The large number of drugs administered simultaneously to neonates and children in hospital results in the formation of particles that are potentially infused. We have investigated the ability of IV in-line filters to eliminate particulate matter from multidrug infusion lines and so prevent contamination. The impact on particle occurrence of the internal volume of the IV line below the in-line filter was then evaluated. The multidrug therapy given to children was reproduced with and without in-line filtration. Three combinations with a filter were tested to vary the internal volume (V) between the filter and the catheter egress. The catheter was then connected to a dynamic particle count to evaluate the particulate matter potentially administered to children during infusion. The introduction of in-line filters led to a significant reduction in overall particulate matter, from 416,974 [208,479–880,229] to 7,551 [1,985–11,287] particles (p < 0.001). Larger particles of ≥10 and 25 µm were also significantly reduced. Adding an extension set to the egress of the in-line filter (V = 1.7 mL) caused a significant increase in particulate contamination for both. This study showed that in-line filtration is an effective tool in preventing particle administration to patients. Their position in the infusion in-line is therefore important because of its impact on internal volume and drug particle formation.
Intravenous (IV) therapy is commonly used in neonatal and pediatric patient populations. They receive many drugs simultaneously through a limited number of venous accesses, which contribute to the risk of drug-drug interactions leading to precipitates. Moreover, contamination of IV fluids can be associated with several contaminants, such as microorganisms, particles or air bubbles. In intensive care units (ICUs), patients can receive up to one million particles daily1. We recently conducted a study to assess the effect of infusion sets on overall particulate contamination exposure during multidrug IV therapy in a pediatric hematology ward2. We showed that, under specific conditions, the use of a multi-lumen infusion set helped to reduce this contamination. The study also indicated however that patients can receive up to 900,000 particles per day during hospitalisation despite these infusion systems, which caused us to turn our attention to in-line filtration. Neonatal and pediatric populations receiving a great number of particles, especially large ones, are particularly affected as shown in a previous study3. These may cause serious damage to the body, especially large particles of about 10 µm in diameter4. Images of particles and conglomerates of considerable size (40 × 20 µm) on the surface of various in-line filters were displayed5. Clinical complications may include injection site reactions, i.e. phlebitis6, granuloma7, arterial embolism, modulation of immune response, microcirculation deterioration8, and even death7,9.
Preventing the infusion of particles to the body is therefore a priority, especially for critically ill patients (i.e. neonate and pediatric populations). A few studies have demonstrated that, under specific conditions, infusion device characteristics could have an impact on preventing physical drug incompatibilities10,11. Nonetheless, even after infusion through multi-lumen catheters, intravascular precipitation has been reported12, indicating that the risk of drug precipitation cannot be completely avoided.
IV in-line filters seem to offer the advantage of preventing particulates, micro-organisms and air from being infused. Despite insufficient evidence to recommend their use to prevent morbidity or mortality in neonates13, infusion-related phlebitis14 or the incidence of sepsis15, several studies have shown they can be useful for critically ill patients. A single-centre prospective and randomised controlled trial (RCT), including 88 newborn infants, found a significant reduction in major complications, such as thrombi and clinical sepsis, compared to the control group16. More recently, a pediatric RCT showed that in-line filters resulted in a significant decrease in overall complication rates and the incidence of systemic inflammatory response syndrome (SIRS)17. The authors also showed that the duration of mechanical ventilation and hospitalisation were reduced in the filter group. In the same RCT, in-line filters could significantly reduce respiratory, renal and hematological complications18. Filters had no effect on the occurrence rates of cardiovascular, hepatic and neurological dysfunctions from one group to the other. The advantage of in-line filtration was analysed in a subgroup of cardiac patients, showing significant decreases in the occurrence of SIRS, renal and hematological complications compared to a control group.
In some studies no benefit was to be found from using in-line filters in infusion sets. In a recent RCT including critically ill adult patients, the authors did not find any significant differences in the number of ICU days with SIRS19. They even showed a higher incidence of SIRS in the in-line filter group.
