested the ability of micro-CT alone, SFDI alone, and the combination of micro-CT and SFDI data to classify malignant and non-malignant image samples extracted from wide field-of-view images of breast tumors.
Radiomics involves the quantification of many image features, mining the features to determine diagnostic signatures not readily discerned by visual inspection, and subsequently using the features to build classification models to inform clinical decision making51,52. The “radiomics approach” is frequently applied to conventional medical imaging modalities (e.g., CT, magnetic resonance imaging, positron emission tomography). Applying it to optical imaging data is termed “optomics” here, extending the “omic” concept to image features extracted from wide field-of-view optical images. The goals of this study were to: first, determine whether combining X-ray micro-CT and optical SFDI image data improves malignancy classification relative to micro-CT alone based on an “omics” approach; and second, identify the most useful radiomic and optomic features for classifying malignant breast tissues.
Portions of the data collected through this imaging protocol have been analyzed in previous studies46,49,53,54. Specifically, SFDI data have demonstrated statistical differences between normal, abnormal benign, and malignant tissue subtypes based on color analysis and diffuse optical properties53, and separately, using texture analysis of monochromatic (i.e., 490 nm), high spatial frequency (1.37 mm−1) data46. SFDI data were also used in a deep learning framework, through which optical properties were approximated directly from raw SFDI data49. Monochromatic, high frequency SFDI and micro-CT data were also used in a comparative study to quantify differences in the wide field-of-view optical and micro-CT images54. The study found that optical imaging revealed intra-tumoral morphology and malignant-fibrous tissue boundaries that were occult to micro-CT scanning. The present work extends these prior contributions significantly with two key advances: first, by incorporating multi-wavelength, multi-spatial frequency SFDI and micro-CT data into the analysis; and second, by evaluating the image data with a supervised machine learning pipeline to quantify binary malignancy classification performance.
Classification using radiomics and optomics
Adipose was the most frequent tissue type in the breast tumor dataset (Supplementary Material Table S1). Relative to fibroglandular and epithelial components, adipose tissue is straightforward to identify by gross surgical inspection and is readily evident with micro-CT scanning29,54. To test the hypothesis that including adipose samples improves performance, the classification pipeline focused on two scenarios: one that included adipose tissue samples and one that excluded them.
Figure 1a shows wide field-of-view optical and micro-CT imagery of a representative tissue specimen. Non-malignant versus malignant classification accuracy is shown for cases when adipose tissue was included (Fig. 1b–d) and excluded (Fig. 1e–g). Accuracies are plotted with respect to the number of optimal features selected by minimum redundancy, maximum relevance (MRMR)55. Sub-image samples of tissue ranged in size from 2 × 2 to 5 × 5 mm and were extracted from wide field-of-view regions of interest (ROIs) defining distinct tissue subtypes in each specimen. Only results from 5 × 5 mm sub-image samples are presented here, because this sub-image size provided the best classification performance overall (Supplementary Material Appendix 2 and Fig. S1). A 1% change in average accuracy determined an appropriate minimum number of features to use in each case. Based on the combined radiomic/optomic classification accuracy (Fig. 1d,g), the 1% change in average accuracy threshold required six features, when adipose tissue was both included and excluded. Therefore, subsequent analysis focused on results derived from 5 × 5 mm sub-image samples and with six radiomic/optomic features. Supplementary Materials Table S2 reports classification performance (i.e., accuracy, recall, precision, receiver operating characteristic (ROC) area under the curve (AUC)) for 5 × 5 mm sub-image samples and six total features. When adipose tissue was included and using six features, mean accuracies achieved by micro-CT features alone, optical features alone, and combined micro-CT and optical features were 82% (Fig. 1b), 72% (Fig. 1c), and 84% (Fig. 1d), respectively. When adipose tissue was excluded, mean accuracies decreased to 74% (Fig. 1e), 70% (Fig. 1f), and 80% (Fig. 1g) for the same respective feature sets.
