Accurate and precise determination of 90Sr at femtogram level in IAEA proficiency test using Thermal Ionization Mass Spectrometry

A novel method for the determination of ultra-trace level 90Sr has been recently developed applying thermal ionization mass spectrometry (TIMS). The method includes the chemical separation of Zr (isobaric interference of 90Zr) from the samples followed by determination of 90Sr/88Sr abundance sensitivity (2.1 × 10−10). The analytical performance of this method was assessed in the IAEA-TEL 2017-3 worldwide open proficiency test. For 90Sr determination, tap water and milk powder samples were distributed amongst the participant laboratories with reference values of 11.2 ± 0.3 Bq kg−1 (2.2 ± 0.1 fg g−1) and 99.9 ± 5.0 Bq kg−1 (19.5 ± 1.0 fg g−1), respectively. The stable Sr concentrations were 39.4 ± 0.9 ng g−1 and 2.5 ± 0.1 µg g−1 while the 90Sr/88Sr isotope ratios were 6.47 ± 0.17 × 10−8 and 9.04 ± 0.45 × 10−9 in the tap water and milk powder samples, respectively. For TIMS measurement, 50 mL water and 1 g milk powder samples were taken for analysis. This TIMS method demonstrated an impressive accuracy (relative bias of 4.2% and −2.1%, respectively) and precision (relative combined uncertainty of 4.1% and 7.6%, respectively) when compared with radiometric techniques. For the first time in the history of inorganic mass-spectrometry, 90Sr analysis using a TIMS instrument is confirmed by an independent proficiency test.

fg g −1 , for contaminated Fukushima soil samples it is around 4 fg g −1 (20 Bq kg −1 ) or lower 9 . In order to successfully measure 90 Sr using inorganic mass spectrometry either a high mass resolution (~30,000) which is not yet available to resolve the isobaric 90 Zr peak or an efficient chemical purification method for removal of Zr from Sr at ultra-trace levels is required 10,11 . For practical purposes, quite a high decontamination factor (in the range of 10 8 to 10 11 ) is required for 90 Zr isobaric interference removal (accepting just a 10% contribution to the 90 Sr signal) in case of soil samples 12 .
The second critical point for the application of mass spectrometry instruments for 90 Sr determination is the peak tailing on the higher mass side from 88 Sr. Although there exists reliable ultra-high vacuum condition (<10 −8 Pa) in the source and analyser zone of the mass spectrometry instruments, scattering of ions occurs during their acceleration. This scattering process causes the energy spread of the ion beam and it is recognized as a peak tail in the mass spectra. In the case of 90 Sr ion beam detection, undesired interference exists because of the peak tailing of the major abundant 88 Sr isotope. To depict this interference the abundance sensitivity term is used and defined as: =

Abundance sensitivity
Ion intensity at mass 90 Ion intensity at mass 88 (1) For reliable 90 Sr analysis in environmental samples, abundance sensitivity in the range of 10 −7 to 10 −11 is required.
Although, there is a new trend to utilize mass spectrometry instruments for 90 Sr analysis, those methods are neither well established nor routinely used. Accreditation to an internal standard applying mass spectrometry method for 90 Sr determination has not been well documented. The main reason for the less development is due to interdisciplinary challenges of this method. Traditionally, the determination of 90 Sr is carried out by radio-chemists with a strong chemistry and nuclear measurement background while the application of mass spectrometry instrument is the domain of mass spectrometry experts with strong physics and/or engineering background.
Considering the need of rapid and precise 90 Sr determination after the Fukushima nuclear accident, a 90 Sr analysis method was developed using the Phoenix X62 TIMS in the laboratory of the National Institutes for Quantum and Radiological Science and Technology (QST), Japan. During the validation process of this new method, detection limit, abundance sensitivity ( 90 Sr/ 88 Sr), accuracy and precision were determined with reference materials e.g. IRMM-426 (wild berry) and NIST-4354 (freshwater lake sediment) assay proving that the method is suitable for environmental level 90 Sr analysis 6 To establish a higher level of quality assurance, it is essential to participate in a proficiency test controlled by an internationally recognised reference laboratory. To fulfil this standard, our laboratory participated in the IAEA-TEL-2017 world-wide open proficiency test organized by the International Atomic Energy Agency. For the first time, an inorganic mass spectrometry method was employed in a proficiency test for 90 Sr analysis, which is the exclusive territory of radiometric methods.

