Abstract
In this paper, we present a novel numerical approach for increasing the resolution of retrieved images of objects after their diffraction patterns are recorded via terahertz pulse timedomain holography (THz PTDH). THz PTDH allows for spectrally resolved imaging with high spatial resolution and does not require the fine alignment of complex optics in the THz path. The proposed data postprocessing method opens up the possibility to reconstruct holograms recorded with spatially restricted THz detectors, and overcome the diffraction limit even for the lowerfrequency spectral components. The method involves an iterative procedure of backwardforward wavefront propagation to simulate the field distribution beyond the initially recorded hologram area. We show significant improvement in both the object reconstruction and contrast across the whole spectrum, with qualitative resolution enhancement at lower frequency spectral components.
Similar content being viewed by others
Introduction
Since its first demonstration back in 1980s^{1}, terahertz (THz) radiation has found numerous spectroscopic and imaging applications^{2}. With the development of compact yet powerful broadband THz transceivers^{3}, THz timedomain spectroscopy (THz TDS) has become a vital tool for numerous diagnostic tasks. Although it can be time consuming, pointbypoint object scanning with the focused THz TDS beam allows for spectrally resolved THz imaging. This approach, first demonstrated in 1996^{4,5}, has been successfully applied to numerous applications, including: (i) in vitro and in vivo tissue characterization on the cellular level for ophtalmology^{6}, dentistry^{7,8} and cancerous cell detection^{9,10,11,12,13}; (ii) noncontact monitoring of parcels and luggage contents for concealed metals or chemical/biological threats^{14,15,16,17}; (iii) quality control and identification of harmful substances in food products^{18,19,20}; (iv) defect testing in materials^{21,22,23,24,25}; (v) chemical composition analysis for oils and fuels^{26,27,28}; and (vi) nondestructive composition analysis of highvalue art objects and antiquities^{29,30,31,32}. All of these examples involve pointbypoint scanning with THz TDS setups, and would greatly benefit from immediate detection of transverse complex amplitude spatiotemporal distribution in the wavefront crosssection.
THz imaging with matrix detectors usually involves the use of narrowband sources, gas lasers^{33,34,35}, free electron lasers^{36}, or (most popularly nowadays) quantum cascade lasers (QCLs)^{37}, with incoherent detection of intensity distribution by microbolometer^{34,37} or pyroelectric^{33,35,36} arrays. Although this type of imaging allows for amplitude and phase image reconstruction, it does not provide any crucial spectroscopic data. By contrast, THz Pulse TimeDomain Holography (THz PTDH)^{38,39} yields full wavefront reconstruction for all spectral components, thus providing immediate information on matter distribution across a larger area crosssection after a single timedomain scan and consecutive numerical reconstruction.
Earlier, a way to detect the whole THz wavefront coherently was demonstrated^{40}, and later applied for pioneering pulsed THz nonholographic^{41} and holographic reconstruction^{38,42,43,44}. However, there are still many unsolved problems in THz PTDH, some of them being fundamental. THz PTDH allows one to obtain a high resolution in all three dimensions. Nevertheless, the capabilities of the method are limited by relatively small measurement areas, because of the detection sensitivity and low availability of the large electrooptic crystals. To overcome the aforementioned limitations, several approaches have been proposed in monochromatic CW THz holography: several exposures of the hologram, while moving the detector^{34}, numerical filtering for signaltonoise ration enhancement^{45}, and iterative phase retrieval method^{33,46,47}.
From traditional holography in the visible range, it is well known that an object can be retrieved from an arbitrary small hologram, with resolution inversely proportional to its size^{48}. In lensless digital holography, an additional restriction is set by the number of pixels in the digitized hologram. The hologram size and number of pixels are even more vital in THz holography. On one hand, the sensor area cannot be immediately extended to cover a wider spatial angle, while on the other hand, the pixel size cannot be reduced since it is already equal or less than THz wavelengths, reaching from ∼100 µm to ∼3 mm.
