Abstract
Realtime quantitative phase imaging has tremendous potential in investigating live biological specimens in vitro. Here we report on a wideband sensitivityenhanced interferometric microscopy for quantitative phase imaging in real time by employing two quadriwave lateral shearing interferometers based on randomly encoded hybrid gratings with different lateral shears. Theoretical framework to analyze the measurement sensitivity is firstly proposed, from which the optimal lateral shear pair for sensitivity enhancement is also derived. To accelerate the phase retrieval algorithm for realtime visualization, we develop a fully vectorized pathindependent differential leveling phase unwrapping algorithm ready for parallel computing, and the framerate for retrieving the phase from each pair of two 4 mega pixel interferograms is able to reach 47.85 frames per second. Experiment results demonstrate that the wideband sensitivityenhanced interferometric microscopy is capable of eliminating all the periodical error caused by spectral leaking problem and reducing the temporal standard deviation to the half level compared with phase directly retrieved by the interferogram. Due to its high adaptability, the wideband sensitivityenhanced interferometric microscopy is promising in retrofitting existing microscopes to quantitative phase microscopes with high measurement precision and realtime visualization.
Similar content being viewed by others
Introduction
Quantitative phase microscopy (QPM), which provides additional phase information about the refractive index distribution of transparent specimens, has drawn much attention in both optical and biomedical research. The main advantage of quantitative phase microscopy (QPM) over conventional intensity imaging or fluorescence microscopy is that it is labelfree and no dye or fluorescent protein is required to enhance the contrast of the microscopic image, which makes it possible to study live biological specimens in vitro ^{1,2,3,4,5}.
Ptychography^{5,6,7,8} is one of the emerging QPM techniques that can retrieve the phase and amplitude of the scattered wavefield when a coherent illumination beam passes through the specimen under test. The diffraction patterns of the specimen are captured by a CCD camera and the phase retrieval algorithm is carried out based on the extended ptychographic iterative engine (ePIE). Ptychographic microscope can obtain high contrast images free of halo artefact using low magnification microscopic objectives^{9, 10}. However, both the acquisition for the tile scan to generate overlapping diffraction patterns and the computation of the ePIE algorithm takes a relatively long time, so it is currently not available to realtime applications. Different from ptychography, the transportofintensity equation (TIE) technique employs focus and defocus images to retrieve the phase variation of the wavefront. It can be applied to a conventional intensity microscope directly and the acquisition rate can be set up to 15Hz if an electrically tunable fluidic lens is employed to vary the equivalent focal length of the optical system automatically^{11}. By introducing partially coherent light source such as LED to avoid speckle noise, the accuracy and signaltonoise ratio (SNR) of the TIE technique can be improved and the rootmeansquare (RMS) error of the reconstructed phase image will decrease to 12%^{12}. Nevertheless, its accuracy still need to be improved if compared with interferometric methods like diffraction phase microscope (DPM)^{1, 13,14,15}, digital holographic microscope (DHM)^{16,17,18,19,20,21} and the offaxis τ interferometer^{22, 23}. For measurement precision, the interferometric technique is now able to reach an optical path stability of about 10 pm with an Adimec camera having larger pixel full well capacity^{24}. Meanwhile, to shorten the computation latency, Nvidia’s compute unified device architecture (CUDA) is recently employed in the DHM to provide realtime processing at 34.8 frames per second (fps) for 2048 × 2048 pixel holograms^{25}. As a result, with interferometric microscopic techniques, the dynamics of red blood cells (RBCs) was studied^{26, 27} and the isovolumetric cell deformation through membrane electromotility has also been investigated^{28}.