Results
In 2017, the IAEA-TEL-2017 World-Wide Open Proficiency Test was organized by the Reference Materials Group of the IAEA Terrestrial Environment Laboratory, Seibersdorf, Austria, in order to monitor the analytical capabilities and reveal gaps in analytical methodologies of different radioanalytical laboratories 13 . For this purpose, five different samples spiked with known amounts of target radionuclides were distributed. Three tap water samples (Sample 1, 2 and 3) each around 500 g, one milk powder (Sample 4) around 180 g and one Ca-carbonate from spring water (Sample 5) around 40 g were allocated. The samples were prepared with various alpha-, betaand gamma-emitter natural (radionuclides of 238 U and 232 Th decay series) and anthropogenic (isotopes of Cs, Ba, Zr, Tc, Mo, Ru, H, I, Ce, La, Nd, Nb, Np, Sr) radionuclides but our laboratory has interest only in the 90 Sr determination.
For 90 Sr determination, Sample 1, 3 and 4 were spiked with 90 Sr. Sample 3 was a quality control sample with described radionuclide content. The details of the samples analysed are summarised in the Table 1.
As discussed in the introduction, Sr and Zr concentration can affect the reliability of 90 Sr analysis using mass spectrometry instruments. Naturally, the water contains less amounts of trace elements compared to any kind of mammal milk. Therefore, the water sample contains significantly lower Sr concentration (39.4 ng g −1 ) than the concentrated milk powder (2.5 µg g −1 ). In the water samples, the 90 Sr concentrations were in the same range, 23.7 Bq kg −1 (4.6 fg g −1 ) and 11.2 Bq kg −1 (2.2 fg g −1 ) respectively. In the milk powder it was higher, 99.9 Bq kg −1 (19.5 fg g −1 ), although it corresponds to 13 Bq kg −1 (2.5 fg g −1 ) in the original milk sample since the average dry matter content of a raw cow milk is around 13% 14 . Thus, stable Sr and 90 Sr concentrations were different in samples, 90 Sr/ 88 Sr isotope ratio also differed and varied in the range from 1.37 × 10 −7 to 9.04 × 10 −9 . The  Tables 2-4. The quality control tap water samples were measured four times while the test samples five times. In the report submitted to IAEA, the mean decay corrected results were reported, 11.7 ± 0.4 Bq kg −1 for the tap water samples and 97.9 ± 5.5 Bq kg −1 for the milk powder (k = 1). The range of TIMS measurement results were very narrow, varied in the range of 11.6-11.8 Bq kg −1 for tap water samples and 96.6-99.9 Bq kg −1 for milk powder samples. Each measurement produced results with excellent accuracy and precision compared to the reference values of 11.2 ± 0.3 (2.2 ± 0.1 fg g −1 ) and 99.9 ± 5 Bq kg −1 (19.5 ± 1.0 fg g −1 ), respectively. These results demonstrated the robust analytical capacity of TIMS for 90 Sr analysis.
Statistical analysis. Water. The reported results of the proficiency test were published in the evaluation report 13 . The results of tap water sample (Sample 1) are shown in Fig. 1. Totally 94 laboratories were involved and 75 (80%) laboratory performances were 'accepted' while 6 (6%) were warned (not acceptable precision) and 13 (14%) were 'not accepted' . The ranges of the results were wide, from 2.9 to 22.2 Bq kg −1 . The TIMS result (marked with black quadrat) proved that the new TIMS method is competitive with radiometric methods for the measurement of 90 Sr in water samples.
TIMS result (11.7 Bq kg −1 ) was slightly higher than the reference value (11.2 Bq kg −1 ) thus the relative bias (RB) statistical parameter, calculated by Eq. 3, was quite good, 4.2% as illustrated in Fig. 2 www.nature.com/scientificreports www.nature.com/scientificreports/ significantly lower than the maximum acceptable relative bias (MARB) for tap water samples (20%) thus the accuracy of TIMS measurement were 'accepted' (acceptance criteria of the accuracy: RB ≤MARB).
