Abstract
Capacitive, inductive and resistive loads of an iontrap system, which can be modelled as LCR circuits, are important to know for building a high accuracy experiment. Accurate estimation of these loads is necessary for delivering the desired radio frequency (RF) signal to an ion trap via an RF resonator. Of particular relevance to the trapped ion optical atomic clock, determination of these loads lead to accurate evaluation of the BlackBody Radiation (BBR) shift resulting from the inaccurate machining of the iontrap itself. We have identified different sources of these loads and estimated their values using analytical and finite element analysis methods, which are found to be well in agreement with the experimentally measured values. For our trap geometry, we obtained values of the effective inductive, capacitive and resistive loads as: 3.1 μH, 3.71 (1) μH, 3.68 (6) μH; 50.4 pF, 51.4 (7) pF, 40.7 (2) pF; and 1.373 Ω, 1.273 (3) Ω, 1.183 (9) Ω by using analytical, numerical and experimental methods, respectively. The BBR shift induced by the excess capacitive load arising due to machining inaccuracy in the RF carrying parts has been accurately estimated, which results to a fractional frequency shift of 6.6 × 10^{−17} for an RF of 1 kV at 2π × 15 MHz and with ±10 μm machining inaccuracy. This needs to be incorporated into the total systematic uncertainty budget of a frequency standard as it is about one order of magnitude higher than the present precision of the trapped ion optical clocks.
Similar content being viewed by others
Introduction
An iontrap system either for guiding^{1,2} or for trapping^{3,4} of ions requires narrow band radio frequency (RF) at the desired parameters. It is equivalent to an LCRcircuit having different impedance than the 50 Ω output impedance of an RF source. Using a directional device following an RF source is a standard technique to avoid unwanted power reflection due to impedance mismatching. So, the RF is generally delivered via inductive coupling through an intermediate resonator whose resonant condition is tuned to obtain the desired RF parameters. Over the last few decades, increasing application of the iontraps for mass spectrometry, precision measurements, generating ionqubit for quantum computation, development of optical clocks, etc. has been demanding high quality RF oscillator and resonator together with improvement in all the associated technologies to achieve an overall high performance.
Focussing on the objectives of this article, here, we briefly review the relevant work on the RF resonator and oscillator. Jones et al. reported novel design of an RF generator that allows for tuning of the resonant frequency which varies with the external capacitive loads^{5,6}. Recently, Reza et al. reported a tunable Colpitts oscillator that also tunes the RF at a desired frequency^{7}. In many cases, the resonant frequency of an RF generator is adjusted by adding extra capacitor in parallel to the iontrap’s intrinsic capacitance but that degrades the quality factor (Qfactor) and efficiency of the power transferred to the trap electrodes. As reported in ref.^{8}, the operating frequency of their indigenously developed RF amplifier got shifted by about 13.6% than it was designed for, which is due to the lack of prior knowledge on the load of their iontrap. In all these abovementioned cases, prior analysis and accurate estimation of the electrical loads will help in proper designing of the RF oscillator. In some applications, feedback controlled RF oscillators are used to enhance the stability^{9}. There, a PhaseLockLoop locks the resonant frequency, which depends on the extra phase added to the RF due to the propagation delay^{10}. Estimation of the loads that are attached to the RF generator and their electrical equivalent circuitry analysis helps in proper designing of the feedback system. For ion ejections, fast discharge of the RF after switching it off is required, otherwise the ions are deflected due to residual oscillations^{11}. As the characteristic discharge time of a RF source depends on the effective inductive and capacitive loads^{12}, estimation of these is required prior to designing of the RF supply.
So far, efforts have been made for novel designing of the RF generator and oscillator for developing high performance iontraps. In this article, we show a pathway for accurate estimation of the capacitive, inductive and resistive loads resulting from an iontrap and its associated system. Towards designing of a precision ion trap that will be used for building of a Ybion optical clock^{13,14}, we have estimated electrical equivalent loads resulting from different sources and verified them experimentally. This helps us to build a desired helical resonator^{15,16} and to estimate the BBR shift resulting from resistive heating of the iontrap.
