Rabies Vaccine Characterization by Nanoparticle Tracking Analysis

There are concerns that effectiveness and consistency of biopharmaceutical formulations, including vaccines, may be compromised by differences in size, concentration and shape of particles in suspension. Thus, a simple method that can help monitor and characterize these features is needed. Here, nanoparticle tracking analysis (NTA) was used to characterize particle concentration and size distribution of a highly-purified rabies vaccine (RABV), produced in Vero cells without raw materials of animal origin (RMAO). The NTA technique was qualified for characterization of RABV particles by assessing the stability profile of vaccine particles over 5–55 °C. Antigenicity of the viral particle was also monitored with the enzyme-linked immunosorbent assay (ELISA) and NTA. RABV particle size diameters were 100–250 nm (mean:150 nm), similar to sizes obtained when labelled with rabies anti-G D1–25 monoclonal antibody, suggesting mainly antigenic virus-like particles, also confirmed by transmission electron microscopy. Thermal stress at 55 °C decreased the concentration of anti-G D1–25-labelled particles from 144 hours, coherent with conformational changes leading to loss of G protein antigenicity without impacting aggregation. Results from RABV antigenicity assessment during the 24 months monitoring of stability showed good correlation between NTA and ELISA. NTA is a suitable approach for the characterization of biopharmaceutical suspensions.


RABV particle visualization by transmission electron microscope (TEM). The RABV particles
were also visualized by TEM (Fig. 2). The particles were not aggregated but uniformly distributed irrespective of whether they were not labelled ( Fig. 2A) or labelled with anti-G D1-25 monoclonal antibody (Fig. 2B). 27 They appear as a homogeneous population of bullet-shaped particle corresponding to the expected taxonomic characteristic of Lyssavirus genus. These results confirm that the structure of the viral particle is not altered by the interaction with the monoclonal antibody, thus supporting the specific labelling of antigenic rabies virus particles required to qualify the NTA results.
RABV drug product particle stability. Although the concentration of unlabelled particles remained relatively unchanged with thermal stress over one week (Fig. 3A,C), there was a decrease in the concentration of anti-G D1-25-labelled particles from 144 hours at 55 °C (Fig. 3B,C). Thermal stress had more of an effect on anti-G D1-25-labelled DP particles as the anti-G D1-25 monoclonal antibody recognizes the conformational glycoprotein epitope,. Thus, thermal stress induces significant conformational changes in the RABV particles, specifically changes that lead to a loss of G protein antigenicity over time without impacting the aggregation state. The decrease in RABV (DP) antigenicity with thermal stress was confirmed by ELISA which showed slightly more sensitivity than NTA anti-G D1-25-labelled samples ( Figure S2). The particle concentration and size distribution (Fig. 4) determined by NTA were stable over 24 months at 5-37 °C. Unlabelled particles have no specific viral epitope targeted for antigenicity, therefore are used to verify the viral particle integrity/intactness under temperature stress. www.nature.com/scientificreports www.nature.com/scientificreports/ Correlation between NTA and ELISA. The monitoring of the stability of the RABV (DP) over 24 months at 5 °C, 6 months at 25 °C and 3 months at 37 °C confirmed the ability of the NTA to count and size antigenic RABV particles. There was a good correlation between antigenic RABV (DP) anti-G D1-25-labelled particle concentrations determined by NTA and ELISA (Fig. 4A-F). In addition to counting and sizing particles, as the ELISA, the NTA detected the rate of change of viral particle properties over time as a consequence of the exposure to temperatures higher than those recommended for storage, most likely accounting for the high correlation between methods (Fig. 4G,H). Furthermore, the constant size distribution and concentration of RABV over time as measured by mode, D10, D50, D90 and Span ( Fig. 4I-N) showed that the viral particle was very stable.
Particle concentration and size distribution qualification by NTA. To evaluate the performance of NTA to determine particle concentration and size distribution, the method was first qualified. The specificity was assessed using matrix alone (no antigen) and showed that the technique was specific since no RABV particles were detected with or without anti-G D1-25 labelling. The particle size distribution of the matrix-only samples had different peaks at different sizes (Fig. 5A), which were not in agreement with the expected size of rabies virus family, but inherent to those present in the diluent and considered negligible. The method is thus considered  www.nature.com/scientificreports www.nature.com/scientificreports/ specific due to the absence of response of the matrices alone. The method was linear for the RABV labelled with anti-G D1-25 over the particle count range 1.62×10 10 -1.19×10 11 with a corresponding coefficient of determination (R 2 ) of 0.97 (Fig. 5B). Accuracy of measurement was demonstrated for the concentration in particles by www.nature.com/scientificreports www.nature.com/scientificreports/ an average recovery between (94% and 96%) for the unlabelled particles and between 93% and 115% for the anti-G D1-25 monoclonal antibody labelled particles. Thus, the method can be considered accurate. Precision was demonstrated as the coefficient of variation showed minimal variation between the six repeats of NTA runs for RABV labelled with and without anti-G D1-25 monoclonal antibody as measured by mode, D10, D50, D90 and Span (Fig. 5C). The NTA technique was thus specific, linear, accurate and precise and considered qualified to quantify particle concentration (unlabelled and labelled particles) and size distribution (unlabelled particles only as antibody interaction can change particle size although this does not impact the state of the particle as verified by TEM).