Most of the interferometric microscopes, including the DHM, the DPM and the offaxis τ interferometer, are based on MachZehnder interferometry (MZI) or point diffraction interferometry (PDI), in which the interferograms are obtained due to the interference between the reference beam and the sample beam. The precision of these interferometers is of significant relevance to the flatness of the reference wavefront in MZI or the fabrication quality and the actual position and orientation of the pinhole in PDI. Thus meticulous alignment of the optical paths should be carried out carefully to ensure high precision when they will be mounted on an existing microscope. In contrast, without the reference beam, lateral shearing interferometers (LSIs) usually have a more compact structure and a higher tolerance to adapt into other optical systems because of its selfreference feature. Quadriwave lateral shearing interferometers (QWLSIs) are capable of transient phase imaging by acquiring shearing wavefronts in two orthogonal shearing directions in only one single interferogram, and exhibit great suppression over environmental vibration as their optical layouts are designed to be rigorously commonpath. The modified Hartmann mask (MHM) for QWLSI consists of a cross grating whose duty ratio is 2/3 and a phase chessboard. The central four orders contains 87% of the output energy^{29, 30}, so it is firstly introduced in QWLSI and has been employed for quantitative phase microscopy of living cells as well^{31, 32}. However, the interferogram of monochromatic light beams diffracted by the MHM varies periodically as the distance from the MHM to the image plane increases, and the available lateral shear amounts related to this observation distance are also discrete. As the lateral shear amount is closely related to both the dynamic range and the testing sensitivity, a flexible shear amount will be beneficial for the quantitative phase imaging in need of higher sensitivity and precision. In our previous work, a randomly encoded hybrid grating lateral shearing interferometer (REHGLSI), which is a novel QWLSI based on the randomly encoded hybrid grating (REHG), has been proposed^{33}. The REHG consists of a randomly encoded binary amplitude grating and a phase chessboard. The randomly encoded binary amplitude grating simulates the transmittance of the ideal quadriwave grating with tiny encoded masking pixels, while the phase chessboard is a transparent substrate on which the phase modulations of 0 and π arrange alternatively. Its Fraunhofer diffractions only contains four orders in the center, which are respectively the +1 and −1 orders in two orthogonal directions. There is no periodical Talbot effect affecting the observation distance^{34} and a continuously variable lateral shear can be obtained with the REHG, which, as a result, makes it possible to control the measurement sensitivity.
In this paper, we describe a novel wideband sensitivityenhanced interferometric microscopy (WSEIM) based on quadriwave lateral shearing interferometry by employing two REHGs with different lateral shears to enhance its measurement sensitivity and the SNR (Fig. 1). Different from most of the conventional interferometric microscopes which are in need of an ideal reference beam, the WSEIM is selfreferenced and only the wavefront signal not the shot noise in quadriwave lateral shearing interferometry is duplicated and sheared to offer shearing wavefronts which may be even larger than the original wavefront signal. Moreover, to obtain the Fourier spectrum of the wavefront under test, the Fourier spectrums of shearing wavefronts in both x and y directions will be integrated by the leastsquare fitting method, which again reduces the contribution of stochastic noise to the testing results. The detailed theoretical framework of the wideband sensitivity enhancement is presented in Methods section. To meet the requirements of realtime application, a vectorized differential leveling phase unwrapping (DLPU) algorithm is proposed in this paper and is implemented on Nvidia’s CUDA platform. In fact, although the time complexity of the phase retrieval in WSEIM is more than 4fold larger than that of the phase retrieval in DHMs, our phase retrieval algorithm for the WSEIM is able to reach a framerate of 47.85 fps by improving the performance of the phase unwrapping algorithm, which usually occupies 50% computation time in conventional phase retrieval algorithm for DHMs on the CPU platform^{35}. In addition, due to the rigorously commonpath feature of QWLSI, no isolation of environmental vibrations is required for our WSEIM. And the retrofit of an existing microscope from the conventional intensity mode to a WSEIM for phase imaging is also very simple, just by inserting two REHGs in front of the CCD cameras. So the WSEIM is highly adaptable for a plenty of potential applications, including quantitative phase imaging for microfluidic and biomedical research.