Figure 2 compares average ROC curves of optimized classifiers for micro-CT alone, optical alone, and combined feature sets. Classification performance decreased when adipose tissue samples were excluded (compare Fig. 2a with d, Fig. 2b with e, and Fig. 2c with f). Micro-CT data alone performed better than optical data alone when adipose samples were included (AUC = 0.88 vs. 0.78, p = 3.30e−13), but the two performed similarly when adipose samples were excluded (AUC = 0.78 vs. 0.75, p = 0.42). Classification performance using both modalities was better than using either modality alone (e.g., when adipose tissue was included: micro-CT vs. combined, AUC = 0.88 vs. 0.90, p = 3.65e−11; optical vs. combined, AUC = 0.75 vs. 0.90, p = 3.60e−36). Improvement in combined feature performance relative to micro-CT alone was more pronounced when adipose samples were excluded (micro-CT vs. combined, AUC = 0.78 vs. 0.85, p = 9.33e−14). ROC curves reflect relatively high variance depicted by one standard deviations (shaded regions) and 95% confidence bands (dashed lines), providing a visual depiction of how individual Monte Carlo CV splits performed.
Selected feature importance
The radiomic and optomic features selected by MRMR in each of n = 1000 splits were tabulated. Figure 3 shows the distribution of these features for combined data classification and highlights the fact that both micro-CT and SFDI features were selected for inclusion in the optimal subset of features. This trend was true both when adipose tissue samples were included (Fig. 3a) and excluded (Fig. 3b). (These features correspond to the classification performances shown in Fig. 2c,f, respectively.) Two trends can be inferred from Fig. 3. First, the most important micro-CT radiomic features were first-order histogram statistics (noted by asterisks in the figure), which contain intensity information alone (i.e., no spatial information). This trend was true independent of whether adipose tissue was included. Second, important SFDI optomic features were generally derived from high spatial frequency reflectance (illumination frequencies of 0.61 and 1.37 mm−1; noted by dots in the figure), particularly when adipose tissue was omitted from the classification task. Notably, all but two of the high spatial frequency optomic features in Fig. 3 were derived from second- and higher-order pixel statistics, which depend on the spatial relationships between multiple pixels, and thus, contain textural information.
A t-distributed stochastic neighbor embedding plot, shown in Fig. 4, reduces the dataset to a two-dimensional embedding of only six optimal radiomic/optomic features (using the six most frequently selected features shown in Fig. 3b). Each sub-image sample is color-coded by tissue subtype, and representative samples are labeled with the associated micro-CT and optical image channels. Figure 4 suggests separability of normal (i.e., connective tissue) and abnormal benign (i.e., fibrocystic disease) samples from malignant tissue subtypes is possible when only six radiomic/optomic features are used. Adipose tissue is also clustered effectively under these conditions.
High-dimensional radiomic datasets are frequently visualized using hierarchically clustered heatmaps51,52,56, and such a visualization is available for the dataset in this study (Supplementary Material Appendix 3 and Fig. S2). Notably, the global structure shown in Fig. S4 is maintained by Fig. 4 using six instead of 1564 radiomic/optomic features. The image channel labels in Fig. 4 exemplify representative tissue subtype image signatures: first, adipose tissue yields relatively low linear attenuation coefficient values in the micro-CT sub-image relative to all other subtypes54; second, adipose tissue and high-grade invasive ductal carcinoma are predominantly characterized by Mie scattering vacuoles (> 25 μm in diameter) and nuclei (~ 5 μm in diameter), respectively, explaining the relatively low high spatial frequency optical reflectance for these samples57; and third, connective tissue and fibrocystic disease contain collagen fibril structures (< 100 μm in diameter) that are strong Rayleigh-type scatterers, giving rise to an increased backscatter signal in these optical sub-images57.
Analyses indicate overall mean accuracy and AUC achieved were 84% and 0.90 (80% and 0.85 without adipose tissue), respectively, using only six optimal radiomic/optomic features. These results appear to be the first application of radiomics to multi-wavelength, multi-spatial SFDI data and represent a novel approach for validating and extending radiomic feature analysis to optical imaging data. They provide a proof of principle for how radiomic and optomic features can be combined to improve overall classification accuracy. Other important contributions from this study include appropriate image data normalization to linear attenuation coefficient and calibrated reflectance values and use of IBSI-compliant image features in the classification pipeline, considerations that lend to study rigor and reproducibility. Finally, use of defined image features within a supervised machine learning pipeline provides increased model interpretability relative to deep learning methods of tissue classification.