Brotschi et al. demonstrated, in an in vitro study, that in-line filters reduce flow irregularities during IV infusion20. The use of filters has become the subject of recommendations, especially for parenteral nutrition (PN). Two types of filters are recommended: 0.2-µm filters for clear PN (i.e. binary PN), and 1.2-µm filters for lipid-containing PN21,22,23,24,25.
Few studies have assessed particulate matter in IV drugs or fluids after filtration by calculating the number of particles contained in different lyophilised preparations (for example ceftriaxone) with and without filtration26. As expected, the use of filters resulted in a reduction of about 96% in particulate matter. Other studies have indirectly evaluated the particulate matter in IV therapies by analysing filtration membranes5.
The purpose of this study was to assess the ability of IV in-line filters to eliminate particulate contamination in multidrug infusions, including highly incompatible drugs, and consequently evaluate the impact of the internal volume of the IV line below the in-line filter on the occurrence of particles.
As expected, a visible and intense precipitate was observed in the IV infusion line during the infusion of the pediatric multidrug protocol, due to the presence of drug-drug incompatibilities (Fig. 1 and Video 1). This precipitate occurred upstream of the infusion line and gradually disappeared along the infusion line. Despite the absence of visible precipitate at the distal end of the IV infusion line, sub-visible particles could be detected by the dynamic camera (Video 2).
An IV in-line filter directly connected to the central venous catheter (CVC) significantly decreased overall particulate contamination during the 24-hour infusion period compared to infusion without filter (Table 1 and Fig. 2). Figure 2B,C also show that filtration significantly reduced the number of particles, sizes ≥10 µm and 25 µm, respectively.
For both total particles and the larger ones, particulate contamination was not significantly different between the egress of the filter (combination 4 “with filter only”; V = 0 mL) and the egress of the catheter directly connected to the filter (combination 2 “with filter and CVC”; V = 0.8 mL) (Table 2 and Fig. 3).
Adding an extension set to the egress of the in-line filter (combination 3 “with filter and extension line”; V = 1.7 mL) caused a significant increase in particulate contamination for both particles as a whole and the larger ones (Table 2 and Fig. 3).
Figure 4 shows that the administration of discontinuous drugs (i.e. acetaminophen, omeprazole and acyclovir) leads to disturbances in particulate contamination. Each intermittent IV infusion resulted in the infusion of particle bolus.
Theoretically, no particles should be detected in the infusion line following in-line filtration. Despite filtration however, drugs are still in contact inside the CVC, which could account for the particles detected. The standard infusion line including a filter and a CVC directly connected to it was seen to be associated with more than 7,000 particles infused per day. Decreasing the internal volume of the IV line below the in-line filter by removing the CVC resulted in a substantial decrease in the total number of particles whereas the addition of a 3-way stopcock with 25-cm infusion line (i.e. 1.70 mL) significantly increased overall particulate matter. These results suggest that the dead-space upstream of the IV in-line filter (i.e. corresponding to CVC volume), contributed to facilitating drug contact inside the CVC, resulting in the presence of sub-visible particles especially when infusing incompatible drugs. Likewise, our results showed differences for larger particles, ≥10 µm and 25 µm, from one combination to another, with or without CVC, with a significant decrease when a filter was added. The increase in internal volume when using a 3-way stopcock with its infusion line also significantly increased the number of larger particles. It has been hypothesised that the formation of larger particles requires prolonged contact time between drugs in the internal volume of the IV line below the in-line filter. Our study highlights the fact that in-line filters should be added, as far as possible, at the egress of each infusion set.