Recently, for visible range holography, Latyshevskaia et al. proposed a selfextrapolation technique^{49} which allows for overcoming the Abbe criterion and resulting resolution enhancement of the reconstructed objects by padding the hologram and iteratively filling the padded area with the numerically propagated wavefront. Theoretical justification (more detailed in^{50}) of the complex wave spatial extrapolation was discussed by J.L. Harris^{51}, who has shown the uniqueness of the problem solution for given spatial spectrum distribution of the analytical signal on the basis of Whittaker Watson law “the spectrum of a sizelimited object is always analytical”^{52} and Guellemin law “a function of a complex variable is determined throughout the entire Zplane from a knowledge of its properties within an arbitrarily small region of analyticity”^{53} Later, this method was extended to phase retrieval^{50,54} and inline narrowband THz holography^{33}.
In this paper, we propose an adaptation of the above mentioned technique for resolution enhancement of broadband THz pulse timedomain holograms. This method is in high demand, due to the absence of large (in the numbers of wavelength) area THz detectors. We show that the enhancement of the holograms occurs only at wavelengths that yield at least one full interference fringe within the recorded hologram. To the best of our knowledge, this is the first demonstration of the method for broadband radiation.
Terahertz Pulse TimeDomain Holography
To record THz pulse time domain hologram one must first record the amplitude and phase of a collimated wavefront at the detection plane, then place the object of interest to record the diffraction pattern with the same detector. As a matter of fact it is a broadband inline holography with coherent detection provided by pulsed THz that gives information on both amplitude and phase distribution upon measurement.
There are two main techniques for coherent THz detection: (i) freespace electrooptical sampling (EOS)^{55} or; (ii) detection in photoconductive antennas (PCAs)^{1}. Each technique has its own advantages and disadvantages^{56}, but they can both be implemented in synchronous and asynchronous^{57,58} layouts, providing reasonably comparable results, the latter being more fast but at the same time requiring significantly more expensive setup. The PTDH resolution enhancement approach we offer can be applied to any coherent wavefront THz detector, and, as there are currently no array PCA detectors, we propose the following layouts for full wavefront EOS using either a synchronous, or asynchronous approach (Fig. 1).
The setup on the left is based on a classical EOS THz detection layout. The more advanced setup on the right uses a retroreflected probe IR beam, thus allowing for closer objecttohologram plane distance, and therefore, a larger spatial angle that positively affects the resolution. Although the full wavefront is detected in both layouts, the size of the hologram is constrained by size of both the matrix detector and the electrooptic crystal. The detection sensitivity also imposes a limit on the hologram size, since commerciallyavailable coherent broadband THz transmitters are not powerful enough to be effectively detected in larger crosssections.
Consider the object \({\mathscr{O}}(x,y,{\rm{\nu }})\) placed in the THz wavefront. After diffraction on the object, the incoming THz field \(E(x,y,t)\) is distorted into \({E}_{{\mathscr{O}}}(x,y,t)\). Further propagation to the registration plane at the distance l forms the diffraction pattern \({E}_{ {\mathcal H} }(\tilde{x},\tilde{y},t^{\prime} )\). By applying a Fourier transform, this pulse wavefront can be converted into the frequency domain
\( {\mathcal H} (\tilde{x},\tilde{y},{\rm{\nu }})\) is the Fourier transform of the hologram recorded in the time domain at the distance L from the object plane, and can be considered as a union of monochromatic holograms calculated for every frequency v.
Generally, we can consider the set of independent monochromatic objects \({\mathscr{O}}(x,y,{\rm{\nu }})\) spatially digitized into M × M pixels to lead to a transverse size of D = (MΔx, MΔy), corresponding to pixel size of Δx × Δy. This set of objects produces a set of independent holograms \( {\mathcal H} (\tilde{x},\tilde{y},{\rm{\nu }})\), each being N × N pixels and the overall transverse size of \(\tilde{D}=(N{\rm{\Delta }}\tilde{x},\,N{\rm{\Delta }}\tilde{y})\). The number of both subobjects and subholograms in the sets is equal to the number of frequency components in the spectrum^{42}. Likewise, both object and hologram can be represented as sets of spatialtemporal field distributions, or spatialspectral intensity charts. This concept is depicted in Fig. 2.
The THz hologram is a threedimensional array of data recorded after diffraction on an object. This array carries information about the spatialtemporal, or spatialspectral distribution of the THz field.