As described in the Introduction, previous studies used portions of the same breast cancer dataset used in this study. The new method revealed in this work relative to previous studies is multimodal radiomic/optomic analysis. The radiomics approach is often deployed with other imaging modalities relevant to the diagnosis and monitoring of breast cancer, mainly magnetic resonance imaging, CT, positron emission tomography, and/or ultrasound. Most studies demonstrate radiomics utility in identifying malignant lesions in the setting of pre-operative breast cancer diagnosis, and recently published classification performance AUCs range from 0.57 to 0.98 for this task58,59. The work described here suggests that wide field-of-view optical images of breast tumors likely contain useful image features, especially textural information, relevant to diagnosis. The optomics approach provides an alternative to optical property quantification, a process that requires tissue model assumptions and can be computationally intensive to implement (e.g., ~ 1 h for a 2.5 cm × 2.5 cm tissue sample57). The study also demonstrates that the combined radiomics/optomics approach is synergistic for automated diagnostic assessment of breast tissues. Additional findings revealed in this work relative to previous studies include the fact that useful optical features can be extracted from a wide wavelength range (i.e., optimal features span 490–700 nm in Fig. 3a and 490–600 nm in Fig. 3b) and the entire spatial frequency range in the dataset (i.e., optimal features span 0.00–1.37 mm−1 in Fig. 3a,b). Notably, the top six features selected by MRMR differed significantly depending on whether adipose tissue was included in the classification task. When adipose was included (Fig. 3a), the top two features were micro-CT first-order histogram statistics. This result is expected given the stark difference in radiodensity between adipose and all other tissues in the breast cancer specimens; purely differences in pixel intensity were sufficient to separate adipose from all other tissues. When adipose was excluded (Fig. 3b), three of the top six features were textural features derived from high spatial frequency optical reflectance, suggesting that this feature type provides value for differentiating tissues that exhibit similar radiodensities. Nevertheless, the second most important feature when adipose tissue was excluded was a micro-CT first-order statistic, reinforcing the claim that micro-CT and optical imaging provide complementary contrast regardless of the presence of adipose tissue.
Importantly, improvements in margin assessment were not directly demonstrated in this study. Rather, freshly resected breast tumor slices were imaged to capture a range of tumor pathologies, and the focus was building differential diagnostic models. It is also necessary to distinguish cancer tissue on the margin that has been thermally denatured or burned by surgical tools during resection, processes known to alter tissue optical properties60. However, this limitation was not addressed in this study. Only invasive cancers were analyzed due to a lack of pre-invasive DCIS specimens in the dataset. This limitation is significant, given that relative to other malignant tissue subtypes, DCIS is responsible for one of the largest shares of positive margins leading to re-excision procedures13. Future studies should include DCIS samples to demonstrate efficacy in classifying this important pre-invasive subtype. Finally, future work should also involve radiomic/optomic analysis of intact BCS specimens, taking advantage of the volumetric sensing of micro-CT in combination with surface-mapped optical reflectance to evaluate the performance of this approach for sensing cancer-positive tissues at the margin. For example, the radiomic feature quantification package used in this study, PyRadiomics56, offers a suite of standardized, IBSI-compliant 3D shape and voxel-based radiomic features that could be extracted from sub-volumes of the micro-CT scan. Connected sub-volumes that exhibit malignant/suspicious radiomic signatures that extend to the tissue margin could then be correlated with optomic signatures from the surface tissue to identify potentially involved margins.
This study introduces an optomics paradigm for analyzing high dimensional optical image data and represents a direct and quantitative assessment of the extent to which coupling micro-CT scanning with optical imaging improves classification of malignant breast tumor tissues. Volumetric X-ray imaging is complimentary to surface SFDI, which provides additional sensing of surface tissue subtypes, especially those that are fibrous and exhibit similar radiodensities. These results warrant further research into the combination of specimen X-ray imaging with wide field-of-view, noncontact optical imaging, such as SFDI, for potentially improving intraoperative margin assessment.