Methodology
The equivalent inductances, L_{i}; capacitances, C_{i} and resistances, R_{i} resulting from various components, as indicated by the subscript (i), of an iontrap appear as loads. Due to impedance mismatch, direct connection of the RF source to an iontrap not only leads to back reflection but also injects intrinsic noise of the RF source to the confining potential, which results to instability of the ions trapped in it. To overcome these, a helical coil resonator is generally used for applying the RF to the iontrap^{17,18}. This also acts as a bandpass filter and suppresses the noise. The resonant frequency, f_{0} and Qfactor, Q_{0} of a helical resonator itself can be estimated as,
where L_{R}, C_{R} and R_{R} are the inductance, capacitance and resistance of the resonator, respectively^{18}. In a loaded condition, i.e., when the iontrap is attached to the resonator via a connector, the f_{0} and Q_{0} changes as,
where L_{L}, C_{L} and R_{L} are the inductive, capacitive and resistive loads, respectively. The ratio of the effective RF voltage, V′ at the trap electrode relative to the input voltage, V is given as,
Our iontrap system, which is attached to an amplified RF source, consists of three parts: a helical resonator R, a connector C and the end cap type Paul trap T. Their arrangements are shown in Fig. 1. Here, we have identified as well as estimated possible sources of loads that contribute to L_{L}, C_{L} and R_{L} which are then validated through experiment.
Helical Resonator (R)
The design parameters of our helical resonator are described in the earlier publications^{15,16}. The antenna and the secondary coil have self inductances, L_{R1} and L_{R2}; resistances, R_{R1} and R_{R2} and a parasitic capacitance, C_{R1}. Capacitance C_{R2} results from the shieldaircoil combination. In the present design, the copper enclosure is silver coated to reduce the electromagnetic radiation loss, which introduces a resistance R_{R3}. Electrical circuitry comprising of these loads are shown in Fig. 2(a). For calculation of the self capacitance of a helical coil of height h_{c} and outer diameter d_{c}, the modified Medhurst equation^{19} as given by D. W. Knight^{20} is,
where ε_{0}, ε_{i} and ε_{x} are permittivities of free space, inside and outside mediums of the coil, respectively and Ψ is the pitch angle. The coefficient k_{c} is expressed as,
where the numerical prefactors are the empirical coefficients^{20}. Assuming the shield and the coil as coaxial cylinders, the shieldtocoil (sc) capacitance is calculated using the relation,
where d_{s} is the inner diameter of the shield and ε_{r} is the relative permittivity of any dielectric material in between. For proper positioning of the coil, the space between the shield and coil is partially filled with polyethylene foam of permittivity 2.22. Hence, following the Eqns (6–8), the capacitances of the coil and shieldtocoil are,
here m = 0.46 and n = 0.54 are the filling factors for air and foam, respectively. Inductance of the coil is estimated as,
where τ is the winding pitch of the coil and the numerical term arises from unit conversion of the formula, as given in ref.^{21} to the SI. The DC resistance of a part of length l and crosssectional area A is given by,
where ρ is the resistivity of the material. AC resistances for the RF carrying parts are different and those are important to be calculated. For a cylindrical rod of diameter D, the AC resistance^{22} is,
where the skin depth of the material is \(\delta =503\,\sqrt{\rho /{\omega }_{RF}\,\mu }\) with ω_{RF} and μ being the applied frequency and the relative permeability of the material, respectively. R_{R1} and R_{R2} are calculated using Eqns (12) and (13).