Methods
Wideband sensitivityenhanced interferometric microscopy (WSEIM)
WSEIM originates from the quadriwave lateral shearing interferometer (QWLSI) based on the randomly encoded hybrid grating (REHG). As is shown in Fig. 1, a collimator is employed to collect the light emitted from an ordinary LED chip (LR CP7P, SSL 80, OSRAM). The typical dominant wavelength of this LED is 623 nm, and the spectral bandwidth at half maximum intensity is 18 nm. The partially coherent feature of the light source helps mitigate speckle noise resulting from parasitic reflections inside the optics and specimens^{12, 13}. Then the light beam is spatially filtered by a combination of a pinhole with a diameter of 100 μm and two lenses L1, L2 to generate a plane wave probe. The refractive index distribution of the specimen will cause wavefront distortion in the plane wave probe, and a 4f telescope system is used to project this wavefront distortion to the image plane. For microscopic imaging of tiny specimens, highNA condenser lens and objective lens with high magnification are employed in the 4f telescope system; otherwise lenses with relatively large focal lengths can be employed as the condenser lens and the objective lens to offer a wide field of view. The probe is then divided into two beams by a beam splitter. The reflected beam is directly diffracted into four replicas by a REHG G1; but the transmitted beam will impinge on a mirror M2 first, which corrects the mirror effect between the reflected beam and the transmitted beam, and then diffract into four replicas by a second REHG G2. Both of the pitches of these two gratings are designed to be 30 μm to ensure a spatial resolution of 501 × 501 pixels in the final measurement results without the need of zero padding. The quadriwave lateral shearing interferograms formed by interference between the wavefront replicas are recorded by two CCD cameras (BM500GE, GigE Vision, JAI), whose image planes are placed in the positions conjugated to the object plane where the specimen locates. As the lateral shear between the wavefront replicas on CCD image plane is proportional to the observation distance from the grating to the image plane^{36}, two different lateral shears s and s′ can be achieved by controlling the distances between the gratings and their corresponding CCDs (Fig. 1b).
In optical metrology, it is generally believed that a large shear amount could help improve the measurement sensitivity and the SNR^{37}, as the wavefront aberrations of optical systems mostly concentrate in low spatial frequencies. As a matter of fact, the relationship between the lateral shear amount and the sensitivity is more complicated than simple monotonicity in quantitative phase imaging. Assuming that a wavefront with onedimensional sinusoidal phase distribution is under test, which is presented in red lines in Fig. 2. The spatial period of the wavefront is D and the measurement sensitivity relative to that of a interferometer with external reference like MZI or PDI is σ. When the wavefront replica shown in blue is slightly shifted to offer a lateral shear amount of 0.1D, the relative sensitivity can only be 0.618. But as the shear amount s increases to 0.5D, the shearing wavefront which is the difference between the original wavefront and its laterally sheared replica will be amplified and the relative sensitivity can reach its maximum value of 2, which will remarkably improves the signal noise ratio (SNR) and also the measurement precision. For a single sinusoidal wavefront under test, the optimal shear amount for sensitivity enhancement equals half of the period of that wavefront.