Materials and methods
Breast tumor imaging protocol
All methods were carried out in accordance with relevant guidelines and regulations. Specimen imaging was performed at the Dartmouth-Hitchcock Medical Center (DHMC) in Lebanon, New Hampshire. The imaging protocol was approved by the Committee for the Protection of Human Subjects, the Institutional Review Board at Dartmouth College, and all aspects of the study followed the approved protocol. Tissue specimens were procured from patients electing BCS at DHMC who participated in the study under informed consent. Imaging was performed post-operatively during standard of care pathological processing in the specimen grossing laboratory and did not impact tissue processing or diagnostic reporting in any way. One representative, ~ 5-mm thick slice from each BCS tumor was selected by an experienced Pathologists’ Assistant for imaging and was de-identified and referenced only by a unique accessioning number. Each imaged slice exhibited a clear cross section of the primary tumor and surrounding tissues, thereby revealing clear regions of several normal, abnormal benign, and malignant breast tissue subtypes. Each slice was firmly positioned between clear acrylic plates, creating a flat tissue surface that mitigated specular reflection and demodulation artifacts in the SFDI data. The top surface of each imaged slice underwent standard of care specimen processing, sectioning, staining with hematoxylin and eosin, and microscopic analysis by a board-certified breast pathologist (WAW). Histologic slides corresponding to the imaged tissue surface underwent whole slide, high resolution digital imaging and were then mosaicked together to confirm wide-field tissue subtype ROIs. These microscopic ROIs were co-registered to the wide field-of-view micro-CT and optical imagery.
Imaging was performed with a customized IVIS SpectrumCT system (PerkinElmer, Hopkinton, MA) containing a cone-beam CT in a “pancake” geometry and retrofitted optical imaging components61. Micro-CT scans were acquired with X-ray tube settings of 50 kVp and 1 mA with an exposure time of 100 ms/projection for a total of 720 projections. The reconstructed scan volume was 12 × 12 × 3 cm3 with 150 μm3 voxels, and the combined acquisition and reconstruction time was ~ 4 min. These settings were the same as those used in a previous micro-CT BCS specimen study29. The SFDI acquisition leveraged the charged coupled device camera native to the IVIS SpectrumCT system (Andor iKon, Andor Technologies Ltd, Belfast, UK) and a digital light projector (CEL5500 Fiber, Digital Light Innovations Inc., Austin, TX) retrofitted in the light-tight, fully shielded imaging cabinet for projecting the structured light patterns. The light source was a supercontinuum laser (SuperK Blue, NKT Photonics, Denmark) with a tunable line filter (SuperK Varia, NKT Photonics, Denmark). SFDI acquisition involved optical wavelengths of 490, 550, 600, and 700 nm, and illumination spatial frequencies of 0.00, 0.15, 0.61, and 1.37 mm−1. Thus, 16 unique wavelength-spatial frequency reflectance images were collected per specimen. SFDI acquisition and reconstruction time was ~ 8 min. Combined with the surface tissue micro-CT slice, every specimen in the dataset had 17 channels of image data (Fig. 5a).
Image data and pre-processing
A total of 70 specimens were imaged under the approved protocol. Sixteen cases failed to meet data analysis eligibility criteria: five were excluded due to inconsistent micro-CT scan settings, six were excluded due to ambiguous histology co-registrations, one was excluded due to a small cross-sectional area (< 2 cm2), and four were omitted, because they presented tissue subtypes that were represented in fewer than three specimens in the dataset. Thus, 54 tumor specimens from 54 BCS patients were analyzed. From these specimens, 177 ROIs were drawn, isolating regions of seven histologically confirmed breast tissue subtypes. Of the 54 specimens, 14 contained only normal and/or abnormal benign tissues, three contained only malignant tissue, and 37 contained both normal or abnormal benign and malignant tissues. Table 1 summarizes the number of breast tumor specimens and the number of ROIs categorized by tissue subtype. The number of sub-images extracted from each ROI was proportional to the size of each ROI, providing an approximate baseline by which all ROIs were equally sampled. Figure 5b illustrates the process of sub-image sampling from the wide field-of-view tissue images. Supplementary Material Table S1 contains totals of sub-image samples categorized by subtype and by sub-image size.