Connector (C)
A connector is used to interface the helical resonator with the iontrap, which is assembly of a Yshaped copper adapter, a grounding rod and an electrical feedthrough (Fig. 1). The Yshaped adapter is used for symmetric distribution of the RF to the iontrap electrode pair. The trap is screwed to two radially symmetric electrical pins of the feedthrough for firm and stable mounting. The capacitance, C_{C1} arises between the adapter and the grounding rod along with the feedthrough pins whereas C_{C2} and \({C^{\prime} }_{C2}\) results from the feedthrough pins and its flange. The adapter, grounding rod, feedthrough pins also carry resistances R_{C1}, R_{C2} and R_{C3}, respectively. The rectangular parts of the connector of width w and thickness t carrying RF has an AC resistance^{23} which is given as,
where x = 2 (1 + t/w) δ/t, current crowding factor K_{c} = 1 + (1 − e^{−0.048p}) [0.06 + 0.22 ln(w/t) + 0.28 (t^{2}/w^{2})] and p = \(\sqrt{A}/1.26\,\delta \). Equivalent electrical circuitry of all the loads described in this section are shown in Fig. 2(b).
Trap (T)
The iontrap electrodes are made of Tantalum, whereas its holding structure is made of Molybdenum and Macor^{24}. The holder mounts the iontrap inside an Ultra High Vacuum (UHV) chamber via an electrical feedthrough. Capacitances in the trap arise out of the inner electrode holder to outer electrode holder with dielectric macor in between. This capacitance, C_{T1} is estimated partbypart as C′_{T1}, C′′_{T1} and C′′′_{T1} as shown in Fig. 1. The capacitances resulting from the mounting bolts and the electrode holders are C_{T2} and \({C^{\prime} }_{T2}\), respectively and from the innertoouter electrodes are C_{T3}, \({C^{\prime} }_{T2}\), C_{T4} and \({C^{\prime} }_{T4}\). Resistances result from the inner electrode holder, R_{T1}; outer electrode holder, R_{T2}; bolts, R_{T3}; inner and outer electrode, R_{T4}, R_{T5} and \({R^{\prime} }_{T4}\) and \({R^{\prime} }_{T5}\), respectively. Trap’s inner and outer electrode holders can be considered as parallel plates spaced apart by macor of thickness d. In that case, the capacitances are calculated as,
Since, the trap electrodes are cylindrical in shape, their inbetween capacitance is calculated from Eqn. (8) by replacing h_{c}, d_{c} and d_{s} with the length, inner and outer diameters of the outer and inner electrodes, respectively. Equation (14) gives R_{T1} whereas Eqn. (13) is utilized for calculating R_{T3} and R_{T4}. The electrical equivalent circuitry of the loads associated to the iontrap is shown in Fig. 2(c).
Numerical and Analytical methods
We perform finite element analysis via COMSOL Multiphysics software (COMSOL) for numerical estimation of various L_{i}, C_{i} and R_{i} values. The geometric designs of the parts are fed into the COMSOL and subdivided into finite size elements as per the user’s choice. The software analyzes them and generates a final solution using variational method. The computational time increases dramatically with finer mesh size, however, after a certain mesh size, the incremental change of the computed value is well within the limit of our desired accuracy. We use the AC/DC electromagnetics module of COMSOL as the subnodes: electrostatics interface, electric current and magnetostatics allow direct computation of capacitances, resistances and inductances, respectively.
For analytical estimation, we use the formulae as described in the Sec. III AC, which rely on certain assumptions. As for example, C_{i} values are estimated assuming parallel plate capacitors. Thus, we do not expect them to be as accurate as for the numerical techniques but this shows inaccuracy of the estimation in case it is done only analytically.
Experiment
Experimental verification of the estimated values are obtained through an indirect measurement as well as via a direct measurement of the loads using an LCR meter. For indirect measurement, the resonant condition is obtained by scanning the RF frequency across f_{0} and f′ as given in Eqns (1 and 3). A small fraction of the RF (≤3%) gets reflected from the resonator. Using a directional coupler, this reflected signal is outcoupled to a spectrum analyzer for further analysis. The resonance condition is ensured by minimum reflection of the RF while tuning its frequency. The outcoupled RF is measured in a spectrum analyzer at different values of fixed capacitive loads and also in the unloaded condition of the resonator.
