Abstract
Loss measurements are at the base of spectroscopy and imaging, thus permeating all the branches of science, from chemistry and biology to physics and material science. However, quantum mechanics laws set the ultimate limit to the sensitivity, constrained by the probe mean energy. This can be the main source of uncertainty, for example when dealing with delicate systems such as biological samples or photosensitive chemicals. It turns out that ordinary (classical) probe beams, namely with Poissonian photon number distribution, are fundamentally inadequate to measure small losses with the highest sensitivity. It is known that quantumcorrelated pair of beams, named “twinbeam state”, allows surpassing this classical limit. Here we demonstrate they can reach the ultimate sensitivity for all energy regimes (even less than one photon per mode) with the simplest measurement strategy. One beam of the pair addresses the sample, while the second one is used as a reference to compensate both for classical drifts and for fluctuation at the most fundamental quantum level. This capability of selfcompensating for unavoidable instability of the sources and detectors allows also to strongly reduce the bias in practical measurement. Moreover, we report the best sensitivity per photon ever achieved in loss estimation experiments.
Similar content being viewed by others
Introduction
The measurement of changes in intensity or in phase of an electromagnetic field, after interacting with matter, is the most simple and effective way to extract relevant information on the properties of a system under investigation, whether a biological sample^{1,2} or a digital memory disc^{3}. Intensity measurements enable absorption/transmission estimation, the base of imaging and spectroscopy, pervasive and fundamental techniques in all science fields, from chemistry^{4} to material science^{5} and physics^{6}. They are routinely employed in biomedical analysis^{7,8,9}, as well as in atmospheric^{10,11,12} and food sciences^{13,14}.
However, the optical transmission losses experienced by a probe beam while interacting with a system cannot be determined with arbitrary precision, even in principle. Quantum mechanics establishes fundamental bounds to the sensitivity^{15,16,17,18}, which is limited, in general, by the mean energy of the probe, or, equivalently, by its mean number of photons. This is in accordance to the intuitive idea that gaining the perfect knowledge on a system would require an infinite amount of physical resources.
The lower bound to the uncertainty, when restricted to the use of classical probe states, coincides with the one achieved by a coherent state, \({{U}}_{{coh}}\simeq {\mathrm{[(1}{\alpha })/\langle {n}_{P}\rangle ]}^{\mathrm{1/2}}\)^{17}, where 〈n_{ P }〉 is the mean number of photons of the probe and 0 ≤ α ≤ 1 is the loss of the sample. Indeed, this limit can be obtained in practice by any probe beam exibiting Poissonian photon statistics, as a laser beam (described theoretically by a coherent state) or even a thermal source like LEDs or incandescent light bulbs in the limit of extremely low photon number per mode. Note that the uncertainty depends on the loss parameter, and can be arbitrary small only in the asymptotic limit of high losses. For a faint loss, \(\alpha \sim 0\), one retrieves the expression U_{ snl } = 〈n_{ P }〉^{−1/2}, usually referred as to “shotnoiselimit” (SNL).
Without restriction on the probe state, it has been shown^{18,19} that the ultimate quantum limit (UQL) in the sensitivity for a single mode interrogation of the sample is \({{U}}_{{uql}}\simeq \sqrt{\alpha }\,{{U}}_{{coh}}\), which scales much more favourably than the classical bound for small losses, a region which is particularly significant in many real applications. It is worth noting that the use of quantum states does not improve the uncertainty scaling with the number of particles. This is different from what happens in phase shift estimation, in which a sensitivity scaling proportional to 〈n_{ P }〉^{−1} is reachable in ideal situations^{15,16}, the so called “Heisenberg limit”. The fundamental difference is that phase shift is a unitary operation, preserving the purity of the state, while a loss is intrinsically non unitary. A loss can be represented as the action of a beam splitter that mixes up the probe state in one port with the vacuum state in the other port, basically spoiling quantum features such as entanglement, which is necessary to approach the Heisenberg limit^{16}.
It is known that single mode squeezed vacuum reaches U_{ uql } for small losses, \(\alpha \sim 0\), and small number of photons \(\langle {n}_{P}\rangle \sim 0\)^{18}. Fock states n〉, having by definition a fixed number of photons, approach U_{ uql } unconditionally, i.e. for all value of α, but they cannot explore the regime of 〈n_{ P }〉 < 1^{19}. The optimal performance of Fock states can be understood by considering that a loss can be easily estimated by comparing the number of photons of the probe before and after the interaction with the sample. The perfect knowledge of the photon number of the unperturbed Fock state allows one to detect better small deviations caused by the sample, which would remain hidden in the intrinsic photon number fluctuation of Poissonian distributed sources.
However it is challenging to produce experimentally true Fock states. A reasonable approximation of a Fock state with n = 1 are the heralded single photons produced by spontaneous parametric down conversion (SPDC)^{20,21}. In this process photons are always emitted in pairs with low probability, but one can get rid of the vacuum component since the detection of one photon of the pair heralds the presence of the other one. This scheme has been demonstrated recently for quantum enhanced absorption measurement both with postselection of the heralded single photons^{22} and, more remarkably, with selection performed by active feedforward enabled by an optical shutter^{23}.
Also quantum correlations of twinbeam (TWB) state have shown the possibility of subSNL sensitivity in absorption/transmission measurement^{24,25,26,27,28,29,30,31}, quantum enhanced sensing^{32,33,34,35}, ghost imaging^{36}, quantum reading of digital memories^{37} and plasmonic sensors^{38,39}. TWB states can be generated by SPDC^{40} as well as by four wave mixing in atomic vapours^{41,42,43,44}, and expose a high level of quantum correlation in the photon number fluctuations between two corresponding modes, for example two propagation directions or two wavelengths. Even if superPoissonian noise characterizes the photon distribution in one mode, the fluctuations are perfectly reproduced in time and space in the correlated mode. Subshot noise correlation of this state has been experimentally demonstrated both in the twomode case^{45,46,47,48,49} and in the case of many spatial modes detected in parallel by the pixels of a CCD camera^{50,51,52}. The exploitation of spatially multimode nonclassical correlation has been proposed for high sensitivity imaging of distributed absorbing object^{53} and a proof of principle of the technique has been reported by Brida et al. in^{28}. Recently our group has realized the first widefield subSNL microscope^{30}, providing 10^{4} pixels images with a true (without postselection) significant quantum enhancement, and a spatial resolution of few micrometers. This represents a considerable advancement towards a real application of quantum imaging and sensing.