Calibrated reflectance maps were generated at each of 16 wavelength-spatial frequency settings using a previously described normalization process with a reflectance standard61. Micro-CT scans were converted to linear attenuation coefficient values based on a 50 kVp X-ray energy. All image data were masked to remove regions with poor tissue coupling to the top acrylic plate, thereby restricting the image analysis to flat, consistent tissue surfaces. Micro-CT scanning and SFDI were completed sequentially, without moving the tissue or acrylic plates between acquisitions, facilitating spatial co-registration between modalities. Tissue subtype ROIs were sampled over a range of square sub-image sizes (2 × 2–5 × 5 mm).
Radiomics package and classification pipeline
The classification pipeline (summarized in Supplementary Material Fig. S3) performed binary stratification of samples as malignant or non-malignant based on radiomic and/or optomic features. Fornacon-Wood et al. recently reviewed several popular radiomic feature quantification software packages, identifying strengths and weaknesses of each package62. Here, the pipeline was implemented in the Python coding language (v3.7.9) using the PyRadiomics package (v3.0.1) for feature quantification, given that it is IBSI-compliant for reproducibility, free of cost, and open-source56. The pipeline involved MRMR feature selection, a random forest classifier, and Monte Carlo cross-validation with n = 1000 splits partitioned on the patient-level. Additional details related to the classification pipeline are available (Supplementary Material Appendix 4 and Fig. S3).
Features were quantified from only micro-CT data (1 channel), only SFDI data (16 channels), and combined data (17 channels) (Fig. 5c). A total of 92 PyRadiomic features were extracted from each image channel. Thus, up to 1,564 features were quantified from each sub-image sample. Quantified features included a range of first-, second-, and higher order pixel statistics. Supplementary Material Appendix 5 contains additional information about the features quantified, including fixed bin width considerations (Appendix 5.1, Fig. S4) and a complete list of features and associated abbreviations (Appendix 5.2).
For every combination of pipeline parameters, ROC curves (n = 1000) were generated through Monte Carlo CV. Average ROC curves were derived by interpolating true positive values to a constant range of false positives across all splits. DeLong’s test determined whether differences between pairs of average ROC AUC values were statistically significant63,64. A p-value ≤ 0.05 was considered significant.
Image data in the form of comma-separated value files, classification pipeline source code, and a PyRadiomics parameter file documenting all feature quantification settings are available with this publication in an open-source repository (https://github.com/optmed/radiomics-optomics).
DeSantis, C. E. et al. Breast cancer statistics, 2019. CA Cancer J. Clin. 69, 438–451 (2019).
Park, C. C. et al. Outcome at 8 years after breast-conserving surgery and radiation therapy for invasive breast cancer: Influence of margin status and systemic therapy on local recurrence. J. Clin. Oncol. 18, 1668–1675 (2000).
Cowen, D. et al. Local and distant failures after limited surgery with positive margins and radiotherapy for node-negative breast cancer. Int. J. Radiat. Oncol. Biol. Phys. 47, 305–312 (2000).
Fisher, B. et al. Twenty-year follow-up of a randomized trial comparing total mastectomy, lumpectomy, and lumpectomy plus irradiation for the treatment of invasive breast cancer. N. Engl. J. Med. 347, 1233–1241 (2002).
Houssami, N., Macaskill, P., Marinovich, M. L. & Morrow, M. The association of surgical margins and local recurrence in women with early-stage invasive breast cancer treated with breast-conserving therapy: A meta-analysis. Ann. Surg. Oncol. 21, 717–730 (2014).
Schnitt, S. J. et al. The relationship between microscopic margins of resection and the risk of local recurrence in patients with breast cancer treated with breast-conserving surgery and radiation therapy. Cancer 74, 1746–1751 (1994).
Smitt, M. C. et al. The importance of the lumpectomy surgical margin status in long-term results of breast conservation. Cancer 76, 259–267 (1995).