Results and Discussions
Simple circuitry analysis using Kirchoff’s current law leads to estimation of the effective loads as following, for the Resonator R:
for the Connector C:
for the Trap T:
where, C_{T1} = (C_{T1}′ + C_{T1}′′ + C_{T1}′′′) as indicated in Fig. 1, A = (R_{T2} + 2R_{T5}) and B = (R_{T1} + 2R_{T3} + 2R_{T4}), for total effective loads E:
The wire that is used for grounding of the trap inside of the UHV chamber results in a resistive load. The extra compensation electrodes, that are generally used for precise positioning of the ion at the trap centre, also results to finite capacitance. These values have been estimated to be orders of magnitude lower than the rest, which are therefore neglected.
The values of the loads as given in Eqns (16–18), estimated via analytical and numerical techniques are given in Table 1. Source of errors in the numerically calculated values are two fold: accuracy a that is obtained from COMSOL for a certain mesh size and their inaccuracy b for ±10 μm machining tolerance. Hence, the total error is estimated to be ±\(\sqrt{{a}^{2}+{b}^{2}}\), which is dominated by b.
In the experiment, we measure resonant frequencies and Qfactors when: (i) the resonator is attached to capacitors and (ii) the resonator is attached to the connector and capacitors. For this purpose, we use capacitors with different values that are calibrated at 0.3% uncertainty. As shown in Fig. 3, the characteristic change of the resonant frequency and Qfactor due to additional capacitive load are fitted to \(f({C}_{L})={f}_{0}\,\sqrt{{C}_{R}/({C}_{L}+{C}_{R})}\) and \(Q({C}_{L})={Q}_{0}\,\sqrt{{C}_{R}/({C}_{L}+{C}_{R})}\), respectively, where f_{0}, Q_{0} and C_{R} are used as the fit parameters. Capacitive load resulting from a component attached to the resonator is obtained by measuring shift of the resonant frequency or Qfactor from the unloaded f_{0}, Q_{0} values and obtaining change of capacitance corresponding to that shift from their respective characteristic curves, as given in Fig. 3. With the known values of f_{0} and f′ for any attached capacitance C_{i}, its unknown value of L_{i} can be estimated by using Eqns (1) and (3). Further, as the Q_{0} and Q′ values are also known from experiments, using the estimated L_{i} values, the unknown R_{i} values can be obtained using the Eqns (2) and (4). The experimental values are obtained by taking mean of the values taken from different techniques and their variance is quoted as the uncertainty. We measure f_{0} = 27.04 (7) MHz and Q_{0} = 600 (8), estimate them as 25.85 (5) MHz and 508 (1) using the numerically estimated loads in Eqns (1–2) and also extract them as 27.5 (1.0) MHz and 590 (7) from the fits. Mutual agreement of these values also validates our calculation in an independent way. The analytical, numerical and experimental values of different loads are compared in Table 2. Using the numerical and experimental values of the loads in Eqn. (5), we estimate the RF voltage transfer factor, k_{RF} ≃ 0.45 and 0.46, respectively. Therefore, voltage at the trap is about half of that is at the output of RF resonator.