The common idea behind these works is that the random intensity noise in the probe beam addressed to the sample can be known by measuring the correlated (reference) beam and subtracted. Note that the twobeams approach is extensively used in standard devices like spectrophotometers, where a classical beam is split in two by a beam splitter and one beam is used to monitor the instability of the source and detectors and to compensate for them. This is particularly effective in practical applications, since unavoidable drifts in the source emission or detector response would lead to strong bias, especially in the estimation of small absorptions. However, in classical correlated beams (CCB) generated in this way, only the superPoissonian component of the fluctuations is correlated (sometimes called classical “excess noise”), whereas the shot noise remains uncorrelated and cannot be compensated. Therefore TWB represent the natural extension to the twobeam approach to the quantum domain, promising to be especially effective for small absorption measurement and when low photon flux is required.
It has been theoretically demonstrated^{54} that using TWB for loss estimation the UQL is in principle attainable; nevertheless the existence of an experimental estimator fit for this purpose is still an open question, as it is its explicit expression.
Here, we show that the answer to this question is unconditionally positive considering TWB generated by SPDC process, for all the energy regime and all values of the loss parameter α. Therefore, TWB overcome the limitations of both single mode squeezed vacuum and Fock states, representing the practical best choice for pure loss estimation. We prove this result by an operative approach: we consider a specific and simple measurement strategy, proposed for the first time by Jakeman and Rarity^{24}, that is to evaluate the ratio between the photon number measured in the probe and in the reference beam. In the ideal lossless detection case this is sufficient to reach the ultimate quantum limit. Taking into account for experimental imperfections, we derive the uncertainty advantage of the twinbeam with respect to the single classical beam (SCB) and to the CCB case in terms of experimental parameters related to the “local” photon statistics of the two beams separately, and the amount of nonclassical correlation of the joint photon number statistics.
In a recent work^{27}, a different optimized estimator which allows improving the sensitivity in case of strong nonideal detection efficiencies has been proposed. The drawback is that this method requires the accurate and absolute characterization of the measurement apparatus, in particular the absolute values of the quantum efficiencies of the detectors and of the excess noise of the source. This aspect places a strong practical limitation, because the determination of quantum efficiency, especially at the few photon level, with uncertainty less than 10^{−3} is extremely challenging, limiting the overall accuracy of the method; then, instabilities could also affect the measurement. We show that the simplest estimator in ref.^{24} behaves almost as good as the optimized one for relatively high values of the efficiencies (the condition of our experiment), but it requires the weakest assumptions on the stationarity of the system and does not require absolute value of any parameter.
Finally we perform the experiment, measuring intensity correlations in the far field of multimode parametric down conversion by a standard low noise and high efficiency CCD camera. For a sample loss of \( \sim \mathrm{2 \% }\), we report an experimental quantum enhancement in the estimation uncertainty of 1.51 ± 0.13 with respect to the single beam classical probe and of 2.00 ± 0.16 compared to the classical twobeam approach, when the same mean energy of the probe and the same detection efficiency are considered.
Theory
In practice, an optical loss α can be easily measured by comparing the number of photons of the probe \({N}_{P}^{^{\prime} }\) after a lossy interaction, with a reference value N_{ R }, which can be evaluated in a previous moment in absence of the sample (Fig. 1a) or by the help of a second beam (Fig. 1d). In particular, one can consider the estimator^{24}:
The factor γ = 〈N_{ R }〉/〈N_{ P }〉 should be introduced in case of unbalancing between the mean energy of probe and reference beams and evaluated in a precalibration phase of the apparatus (Fig. 1c). A loss is a random process modelled by the action of a beam splitter of transmission 1 − α, so that the statistics of the photon counting of the probe beam is modified in this way^{40}:
Here N_{ P } is the measured photon number without the sample. Its fluctuation is represented by the Fano factor F_{ P } = 〈Δ^{2}N_{ P }〉/〈N_{ P }〉 ≥ 0 which quantifies the nonclassicality of the photon statistics. In particular F_{P} < 1 indicates subPoissonian noise^{55} and in general the possibility to surpass the SNL.
By expanding the photon number operators in Eq. (1) at the first order around their mean value, the expected value of the estimator becomes^{24}:
An unbiased estimation of the loss can be obtained solving the Eq. (4) with respect to α. By propagating the uncertainty of the quantities \({N}_{P}^{^{\prime} }\) and N_{ R } on S_{ α }, and rewriting the terms using the unperturbed variance 〈Δ^{2}N_{ P }〉, the quantum expectation value of fluctuation is:
Note that \({U}_{uql,\langle {N}_{P}\rangle }\) has the form of the UQL but refers to the number of detected photons. Considering the probes photons 〈n_{ P }〉 incident on the sample, one has \({U}_{uql,\langle {n}_{P}\rangle }={U}_{uql,\langle {N}_{P}\rangle }\sqrt{{\eta }_{d}}\), where η_{ d } represents the detection efficiency, i.e. the losses experienced after the sample. The most relevant quantity appearing in Eq. (5) is the positive factor:
In the case of γ = 1 it represents the quantifier of the nonclassical correlation known as noise reduction factor (NRF), σ = σ_{γ=1}, where the bound between classical and quantum correlations is set by σ = 1. Thus, the uncertainty is expressed in terms of simple measurable quantities related to the photon number statistics, i.e. the intensity fluctuations. Eq. (5) shows that whenever γ = 1 and σ = 0 the UQL is retrieved, \({{\rm{\Delta }}}^{2}{S}_{\alpha }(\gamma =1,\,\sigma =0)\simeq {U}_{uql,\langle {N}_{P}\rangle }^{2}\).
In the following we consider different states for the probe and the reference beam to establish the limit to the sensitivity in relevant scenarios.