To satisfy the performance criteria of the precision of an applied measurement method, the reported results must overlap with the reference value within their uncertainties. Additionally, the precision (P) must be below (or equal to) the MARB and over (or equal to) the relative bias (k = 2.58) as interpreted in Eq. 4 15 . In this study, the uncertainty of TIMS measurements was 3.1%, the calculated P value 4.1% thus the precision attained was satisfactory and certified as 'accepted' . The precisions of the tap water measurements are shown in Fig. 3. The precision of the measurements fluctuated between 2.8 and 30.6% and only in three cases was too high to be accepted. In the Fig. 3 only the MARB can be seen as performance criteria since the RB varies individually thus not presentable. In three more cases, the precision below the MARB level, was not accepted because of high RB (RB ˃2.58 × P) of the measurement results (which is not shown in Fig. 3). The major reason of failure in the proficiency test is the low accuracy (high RB) of the values reported by the participant laboratories as can be seen from the results of Figs 2 and 3.
The z-score statistics has been used to evaluate the performances of the participant laboratories and are shown in Fig. 4. The z-score calculation was carried out using the robust standard deviation described in Eqs 5 and 6. As it can be seen from the calculation method, the z-score value does not give exact information about the precision only the accuracy. Moreover, the calculated z-score is a relative parameter, as robust standard deviation originates from the reported results and is affected by the performance of the participant laboratories. The reported results with a higher z-score of 2 not acceptable. The range of z-scores was from −7.7 to 10.3. Four laboratories reported exactly the same value as the reference one (z-score = 0) and seven more laboratories presented values with almost the same z-score as in the range from −0.01 to 0.01. In our case, the z-score of 0.44 was in the range from −0.5 to 0.5 which is quite impressive since only 37 laboratories of the 94 were able to accomplish this level of accuracy. Totally 80 laboratories were involved, 54 (67.5%) laboratories gained the "accepted" status while 26 (32.5%)  www.nature.com/scientificreports www.nature.com/scientificreports/ laboratories were not able to fulfil the performance criteria of proficiency test (MARB = 15%). The ranges of the reported results were also very wide, from 3.9 to 680 Bq kg −1 . Compared to tap water results, reliable 90 Sr analysis in milk powder seems a more challenging task.
The TIMS result is marked here with a black quadrat and confirmed that this novel TIMS method is competitive with radiometric methods to determine 90 Sr in milk powder samples. In this case, TIMS result (97.9 Bq kg −1 ) was slightly lower than the reference value (99.9 Bq kg −1 ) thus the relative bias was also very good, −2.5% as presented in Fig. 6 where the range of the RB was very wide, from −96.1 to 580.7%. Since the relative bias of the   www.nature.com/scientificreports www.nature.com/scientificreports/ TIMS result was significantly lower than the MARB value (15%), the "accepted" certificate was obtained for the accuracy.
The uncertainty of this TIMS measurement was 5.7% and the calculated P value was 7.6% therefore the precision of the TIMS method complied with the standards assigned by the IAEA proficiency test. The precision of the milk powder measurement is given in Fig. 7 and it was found to range between 5.1 and 21.8%. There were only two measurements reported where the precisions were over the 15% MARB as can be seen in Fig. 7 and 19 more measurements (not presented in Fig. 7) where the RB was too high (low accuracy) to be covered by the expanded relative combined uncertainty (2.58 × P).
The calculated z-scores of the milk powder measurements are presented in the Fig. 8. Here the z-score acceptance limit (1.1) was much lower than in case of the water sample (2) because the MARB level for milk powder (15%) was lower, than for tap water (20%). In this case, the range of z-scores was quite wide, from −7.7 to 46.7. However, the z-score values of 31out of 80 laboratories reported were in the range from −0.5 to 0.5 as well as our laboratory with the excellent result of −0.17.

Discussion
Precision of the TIMS method. TIMS is a highly precise instrument to measure accurately stable isotope ratios. For instance, stable Sr isotope ratios can be determined with a relative standard deviation of around 0.003% 16,17 . This precision performance is not possible for 90 Sr analysis because the sample amount of 90 Sr loaded onto a single degassed zone refined Re filament is very low, only a few fg 6 . Therefore, the generated 90 Sr ion beam is quite weak and only few counts are detectable with a relatively high measurement error. In case of tap water sample (Sample 1), around 400 ng stable strontium was loaded along with 20 fg 90 Sr, while for milk powder samples (Sample 4) around 1000 ng stable strontium was loaded along with 8 fg 90 Sr. Consequently, the 90 Sr beam intensity was higher for tap water samples, resulting in better relative standard deviation of 90 Sr/ 88 Sr isotope ratio (~2%) than for the milk powder (~5%) as can be seen in Tables 3 and 4. The total number of cycles measurement were 50 for both samples, consequently the final result was derived from the average of 50 measurement cycles.