Our results are in accordance with a previous study that showed that, despite the gradual blackening of in-line filters during PN bag infusion, the filter membrane continued to filter solutions efficiently for up to 72 hours with an acceptable rate of released particles27. Few studies have assessed particulate matter in IV drugs or fluids after filtration. Kuramoto et al., mentioned above, evaluated the number of particles contained in different lyophilised preparations with and without filtration and showed, as expected, that filters produced a reduction of about 96% in particulate matter26. Other studies have indirectly evaluated the particulate matter in IV therapies by analysing filtration membranes5. This study however is the first to demonstrate that zero threshold of particulate contamination does not exist in real and dynamic infusion conditions. The use of in-line filters helps to significantly reduce overall particulate contamination during drug infusion but whatever the infusion conditions, we have to consider that patients will receive particles even with the use of in-line filters. The objective now is to reduce this particulate contamination as far as possible, especially by limiting delivery interaction between co-infused medications (i.e. by adapting infusion flow rate, concentration, separation of incompatible drugs, etc.) so as to minimise drug contact inside the CVC. In this context, the use of specific multi-lumen infusion devices associated with filters may be the solution2,10,11.
On the basis of our study findings, particles <10 µm represent more than 99% of the particles counted. Although they have a smaller diameter than pulmonary capillaries (i.e. 2 µm to 15 µm), these sub-visible particles are not devoid of toxicity for the body. About one third of the injected microparticle dose may be localised in the lungs. Similarly, Ilium et al. showed that small particles can also diffuse into various organs such as the liver or spleen, leading to deleterious effects on their functions4. Larger particles >10 µm were also counted during IV filtered infusions. In studies cited above, it has been shown that these particles were trapped in the lungs, whatever their nature, leading to obstruction of the pulmonary vessels. Recently, a German research group assessed the clinical effects of particulate contamination during IV drug infusion into the human body17,18,28. Delivery of particles was responsible for an increase in the incidence of SIRS, deleterious effects on microcirculation and organ failures.
However, there are limitations to this study. The particulate counter analysis has a detection limit of 1 µm, which probably underestimates the number of particles really administered to patients. It is quite conceivable that a large proportion of particles of between 0.2 µm and 1 µm have been disregarded in overall particulate matter. Furthermore, experiments were performed in static conditions, without any disturbance along the infusion line. As demonstrated in our previous study2, this condition favours a basal level of particulate contamination, whereas it is known that any disturbance to balance (especially changes in drug or fluid flow rates) may affect particle exposure. This study shows that, despite in-line filtration, particles may be administered to patients. No chemical analysis was performed to identify the nature of the precipitate which is probably the result of incompatibility between vancomycin and piperacillin-tazobactam, as already observed in our experience and mentioned in a previous publication2. Nevertheless, it is necessary to take into account the presence of other drugs which are potential vectors of incompatibility (i.e. acyclovir or omeprazole). It would be interesting to specify the nature of particle exposure so as to analyse the effects of drug-drug interaction or identify specific particles of glass or plastic. Chemical analysis could also determine any interactions between drugs and filter membrane, in relation to drug retention. A limited number of cases concord with this29, whereas most studies have revealed no variations in drug concentration when using in-line filters30,31,32,33. Room temperature should also be taken into account because of its impact on particle formation.
In conclusion, this study is the first step towards evaluating particulate matter in pediatric multidrug IV therapy with an in-line filter, especially when infusing two incompatible drugs that lead to precipitate. It validates the concept that in-line filtration is an effective tool in limiting particle administration to patients, but it is not sufficient to eliminate particle infusion. Particulate contamination is never non-existent notably when using an infusion line where the internal volume of the IV line below the in-line filter is high. This study demonstrates that the in-line filter should be positioned as close as possible to the patient to be most effective by minimising internal volume. It would therefore be appropriate to set in motion the means required to establish a clinical protocol.
Pediatric Multidrug Infusion Protocol
This study exactly reproduced the conditions of drug use during therapy administered to children receiving hematopoietic stem cell transplantation for leukemia, in our pediatric ward at the Lille University Hospital (France).