To reconstruct an object from a hologram, the recorded complex field is backpropagated into the object plane:
where \({ {\mathcal F} }_{L,\nu }^{1}[{\mathbb{H}}]\to {\mathbb{O}}\) is the inverse wavefront propagation operator, here, \({\mathbb{H}}\) and \({\mathbb{O}}\) are the sets of all holograms and objects, correspondingly. Depending on the distance between the object and the hologram, it can be either a RayleighSommerfeld integral, 2D Fourier or 2D Fresnel transforms, or angular spectrum calculation^{39,59}.
Iterative selfextrapolation algorithm for PTDH
The principle for object selfhealing in the case of broadband THz PTDH is quite similar to the one described for monochromatic holography^{49}. Proposed technique is based on the assumption that the diffraction pattern, recorded in the hologram, is created by waves that exist not only within, but also outside the registration area. Therefore, selfextrapolation of the digital hologram to the area outside of the experimentally recorded diffraction pattern can be performed. The principal scheme for the algorithm in the case of broadband THz PTDH is shown in Fig. 3. In general words, the hologram is backpropagated to reconstruct an object from the data ‘as is’, as described in equation (2). Then, after the application of the mask \(\tilde{{\mathscr{O}}}(x,y,{\rm{\nu }})={ {\mathcal M} }_{x,y}[{\mathscr{O}}^{\prime} (x,y,{\rm{\nu }})]\) that restricts the energy distribution in the object plane and helps to avoid the ghost image appearance, the reconstructed object wavefront is numerically propagated to form a synthesized hologram
\( {\mathcal F} [{\mathbb{O}}]\to {\mathbb{H}}\) is the forward propagation operator, that, similarly to inverse operator \({ {\mathcal F} }^{1}\) is chosen depending on the distance between object and hologram. Thus, we can introduce an iteration operator \( {\mathcal I} [{\mathbb{H}}]\to {\mathbb{H}}\), describing the reconstruction of an object \(\tilde{{\mathscr{O}}}\), and consecutive generation of a synthesized hologram \(\tilde{ {\mathcal H} }^{\prime} \) as follows:
When this operator is applied, one should keep in mind that any region of the object plane that is involved in calculations, but does not contain meaningful information, must be zeroed to avoid ghost images^{60}. While calculation, it is achieved by multiplication of the recovered object by the mask \({ {\mathcal M} }_{(x,y)}[{\mathscr{O}}]\) that removes all unwanted areas by zeroing them.
The iterative algorithm can be described by the following scheme: if it is a zerostep, zeropad the recorded hologram \( {\mathcal H} (\tilde{x},\tilde{y},{\rm{\nu }})\) to achieve a larger area wavefront in the hologram plane \({\tilde{ {\mathcal H} }}_{0}(\tilde{x},\tilde{y},{\rm{\nu }})\), for other steps, replace the central part of the synthesized hologram \({ {\mathcal H} }_{i}^{^{\prime} }(\tilde{x},\tilde{y},{\rm{\nu }})= {\mathcal I} [{\tilde{ {\mathcal H} }}_{i1}(\tilde{x},\tilde{y},{\rm{\nu }})]\), corresponding to originally recorded hologram \(({\rm{rect}}(\frac{\tilde{x}}{N{\rm{\Delta }}\tilde{x}},\frac{\tilde{y}}{N{\rm{\Delta }}\tilde{y}}) > 0)\), with the originally recorded hologram \( {\mathcal H} (\tilde{x},\tilde{y},{\rm{\nu }})\):
After a reasonable number of iterations, the reconstructed object is recovered, along with its spectral features.
Results and Discussion
Since the spectrum of THz pulsed radiation is more that 1 octave wide, depending on the object type, the algorithm can be implemented in a different manner. For spectrally uniform objects, like binary amplitude opaque (e.g. metal) objects, reconstructed data obtained with one of the wavelengths can be used as a synthetic object in the next iteration for forward propagation simulation with a different wavelength. Thus, an enhancement across the whole spectrum can be achieved. On the other hand, for an object with numerous spectral features, each feature can be reconstructed independently, provided that the numerical aperture of the hologram gives the sufficient amount of wavefront data for reconstruction. Indeed, if the full contrast interference fringe is not recorded, the object will not be retrieved^{49}.