Discussion
In this study, NTA was established as a versatile technique for characterizing RABV particle size, concentration, and antigenicity. RABV particle sizes ranged from 100 nm to 250 nm consistent with the expected size of rabies virus family 28,29 . The concentration was unaffected by anti-G D1-25 monoclonal antibody labelling, suggesting that the particles were mainly comprised of antigenic virus particles. Moreover, anti-G D1-25-labelled particle concentrations and stability measured by NTA correlated with RABV antigenicity determined by ELISA, confirming the ability of the former to characterize antigenic particles. NTA was also able to monitor the stability of antigenic RABV particles over 24 months (as with the ELISA) confirming the size, concentration and antigenicity of the RABV was constant over time, thus indicating vaccine stability. Thermal stress at 55 °C caused a decrease in the concentration of anti-G D1-25 -labelled particles from 144 hours suggesting conformational changes in the viral particles leading to a loss of G protein antigenicity without impacting the aggregation state, which directly correlated to results from the ELISA. In addition, NTA was qualified for characterization of RABV particle count and size measurements and lot-to-lot consistency was demonstrated.
The RABV particle size distribution determined in this study is consistent with that reported previously with NTA by Clénet et al. 21 . In addition, they showed the loss of RABV antigenicity during thermal stress assessments performed between 5 °C and 60 °C over one month by NTA mirrored the loss determined by ELISA up to 45 °C. Clénet et al. also observed the emergence of larger particle population sizes concomitant with decreased antigenicity during thermal stress, suggesting RABV aggregation and demonstrated that RABV particle polydispersity inversely correlated with antigenicity. Of note, in our study, at high thermal stress temperatures (55 °C) with RABV (DP) there was not a complete loss of antigenicity up to 144 hours, whereas at similar high temperatures (60 °C) there was an apparent complete loss of antigenicity within the first day, as shown previously with RABV (DS) in the Clénet study 21 . This suggests that the formulated RABV (DP) in the study has a protective element.
Although NTA may be considered the technique of choice for the characterization of polydispersed aggregate suspensions, such as with RABV, operational settings during analysis and recording may influence estimation of particle size and care should be taken to ensure that these are optimized 20,24 . It is possible that even when identical operational settings are used there may be significant divergence between instruments in estimated mean sizes and size distribution for the same biological suspension 30 . Thus, longitudinal variation in operational settings may need to be assessed using both synthetic beads and samples to validate the accuracy of the NTA equipment over time. Other potential limitations with NTA included that non-spherical particles above 500 nm may not be tracked properly by the software which could lead to smaller size estimation of the particles 24 , and that particle morphology cannot be ascertained. www.nature.com/scientificreports www.nature.com/scientificreports/ In this study, as suggested by TEM, the RABV particles were relatively homogenous in terms of morphology, presenting particle sizes compatible with that expected for rabies virus family when detected by NTA. As well, results using the TEM indicate that the interaction between the antibody and the G-protein from the rabies does not alter the RABV particle and no aggregates are observed, confirming that labelled particles are in the correct conformation and suitable to be counted using the NTA method.
In conclusion, the NTA measurements in our study showed good repeatability, as demonstrated by the specificity, linearity, accuracy and precision in repeated particle count and size measurements. We also demonstrated the utility of NTA to simultaneously count and size antigenic RABV particles with the use of anti-G D1-25 monoclonal antibodies at different states of manufacturing and at different formulations. Moreover, there was good correlation between RABV antigenicity determined by NTA and that determined by the ELISA reference method, confirming the ability of NTA to characterize antigenic particles and to be a complementary technique to ELISA. We propose the use of NTA as a simple method to facilitate the characterization of RABV particle suspensions throughout the shelf life of the product.

Material and Methods cells and viruses.