Let us first focus on the states which do not present correlation between probe and reference (e.g. the measurements on the probe and reference beam are performed in two different moments, refer to Fig. 1 a), b), so that 〈ΔN_{ P }ΔN_{ R }〉 = 0.

Fock states. It is clear that the only chance for uncorrelated states to achieve the condition σ_{ γ } = 0 and hence the UQL according to Eq. (5) is to have null fluctuation in the photon number both for the reference and probe beam, 〈Δ^{2}N_{ R }〉 ≡ 〈Δ^{2}N_{ P }〉 ≡ 0. This means that the state must be the product of two unperturbed Fock states, \(n{\rangle }_{P}\otimes n{\rangle }_{R}\) detected with unitary efficiency. Thus, as anticipated, Fock states reaches the UQL unconditionally, i.e. for all the value of the parameter, with the only limitation that the mean photon number cannot be arbitrarily small^{19} (i.e. 〈n_{ P }〉 ≥ 1).

Coherent states. Let us now consider the state \(coh{\rangle }_{P}\otimes coh{\rangle }_{R}\), particularly interesting for its simple experimental implementation. In the photon number basis, coherent states have the form \(coh\rangle ={e}^{\frac{1}{2}\langle n\rangle }{\sum }_{n\mathrm{=0}}^{\infty }\frac{{\langle n\rangle }^{n\mathrm{/2}}}{\sqrt{n!}}n\rangle \), following the Poissonian photon number distribution P_{ coh }(n) = e^{−〈n〉}〈n〉^{n}/n!, which has the property 〈Δ^{2}n〉 = 〈n〉. Thus, substituting the variances with the mean values in the right hand side of Eq. (6) one get σ_{ γ } = (1 + γ)/2, and accordingly:
The lower limit for a pair of coherent states is reached under the condition of \(\gamma \gg 1\), i.e. when the reference beam has much more energy than the transmitted probe, and the relative fluctuation on its photon number becomes negligible. In this case \({{\rm{\Delta }}}^{2}{S}_{\alpha }^{(coh)}\) equals the classical lower bound, detection efficiency apart, \({{\rm{\Delta }}}^{2}{S}_{\alpha }^{(coh)}=\) \((1\alpha )/\langle {N}_{P}\rangle ={\eta }_{d}^{1}{U}_{coh,\langle {n}_{P}\rangle }^{2}\). In practice, one can also consider an equivalent situation, in which the reference uncertainty has been statistically reduced to a negligible contribution by a long acquisition time in the calibration phase (Fig. 1a), namely a time much longer than the one used for the measurement of the probe beam in presence of the sample (Fig. 1b). Indeed, replacing the variable N_{ R } with its mean value 〈N_{ R }〉 in the definition of S_{ α } and of σ_{ γ } in Eq. (6) leads to the an identical limit of the sensitivity.
More in general, it is convenient to rewrite the noise reduction factor for uncorrelated states in terms of the measurable Fano factor of each beam in absence of the sample, i.e. σ_{ γ } = (F_{ R } + γF_{ P })/2. With this substitution, Eq. (5) becomes:
The measured Fano factors account for the statistics of light sources and for transmission inefficiency and detection losses. If 0 ≤ η_{ j } ≤ 1(j = P, R) is the overall channel efficiency, including the detection one η_{ d } and the losses between the source and the sample, the Fano factor can be written as \({F}_{j}={\eta }_{j}{F}_{j}^{\mathrm{(0)}}+1{\eta }_{j}\), where \({F}_{j}^{\mathrm{(0)}}\) refers to the one of the unperturbed state of the source. As expected, detection losses deteriorate the non classical signature of the probe and reference beams, preventing the real possibility to reach the UQL even with Fock states.
Considering now joint states where a correlation between probe and reference is present, i.e. 〈ΔN_{ P }ΔN_{ R }〉 ≠ 0 (Fig. 1c,d) we have:

TWB state. Two mode twin beam state, generated by SPDC, is represented by the following entangled state in the photon number basis {n〉}^{56}:
The two modes, separately, obey to a thermal statistics each, where 〈Δ^{2}n〉 = 〈n〉(1 + 〈n〉). However, they are balanced in the mean energy, 〈n_{ P }〉 = 〈n_{ R }〉, and their fluctuations are perfectly correlated, 〈Δn_{ P }Δn_{ R }〉 = 〈Δ^{2}n〉. This leads to γ = 1 and σ = 0, thus demonstrating that TWB detected with unitary efficiency reaches the U_{ uql }, according to Eq. (5). Note that this result is independent on the value of the parameter α and on the energy of the probe beam which can contain less than one photon per mode on average. Indeed, this is usually the case in experiments.

Classical correlated beams (CCB). Let us consider a bipartite correlated state produced by a unitary splitting of a single beam. Given a splitting ratio 0 ≤ τ ≤ 1, it turns out that the statistics of the two outcoming beams, the probe and the reference, is characterized by γ = τ^{−1} − 1 and σ_{ γ } = (2τ)^{−1}, which are remarkably independent on the photon number distribution of the initial beam. Substituting these values in Eq. (5) leads to the same uncertainty of two uncorrelated coherent beams \({{\rm{\Delta }}}^{2}{S}_{\alpha }^{(CCB)}={{\rm{\Delta }}}^{2}{S}_{\alpha }^{(coh)}\), reported in Eq. (8). It shows that classical correlation can never approach the UQL, and that the lower uncertainty is achieved for a splitting ratio \(\tau \simeq 0\) corresponding to a strong unbalancing of beam energies, \(\langle {N}_{P}\rangle \ll \langle {N}_{R}\rangle \). Therefore, for the specific measurement strategy considered here and whatever the input state, it is convenient to use a highly populated reference beam and a weak prope beam. This result is in agreement with the behaviour reported by Spedalieri et al.^{57} in the complementary situation in which the input state is a thermal one while the measurement strategy is the most general one allowed by quantum mechanics.