Image reconstruction by low frequency components of the THz range
Let us consider the problem of increasing the resolution of a binary amplitude object. In accordance with the Abbe criterion, the resolution of the reconstructed image for each spectral component is limited by the numerical aperture NA of the hologram, hence it depends on the distance between the object and the registration plane, as well as the linear dimensions of the hologram as
where R_{Abbe} is the resolution in m, c is the speed of light, D and l are the transverse size and distance to the hologram. Thus, after a size restricted hologram \({ {\mathcal H} }_{0}(\tilde{x},\tilde{y},{\rm{\nu }})\) is recorded, application of the iterative algorithm and hologram self extrapolation obtains the field \({\tilde{ {\mathcal H} }}_{i}(\tilde{x},\tilde{y},{\rm{\nu }})\), which has a larger transverse size, thus enhancing the reconstructed image resolution.
Figure 4 illustrates the results of the algorithm applied to the binary object of the two apertures separated by ∼15 mm. The object itself, 25 × 25 mm^{2} in size, is shown in the inset of Fig. 4(a). The central part of Fig. 4(a) shows the simulated 25 × 25 mm^{2} hologram at 0.4 THz, recorded at 600 mm from the object. This frequency is selected, as in this frequency range the algorithm application demonstrates the highest effect. Indeed, if we calculate the initial resolution at 0.4 THz, it is equal to \({R}_{0}^{{\rm{A}}bbe}=18\) mm, and the result of the object reconstruction from this hologram is shown in Fig. 4(b). Then, this reconstruction is used to simulate the hologram \({\tilde{ {\mathcal H} }}_{1}(\tilde{x},\tilde{y},{\rm{\nu }})\) shown in Fig. 4(c), with its central part replaced with initially recorded \({ {\mathcal H} }_{0}(\tilde{x},\tilde{y},{\rm{\nu }})\), as it was described in Fig. 3. The wavefront in this hologram is wider than the area of the initially recorded hologram, hence the NA and the reconstruction resolution is higher (\({R}_{1}^{{\rm{A}}bbe}=4.5\) mm). The reconstruction after first iterative procedure shown in Fig. 4(d), already reveals the start of the two dots separation. After the 10th algorithm iteration, the theoretical resolution reaches \({R}_{10}^{{\rm{A}}bbe}=3\) mm, and the two dots in the reconstruction are clearly resolved. Holograms and reconstructed objects at further iteration steps (5th and 10th), are shown in Fig. 4(e,g) and (f,h), respectively.
Spectra of the reconstructed images crosssections (Fig. 4(i–l)) demonstrate the effectiveness of the proposed method. While slightly enhancing the reconstructed images in the higher frequency range, the crucial change is revealed in the spectral area between 0.35–0.45 THz (670–850 μm), where the wavelength is approximately equal to the detector pixel size (780 μm) and the theoretical resolution changes from \({R}_{0}^{{\rm{A}}bbe}\approx 23\) mm to \({R}_{10}^{{\rm{A}}bbe}\approx 3\) mm upon iterations, i.e. from below to above the dots separation, and the dots become resolvable. At higher frequencies, the dots are being resolved even after the direct hologram reconstruction, and at longer wavelengths, the NA of the hologram is still not enough to cover at least one interference fringe and thus provide enough contrast in the object reconstruction. As a result, the apertures, although they are larger than the wavelength, can not be recovered through the proposed algorithm. Such a fast convergence of the algorithm, as compared with results in^{49}, occurs due to longer THz wavelengths involved (hence smaller objects, holograms, and distances between them in wavelength units) and higher impact of every iteration step. Note that reconstruction of the image at the frequency of 0.42 THz require only 10 iteration, while at lower frequencies (e.g. 0.38 THz) more iterations should be performed for significant improvement of reconstructed image spatial resolution.
Application of the technique to spectrally selective object reconstruction
Next, we study a spectrally selective spatially inhomogeneous object (Fig. 5). As an object, we used opaque mask 44.8 × 44.8 mm^{2} in size with ‘P’, ‘T, ‘D’ and ‘H’ letters transparent in different spectral windows.