The detection capacity limit of the Faraday cups during TIMS measurement for abundant 88 Sr or 86 Sr peaks play a significant role that needs special attention. At the beginning, a direct measurement of the 88 Sr peak on a  www.nature.com/scientificreports www.nature.com/scientificreports/ Faraday cup and simultaneously 90 Sr on a Daly ion-counter was tested. 88 Sr ion beam detection was not possible over 10 V using Faraday cup with resistor of 10 11 Ω and the 90 Sr ion beam remained undetectable on the Daly ion-counter if the 90 Sr/ 88 Sr isotope ratio was lower than 10 −8 . Since there was no way to expand the detection range of the Faraday cups, a less abundant Sr isotope, 86 Sr was used to detect the reference ion beam instead of 88 Sr. In Sr element the abundance of 88 Sr is 82.58%, whereas the abundance of 86 Sr is 9.86%. Moreover, the 86 Sr/ 88 Sr isotope ratio (0.1194) is a naturally stable ratio with fluctuation less than 0.1% 18 . Considering that the uncertainty of the 90 Sr determination is more than 1% for the TIMS method, the uncertainty of 86 Sr/ 88 Sr isotope ratio has to be compared always with the overall measurement uncertainty for all kinds of environmental samples. The 87 Sr isotope or 87 Sr/ 88 Sr isotope ratio cannot be utilized due to the radiogenic origin of the 87 Sr (decay product of the 87 Rb).
Detection limit of the TIMS method. The detection limit (DL) of 90 Sr analysis for radiometric methods is affected by two major parameters. One is the background count rate and second the sample amount. Basically, the background count rate is a constant, caused by background radiation derived from 40 K, U-and Th-decay series (these natural radionuclides occurs in building materials, metal parts of the instrument, e.g. lead shield, etc.) and electric noise of the measurement system 19 . With increments of the sample intake, 90 Sr concentration is increased in the analyte and the DL can be improved. However, the limitation lies in expenditure (necessity of more chemicals, resins, etc.) and increase in preparation time along with the sample amount.
There are two affecting parameters also for the mass spectrometry method. One is the 90 Sr/ 88 Sr abundance sensitivity, corresponding to a background effect of the peak tail of 88 Sr and electric noise of the instrument. The abundance sensitivity is a kind of quality of a given mass spectrometry instrument which may decline with time as the pressure degrades mainly in the analyser house. However, the abundance sensitivity must be considered as a constant and needs to be checked regularly as a protocol of quality control.
The other parameter is not the sample intake itself but the stable Sr concentration of sample. For example, if one litre water and milk have the same level of 90 Sr contamination, the 90 Sr/ 88 Sr isotope ratio will be higher in the water sample than in the milk sample because water has a significantly lower stable Sr concentration than the milk. Therefore, if 90 Sr/ 88 Sr abundance sensitivity is a constant, the detection limit will be lower for water than milk samples. In this proficiency test, as it can be observed in Table 2, tap water (39.4 ± 0.9 ng g −1 ) had two magnitudes lower stable Sr concentration than milk powder (2.5 ± 0.1 µg g −1 ). This explains the detection limit data in Tables 3 and 4, where the detection limit is two magnitudes lower for the tested water (0.04 Bq kg −1 ) than for milk powder sample (2.3 Bq kg −1 ). Consequently, mass spectrometry is competitive with radiometric method if the stable Sr concentration is in the range of ng g −1 in environmental samples, such as in fresh water samples.
Sample intake. The stable Sr concentration also influences the sample size that is necessary for 90 Sr analysis with TIMS. Typically, 400 ng of stable Sr loaded on the filament is adequate for reliable analysis but larger sample amounts result in a more controllable beam and more measurement cycles. Obviously, to get the required amount of stable Sr, more water sample was used than milk powder for 90 Sr analysis. Counting with some strontium loss during the sample preparation and separation, the sample intake was calculated with 2000 ng stable strontium. This stable strontium amount corresponded to 50 mL water sample and 1 g milk powder. The homogenous distribution of the added 90 Sr in water sample is not an issue but in case of the milk powder non-homogeneity may occurs. To decrease the uncertainties derived from non-homogeneity, 50 g milk powder sample was recommended by the organizers of the proficiency test as a minimum sample intake. Since the sample preparation and strontium separation procedures of the TIMS method is optimized for 2000-4000 ng stable strontium, the risk was taken and only 1 g milk powder sample was digested to analyze 90 Sr. As our measurement results demonstrated, the homogeneity of 90 Sr in the milk powder sample was excellent and 1 g sample represented 90 Sr distribution perfectly.