Finally, to better understand how losses or excess noise of the source influence the final accuracy in real experiment we note that the parameter σ_{ γ } can be rewritten as \({\sigma }_{\gamma }=\frac{\gamma +1}{2}\sigma +\frac{\gamma 1}{2}({F}_{R}\gamma {F}_{P})\). In presence of equal losses in both the branches η_{ R } = η_{ P } = η, the noise reduction factor, expressed in terms of the ideal unperturbed one σ^{(0)}, is σ = ησ^{(0)} + 1−η. For the relevant case of a TWB state, it is F_{ R } = F_{ P }, γ = 1 and σ^{(0)} = 0, leading to:
This expression shows how the degradation of the accuracy in presence of losses prevents reaching the UQL in practice.
On the other side, for γ = 1, balanced CCB (bCCB) fulfills the lower classical bound σ_{ γ } = σ = σ^{(0)} = 1, thus using Eq. (5) we obtain:
Note that in case of bCCB, the accuracy is immune from the detection losses but it is always worse than in the case of TWB reported in Eq. (11).
Up to now we have analyzed the performance of the specific estimator in Eq. (1), showing that it allows reaching the optimal limits both for classical and quantum states, in particular using TWB state the UQL is retrieved. However, other estimators have been considered in literature for absorption measurement with TWB. An interesting alternative is the estimator used in the recent experiment by Moreau et al.^{27},
where the weight factor k can be determined in order to minimize the uncertainty on \({S}_{\alpha }^{^{\prime} }\), while δE is a small correction introduced to render the estimator unbiased. However, k and δE need to be estimated in a phase of precalibration of the apparatus. In particular it turns out that k_{ opt } is a function of the detection efficiencies of the channels and the local excess noise k_{ opt } = f(η_{ P }, η_{ R }, F_{ P }, F_{ R }) while δE depends also from the measured covariance 〈ΔN_{ P }ΔN_{ R }〉. We have evaluated analytically in the general case, with the only hypothesis of balanced sources, the expected uncertainty of the estimator in Eq. (13) when k = k_{ opt }. For the sake of simplicity, here we report the expression obtained in case of symmetric statistical properties of the channels, γ = 1 and F_{ P } = F_{ R } = F:
For TWB and lossless detection, the noise reduction factor σ is identically null and the UQL is retrieved also with this estimator. Taking into account balanced detection losses, and the common experimental case of a mean photon number per mode much smaller than one, one can substitute in Eq. (14) σ = 1 − η and \(F\simeq 1\). Therefore, the uncertainty becomes:
Comparing the uncertainty in Eq. (15) with the one reported in Eq. (11) makes clear that the estimator \({S}_{\alpha }^{^{\prime} }\) proposed in^{27} performs better than S_{ α }, especially when detection losses are considerable.
Finally, in Brambilla et al.^{53} it is suggested to measure the absorption by a differential measurement, considering the following estimator:
For a source producing a pairs of beams with the same local statistical properties, the variance of \({S}_{\alpha }^{^{\prime\prime} }\) can be calculated as:
However, this choice is not optimal and depends on the value of the measured local statistics: in the best case of unperturbed TWB, in which σ_{ γ } = 0 and γ = 1, it approaches U_{ uql } only asymptotically for \({F}_{R}{\alpha }^{2} \sim 0\). In TWB, produced experimentally by SPDC, the statistics of each mode is thermal with a photon number per mode much smaller than one, thus \({F}_{R}\simeq 1\) and the condition reduces to \(\alpha \sim 0\). Conversely, for high value of the estimated losses, \(\alpha \sim 1\), the performance of this estimator is much worse than the one of S_{ α } and \({S}_{\alpha }^{^{\prime} }\).
Experiment
A scheme of the experimental setup is reported in Fig. 2.
A CW laserbeam (10 mW at λ_{ pump } = 405 nm) pumps a 1 cm TypeIIBetaBariumBorate (BBO) non linear crystal, where SPDC occurs and two beams with perfect correlation in the photon number are generated. Note that the state Ψ〉 produced by SPDC process is intrinsically multimode and can be expressed, in the planewave pump approximation, as a tensor product of twomodes TWB states of the form in Eq. (10) as: Ψ〉 = ⊗_{q,λ}TWB〉_{q,λ}, where q and λ are respectively the transverse momentum and the wavelength of one of the two photons produced, while momentum and wavelength of the other photon are fixed by energy and momentum conservation.
The far field of the emission is realized at the focal plane of a lens with f_{ FF } = 1 cm focal length. Then a second lens, with f_{ IM } = 1.6 cm, images the far field plane to the detection plane. The magnification factor is M = 7.8. The detector is a chargecoupleddevice (CCD) camera Princeton Inst. Pixis 400BR Excelon operating in linear mode and cooled down to −70 °C. It presents high quantum efficiency (nominally > 95% at 810 nm), 100% fill factor and low noise (readnoise has been estimated around 5 e^{−}/(pixel⋅second)). The physical pixel of the camera measures 13 μm, nevertheless, not being interested in resolution, we group them by 24 × 24 hardware binning. This allows us to reduce the acquisition time and the effects of the readout noise. Just after the crystal an interference filter ((800 ± 20) nm, 99% transmittance) is positioned to select only the modes of frequencies around the degeneracy, λ_{ d } = 2λ_{ pump }. This choice allows the presence of different spatial modes, in our case we have \({M}_{sp} \sim 2500\) spatial modes impinging on each detection area, S_{ P } and S_{ R }, where P and R subscripts refer to the probe and reference beam, respectively. We integrate the signals in S_{ R } and in S_{ P }. The sample consists in a coated glassslide with a deposition of variable absorption coefficient α intercepting the probe beam in the focal plane. We consider values of α from 1% to 70%. Finally, in order to check the theoretical model at varying η_{ R } and η_{ P }, neutral filters of different absorption can be eventually positioned on the beams path.
The acquisition time of a single frame is set to 100 ms, whilst the coherence time of the SPDC process is around 10^{−12} s, thus the number of the detected temporal modes is approximatively \({M}_{t} \sim {10}^{11}\). Since in each detection area we register around \(\langle {N}_{P}\rangle \sim 50\cdot {10}^{4}\) photons per frame, it follows that the occupation number of the single spatiotemporal mode is \(\mu \sim 2\cdot {10}^{9}\) photons/mode. Being \(\mu \ll 1\), this implies that the statistic of a single mode is well modeled by a Poissonian statistic: it follows that if only one beam is considered the measurements are shotnoise limited.