The spectralselective amplitude transmission of an object is given by the function:
Here R(x, y), G(x, y) and B(x, y) are spatial normalization coefficients of the amplitude for the Gaussian distribution; \({{\rm{\nu }}}_{r},{{\rm{\nu }}}_{g},{{\rm{\nu }}}_{b}\) are fixed central frequencies of the normal distribution 0.4 THz, 0.85 THz and 1.3 THz respectively; \({{\rm{\Delta }}{\rm{\nu }}}_{r}\), \({{\rm{\Delta }}{\rm{\nu }}}_{g}\), \({{\rm{\Delta }}{\rm{\nu }}}_{b}\) are average standard deviations from fixed frequencies \({{\rm{\nu }}}_{r},{{\rm{\nu }}}_{g},{{\rm{\nu }}}_{b}\) respectively. The transmission spectra are shown in Fig. 5(c). The letters P, T and D in Fig. 5(a) are transparent in the lower (red), medium (green) and high (blue) frequency ranges, respectively. Meanwhile, the letter H is transparent in all three frequency ranges. The size of the square object in Fig. 5(a) is 44.8 × 44.8 mm^{2}. The size of the simulated broadband timedomain hologram (marked on the figure by the white dashed square) is 22.4 × 22.4 mm^{2}, and the diffraction pattern was simulated 100 mm away from the object using a flat broadband wavefront. Figure 5(d) and Table 1 present the results of the iterative object reconstruction. Both spectral and spatial features eventually get resolved after ∼10 iterations, and the object gains larger area of fine reconstruction, which initially was restricted by the smaller size of the hologram and very low scattering introduced by the objects. As a result, the mean root square error (MRSE) of reconstruction quality (normalized error between reconstructed image compared to initial object) drops from 0.66 to 0.34 (see Table 1). Due to the fact that MRSE depends mostly on the general shape of the object rather than on fine structures of the object (in this case sharp edges of the letters), significant decrease of MRSE can be observed right after the first iteration, which provides overall shape of the letters. On the other hand, addition of higher spatial frequency components during further iterations, which results in increase of letter edge reconstruction quality, does not introduce significant contribution to decrease of MRSE.
Conclusion
In this paper, we proposed the solution for resolution and field of view enhancement in THz pulse time domain holography (THz PTDH). THz PTDH allows for spectrally resolved images to be obtained without any additional optics in the THz beam, as the signal is coherently detected after diffraction on the object of study. The resolution enhancement method involves an iterative procedure of backwardforward wavefront propagation, with numerical extension of the hologram area. The method is especially important for THz holography, as the detectors are usually restricted in size, and the relatively large wavelength means that a very low number of interference fringes can be recorded. We have shown the method to enhance both the resolution and the field of view for the reconstructed images after performing only 10 iterations. We successfully applied the method for both spectrally uniform and spectrally selective objects, and revealed significant enhancement at all frequencies that produce at least one interference fringe in the diffraction pattern of the recorded hologram. At the frequencies between 0.35–0.45 THz, the Abbe criterion for the object and hologram under study is reduced from the value of 23 mm (above the distance between the apertures used as an object) at the first iteration, to the value of 3 mm (well below the distance between the apertures) at the last iteration. Spectrally selective test object also revealed significant reconstruction quality enhancement resulting in double reduction of reconstruction mean root square error. Thus, the proposed method can be can be successfully applied to all kinds of pulsed THz holographic imaging tasks, a well as to spatiotemporal metrology of broadband complex wavefronts^{59,61}. Field of view broadening, in all these tasks, when THz detection is performed with relatively small crystal, will push the boundaries of the method experimental applicability.
Methods
Numerical simulations were performed with inhouse software, written in National Instruments LabView development framework. The software follows mathematical procedures described in this paper, and our preceeding works^{39,59,61}.
References
Auston, D. H., Cheung, K. P. & Smith, P. R. Picosecond photoconducting Hertzian dipoles. Appl. Phys. Lett. 45, 284–286 (1984).
Jepsen, P., Cooke, D. & Koch, M. Terahertz spectroscopy and imaging  Modern techniques and applications. Laser Photon. Rev 5, 124–166 (2011).
Lepeshov, S., Gorodetsky, A., Krasnok, A., Rafailov, E. & Belov, P. Enhancement of terahertz photoconductive antenna operation by optical nanoantennas. Laser Photon. Rev 11, 1600199 (2017).
Hu, B. B. & Nuss, M. C. Imaging with terahertz waves. Opt. Lett. 20, 1716 (1995).
Chan, W. L., Deibel, J. & Mittleman, D. M. Imaging with terahertz radiation. Reports on Progress in Physics 70, 1325–1379 (2007).