Actual wavefronts always contain numerous sinusoidal components with different spatial frequencies. Assume that the wavefront under test is W(x, y), which can be decomposed as follows using inverse Fourier transform,
By assuming the bandwidth of the sampling system [−Ω, Ω], the wavefront under test W(x, y) can be expressed as an integral of its Fourier spectrum F(u, v) multiplied by an exponential function,
In quadriwave lateral shearing interferometry, the wavefronts measured directly are shearing wavefronts in the x and y directions W _{ x }(x, y) and W _{ y }(x, y), which are,
where s is the lateral shear. The Fourier transform of the shearing wavefronts in the x and y directions F _{ x }(u, v) and F _{ y }(u, v) can be derived from Eqs (2)–(4),
Note that some spectrums are amplified in the shearing wavefront spectrums while some decrease even to 0. The Fourier spectrum of the wavefront under test F(u, v) can then be reconstructed using these shearing wavefront spectrums,
Assume that the spectrum uncertainties of the shearing wavefronts in the x and y directions are Δ_{ x }(u, v) and Δ_{ y }(u, v), respectively. The overall uncertainty of the Fourier spectrum of the wavefront to be retrieved is,
where s is the lateral shear of the QWLSI. Since the shearing wavefronts are measured directly using the same spatial carrier modulation method as the MZI or PDI, their uncertainties are also related to the photon shot noise^{24}. The diffraction efficiency of the REHG is 22.17% on its central four orders, and the illumination power should be adjusted so that the maximum intensity of the interferogram captured by CCDs will get close to saturation. If the same type of CCD camera with the same full electron well capacity is employed, the spectrum uncertainties of the shearing wavefronts in the x and y directions Δ_{ x }(u, v) and Δ_{ y }(u, v) will be equal to the uncertainty Δ_{0}(u, v) of a conventional interferometric microscope under the same intensity of interferogram. And the relationship between the uncertainty of a QWLSI Δ(u, v) and the uncertainty of a conventional interferometric microscope Δ_{0}(u, v) is obtained,
Therefore the sensitivity σ(u, v) of a QWLSI relative to that of a conventional interferometric microscope can be defined as,
Compared with a conventional interferometric microscope with external reference, the uncertainty of a QWLSI is only \(1/2\sqrt{2}\) at the frequencies like,
At these frequencies, not the noise but only the signal is amplified by a factor of 2 to achieve higher sensitivity and SNR. However, the sensitivity of the QWLSI goes down again near the following frequencies,
Due to the noise in the measured shearing wavefronts, the reduction of sensitivity at these frequencies usually results in a large error in the Fourier spectrum at these frequencies, which is also referred to as spectral leaking problem. This problem can be solved by introducing another lateral shear^{38,39,40}. As is shown in Fig. 1a, if the beam under test is firstly divided by a beam splitter and one more QWLSI interferogram with a different lateral shear s′ is obtained, the relative sensitivity σ(u, v) will be obtained as follows taking into consideration the increase of the shot noise by a factor of \(\sqrt{2}\),
By altering different pairs of lateral shears (s, s′), the sensitivity performance of the multishear QWLSI will also change. To enhance the sensitivity of the multishear QWLSI on a wide band of frequencies, the variation of σ′(u, v) should be gentle given the pair of lateral shears, which means the standard deviation of σ′(u, v) in the actual frequency range [−Ω, Ω] × [−Ω, Ω] should be minimized. This criterion and the detailed analysis of the QWLSI sensitivity above lay a theoretical framework in finding an optimal lateral shear pair to achieve wideband sensitivity enhancement.
Differential leveling phase unwrapping algorithm for realtime applications
The phase retrieval algorithm for a QWLSI consists of two steps, the extraction of shearing wavefronts in the x and y directions from the quadriwave lateral shearing interferogram and the final retrieval of the wavefront under test with these shearing wavefronts^{41}. The first step is similar with the phase retrieval in the DPM^{14, 42} and the DHM^{15}, but one more wavefront which is the shearing wavefront in the y direction needs to be processed in the QWLSI. The WSEIM, which employs two QWLSIs with different lateral shears, is designed to retrieve the wavefront under test with two quadriwave lateral shearing interferograms, and both of them contain two shearing wavefronts. Therefore, the time complexity of the phase retrieval algorithm for WSEIM is at least 4fold larger than that of the phase retrieval algorithm for DHMs. The detailed phase retrieval steps in Fig. 3 can be generalized as the following three operations, fast Fourier transform (FFT), inverse fast Fourier transform (IFFT) and the phase unwrapping. Among them, both of the FFT and IFFT algorithms with great performance have been developed on Nvidia’s CUDA, but the widely used Goldstein’s branchcut (GOLD)^{15, 43} and noncontinuous qualityguided path (NCQUAL) phase unwrapping algorithm^{44, 45} can hardly be implemented here by parallel computation because they depend on sequential searching and joining path, though noncontinuous. A robust and fully vectorized pathindependent phase unwrapping algorithm with elementwise matrix operations will then help realize all the potentials of the parallel computing system. As a result, the differential leveling phase unwrapping (DLPU) algorithm is proposed to solve this problem.