However, it is possible to go beyond the shot noise limit exploiting the photon number correlation between pairs of correlated modes. In the plane wave pump approximation with transverse momentum q_{ pump } = 0, in the far field region any mode with transverse momentum q is associated with a single position x according to the relation: \({\bf{x}}=\frac{2c{f}_{FF}}{{\omega }_{pump}}{\bf{q}}\), where c is the speed of light, f_{ FF } the focal length of the first lens and ω_{ pump } the laser frequency. The exact phasematching condition for correlated modes q_{ P } + q_{ R } = q_{ pump } = 0 becomes in the far field, for degenerate wavelengths λ_{ P } = λ_{ R } = 2λ_{ pump }, a condition on their position: x_{ P } + x_{ R } = 0. Under the hypothesis of plane wave pump it is therefore expected that two symmetric pixels of the camera, respect to the pump direction, always detect the same number of photons. For a realistic pump with a certain spread Δq it follows: \({{\boldsymbol{x}}}_{P}+{{\boldsymbol{x}}}_{R}=0\pm {\rm{\Delta }}{\boldsymbol{x}}=\pm \,\frac{2c{f}_{FF}}{{\omega }_{pump}}{\rm{\Delta }}{\boldsymbol{q}}\). Δx represents the size in the far field of the so called coherence area, A_{ coh }, area in which photons from correlated modes can be collected. Moreover, the nonnull frequency bandwidth (about 40 nm in our experiment) determines a further broadening of the spot in which correlated detection events occur. To experimentally measure the size of A_{ coh } the spatial crosscorrelation between the two beams can be considered^{30}. Its evaluation is important to compare it with the detection area A_{ det } since, to detect a significant level of correlation, it is necessary that A_{ det } ≥ A_{ coh }. In our case, integrating on the two regions of interest this condition is fully fulfilled, indeed it holds \({A}_{det}\gg {A}_{coh}\). In general the measured NRF can be modeled as^{58}:
where two contributions are present.

0 ≤ η_{ R } ≤ 1, the total efficiency of the reference optical path.

0 ≤ η_{ coll } ≤ 1, the collection efficiency of correlated photons. This factor represents approximatively the probability that given a detected photon in S_{ R }, its “twin” is expected to fall in S_{ P }.
In our experimental situation, since \({S}_{P}={S}_{R}\gg {A}_{coh}\) it follows η_{ coll } → 1 and consequently \({\sigma }_{\gamma }=\frac{1+\gamma }{2}{\eta }_{R}\). Inverting this relation offers a useful way to measure the total efficiencies (Klyshko heralding efficiency) of the two channels, without the need of comparing with calibrated devices^{59}. In the experimental situation corresponding to Fig. 3 we measured σ_{ γ } = 0.24 ± 0.03 and γ = 1.006, which implies overall heralding efficiencies η_{ R } = η_{ P } = 0.76, as reported in the caption. The same method has been adopted to evaluate the efficiencies in the other cases, reported in Figs 4 and 5.
In all these figures the mean values of α (xaxis) and their corresponding uncertainties Δα (yaxis) have been obtained acquiring 200 frames with the absorbing sample inserted. Repeating each measurement 10 times the error bars have been estimated. In particular, for each frame, we integrate the data on S_{ R } and S_{ P }, opportunely corrected for the background, obtaining N_{ R } and N′_{ P }, necessary for the estimation of the mean absorption α according to the different estimators considered, in Eqs (1–16).
To reproduce the singlemode classical strategy we performed a calibration measurement without the sample obtaining 〈N_{ R }〉; we then estimate α as:
For ideal Poissonian statistics of the probe, this strategy leads to the classical lower bound U_{ coh } (see Theory section). In our experiment, the Poissonian behavior is guaranteed by the condition \(\mu \ll 1\), as discussed before.
Finally to reproduce the bCCB case we consider a different region of the detector \({S}_{R}^{^{\prime} }\), displaced from S_{ R } and only classically correlated with S_{ P }.
Note that from the calibration measurement also γ, σ_{ γ }, F_{ P } and F_{ R } can be simply evaluated.
Results and Discussion
In Eqs (11) and (15) we have explicitly reported the uncertainty achieved by TWB for the estimators \({S}_{\alpha }^{(TWB)}\) and \({S}_{\alpha }^{^{\prime} (TWB)}\) respectively, in case of balanced total efficiencies in the probe and reference beam. The unbalanced case leads to cumbersome analytical expressions, so we report this situation graphically in Fig. 6. The uncertainties on these two estimators are compared at varying η_{ R } and fixed η_{ P } with respect to the classical lower bound \({U}_{coh,\langle {N}_{P}\rangle }\), evaluated for the same number of detected photons. It emerges that for η_{ R } = 1 the two estimators offer exactly the same quantum enhancement, maximum for \(\alpha \ll 1\). Nonetheless, for η_{ R } ≠ 1 and sufficiently large, the performances of the two estimators remain comparable. Instead when η_{ R } < 0.5 the uncertainty on \({S}_{\alpha }^{(TWB)}\) becomes greater than the classical one; on the contrary \({\rm{\Delta }}{S}_{\alpha }^{^{\prime} (TWB)}\) maintains always below it. Note that in Fig. 6 we fix η_{ P } = 0.76 (the value of our experiment) and we considered the dependence from η_{ R }. The opposite situation, where η_{ R } is kept fix is not reported. In this case ΔS_{ α } and Δ\({S}_{\alpha }^{^{\prime} }\) behave similarly for all the variability range of η_{ P }, and are always below \({U}_{coh,\langle {N}_{P}\rangle }\).