Bennett, D. B. et al. Terahertz sensing in corneal tissues. Journal of Biomedical Optics 16, 057003 (2011).
Crawley, D. et al. Threedimensional terahertz pulse imaging of dental tissue. Journal of Biomedical Optics 8, 303 (2003).
Smirnov, S. V., Grachev, Y. V., Tsypkin, A. N. & Bespalov, V. G. Experimental studies of the possibilities of diagnosing caries in the solid tissues of a tooth by means of terahertz radiation. J. Opt. Technol 81, 464 (2014).
Yu, C., Fan, S., Sun, Y. & PickwellMacpherson, E. The potential of terahertz imaging for cancer diagnosis: A review of investigations to date. QIMS 2, 33–45 (2012).
Ruth, M. W. et al. Terahertz pulse imaging in reflection geometry of human skin cancer and skin tissue. Phys. Med. Biol. 47, 3853 (2002).
Fitzgerald, A. et al. Terahertz Pulsed Imaging of human breast tumors. Radiology 239, 533–540 (2006).
Wallace, V. P. et al. Terahertz pulsed imaging of basal cell carcinoma ex vivo and in vivo. British Journal of Dermatology 151, 424–432 (2004).
Smolyanskaya, O. et al. Terahertz biophotonics as a tool for studies of dielectric and spectral properties of biological tissues and liquids. Prog. Quantum Electron., https://doi.org/10.1016/j.pquantelec.2018.10.001 (2018).
Federici, J. F. et al. THz imaging and sensing for security applications—explosives, weapons and drugs. Semiconductor Science and Technology 20, S266–S280 (2005).
Shen, Y. C. et al. Detection and identification of explosives using terahertz pulsed spectroscopic imaging. Applied Physics Letters 86, 1–3 (2005).
Kemp, M. C. Explosives Detection by Terahertz Spectroscopy—A Bridge Too Far? IEEE T. THz Sci. Techn 1, 282–292 (2011).
Zhang, L., Zhong, H., Deng, C., Zhang, C. & Zhao, Y. Terahertz wave referencefree phase imaging for identification of explosives. Applied Physics Letters 92 (2008).
Gowen, A., O’Sullivan, C. & O’Donnell, C. Terahertz time domain spectroscopy and imaging: Emerging techniques for food process monitoring and quality control. Trends in Food Science & Technology 25, 40–46 (2012).
Ok, G., Park, K., Kim, H. J., Chun, H. S. & Choi, S.W. Highspeed terahertz imaging toward food quality inspection. Applied Optics 53, 1406 (2014).
Massaouti, M., Daskalaki, C., Gorodetsky, A., Koulouklidis, A. D. & Tzortzakis, S. Detection of Harmful Residues in Honey Using Terahertz TimeDomain Spectroscopy. Appl. Spectrosc. 67, 1264–1269 (2013).
Stoik, C. D., Bohn, M. J. & Blackshire, J. L. Nondestructive evaluation of aircraft composites using transmissive terahertz time domain spectroscopy. Optics Express 16, 17039 (2008).
Balbekin, N. S., Novoselov, E. V., Pavlov, P. V., Bespalov, V. G. & Petrov, N. V. Nondestructive monitoring of aircraft composites using terahertz radiation. In Proc. of SPIE, vol. 9448, 94482D (2015).
Ospald, F. et al. Aeronautics composite material inspection with a terahertz timedomain spectroscopy system. Optical Engineering 53, 031208 (2013).
Amenabar, I., Lopez, F. & Mendikute, A. In introductory review to THz nondestructive testing of composite mater. Journal of Infrared, Millimeter, and Terahertz Waves 34, 152–169 (2013).
Yakovlev, E. V., Zaytsev, K. I., Dolganova, I. N. & Yurchenko, S. O. NonDestructive Evaluation of Polymer Composite Materials at the Manufacturing Stage Using Terahertz Pulsed Spectroscopy. IEEE T. THz Sci. Techn. 5, 810–816 (2015).
Zhao, H., Zhao, K. & Bao, R. Fuel property determination of biodieseldiesel blends by terahertz spectrum. J. Infrared Millim. Te 33, 522–528 (2012).