Smitt, M. C., Nowels, K., Carlson, R. W. & Jeffrey, S. S. Predictors of reexcision findings and recurrence after breast conservation. Int. J. Radiat. Oncol. Biol. Phys. 57, 979–985 (2003).
Leong, C. et al. Effect of margins on ipsilateral breast tumor recurrence after breast conservation therapy for lymph node-negative breast carcinoma. Cancer 100, 1823–1832 (2004).
McCahill, L. E. et al. Variability in reexcision following breast conservation surgery. JAMA 307, 467–475 (2012).
Landercasper, J., Whitacre, E., Degnim, A. C. & Al-Hamadani, M. Reasons for re-excision after lumpectomy for breast cancer: Insight from the American Society of Breast Surgeons Mastery(SM) database. Ann. Surg. Oncol. 21, 3185–3191 (2014).
Kaczmarski, K. et al. Surgeon re-excision rates after breast-conserving surgery: A measure of low-value care. J. Am. Coll. Surg. 228, 504-512.e2 (2019).
Landercasper, J. et al. Factors associated with reoperation in breast-conserving surgery for cancer: A prospective study of American society of breast surgeon members. Ann. Surg. Oncol. 26, 3321–3336 (2019).
Moran, M. S. et al. Society of Surgical Oncology-American Society for Radiation Oncology consensus guideline on margins for breast-conserving surgery with whole-breast irradiation in stages I and II invasive breast cancer. Int. J. Radiat. Oncol. Biol. Phys. 88, 553–564 (2014).
Wazer, D. E. et al. Factors influencing cosmetic outcome and complication risk after conservative surgery and radiotherapy for early-stage breast carcinoma. J. Clin. Oncol. 10, 356–363 (1992).
King, T. A. et al. Clinical management factors contribute to the decision for contralateral prophylactic mastectomy. JCO 29, 2158–2164 (2011).
Heil, J. et al. Do reexcisions impair aesthetic outcome in breast conservation surgery? Exploratory analysis of a prospective cohort study. Ann. Surg. Oncol. 19, 541–547 (2012).
Abe, S. E. et al. Margin re-excision and local recurrence in invasive breast cancer: A cost analysis using a decision tree model. J. Surg. Oncol. 112, 443–448 (2015).
Cabioglu, N. et al. Role for intraoperative margin assessment in patients undergoing breast-conserving surgery. Ann. Surg. Oncol. 14, 1458–1471 (2007).
Nunez, A., Jones, V., Schulz-Costello, K. & Schmolze, D. Accuracy of gross intraoperative margin assessment for breast cancer: Experience since the SSO-ASTRO margin consensus guidelines. Sci. Rep. 10, 17344 (2020).
Landercasper, J. et al. Toolbox to reduce lumpectomy reoperations and improve cosmetic outcome in breast cancer patients: The American society of breast surgeons consensus conference. Ann. Surg. Oncol. 22, 3174–3183 (2015).
Olson, T. P., Harter, J., Muñoz, A., Mahvi, D. M. & Breslin, T. Frozen section analysis for intraoperative margin assessment during breast-conserving surgery results in low rates of re-excision and local recurrence. Ann. Surg. Oncol. 14, 2953–2960 (2007).
Garcia, M. T. et al. Accuracy of frozen section in intraoperative margin assessment for breast-conserving surgery: A systematic review and meta-analysis. PLoS ONE 16, e0248768 (2021).
Bakhshandeh, M., Tutuncuoglu, S. O., Fischer, G. & Masood, S. Use of imprint cytology for assessment of surgical margins in lumpectomy specimens of breast cancer patients. Diagn. Cytopathol. 35, 656–659 (2007).
Esbona, K., Li, Z. & Wilke, L. G. Intraoperative imprint cytology and frozen section pathology for margin assessment in breast conservation surgery: A systematic review. Ann. Surg. Oncol. 19, 3236–3245 (2012).
Schnitt, S. J. & Connolly, J. L. Processing and evaluation of breast excision specimens: A clinically oriented approach. Am. J. Clin. Pathol. 98, 125–137 (1992).
Sahin, A. A. Chapter 34: Surgical Margin Evaluation in Patients Treated with Breast-Conserving Therapy. in Advanced Therapy of Breast Disease 341–384 (BC Decker Inc., 2004).