These different regimes at varying η_{ R } have been experimentally explored with our setup and the results are shown in Figs 3–5. In these figures, considering different estimators, the dependence of the uncertainty on α in function of its mean value is reported. The three situations only differ from the value of η_{ R } considered. The solid lines are the theoretical curves in Eqs (9, 5 and 17) and the equivalent of Eq. (14) in the general case of γ ≠ 1 where the experimental values of the quantities σ_{ γ }, F_{ P }, F_{ R }, γ have been substituted. The markers represent the experimental data which are in a good agreement with our theoretical model describing experimental imperfections. The black curves stand for significant limits, obtained with ideal states. The dotteddashed line is the fundamental quantum limit U_{ uql } = [α(1 − α)/〈n_{ P }〉]^{1/2}, achievable by TWB and unitary efficiencies. The dashed line is the classical lower bound calculated for the actual number of detected photons, \({U}_{coh,\langle {N}_{P}\rangle }\), while the dotted line is the classical limit in the twomode balanced case, \({\rm{\Delta }}{S}_{\alpha }^{(bCCB)}\). Figure 3 reports also the classical lower bound assuming no losses occurring after the sample, \({U}_{coh,\langle {n}_{P}\rangle }\mathrm{=[(1}\alpha )/\langle {n}_{P}\rangle {]}^{\mathrm{1/2}}\), where 〈n_{ P }〉 is the number of photons of the probe interacting with the sample. This quantity can be easily estimated as \(\langle {n}_{P}\rangle =\langle {N}_{P}\rangle {\eta }_{d}^{1}\), where η_{ d } represents the detection efficiency after the sample. The obtained value of η_{ d } = 0.80 ± 0.01 takes into account transmission and collection losses trough all the optical elements after the sample (a lens, an interference filter and the quantum efficiency of our CCD camera). The efficiency of the camera with the filter placed in front of it has been experimentally measured using the technique presented in^{58} (η_{ CCD } = 0.84 ± 0.01).
Although experimental not unitary efficiencies lead to a remarkable detachment from the UQL, for \(\alpha \sim \mathrm{2 \% }\), we still obtain a significant quantum enhancement: \({U}_{coh,\langle {N}_{P}\rangle }/{\rm{\Delta }}{S^{\prime} }_{\alpha }=1.51\pm 0.13\) and \({\rm{\Delta }}{S}_{\alpha }^{(bCCB)}/{\rm{\Delta }}{S^{\prime} }_{\alpha }=2.00\pm 0.16\). The comparison respect to the classical lower bound assumed with ideal detection efficiency leads to \({U}_{coh,\langle {n}_{P}\rangle }/{\rm{\Delta }}{S^{\prime} }_{\alpha }=1.32\pm 0.14\).
The comparison with the twomode classical strategy (\({S}_{\alpha }^{(bCCB)}\)) is of particular interest since the twobeam approach allows compensating unavoidable drifts and instability of source and detectors, leading to an unbiased estimation of α, i.e. not affected by temporal drifts of the experimental setup. Estimators S_{ α } and \({S}_{\alpha }^{^{\prime\prime} }\) do not require the knowledge of the individual absolute power of the source or detector response but a measurement of the average arms unbalance in absence of the object \(\gamma =\frac{\langle {N}_{R}\rangle }{\langle {N}_{P}\rangle }\), and the condition for having an unbiased estimator is the stability of this parameter. Experimentally, this is much less demanding than controlling the power stability of the individual probe beam (i.e. 〈N_{ P }〉 constant over time) and detector response for the direct/single beam case. Indeed, it is expected that the factors affecting the source and the detectors act in the same way on the probe and on the reference channels.
On the other side \({S}_{\alpha }^{^{\prime} }\), in particular the calculation of k_{ opt } and δE, requires the knowledge of the two absolute values of both the efficiencies η_{ R } and η_{ P }, which include optical transmission and detectors quantum efficiency. The last one is usually obtained by comparison with calibrated radiometric standards. Alternatively, they can be determined from the same SPDC setup by using some extensions of the Klyshko’s method^{58,59,60}. This second approach is the one used in the present paper: as described after the Eq. 18, absolute arms efficiencies can be extracted from the measured value of σ_{ γ }. In any case, uncertainty smaller than 10^{−3} is quite challenging in the calibration of detector operating at low optical power. Inaccuracy in the determination of these parameters, although does not propagate directly to the loss estimation, could somehow affect the optimality of \({S}_{\alpha }^{^{\prime} }\). Furthermore, \({S}_{\alpha }^{^{\prime} }\) could be affected by drift in the mean value of 〈N_{ P }〉, as it happens for the single mode strategy.
Conclusion
We address the question of loss estimation and analyze different measurement strategies. In particular we show that with a simple photon number measurement of TWB state it is possible to approach the ultimate quantum limit of the sensitivity in case of perfect detection efficiency. The experiment reports the best sensitivity per photon ever achieved in loss estimation without any kind of data postselection. Indeed, as far as we know the best reported result is a quantum enhancement of 1.21 ± 0.02, recently achieved by Moreau et al.^{27}. Also other transmission based experiments demonstrating significant quantum enhanced sensitivity are present in literature, as^{39}, however their results are not directly comparable with ours since the uncertainty on the absorption coefficient is not reported.
In particular we double the sensitivity of the conventional classical twobeam approach and we overtake of more than 50% the sensitivity of the coherent case. The advantage, considering perfect detection efficiency of the classical beam after the sample, reduces to 32%. At the same time these results accurately confirm the theoretical model accounting for experimental imperfections.
The estimator represented by S_{ α } in Eq. (1)^{24}, is compared both theoretically and experimentally, with other estimators in literature (see Eqs (13) and (16)) in presence of experimental imperfections (e.g. not unitary detection efficiency). Despite in case of high detection losses the estimator \({S}_{\alpha }^{^{\prime} }\) in Eq. (13) has the smallest uncertainty, it turns out that where the quantum enhancement is significant, i.e. for sufficiently high efficiencies, S_{ α } and \({S}_{\alpha }^{^{\prime} }\) approximately offer the same quantum enhancement. Moreover, we argue that S_{ α }, beside its simple form, has several practical advantages. On the one side, it is robust to experimental unavoidable drifts of the sources and detectors, leading to unbiased estimate. On the other side, it does not require absolute detection efficiency calibration. These features are of the utmost importance in view of real applications.
References
Cone, M. T. et al. Measuring the absorption coefficient of biological materials using integrating cavity ringdown spectroscopy. Optica 2(2), 162–168 (2015).