In the DLPU illustrated in Fig. 3, the wrapped phase is firstly differentiated into the phase differentials in the x and y directions, respectively. Due to the 2π phase jumps in the wrapped phase, some pixels of the differentials will be 2π and others will be −2π. The next step which is leveling is to restore these pixels to the correct differentials by subtracting or adding 2π. Then a phase distribution can be retrieved by the lateral shearing phase retrieval algorithm based on the discrete Fourier transform^{46} using these leveled differentials. To avoid the error produced by this phase retrieval, however, this phase distribution cannot be regarded as the final unwrapped result, but only serves as a reference for the following unwrapping. Tilts are then removed from the wrapped phase and the reference phase, and the difference between these two phases at every pixel determines how many 2π jumps the wrapped phase should be added or subtracted at this pixel. The phase unwrapped can finally be obtained by adding or subtracting the corresponding 2π jumps at each pixel to the wrapped phase. The lateral shearing phase retrieval in the DLPU only contains FFT, IFFT, adding, elementwise multiplication and division of matrix. Other operations in the DLPU are also independent among the pixels and ready for parallel computing. So the advantage of the DLPU algorithm is that all operations are now vectorized which can be implemented with high performance on a parallel computing platform.
Results and Discussion
Wideband sensitivityenhanced interferometric microscopy (WSEIM)
For conventional single shear QWLSI, although there are more than one half of the spatial frequencies where the sensitivities relative to the interferometer with external reference are greater than 1, the relative sensitivities can also be as low as zero at some frequencies, as detailed in Methods Section. To enhance the sensitivity on a wide band of spatial frequencies, two quadriwave lateral shearing interferograms with different lateral shears are introduced to complement each other’s zero points on the spectrums. In fact, the optimal pair of lateral shears comes with the equilibrium of the spectral sensitivities, which, in others words, means the standard deviation (SD) of spectral sensitivity distribution should be minimized. As a result, a map of the SD values of spectral sensitivity distribution when the two lateral shear ratio β and β′, which were respectively the ratios of lateral shear amounts s and s′ to the size 7.066 mm of the beam incident on the CCD image plane, change from 1% to 3% was firstly obtained (Fig. 4a). The twodimensional spatial frequencies were calculated using the dimension of 7.066 mm × 7.066 mm and the resolution of 501 × 501 pixels. The diagonal of this map represented the circumstances when the two lateral shears equaled to each other, which could also be regarded as single shear QWLSI. In these cases, the spectral sensitivities were periodical and multiple zero points were located in the sensitivity distribution (Fig. 4b). When the lateral shears went slightly different from each other, there would be a decrease in the SD of spectral sensitivity distribution. In the SD map, two valleys lied along each side of the diagonal peak, and an optimal lateral shear pair could be selected from these two sections. Figure 4c shows the spectral sensitivity distribution when the lateral shear ratio β and β′ were 1.88% and 1.57%, respectively. Only one point which was the zero frequency exhibited a sensitivity of zero and almost all the spectral relative sensitivities were over 1, compared with the interferometer with external reference. The average spectral relative sensitivity of our WSEIM was around 2, which resulted in the achievement of wideband sensitivity enhancement.