All components were standard clinical materials and devices. Solutions and medications were prepared according to manufacturers’ instructions (Table 3). In our laboratory the infusion set was replicated and consisted of a standard single-lumen IV infusion set, with a 4-port manifold and 150-cm extension line (ref. RPB4320, 14A20-T, Cair LGL, Lissieu, France). The carrier fluid consisted of a solution of 5% glucose with electrolytes (3 g/L hypertonic saline solution, 2 g/L potassium chloride and 10% 3 g/L magnesium chloride). All medications and fluids were infused via an in-line 0.22 µm pore size filter (ref. AEF1NTE, 14–658; Pall Medical, Fribourg, Switzerland). Piperacillin-tazobactam, acetaminophen, acyclovir and the carrier fluid were infused with a volumetric pump (MVP module Orchestra, Fresenius Vial, Brezins, France). Vancomycin, cyclosporine-A and omeprazole were infused through syringe pumps (DPS module Orchestra, Fresenius Vial, Brezins, France). All extension lines were flushed with the carrier fluid before use, as is routinely performed by nurses.
Four measuring locations and combinations were compared (Fig. 5):
At the egress of the catheter of a standard infusion set and catheter as described in a previous study (2), with a CVC and without any IV filter (i.e. combination 1 “without filter and with CVC”).
At the egress of the catheter of an infusion line with a Broviac 6.6-fr single-lumen CVC, length = 57 cm; internal diameter = 1.0 mm (ref. 0600540, HUXE0422, Bard, Salt Lake City, Utah), added directly after the in-line filter (internal volume of the IV line below the in-line filter V = 0.80 mL; i.e. combination 2 “with filter and CVC”).
At the egress of an infusion line with a 3-way stopcock and a 25-cm extension set connected directly after the in-line filter (V = 1.70 mL; i.e. combination 3 “with filter and extension set”).
At the egress of the in-line filter. In this case, the IV in-line filter was directly connected to the particle counter by removing the CVC (V = 0 mL; i.e. combination 4 “with filter only”).
Analysis of particulate matter
All drug combinations were tested with extemporaneous mixtures. A visual inspection of the extension line against a black background and an obscured-light sub-visible particle count test were made to highlight the model of drug incompatibility.
For all experiments with and without in-line filter, the distal end of each IV administration set was directly connected to the Qicpic analyser through Luer-lock connections (Sympatec Inc., Clausthal-Zellerfeld, Germany). The internal volume of the Qicpic instrument was about 1 mL. The particulate counter dynamically detected all particles infused during therapy over a 24-hour period.
Experimental infusion was repeated ten times for each combination, over 24 hours. Visual inspection of the IV infusion line was performed to detect drug-drug incompatibility. For all combinations, particle size analyses were based on overall particulate contamination plus larger particles (i.e. 10 µm and 25 µm), as defined by the European Pharmacopoeia34. All experiments were made at 18 ± 2 °C ambient temperature.
Comparing the number of particles infused over a 24-hour period with measuring combinations 1 and 2 indicates the ability of in-line IV filters to eliminate particulate contamination during multidrug infusion. Comparing the data obtained with measuring combinations 2, 3 and 4 assesses the impact on particle occurrence of the internal volume of the IV line below the in-line filter.
The normality of all data collection was assessed by the D’Agostino & Pearson test (p value >0.05). The Student t test was used to compare the number of particles obtained with combinations 1 and 2. Values obtained with combinations 2, 3 and 4 were compared using an analysis of variance (ANOVA). When this revealed a significant p value (p < 0.05), an analysis using Tukey’s test was performed to detect significant differences between couples of infusion sets. All statistical tests were performed using GraphPad Prism version 6.00 for MacOS X (GraphPad Software, La Jolla California USA, www.graphpad.com). Data is presented as median values and range (min – max), if not otherwise specified. All histograms are presented as mean values and standard error of the mean. For all analyses, statistical significance was considered as a p-value <0.05.
Oie, S. & Kamiya, A. Particulate and microbial contamination in in-use admixed parenteral nutrition solutions. Biol Pharm Bull. 28, 2268–70 (2005).
Perez, M. et al. In vitro analysis of overall particulate contamination exposure during multidrug IV therapy: impact of infusion sets. Pediatr Blood Cancer. 62, 1042–7 (2015).