The described electrical load analysis also leads to estimation of the excess BBR shift that results from heating of the iontrap. The resistive heating of the iontrap electrodes due to absorption of the RF power propagating through it has already been reported elsewhere^{25,26,27}. Here, we are estimating the BBR shift associated to heating of the trap due to current flowing through it, which originates due to finite capacitance introduced by inaccurate machining of the trap parts. The RF appears at a different phase with a relative phase difference ϕ on the trap electrodes, that are aligned axially facing each other, due to a path difference x resulting from nonnegligible machining tolerances. Due to this, at any instant of time, the RF, V_{o}sinω_{RF}t, appears at different voltage level, which can be modelled as a parallel plate capacitor with an unwanted excess capacitance C_{ex} kept at a potential V_{o}ϕ^{28}. The phase difference, resulting from the inaccurate machining, is
where p is the integer number of RF wavelength λ_{RF} that gets accommodated within x. Figure 4(a) shows numerically estimated values of C_{ex} for our iontrap and calculated ϕ following the Eqn. (20) as a function of the machining inaccuracy. Even though there is no direct relation, the capacity factor in our iontrap geometry is α_{c} = C_{ex}/ϕ = 1.9, where C_{ex} and ϕ are in fF and millidegree, respectively. Although, in an ideal condition (ϕ = 0), the RF electrodes stay in an open circuit condition, a finite current I_{ex} = V_{o}ϕ/R_{RF} flows through in a real trap that has certain capacity factor. A simple minded physical picture to understand this is by considering a voltage source of V_{o}ϕ V that is present in between the electrodes. In that case, the resulting I_{ex} flows through the resistance R_{RF} that is present in the RF path of the iontrap system. In a thermal equilibrium, the resistive heating due to I_{ex} will elevate temperature of the trap, which is nonnegligible in many precision experiments particularly in case of an optical clock as that significantly contributes to the BBR shift^{29}. The amount of temperature rise during the time of current flow (i.e. cycle time of the experiment and we have considered that to be 1 s just for simplicity) can be estimated following the conservation of energy as
where m_{j} and s_{j} are masses and specific heats of the elements j, respectively. Corresponding BBR shift due to the electromagnetic radiation at an elevated temperature T^{30} can be estimated as,
where h is the Planck’s constant, δα_{o} is differential scalar polarizability of the transition and η(T) is the dynamic correction since atomic transitions other than the E1transitions are neglected in δα_{o}. Here, we have neglected this factor since η < 0.01 at room temperature. As an example, here we estimate Δν^{BBR} at T = (296 + ΔT) that is from the machining inaccuracy for the Ytterbiumion (^{171}Yb^{+}) octupole (E3) clock transition at frequency ν_{E3} = 642121496772645.150 Hz^{31} and having δα_{o} = 0.859 × 10^{−40} Jm^{2}V^{−2} ^{32}. We use our estimated values of resistances, as given in Table 1, to get \({R}_{RF}=({R}_{T1}+{R}_{T3}+{R^{\prime} }_{T3}+{R}_{T4}+{R^{\prime} }_{T4}+{R}_{C1}+{R}_{C3}+{R^{\prime} }_{C3})\). Figure 4(b) shows expected fractional BBRshift Δν^{BBR}/ν_{E3} resulting from the machining inaccuracy of the electrodes at different amplitudes of RF of ω_{RF} = 2π × 15 MHz. Accurate estimation of this systematic shift carries merit for developing optical clocks with increasing accuracies as it is about two orders of magnitude higher than the present best trapped ion optical frequency standard^{31}.
Conclusion
We made detailed analysis of the resistive, inductive and capacitive loads that results from each part of the iontrap system. We also performed an experiment to measure the effective loads to validate our theoretical estimations. Considering the estimated load values, the design parameters of the resonator can be chosen such that it operates as desired. This analysis shows a pathway for predicting output values, e.g., resonant frequency, Qfactor of an RF resonator both in loaded and unloaded conditions prior to its construction. The RF phase difference at the tip of the electrodes resulting from the unignorable machining inaccuracy, has been estimated accurately. Its relation with one of the dominant systematic shifts in the atomic clock experiments: BBR shift is obtained, which is important to be considered in the overall systematic budget. As for example, at an RF amplitude of 1 kV at 2 π × 15 MHz and with ±10 μm machining inaccuracy will result to ±0.42 K heating of the trap. A temperature increase (decrease) will result to 43 (+43) mHz shift for the Yb ion E3clock transition that corresponds to fractional frequency uncertainty 6.6 × 10^{−17} due to the BBR effect. This exercise for precise estimation of the machining inaccuracy assisted BBR shift helps in building highly accurate optical frequency standards that are approaching to 10^{−18} level even without any cryogenic environment.
References
Teloy, E. & Gerlich, D. Integral cross sections for ion–molecule reactions. I. The guided beam technique. Chem. Phys. 4, 417–427 (1974).