Realtime phase visualization using WSEIM
Realtime visualization demand that the computation latency should be less than the time interval between every two frames recorded by the camera. For CCDs (BM500GE, GigE Vision, JAI) employed in the experiment, the framerate of full frame is 15 fps, so the computation latency of phase retrieval should not exceed 66.6 ms. Although the fast and robust phase unwrapping algorithm based on noncontinuous qualityguided path^{44} (NCQUAL) proposed by Herráez et al. was employed, the processing of a 2048 × 2048 interferogram pair on CPU (Intel i76700H, 2.60 GHz) still took 1417.92 ms (Table 1), which is obviously not available for realtime applications. To accelerate the processing speed, open source ArrayFire library based on CUDA technology which could enable the parallel computing on GPU was introduced. The time cost for the FFT of a single 2048 × 2048 interferogram which was 117.45 ms on CPU had decreased to 1.15 ms on a laptop GPU (GeForce GTX 960 M, 640 CUDA cores), and could further reach 0.49 ms on a highperformance desktop GPU (GeForce GTX Titan Black Edition, 2880 CUDA cores). However, the NCQUAL algorithm for phase unwrapping could hardly be implemented on GPU in parallelism, and it became a handicap for realtime visualization. As a result, a fully vectorized pathindependent phase unwrapping algorithm with elementwise matrix operations named differential leveling phase unwrapping (DLPU) algorithm, which is detailed in Methods Section, was proposed and implemented on CUDA platforms. The time cost for phase unwrapping then decreased to 5.54 ms on the GTX 960M and to 3.77 ms on the GTX Titan Black. Moreover, lazy execution (LE) provided by ArrayFire library would automatically construct optimallysized kernels from the algorithms to be executed. Without the contribution of LE, the total processing time was slightly over the sum of the duration of each step. But when LE was switched on, an even less total processing time could be finally achieved. The phase retrieval framerate of our WSEIM was 24.81 fps on the laptop GTX 960 M and 47.85 fps on the highend GTX Titan Black, which well met the realtime requirement over the framerate of 15 fps.
Error analysis and calibration
The error sources of the WSEIM include the wavefront distortion of the probe, the wavelength difference due to the partial coherent light illumination, the translational offset between two cameras and the vibration in the two beams transmitted or reflected by the beam splitter. The wavefront distortion of the probe can be easily eliminated by measuring the background phase image and storing it as the systematic error data for the subtraction in the future measurement. For the wavelength difference, one of the advantages of the ideal QWLSI is that it is theoretically achromatic^{30}. Although the difference in the lateral shear amounts of different wavelengths will result in the difference between shearing wavefronts, it will be compensated by the wavelength itself in the phases of these shearing wavefronts, which are exactly what we obtained from a quadriwave lateral shearing interferogram at the first stage. In detail, the wavefront under test W(x, y) is caused by the optical path difference (OPD) of the sample, which is independent of the wavelength. As the lateral shear for QWLSI measurement is usually very small, the shearing wavefronts in Eqs (3) and (4) can be expressed as follows by firstorder approximation,
where the lateral shear amount s can be derived as \(2\sqrt{2}l\lambda /d\), l is the distance from the grating to the image plane and d is the grating pitch. Then the phases of the shearing wavefronts to form and to be retrieved from the interferogram can be obtained as,
which show no dependence on the wavelength actually. For REHG which employs a phase chessboard designed for the specific center wavelength to simulate the 0 and π phase steps, unexpected diffraction orders may emerge by broadband wavelength illumination, but as the phase step is only changing from 0.986π to 1.014π using the red LED chip (LR CP7P, SSL 80, OSRAM) in our WSEIM, the retrieval error caused by this phase step change can also be neglected^{34}.
For the analysis of the translational offset and the vibration, another advantage of employing the QWLSIs is that each of the two beams transmitted or reflected by the beam splitter for quadriwave lateral shearing interference is commonpath, so either the translational offset or the vibration doesn’t affect the test results at the sensitive interferometric level, all they will influence is the lateral shift of the phase images obtained by the two QWLSIs, which can be also regarded as the same lateral shift of the brightfield images on the two cameras when the REHGs haven’t been set. As a result, the alignment of the two QWLSIs always begins with the alignment of the only two cameras with no REHGs in front, and the translational offset can be reduced by laterally moving one camera on a precise translational stage to match the two brightfield images. However, this error can hardly be fully eliminated when the lateral shift is under one camera pixel. For quantitative analysis, circular phase targets with the same diameter of 12 μm and the same etching depth of 136 nm on a fused quartz substrate in modeling and also in reality were introduced (Fig. 5a). Figure 5b shows the simulation result of the relative phase retrieval error of this phase target due to the translational offset. When the translational offset between the cameras reached one pixel, the maximum error relative to the total range of the phase target was 1.127% and the relative RMS error was 0.049%, and the relationship between these errors and the offset were both linear. Vibration error, considered at the imaging level, can be regarded as the translational offset in the two arms changing with time. By measuring the actual phase target for 20 seconds, the translational offsets in the x and y directions due to the environmental vibration could be obtained from the centroid displacements of the binarized phase images retrieved from the two QWLSIs (Fig. 5c). The histogram of the offset distance taking both the translational offsets in the x and y directions into account shows that the largest group of offset distances ranged from 0.2 to 0.4 pixel, and their corresponding phase retrieval errors in Fig. 5b are also acceptable.