Perez, M. et al. Dynamic Image Analysis To Evaluate Subvisible Particles During Continuous Drug Infusion In a Neonatal Intensive Care Unit. Sci Rep. 24, 9404 (2017).
Ilium, L. et al. Blood clearance and organ deposition of intravenously administered colloidal particles. The effects of particle size, nature and shape. Int J Pharm. 12, 135–46 (1982).
Jack, T. et al. Analysis of particulate contaminations of infusion solutions in a pediatric intensive care unit. Intensive Care Med. 36, 707–11 (2010).
Falchuk, K. H., Peterson, L. & McNeil, B. J. Microparticulate-induced phlebitis. Its prevention by in-line filtration. N Engl J Med. 312, 78–82 (1985).
McNearney, T., Bajaj, C., Boyars, M., Cottingham, J. & Haque, A. Total parenteral nutrition associated crystalline precipitates resulting in pulmonary artery occlusions and alveolar granulomas. Dig Dis Sci. 48, 1352–4 (2003).
Lehr, H.-A., Brunner, J., Rangoonwala, R. & Kirkpatrick, C. J. Particulate matter contamination of intravenous antibiotics aggravates loss of functional capillary density in postischemic striated muscle. Am J Respir Crit Care Med. 165, 514–20 (2002).
Reedy, J. S., Kuhlman, J. E. & Voytovich, M. Microvascular pulmonary emboli secondary to precipitated crystals in a patient receiving total parenteral nutrition: a case report and description of the high-resolution CT findings. Chest. 115, 892–5 (1999).
Foinard, A., Décaudin, B., Barthélémy, C., Debaene, B. & Odou, P. The impact of multilumen infusion devices on the occurrence of known physical drug incompatibility: a controlled in vitro study. Anesth Analg. 116, 101–6 (2013).
Perez, M. et al. Compatibility of medications during multi-infusion therapy: A controlled in vitro study on a multilumen infusion device. Anaesth Crit Care Pain Med (2015).
Bradley, J. S., Wassel, R. T., Lee, L. & Nambiar, S. Intravenous ceftriaxone and calcium in the neonate: assessing the risk for cardiopulmonary adverse events. Pediatrics. 123, e609–13 (2009).
Foster, J., Richards, R. & Showell, M. Intravenous in-line filters for preventing morbidity and mortality in neonates. Cochrane Database Syst Rev. CD005248 (2006).
Niël-Weise, B. S., Stijnen, T. & van den Broek, P. J. Should in-line filters be used in peripheral intravenous catheters to prevent infusion-related phlebitis? A systematic review of randomized controlled trials. Anesth Analg. 110, 1624–9 (2010).
van den Hoogen, A. et al. In-line filters in central venous catheters in a neonatal intensive care unit. J Perinat Med. 34, 71–4 (2006).
van Lingen, R. A., Baerts, W., Marquering, A. C. M. & Ruijs, G. J. H. M. The use of in-line intravenous filters in sick newborn infants. Acta Paediatr. 93, 658–62 (2004).
Jack, T. et al. In-line filtration reduces severe complications and length of stay on pediatric intensive care unit: a prospective, randomized, controlled trial. Intensive Care Med. 38, 1008–16 (2012).
Boehne, M. et al. In-line filtration minimizes organ dysfunction: new aspects from a prospective, randomized, controlled trial. BMC Pediatr. 13, 21 (2013).
Gradwohl-Matis, I. et al. Influence of in-line microfilters on systemic inflammation in adult critically ill patients: a prospective, randomized, controlled open-label trial. Ann Intensive Care. 5, 36 (2015).
Brotschi, B., Grass, B., Weiss, M., Doell, C. & Bernet, V. In-line filter included into the syringe infusion pump assembly reduces flow irregularities. Intensive Care Med. 38, 518–22 (2012).
Safe Practices for Parenteral Nutrition Formulations. National Advisory Group on Standards and Practice Guidelines for Parenteral Nutrition. JPEN J Parenter Enteral Nutr. 22, 49–66 (1998).