Cheong, W. F., Prahl, S. A. & Welch, A. J. A review of the optical properties of biological tissues. IEEE Journal of Quantum Electronics 26(12), 2166–2185 (1990).
Gu, M., Li, X. & Cao, Y. Optical storage arrays: a perspective for future big data storage. Light: Science & Applications 3, e177 (2014).
Koningsberger, D. C. & Prins, R. Xray absorption: principles, applications, techniques of EXAFS, SEXAFS, and XANES. New York: Wiley (1988).
Weller, H. Quantized semiconductor particles: a novel state of matter for materials science. Advanced Materials 5(2), 88–95 (1993).
Savage, B. D. & Sembach, K. R. Interstellar abundances from absorptionline observations with the Hubble Space Telescope. Annual Review of Astronomy and Astrophysics 34(1), 279–329 (1996).
Hebden, J. C., Arridge, S. R. & Delpy, D. T. Optical imaging in medicine: I. Experimental techniques. Physics in Medicine and Biology 42(5), 825 (1997).
Jacques, S. L. Optical properties of biological tissues: a review. Physics in Medicine and Biology 58(11), R37 (2013).
Zonios, G. et al. Melanin absorption spectroscopy: new method for noninvasive skin investigation and melanoma detection. Journal of Biomedical Optics 13(1), 014017–014017 (2008).
Edner, H., Ragnarson, P., Spännare, S. & Svanberg, S. Differential optical absorption spectroscopy (DOAS) system for urban atmospheric pollution monitoring. Applied Optics 32(3), 327–333 (1993).
Schiff, H. I., Mackay, G. I. & Bechara, J. The use of tunable diode laser absorption spectroscopy for atmospheric measurements. Research on Chemical Intermediates 20(35), 525–556 (1994).
Bowling, D. R., Sargent, S. D., Tanner, B. D. & Ehleringer, J. R. Tunable diode laser absorption spectroscopy for stable isotope studies of ecosystematmosphere CO 2 exchange. Agricultural and Forest Meteorology 118(1), 1–19 (2003).
Mehrotra, R. Infrared Spectroscopy, Gas Chromatography/Infrared in Food Analysis. John Wiley & Sons, Ltd (2000).
Nicolai, B. M. et al. Nondestructive measurement of fruit and vegetable quality by means of NIR spectroscopy: A review. Postharvest Biology and Technology 46(2), 99–118 (2007).
Giovannetti, V., Lloyd, S. & Maccone, L. Advances in quantum metrology. Nature Photonics 5(4), 222–229 (2011).
DemkowiczDobrzański, R., Jarzyna, M. & Kołodyński, J. Quantum Limits in Optical Interferometry. Progress in Optics 60, 345–435 (2015).
Braun, D. et al. Quantum enhanced measurements without entanglement. arXiv preprint arXiv:1701.05152 (2017).
Monras, A. & Paris, M. G. Optimal quantum estimation of loss in bosonic channels. Physical Review Letters 98(16), 160401 (2007).
Adesso, G., Dell’Anno, F., De Siena, S., Illuminati, F. & Souza, L. A. M. Optimal estimation of losses at the ultimate quantum limit with nonGaussian states. Physical Review A 79(4), 040305 (2009).
Brida, G. et al. Experimental realization of a lownoise heralded singlephoton source. Optics Express 19(2), 1484–1492 (2011).
Krapick, S. et al. An efficient integrated twocolor source for heralded single photons. New Journal of Physics 15, 033010 (2013).
Whittaker, R. et al. Absorption spectroscopy at the ultimate quantum limit from singlephoton states. New Journal of Physics 19(2), 023013 (2017).
SabinesChesterking, J. SubShotNoise Transmission Measurement Enabled by Active FeedForward of Heralded Single Photons. Physical Review Applied 8, 014016 (2017).
Jakeman, E. & Rarity, J. G. The use of pair production processes to reduce quantum noise in transmission measurements. Optics Communications 59(3), 219–223 (1986).
Tapster, P., Seward, S. & Rarity, J. Subshotnoise measurement of modulated absorption using parametric downconversion. Physical Review A 44, 3266 (1991).
Hayat, M. M., Joobeur, A. & Saleh, B. E. Reduction of quantum noise in transmittance estimation using photoncorrelated beams. The Journal of the Optical Society of America A 16, 348–358 (1999).
Moreau, P. A. et al. Demonstrating an absolute quantum advantage in direct absorption measurement. Scientific Reports 7, 6256 (2017).
Brida, G., Genovese, M. & RuoBerchera, I. Experimental realization of subshotnoise quantum imaging. Nature Photonics 4(4), 227–230 (2010).
Brida, G., Genovese, M., Meda, A. & RuoBerchera, I. Experimental quantum imaging exploiting multimode spatial correlation of twin beams. Physical Review A 83, 033811 (2011).
Samantaray, N., RuoBerchera, I., Meda, A. & Genovese, M. Realisation of the first sub shot noise wide field microscope. Light: Science & Applications 6, e17005 (2017).
Genovese, M. Real applications of quantum imaging. Journal of Optics 18, 073002 (2016).
Zhang, Z., Mouradian, S., Wong, F. N. C. & Shapiro J. H. Entanglementenhanced sensing in a lossy and noisy environment. Physical Review Letter 114, 110506 (2015).
Lopaeva, E. D. et al. Experimental Realization of Quantum Illumination. Physical Review Letter 110, 153603 (2013).
Pooser, R. C. & Lawrie, B. Ultrasensitive measurement of microcantilever displacement below the shotnoise limit. Optica 2, 393–399 (2015).
Clark, J. B., Zhou, Z., Glorieux, Q., Marino, M. A. & Lett, P. D. Imaging using quantum noise properties of light. Optics Express 20(15), 17050 (2012).
Brida, G. et al. Systematic analysis of signaltonoise ratio in bipartite ghost imaging with classical and quantum light. Physical Review A 83, 063807 (2011).
Pirandola, S. Quantum Reading of a Classical DigitalMemory. Physical Review Letter 106, 090504 (2011).
Lawrie, B. J., Evans, P. G. & Pooser, R. C. Extraordinary optical transmission of multimode quantum correlations via localized surface plasmons. Physical Review Letter 110, 156802 (2013).