Yin, M., Tang, S. & Tong, M. The application of terahertz spectroscopy to liquid petrochemicals detection: A review. Applied Spectroscopy Reviews 51, 379–396 (2016).
Watanabe, Y. et al. Component spatial pattern analysis of chemicals using terahertz spectroscopic imaging. Applied Physics Letters 83, 800–802 (2003).
Skryl, A. S., Jackson, J. B., Bakunov, M. I., Menu, M. & Mourou, G. A. Terahertz timedomain imaging of hidden defects in wooden artworks: application to a Russian icon painting. Applied Optics 53, 1033 (2014).
Jackson, J. B. et al. Terahertz pulse imaging in archaeology. Frontiers of Optoelectronics 8, 81–92 (2014).
Öhrström, L. et al. Terahertz Imaging Modalities of Ancient Egyptian Mummified Objects and of a Naturally Mummified Rat. Anatomical Record 298, 1135–1143 (2015).
Krügener, K. et al. Terahertz meets sculptural and architectural art: Evaluation and conservation of stone objects with tray technology. Sci. Rep 5, 14842 (2015).
Rong, L. et al. Terahertz inline digital holography of dragonfly hindwing: amplitude and phase reconstruction at enhanced resolution by extrapolation. Opt. Express 22, 17236–17245 (2014).
Zolliker, P. & Hack, E. THz holography in reflection using a high resolution microbolometer array. Optics express 23, 10957–67 (2015).
Rong, L. et al. Terahertz inline digital holography of human hepatocellular carcinoma tissue. Scientific reports 5, 8445 (2015).
Choporova, Y. Y., Knyazev, B. A. & Mitkov, M. S. Classical Holography in the Terahertz Range: Recording and Reconstruction Techniques. IEEE Trans. Terahertz Sci. Technol. 5, 836–844 (2015).
Hack, E. & Zolliker, P. Terahertz holography for imaging amplitude and phase objects. Opt. Express 22, 16079 (2014).
Gorodetsky, A. A. & Bespalov, V. G. THz pulse timedomain holography. Proc. of SPIE 7601, 760107–760107 (2010).
Petrov, N. V., Kulya, M. S., Tsypkin, A. N., Bespalov, V. G. & Gorodetsky, A. Application of terahertz pulse timedomain holography for phase imaging. IEEE T. THz Sci. Techn 6, 464–472 (2016).
Wu, Q., Hewitt, T. & Zhang, X.C. Twodimensional electrooptic imaging of THz beams. Applied Physics Letters 69, 1026–1028 (1996).
Ushakov, A., Chizhov, P., Bukin, V., Savel’ev, A. & Garnov, S. Broadband inline terahertz 2D imaging: comparative study with timeofflight, crosscorrelation, and Fourier transform data processing. J. Opt. Soc. Am. B 35, 1159 (2018).
Bespalov, V. G. & Gorodetsky, A. Modeling of referenceless holographic recording and reconstruction of images by means of pulsed terahertz radiation. Journal of Optical Technology 74, 745 (2007).
Zhang, Y., Zhou, W., Wang, X., Cui, Y. & Sun, W. Terahertz Digital Holography. Strain 44, 380–385 (2008).
Wang, X., Cui, Y., Sun, W. & Zhang, Y. Terahertz digital holography. Proc. of SPIE 8330, 833003–833003 (2012).
Chen, G. & Li, Q. Markov chain Monte Carlo sampling based terahertz holography image denoising. Applied Optics 54, 4345–4351 (2015).
Petrov, N. V., Gorodetsky, A. & Bespalov, V. G. Holography and phase retrieval in terahertz imaging. In Proc. SPIE, 8846, 88460S (2013).
Petrov, N. V., Bespalov, V. G. & Volkov, M. V. Phase retrieval of THz radiation using set of 2D spatial intensity measurements with different wavelengths. In Proc. of SPIE, 8281, 82810J–82810J–7 (2012).
Gabor, D. Microscopy by Reconstructed WaveFronts. Proceedings of the Royal Society A: Mathematical, Physical and Engineering Sciences 197, 454–487 (1949).
Latychevskaia, T. & Fink, H.W. Resolution enhancement in digital holography by selfextrapolation of holograms. Optics Express 21, 7726–33 (2013).