Mirtallo, J. et al. Safe practices for parenteral nutrition. JPEN J Parenter Enteral Nutr. 28, S39–70 (2004).
Koletzko, B. et al. 1. Guidelines on Paediatric Parenteral Nutrition of the European Society of Paediatric Gastroenterology, Hepatology and Nutrition (ESPGHAN) and the European Society for Clinical Nutrition and Metabolism (ESPEN), Supported by the European Society of Paediatric Research (ESPR). J Pediatr Gastroenterol Nutr. 41, S1–87 (2005).
Boullata, J. I. et al. A.S.P.E.N. clinical guidelines: parenteral nutrition ordering, order review, compounding, labeling, and dispensing. JPEN J Parenter Enteral Nutr. 38, 334–77 (2014).
Bethune, K., Allwood, M., Grainger, C. & Wormleighton, C. & British Pharmaceutical Nutrition Group Working Party. Use of filters during the preparation and administration of parenteral nutrition: position paper and guidelines prepared by a British pharmaceutical nutrition group working party. Nutr. 17, 403–8 (2001).
Kuramoto, K., Shoji, T. & Nakagawa, Y. Usefulness of the final filter of the IV infusion set in intravenous administration of drugs–contamination of injection preparations by insoluble microparticles and its causes. Yakugaku Zasshi. 126, 289–95 (2006).
Foinard, A. et al. In Vitro Assessment of Interaction Between Amino Acids and Copper in Neonatal Parenteral Nutrition. JPEN J Parenter Enteral Nutr 40, 827–34 (2016).
Sasse, M. et al. In-line Filtration Decreases Systemic Inflammatory Response Syndrome, Renal and Hematologic Dysfunction in Pediatric Cardiac Intensive Care Patients. Pediatr Cardiol. 36, 1270–8 (2015).
Böhrer, H., Zhang, C. H. & Krier, C. Decrease in the concentration of tobramycin, vancomycin and phenobarbital in administration with infusion filter. Infusionstherapie Basel Switz. 18, 96–100 (1991).
Bononi, I., Forgetta, C., Baldisserotto, A. & Tognon, M. Simulated infusion of paclitaxel with in-line filters. Chemotherapy. 57, 204–8 (2011).
Foinard, A., Décaudin, B., Barthélémy, C., Debaene, B. & Odou, P. Impact of physical incompatibility on drug mass flow rates: example of furosemide-midazolam incompatibility. Ann Intensive Care. 2, 28 (2012).
Kennedy, R. A., Kennedy, M. L., Morrissey, H. & Ball, P. A. In vitro analysis of the effect of in-line 1.2 micron filters on two formulations of propofol (2,6-diisopropyl phenol). PDA J Pharm Sci Technol. 69, 297–306 (2015).
Hirakawa, M., Makino, K., Nakashima, K., Kataoka, Y. & Oishi, R. Evaluation of the in-line filters for the intravenous infusion of amphotericin B fluid. J Clin Pharm Ther. 24, 387–92 (1999).
Ph. Eur., General, particulate contamination: sub-visible particles, in: The European Pharmacopoeia, ninth ed., Chapter 2.9.19 (2015).
We thank Pall for providing technological expertise, as well as filters. All phases of this study were supported by intramural funds from the Department of Biopharmacy, Galenic and Hospital Pharmacy, Univ. Lille, CHU Lille, EA 7365 – GRITA – Groupe de Recherche sur les Formes Injectables et les Technologies Associées, F-59000 Lille, France. English language editing assistance was provided by Alexandra Tavernier (M.A. University of Glasgow U.K., Professeur Agrégée, France).
The authors declare no competing interests.
Publisher's note: Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Perez, M., Décaudin, B., Abou Chahla, W. et al. Effectiveness of in-Line Filters to Completely Remove Particulate Contamination During a Pediatric Multidrug Infusion Protocol. Sci Rep 8, 7714 (2018) doi:10.1038/s41598-018-25602-6