Kopans, D. Breast Imaging (Lippincott Williams & Wilkins, 2006).
McClatchy, D. M. 3rd., Zuurbier, R. A., Wells, W. A., Paulsen, K. D. & Pogue, B. W. Micro-computed tomography enables rapid surgical margin assessment during breast conserving surgery (BCS): Correlation of whole BCS micro-CT readings to final histopathology. Breast Cancer Res. Treat 172, 587–595 (2018).
DiCorpo, D. et al. The role of Micro-CT in imaging breast cancer specimens. Breast Cancer Res. Treat https://doi.org/10.1007/s10549-020-05547-z (2020).
Kulkarni, S. A. et al. High-resolution full-3D specimen imaging for lumpectomy margin assessment in breast cancer. Ann. Surg. Oncol. https://doi.org/10.1245/s10434-021-10499-9 (2021).
Pradipta, A. R. et al. Emerging technologies for real-time intraoperative margin assessment in future breast-conserving surgery. Adv. Sci. (Weinh) 7, 1901519–1901519 (2020).
St John, E. R. et al. Diagnostic accuracy of intraoperative techniques for margin assessment in breast cancer surgery: A meta-analysis. Ann. Surg. 265, 300–310 (2017).
Chagpar, A. B. et al. A randomized, controlled trial of cavity shave margins in breast cancer. N. Engl. J. Med. 373, 503–510 (2015).
Wang, K., Ren, Y. & He, J. Cavity shaving plus lumpectomy versus lumpectomy alone for patients with breast cancer undergoing breast-conserving surgery: A systematic review and meta-analysis. PLoS ONE 12, e0168705–e0168705 (2017).
Kobbermann, A. et al. Impact of routine cavity shave margins on breast cancer re-excision rates. Ann. Surg. Oncol. 18, 1349–1355 (2011).
Marudanayagam, R. et al. Effect of cavity shaving on reoperation rate following breast-conserving surgery. Breast J. 14, 570–573 (2008).
Unzeitig, A. et al. Influence of surgical technique on mastectomy and reexcision rates in breast-conserving therapy for cancer. Int. J. Surg. Oncol. 2012, 725121 (2012).
Janes, S. E. J., Stankhe, M., Singh, S. & Isgar, B. Systematic cavity shaves reduces close margins and re-excision rates in breast conserving surgery. Breast 15, 326–330 (2006).
Huston, T. L., Pigalarga, R., Osborne, M. P. & Tousimis, E. The influence of additional surgical margins on the total specimen volume excised and the reoperative rate after breast-conserving surgery. Am. J. Surg. 192, 509–512 (2006).
Jones, V. et al. Excising additional margins at initial breast-conserving surgery (BCS) reduces the need for re-excision in a predominantly african american population: A report of a randomized prospective study in a public hospital. Ann. Surg. Oncol. 23, 456–464 (2016).
Maloney, B. W. et al. Review of methods for intraoperative margin detection for breast conserving surgery. J. Biomed. Opt. 23, 1–19 (2018).
Cuccia, D. J., Bevilacqua, F., Durkin, A. J. & Tromberg, B. J. Modulated imaging: Quantitative analysis and tomography of turbid media in the spatial-frequency domain. Opt. Lett. 30, 1354–1356 (2005).
Angelo, J. P. et al. Review of structured light in diffuse optical imaging. J. Biomed. Opt. 24, 1–20 (2018).
McClatchy, D. M. et al. Light scattering measured with spatial frequency domain imaging can predict stromal versus epithelial proportions in surgically resected breast tissue. J. Biomed. Opt. 24, 1–11 (2018).
Streeter, S. S. et al. Structured light imaging for breast-conserving surgery, part II: Texture analysis and classification. J. Biomed. Opt. 24, 096003 (2019).
Zhao, Y. et al. Deep learning model for ultrafast multifrequency optical property extractions for spatial frequency domain imaging. Opt. Lett. 43, 5669–5672 (2018).