Latychevskaia, T., Chushkin, Y. & Fink, H. W. Resolution enhancement by extrapolation of coherent diffraction images: a quantitative study on the limits and a numerical study of nonbinary and phase objects. Journal of Microscopy 264, 3–13 (2016).
Harris, J. L. Diffraction and resolving power. JOSA 54, 931–936 (1964).
Whittaker, E. T. & Watson, G. N. A course of modern analysis (Cambridge university press, 1996).
Guillemin, E. A. The mathematics of circuit analysis (John Wiley New York, 1949).
Katkovnik, V., Shevkunov, I., Petrov, N. V. & Egiazarian, K. Computational superresolution phase retrieval from multiple phasecoded diffraction patterns: simulation study and experiments. Optica 4, 786 (2017).
Wu, Q. & Zhang, X. Freespace electrooptic sampling of terahertz beams. Applied Physics Letters 67, 3523–3525 (1995).
Grachev, Y. V., Osipova, M. O. & Bespalov, V. G. Comparison of an electrooptical system and photoconducting antenna employed as detectors of pulsed terahertz radiation by means of a new method for measuring spectral width. Quantum Electronics 44, 1170–1172 (2014).
Yasui, T., Saneyoshi, E. & Araki, T. Asynchronous optical sampling terahertz timedomain spectroscopy for ultrahigh spectral resolution and rapid data acquisition. Applied Physics Letters 87, 061101 (2005).
Brown, M. S. et al. Watervapor detection using asynchronous THz sampling. Appl. Spectrosc. 60, 261–265 (2006).
Kulya, M. S., Semenova, V. A., Bespalov, V. G. & Petrov, N. V. On terahertz pulsed broadband gaussbessel beam freespace propagation. Sci. Rep 8, 1390 (2018).
Latychevskaia, T. & Fink, H. W. Solution to the twin image problem in holography. Physical Review Letters 98 (2007).
Kulya, M., Semenova, V., Gorodetsky, A., Bespalov, V. G. & Petrov, N. V. Spatiotemporal and spatiospectral metrology of terahertz broadband uniformly topologically charged vortex beams. Applied Optics 58(5), (2019).
Acknowledgements
Authors thank Mr. Thomas R. Hopper for fruitful discussions. This work was supported by the Ministry of Education and Science of the Russian Federation, Project No. 3.1893.2017/4.6. A.G. acknowledges Magicplot LLC for providing a copy of Magilplot Pro used for preparation of the figures in the manuscript and Russian Foundation for Basic Research (Support under project #180701492A).
Author information
Authors and Affiliations
Contributions
N.V.P. and A.G. conceived the research. M.S.K. developed the software, N.S.B. performed the numerical simulations, A.G., N.S.B., A.V.B. and N.V.P. analyzed the results, A.G. and N.S.B. wrote the manuscript with input from all coauthors. N.V.P. supervised the project.
Corresponding author
Ethics declarations
Competing Interests
The authors declare no competing interests.
Additional information
Publisher’s note: Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The images or other third party material in this article are included in the article’s Creative Commons license, unless indicated otherwise in a credit line to the material. If material is not included in the article’s Creative Commons license and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this license, visit http://creativecommons.org/licenses/by/4.0/.
About this article
Cite this article
Balbekin, N.S., Kulya, M.S., Belashov, A.V. et al. Increasing the resolution of the reconstructed image in terahertz pulse timedomain holography. Sci Rep 9, 180 (2019). https://doi.org/10.1038/s41598018366423
Received:
Accepted:
Published:
DOI: https://doi.org/10.1038/s41598018366423
This article is cited by

Highresolution terahertz digital holography based on frequencydomain diagonal extension imaging
Optical and Quantum Electronics (2023)

Singlescan multiplane phase retrieval with a radiation of terahertz quantum cascade laser
Applied Physics B (2022)

Resolution Enhancement in Terahertz Digital Inline Holography by SparsityBased Extrapolation
Journal of Infrared, Millimeter, and Terahertz Waves (2021)

Fast Terahertz Spectroscopic Holographic Assessment of Optical Properties of Diabetic Blood Plasma
Journal of Infrared, Millimeter, and Terahertz Waves (2020)
Comments
By submitting a comment you agree to abide by our Terms and Community Guidelines. If you find something abusive or that does not comply with our terms or guidelines please flag it as inappropriate.