Gerlich, D. Inhomogeneous RF fields. A versatile tool for the study of processes with slow ions. Adv. Chem. Phys. 1, 82 (1992).
Paul, W. Electromagnetic traps for charged and neutral particles. Rev. Mod. Phys. 62, 531–540 (1990).
Penning, F. M. Die glimmentladung bei niedrigem druck zwischen koaxialen zylindern in einem axialen magnetfeld. Physica 3, 873–894 (1936).
Jones, R. M., Gerlich, D. & Anderson, S. L. Simple radiofrequency power source for ion guides and ion traps. Rev. Sci. Instrum. 68, 3357–3362 (1997).
Jones, R. M. & Anderson, S. L. Simplified radiofrequency generator for driving ion guides, traps, and other capacitive loads. Rev. Sci. Instrum. 71, 4335 (2000).
Reza, A. et al. An in situ trap capacitance measurement and iontrapping detection scheme for a Penning ion trap facility. Rev. Sci. Instrum. 88, 034705 (2017).
Noriega, J. R., GarcíaDelgado, L. A., GómezFuentes, R. & GarcíaJuárez, A. Low power RF amplifier circuit for ion trap applications. Rev. Sci. Instrum. 87, 094704 (2016).
Johnson, K. G. Active stabilization of ion trap radiofrequency potentials. Rev. Sci. Instrum. 87, 053110 (2016).
Jau, Y., Benito, F. M., Partner, H. & Schwindt, P. D. D. Low power highperformance radio frequency oscillator for driving ion traps. Rev. Sci. Instrum. 82, 0231181–0231185 (2011).
Marmillod, P., Antonioni, S. & Lorenz, U. J. A radio frequency/high voltage pulse generator for the operation of a planar multipole ion trap/timeofflight mass spectrometer. Rev. Sci. Instrum. 84, 044707 (2013).
Cermak, I. Compact radiofrequency power supply for ion and particle guides and traps. Rev. Sci. Instrum. 76, 063302 (2005).
De, S., Batra, N., Chakraborty, S., Panja, S. & Gupta, A. Sen. Design of an ion trap for trapping single ^{171}Yb^{+}. Curr. Sci. 106, 1348–1352 (2014).
Batra, N., Sahoo, B. K. & De, S. An optimized ion trap geometry to measure quadrupole shifts of ^{171}Yb^{+} clocks. Chin. Phys. B 25, 11 (2016).
Batra, N. Design and Construction of a Helical Resonator for Delivering Radio Frequency to an Ion Trap. MAPAN  Journal of Metrology Society of India, https://doi.org/10.1007/s1264701702095123 (2017).
Panja, S., De, S., Yadav, S. & Sen Gupta, A. Measuring capacitance and inductance of a helical resonator and improving its quality factor by mutual inductance alteration. Rev. Sci. Instrum. 86, 056104 (2015).
Raizen, M. G., Gilligan, J. M., Bergquist, J. C., Itano, W. M. & Wineland, D. J. Ionic crystals in a linear Paul trap. Phys. Rev. A 45, 6493 (1992).
Siverns, J. D., Simkins, L. R., Weidt, S. & Hensinger, W. K. On the application of radio frequency voltages to ion traps via helical resonators. J. Appl. Phys. B 107, 921–934 (2012).
Medhurst, R. G. H. F. R. and SelfCapacitance of Sigle Layer Solenoids.Wireless. Engineer 24, 80 (1947).
Knight, D. The selfresonance and selfcapacitance of solenoid coils. applicable theory, models and calculation methods. G3YNH info, https://doi.org/10.13140/RG.2.1.1472.0887 (2016).
Roper, D. C. & Briggs, L. J. Radio Instruments and Measurements. Bureau of Standards circular C74; U.S. Government Printing Office; Washington; D.C., 252 (1937).
Payne, A. N. The HF Resistance of Single Layer Coils. G3RBJ, http://g3rbj.co.uk/ (2013).
Payne, A. N. The AC Resistance of Rectangular Conductors. G3RBJ, http://g3rbj.co.uk/ (2017).