Chen, M. T., Mahmood, F., Sweer, J. A. & Durr, N. J. GANPOP: Generative adversarial network prediction of optical properties from single snapshot wide-field images. IEEE Trans. Med. Imaging 39, 1988–1999 (2020).
Pardo, A. et al. Modeling and synthesis of breast cancer optical property signatures with generative models. IEEE Trans. Med. Imaging PP, (2021).
Zwanenburg, A. et al. The image biomarker standardization initiative: Standardized quantitative radiomics for high-throughput image-based phenotyping. Radiology 295, 328–338 (2020).
Aerts, H. J. W. L. et al. Decoding tumour phenotype by noninvasive imaging using a quantitative radiomics approach. Nat. Commun. 5, 4006 (2014).
Gillies, R. J., Kinahan, P. E. & Hricak, H. Radiomics: Images are more than pictures, they are data. Radiology 278, 563–577 (2016).
Maloney, B. W. et al. Structured light imaging for breast-conserving surgery, part I: Optical scatter and color analysis. J. Biomed. Opt. 24, 096002 (2019).
Streeter, S. S. et al. Optical scatter imaging of resected breast tumor structures matches the patterns of micro-computed tomography. Phys. Med. Biol. 66, 115021 (2021).
Radovic, M., Ghalwash, M., Filipovic, N. & Obradovic, Z. Minimum redundancy maximum relevance feature selection approach for temporal gene expression data. BMC Bioinform. 18, 9 (2017).
van Griethuysen, J. J. M. et al. Computational radiomics system to decode the radiographic phenotype. Cancer Res. 77, e104–e107 (2017).
McClatchy, D. M. et al. Wide-field quantitative imaging of tissue microstructure using sub-diffuse spatial frequency domain imaging. Optica 3, 613–621 (2016).
Conti, A., Duggento, A., Indovina, I., Guerrisi, M. & Toschi, N. Radiomics in breast cancer classification and prediction. Semin. Cancer Biol. 72, 238–250 (2021).
Sollini, M. et al. PET/CT radiomics in breast cancer: Mind the step. Methods 188, 122–132 (2021).
Nguyen, J. Q. et al. Spatial frequency domain imaging of burn wounds in a preclinical model of graded burn severity. J. Biomed. Opt. 18, 66010 (2013).
McClatchy, D. M. et al. Calibration and analysis of a multimodal micro-CT and structured light imaging system for the evaluation of excised breast tissue. Phys. Med. Biol. 62, 8983–9000 (2017).
Fornacon-Wood, I. et al. Reliability and prognostic value of radiomic features are highly dependent on choice of feature extraction platform. Eur. Radiol. 30, 6241–6250 (2020).
DeLong, E. R., DeLong, D. M. & Clarke-Pearson, D. L. Comparing the areas under two or more correlated receiver operating characteristic curves: A nonparametric approach. Biometrics 44, 837–845 (1988).
Sun, X. & Xu, W. Fast implementation of delong’s algorithm for comparing the areas under correlated receiver operating characteristic curves. IEEE Signal Process. Lett. 21, 1389–1393 (2014).
This work has been funded by NIH research Grants R01CA192803 and F31CA257340.
B.P., S.S., and K.P. have a patent pending (US Application No.: 17/076,788) related to this study. K.P. is Co-Founder of CairnSurgical, Inc. B.P. is President and Co-Founder of DoseOptics, LLC. Authors in their roles in the medical industry did not in any way impact this study. B.H., R.Z., and W.W. declare no potential conflict of interest.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons licence, and indicate if changes were made. The images or other third party material in this article are included in the article's Creative Commons licence, unless indicated otherwise in a credit line to the material. If material is not included in the article's Creative Commons licence and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this licence, visit http://creativecommons.org/licenses/by/4.0/.
About this article
Cite this article
Streeter, S.S., Hunt, B., Zuurbier, R.A. et al. Developing diagnostic assessment of breast lumpectomy tissues using radiomic and optical signatures. Sci Rep 11, 21832 (2021). https://doi.org/10.1038/s41598-021-01414-z
By submitting a comment you agree to abide by our Terms and Community Guidelines. If you find something abusive or that does not comply with our terms or guidelines please flag it as inappropriate.