The sample homogeneity and stability are fundamental requirements for reference materials and proficiency test materials. In case of homogeneity test, the advantage of the mass spectrometry method is to be recognized. For example, to test the homogeneity of 90 Sr in soil matrix, 100 mg or less sample amount is sufficient for reliable 90 Sr analysis using the TIMS instrument, however the detection limit must be considered.

Limitations of the proficiency test and quality control.
The TIMS method demonstrated a very precise and accurate 90 Sr measurement capacity but the limitations of the proficiency test must be considered. In case of the mass spectrometry methods, both the 90 Sr activity concentration and the 90 Sr/ 88 Sr isotope ratios are www.nature.com/scientificreports www.nature.com/scientificreports/ important. As it can be observed in Fig. 9, 90 Sr/ 88 Sr isotope ratio of the water and milk powder samples distributed for the IAEA proficiency test are relatively high, varies between 1.4 × 10 −7 and 9.0 × 10 −9 therefore 90 Sr detection was rendered possible using TIMS. Moreover, the Zr interference was not a real challenge since Zr concentrations in the samples were quite low (pg g −1 range). As Fig. 9 presents, soil sample analysis is more demanding due to higher concentrations of stable Sr (µg g −1 range) and Zr (µg g −1 range) concentration in soil samples. In case of Fukushima soil samples wherein stable Sr concentration is over 100 µg g −1 , 90 Sr is not detectable around 100 Bq kg −1 but could be confidently measured at around 1000 Bq kg −1 . Consequently, TIMS 90 Sr measurement capacity is out of range to use the most preferred and frequently used IAEA-375 reference material (Chernobyl affected soil sample) for internal quality control.
As Fig. 9 demonstrates there are adequate reference materials containing 90 Sr for mass spectrometry analysis (wide range of 90 Sr/ 88 Sr isotope ratios and Zr concentration) but interestingly for radiometric analysis it is lacking, considering the characteristics of the Fukushima accident. Fukushima accident affected environmental samples are mainly contaminated with volatile beta emitter fission products such as, I, Te and Cs isotopes. The contamination of non-volatile (Sr, Ru isotopes etc.) and refractory elements (Pu, U isotopes etc.) are very low in comparison because the primary concrete containment system mounted at the Fukushima Daichi reactors and the reactor pressure vessels were not damaged significantly by hydrogen explosions [20][21][22] . The 90 Sr (and 89 Sr) is a pure beta emitter radionuclide and beta particles have continuous spectrum unlike gamma rays therefore efficient separation of interfering beta emitter radionuclides is crucial before 90 Sr measurement with radiometric instruments (every fission product is a beta emitter as well as gamma emitter except 90 Sr and 89 Sr). In Fukushima environmental samples the 137 Cs (T 1/2 = 30.2 y) has the highest significance for long-term radioactive contamination 23 . The 90 Sr/ 137 Cs activity ratio in Fukushima contaminated samples is around 1 × 10 −4 which corresponds to a Cs decontamination factor of 10 5 24 . As can be seen in Fig. 10, reference materials with this 90 Sr/ 137 Cs activity ratio are not available, wherein presence of 137 Cs significantly dominates 90 Sr. The 90 Sr/ 137 Cs activity ratio in certified reference materials varies between 3.6 × 10 −3 and 1.8 × 10 1 (only the IAEA-152 milk sample was in the range of 10 −3 but acquisition of this sample is not possible as it is out of stock). Consequently, applicable internal quality control is not possible and valid proficiency tests are not available for laboratories where 90 Sr determination is carried out in Fukushima samples using radiometric methods. One possible way to solve this problem, until the production of proper reference materials, is the cooperation of laboratories where mass spectrometry methods  www.nature.com/scientificreports www.nature.com/scientificreports/ are applied for 90 Sr analysis and selected Fukushima samples could be determined in both laboratories for the performance estimation (inter-laboratory comparison).

Materials and Methods
Sample preparation. 50 mL water sample was evaporated and dissolved with 5 mL 8 M HNO 3 . 1 g milk powder was ashed at 600 °C and dissolved with 5 mL 8 M HNO 3 .