Rastogi, A. et al. Design of the Ion Trap and Vacuum System for 171Ybion Optical Frequency Standard. MAPAN  Journal of Metrology Society of India 30, 169–174 (2015).
Doležal, M. et al. Analysis of thermal radiation in ion traps for optical frequency standards. Metrologia 52, 842 (2015).
Keller, J., Burgermeister, T., Kalincev, D., Kiethe, J. & Mehlstäubler, T. E. Evaluation of trapinduced systematic frequency shifts for a multiion optical clock at the 10^{−19} level. J. Phys. Conf. Ser. 723, 012027 (2016).
Zhang, P. et al. Evaluation of blackbody radiation shift with temperatureassociated fractional uncertainty at 10^{−18} level for _{40}Ca^{+} ion optical clock. J. Appl. Phys. B 50, 015002 (2016).
Berkeland, D. J., Miller, J. D., Bergquist, J. C., Itano, W. M. & Wineland, D. J. Minimization of ion micromotion in a Paul trap. J. Appl. Phys. 83, 502 (1998).
Roy, A., De, S., Bindiya, A. & Sahoo, B. K. Accurate determination of blackbody radiation shift, magic and tuneout wavelengths for the ^{6}S_{1/2} → ^{5}D_{3/2} clock transition in Yb^{+}. J. Appl. Phys. B 50, 205201 (2017).
Farley, J. W. & Wing, W. H. Accurate calculation of dynamic Stark shifts and depopulation rates of Rydberg energy levelsinduced by blackbody radiation. Hydrogen, helium and alkalimetal atoms. Phys. Rev. A 23, 2397 (1981).
Huntemann, N., Sanner, C., Lipphardt, B., Tamm, C. & Peik, E. SingleIon Atomic Clock with 3 × 10^{−18} Systematic Uncertainty. Phys. Rev. Lett. 116, 063001–063006 (2016).
Baynham, C. F. A. et al. Measurement of differential polarizabilities at a midinfrared wavelength in ^{171}Yb^{+} Preprint at arXiv:1801.10134v2 (2018).
Acknowledgements
Authors thank Tanja Mehlstäubler for fruitful discussions. We acknowledge CSIRNational Physical Laboratory, Department of Science and Technology (DST) grant no. SB/S2/LOP/033/2013 and Board of Research in Nuclear Sciences grant no. 34/14/19/2014BRNS/0309 for supporting this work. LS and AR thank DST and CSIR for providing Inspire fellowship and Senior research fellowship, respectively.
Author information
Authors and Affiliations
Contributions
S.D. was conceptualizing the entire work reported here. L.S. and S.D. was performing the theoretical analysis. L.S. has performed the numerical simulations using COMSOL Multiphysics software. L.S., A.R., S.P. and S.D. designed and performed the experimental work reported here. L.S., A.R., S.P., V.N.O. and S.D. discussed the result altogether, interpreted the physics behind it and contributed for writing the paper. S.D. supervised the reported work.
Corresponding author
Ethics declarations
Competing Interests
The authors declare no competing interests.
Additional information
Publisher’s note: Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The images or other third party material in this article are included in the article’s Creative Commons license, unless indicated otherwise in a credit line to the material. If material is not included in the article’s Creative Commons license and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this license, visit http://creativecommons.org/licenses/by/4.0/.
About this article
Cite this article
Sharma, L., Roy, A., Panja, S. et al. Estimation of the iontrap assisted electrical loads and resulting BBR shift. Sci Rep 8, 16884 (2018). https://doi.org/10.1038/s41598018352345
Received:
Accepted:
Published:
DOI: https://doi.org/10.1038/s41598018352345
Keywords
This article is cited by

Reflected SignalBased Feedback Control of a Helical Resonator for Continuous and Efficient Delivery of High Voltage RF to an Ion Trap
MAPAN (2023)

An easy to construct submicron resolution imaging system
Scientific Reports (2020)

Optical Atomic Clocks for Redefining SI Units of Time and Frequency
MAPAN (2020)