Pooser, R. C. & Lawrie, B. Plasmonic trace sensing below the photon noise limit. ACS Photonics 3(1), 8–13 (2016).
Meda, A. et al. Photonnumber correlation for quantum enhanced imaging and sensing. Journal of Optics 19, 094002 (2017).
Glorieux, Q., Guidoni, L., Guibal, S., Likforman, J. P. & Coudreau, T. Quantum correlations by fourwave mixing in an atomic vapor in a nonamplifying regime: Quantum beam splitter for photons. Physical Review A 84, 053826 (2011).
Embrey, C. S., Turnbull, M. T., Petrov, P. G. & Boyer, V. Observation of Localized MultiSpatialMode Quadrature Squeezing. Physical Review X 5, 031004 (2015).
Cao, L. et al. Experimental observation of quantum correlations in fourwave mixing with a conical pump. Optics Letter 42(7), 1201–1204 (2017).
Boyer, V., Marino, A. M., Pooser, R. C. & Lett, P. D. Entangled images from fourwave mixing. Science 321, 544 (2008).
Heidmann, A., Horowicz, R. J., Reynaud, S., Giacobino, E. & Fabre, C. Observation of quantum noise reduction on twin laser beams. Physical Review Letters 59, 2555–2557 (1987).
Mertz, J., Heidmann, A., Fabre, C., Giacobino, E. & Reynaud, S. Observation of highintensity subpoissonian light using an optical parametric oscillator. Physical Review Letters 64, 2897 (1990).
Agafonov, I. N. et al. Absolute of photodetectors: photocurrent multiplication versus photocurrent subtraction. Optics Letters 36(8), 1329–1331 (2011).
Bondani, M., Allevi, A., Zambra, G., Paris, M. & Andreoni, A. Subshotnoise photonnumber correlation in a mesoscopic twin beam of light. Physical Review A 76, 013833 (2007).
Iskhakov, T. S. et al. Heralded source of bright multimode mesoscopic subPoissonian light. Optics Letters 41, 2149–2152 (2016).
Jedrkiewicz, O. et al. Detection of SubShotNoise Spatial Correlation in HighGain Parametric Down Conversion. Physical Review Letter 93, 243601 (2004).
Brida, G. et al. Measurement of sub shotnoise spatial correlations without background subtraction. Physical Review Letter 102, 213602 (2009).
Blanchet, J. L., Devaux, F., Furfaro, L. & Lantz, E. Measurement of subshotnoise correlations of spatial fluctuations in the photoncounting regime. Physical Review Letter 101, 233604 (2008).
Brambilla, E., Caspani, L., Jedrkiewicz, O., Lugiato, L. A. & Gatti, A. Highsensitivity imaging with multimode twin beams. Physical Review A 77(5), 053807 (2008).
Monras, A. & Illuminati, F. Measurement of damping and temperature: Precision bounds in Gaussian dissipative channels. Physical Review A 83(1), 012315 (2011).
Palms, J. M., Rao, P. V. & Wood, R. E. A Fano factor measurement for silicon using low energy photons. Nuclear Instruments and Methods 76(1), 59–60 (1969).
Lvovsky, A. I. Squeezed light. arXiv:1401.4118v1 (2014).
Spedalieri, G., Braunstein, S. L. & Pirandola, S. Thermal Quantum Metrology, arXiv:1602.05958 (2016).
Meda, A. et al. Absolute calibration of a chargecoupled device camera with twin beams. Applied Physics Letter 105, 101113 (2014).
Brida, G., Degiovanni, I. P., Genovese, M., Rastello, M. L. & RuoBerchera, I. Detection of multimode spatial correlation in PDC and application to the absolute calibration of a CCD camera. Optics Express 18, 20572–20584 (2010).
Avella, A., RuoBerchera, I., Degiovanni, I. P., Brida, G. & Genovese, M. Absolute calibration of an EMCCD camera by quantum correlation, linking photon counting to the analog regime. Optics Letters 41, 1841–4 (2016).
Acknowledgements
This work has received funding from the European Union’s Horizon 2020 and the EMPIR Participating States in the context of the project 17FUN01 BeCOMe. The Authors thank I.P. Degiovanni, S. Pirandola and C. Lupo for elucidating discussion and N. Samantaray for his help in setting the preliminary phase of the experiment.
Author information
Authors and Affiliations
Contributions
I.R.B. and A.M. conceived the idea of the experiment, which were designed and discussed with input from all authors. I.R.B. and E.L. developed the theoretical model. E.L. and A.M. realized the experimental setup and collected the data in INRIM quantum optics labs (coordinated by M.G.). All authors discussed the results and contributed to the writing of the paper. All authors reviewed the manuscript.
Corresponding author
Ethics declarations
Competing Interests
The authors declare no competing interests.
Additional information
Publisher's note: Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The images or other third party material in this article are included in the article’s Creative Commons license, unless indicated otherwise in a credit line to the material. If material is not included in the article’s Creative Commons license and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this license, visit http://creativecommons.org/licenses/by/4.0/.
About this article
Cite this article
Losero, E., RuoBerchera, I., Meda, A. et al. Unbiased estimation of an optical loss at the ultimate quantum limit with twinbeams. Sci Rep 8, 7431 (2018). https://doi.org/10.1038/s4159801825501w
Received:
Accepted:
Published:
DOI: https://doi.org/10.1038/s4159801825501w
This article is cited by

Quantum enhanced noninterferometric quantitative phase imaging
Light: Science & Applications (2023)

Theoretical studies on quantum imaging with timeintegrated singlephoton detection under realistic experimental conditions
Scientific Reports (2022)

Overcoming detection loss and noise in squeezingbased optical sensing
npj Quantum Information (2021)

Twin beam quantumenhanced correlated interferometry for testing fundamental physics
Communications Physics (2020)

Imaging with quantum states of light
Nature Reviews Physics (2019)
Comments
By submitting a comment you agree to abide by our Terms and Community Guidelines. If you find something abusive or that does not comply with our terms or guidelines please flag it as inappropriate.