For HNO 3 solution preparation, analytical grade Tamapure AA-100 reagents (68% w/w) and Milli-Q2 purified water (>18 MΩ cm at 25 °C) were used. The detailed information of reagents and materials, sample preparation, strontium separation and 90 Sr analysis using TIMS are given elsewhere 6,12 .  www.nature.com/scientificreports www.nature.com/scientificreports/ Instruments. The TIMS instrument used are described elsewhere 6 but schematic diagram of the 90 Sr measurement method (Fig. 11), layout of the Phoenix X62 TIMS (Fig. 12) and 90 Sr detection in the TIMS analyser section (Fig. 13) are presented here: During the measurement, 86 Sr ion beam is detected, while 88 Sr ion beam is not collected by any Faraday cups and directed into the analyser wall. The 90 Sr/ 88 Sr isotope ratio is calculated from the measured 90 Sr/ 86 Sr ratio 6 . With this method, 90 Sr ion beam is detectable on the Daly ion-counting detector in samples where 90 Sr/ 88 Sr isotope ratio is in the range of 10 −9 and 10 −10 as it occurs in many environmental samples.
For stable strontium concentration determination, Agilent-8800 inductively coupled plasma mass spectrometer (Agilent Technologies, USA) was used. The detailed strontium analysis method is described elsewhere 25 . interference removal of 90 Zr. Owing to the evaporation temperature and ionization potential difference between Sr (~1380 °C and 5.7 eV) and Zr (~1900 °C and 6.6 eV), Zr separation can be accomplished during the thermal ionization process, however to reach higher decontamination level (10 8 to 10 11 ), additional chemical separation of Zr is necessary. Applying two consecutive DGA and Sr resin extraction chromatography separations on environmental samples prior to the TIMS measurement, Zr decontamination factor of 10 11 can be accomplished 6,12 . Abundance sensitivity. The abundance sensitivity of the Phoenix X62 TIMS is 2.1 × 10 −10 , which is excellent due to very stable ion-beam generation via thermal ionization and ultra-high vacuum condition in the source (10 −8 Pa or lower) and the analyzer (10 −9 Pa or lower). The abundance sensitivity of the TIMS has been regularly checked and determined measuring NIST-987 standard reference material without any traces of 90 Sr 6 . The NIST-987 has been purified by extraction chromatography to avoid 90 Zr interference. 90 Sr activity concentration calculation. The 90 Sr activity concentration was calculated as the follows: where, P is the precision of the measurement, u m is the combined uncertainty (k = 1) of the reported 90 Sr activity concentration (Bq·kg −1 ) and u ref is the combined uncertainty (k = 1) of the reference 90 Sr activity concentration (Bq·kg −1 ).
where, z is the z-score value, A m is the reported 90 Sr activity concentration (Bq·kg −1 ) and A ref is the reference 90 Sr activity concentration (Bq·kg −1 ), Robust sd is the robust standard deviation (Bq·kg −1 ). The robust standard deviation was calculated as follows: = . × − Robust sd A A 1 483 median of (6) m r ef

conclusion
The results of this work demonstrate that our scientific efforts, to push the boundaries of inorganic mass spectrometry into a higher level, were successful. This is the first time that an inorganic mass spectrometry method fulfilled the requirements of a proficiency test in environmental level 90 Sr determination. Considering the accuracy, precision and detection limit, the new mass spectrometry method is very competitive with the traditional radiometric methods and ready for accreditation. The outstanding sensitivity of Daly ion-counting detector and improved abundance sensitivity of the thermal ionization mass spectrometry widen the application not only stable isotope ratio but also radioactive isotope ratio analysis of radionuclides with a half-life less than 100 years. In case of a nuclear accident when the rapid analysis of 90 Sr is critical, the usage of this new method is recommended.
The new method needs further improvement to decrease the strontium amount necessary for analysis. This step can significantly reduce the sample intake along with the corrosive acid usage. Furthermore, the separation time can be also shortened promising the most rapid 90 Sr analysis method in the future. It can also be a useful tool for reference material characterization, where employment of different methods is important for reliable results. Additionally, the low sample intake makes the new method capable to test radionuclide homogeneity in reference material.
Considering the 90 Sr/ 88 Sr isotope ratio, the availability of reference materials is adequate to support the internal quality control process of laboratories where mass spectrometry method used for 90 Sr analysis. However, from the viewpoint of 90 Sr/ 137 Cs activity ratio, reference material related to the Fukushima accident is not accessible.