Samples of purified Vero Cell Rabies Vaccine (PVRV, Verorab) and PVRV-NG2 (VRVg 2.0, Sanofi Pasteur, Lyon, France) were obtained inactivated by β-propiolactone at the end of substance manufacturing process (bulk stage; drug substance [DS]). The drug product (DP) batches were obtained formulated from the inactivated bulk in three lots S4483, S4491 and S4497. RABV was formulated in stabilizer containing a mixture of excipients (amino acids, sugar, surfactant and chelator) in a phosphate buffer (pH 8) that had been reconstituted with water. DP rabies virus was incubated at 5 °C, 25 °C and 37 °C over 24 months before NTA and ELISA analysis and comparison. The thermal stability of RABV (DP) particles was also assessed in samples incubated at 55 °C for up to seven days. nanoparticle tracking analysis (ntA). All analyses were performed using a NanoSight NS300 instrument (Malvern Instruments Ltd, Worcestershire, UK), equipped with a 488 nm laser and a high sensitivity scientific complementary metal-oxide-semiconductor (sCMOS) camera from Malvern Instruments Ltd, Worcestershire, UK 21 .
Each DP sample was added to 0.5 mL of 0.4% sodium chloride, then diluted in PBS (CXXPBS00-01 Eurobio, les Ulis, France) before being analyzed by NTA. Three independent replicates for each sample (DS and DP) were analyzed and three videos of 60 seconds recorded for each. A minimum of 1,000 valid tracks were required to validate each experimental test sample. Live monitoring NTA acquisition was performed using a syringe loading system. Detection threshold (four for labelled and unlabelled particles), limiting background noise, and Camera Level (12 for unlabelled particle and 16 for labelled particle acquisitions) settings were maintained unchanged between sample acquisitions. Data were processed with NanoSight NTA 3.1 software (Malvern Instruments Ltd, Worcestershire, UK). These parameters/settings were defined in collaboration with Malvern Instruments Ltd, Worcestershire, UK.
All samples were assessed with or without anti-G D1-25 to the rabies glycoprotein conformational epitope, as previously described 25 . In brief, for the anti-G D1-25 labelled particles, RABV samples were incubated for 1 hour at 37 °C with anti-G D1-25 monoclonal antibody directly conjugated to Alexa Fluor 488 by BIOTEM (Apprieu, France), at 1:200 dilution in phosphate buffered saline. NTA data acquisition was performed using a 500 nm long pass filter to count fluorescent virus particles with bound D1-25 monoclonal antibody. Analyses of unlabelled and anti-G D1-25 labelled particles were two distinct assessments.All measurements were made according to the American Society for Testing and Materials (ASTM) 31 .
Enzyme-linked immunosorbent assay (ELISA). RABV antigenicity was also assessed by ELISA using two monoclonal anti-G antibodies: D1-25 and WI 1112. Monoclonal antibody W1112 was coated on ELISA plates. Plates were washed and saturation buffer added to each well and plates incubated at 37 °C for 1 h. The rabies vaccine lots were added, and plates incubated at 37 °C for 1 h. Plates were then washed and buffer containing biotin-labelled monoclonal antibody D1-25 added to each well. Plates were then washed and buffer containing streptavidin peroxidase added to each well. Plates were then washed and buffer containing revelation substrate added to each well. The OD 492 nm was determined using an ELISA reader (Molecular Devices) as previously described 25 . This ELISA was proposed to the European Directorate for the Quality of Medicines and Healthcare (EDQM) and accepted for an international collaborative study to replace the NIH activity tests for rabies vaccines 32 .
Transmission electron microscopy (TEM). Samples, with or without anti-G D1-25 monoclonal antibody, were negatively stained by 2% uranyl acetate as previously described 23 and observed on a FEI Tecnai F20 electron microscope operated at 200 kV. Images were recorded with a 4k x 4k USC 4000 slow-scan CCD camera (GATAN, Inc. Pleasanton, California, United States).
NTA qualification. The specificity, linearity, accuracy and precision of NTA for determining particle concentration and size distribution were qualified based on the Sanofi Pasteur internal procedure. Specificity was assessed using the vaccine final product matrix (i.e. not containing antigen) with or without anti-G D1-25. Linearity and accuracy was determined through triplicate NTA runs (k = 3) performed by two operators on different days. Intermediate precision was evaluated through six NTA runs (k = 6), by two operators on different days, repeated three times in order to evaluate repeatability. Particle size distributions for non-labelled particle only (as these show the particle size and the sample homogeneity), were described using five size parameters: mode (nm): size of the majority population, D10 (nm): particle diameter corresponding to 10% cumulative undersize particle size distribution, D50 (nm): the median diameter of particle size distribution, D90 (nm): the particle diameter corresponding to 90% cumulative undersize particle size distribution and Span: the distribution width which is an indicator of aggregation.