Characterization of etched substrate
To demonstrate the precision improvement of our WSEIM, a fused quartz substrate etched with the letter “s” was characterized and the testing results were later compared with the results from a Wyko NT9100 optical profiling system (Fig. 6). As the size of the letter was about 2.5 mm, two aplanatic lenses whose focal lengths are 90 mm were employed as the condenser lens and the objective lens to provide a wide field of view. In the shearing wavefronts in the x and y directions obtained in Fig. 6a,b, an amplification from the original amplitude of the wavefront signal to its double was observed, which is key to the sensitivity enhancement of WSEIM. The testing results of the etching depth of the letter is shown in Fig. 6c, where the OPD obtained by WSEIM has been converted into the etching depth z using the following equation,
where n is the refractive index of the fused quartz at the wavelength of 623 nm. In comparison with the testing result by the Wyko NT9100 in Fig. 6d, the detailed onedimensional etching depth distribution in the central cross sections illustrated in Fig. 6c are shown in Fig. 6e. As our WSEIM was designed to measure the wavefront distortion of the fused quartz substrate in transmittedlight configuration, which was different from the direct reflectedlight measurement of the etching depth by Wyko NT9100, there existed small differences whose standard deviation is 4.09 nm between the two results. Figure 6f exhibits the histogram of temporal standard deviations of phase which indicate the measurement sensitivities. Due to the relatively low brightness of the LED light filtered by the pinhole, the maximum intensity deviation between every two frames recorded by the CCD was as much as 27 in 8bit gray scale, while the maximum intensity of the interferogram was only 132 in 8bit gray scale. As a result, the temporal standard deviation of the shearing wavefronts ranged from 0.024 rad to 0.050 rad with a mean value of 0.034 rad, which also represents the temporal stability of conventional interferometric microscope with external reference employing the same CCD camera under the same intensity of interferogram. In contrast, the temporal standard deviation of the phase results by WSEIM only ranged from 0.013 rad to 0.021 rad, and their mean value of 0.017 rad was only half of the shearing wavefronts’ temporal standard deviation, which agreed well with the theoretical prediction that the average relative sensitivity of the WSEIM is around 2.
Quantitative phase imaging of the dynamics of RBCs
The purpose of realtime quantitative phase imaging based on the WSEIM is to observe and study the dynamics of living cells, as a result, samples of sluggishly flowing RBCs sandwiched in two cover glasses were tested by the WSEIM. Two oilimmersed microscopic objectives (UPLSAPO 100X, NA 1.4, Olympus) were employed to project high resolution images onto the CCD cameras through two REHGs. The resolution of the quadriwave lateral shearing interferograms was 2048 × 2048 pixels. The distances from the REHGs to the CCD image planes were adjusted so that the two lateral shear ratios β and β′ were set to 1.93% and 1.55%, respectively. This lateral shear pair were selected in the optimal section and was able to enhance the sensitivity on a wide band of frequencies. Comparison of the phase imaging results obtained by the WSEIM and conventional single shear QWLSI is shown in Fig. 7a. Due to the zero sensitivities at multiple frequencies, the SNR went down and the noise was then amplified by the small denominator in Eq. (7) around these frequencies, which resulted in a latticelike error emerged in the Fourier spectrum of the phase image obtained by single shear QWLSI, and also the corresponding periodical error in the spatial domain. In comparison, these errors caused by spectral leaking problem were eliminated in the testing result of the WSEIM, and the temporal standard deviation of background and the spatial noise were also greatly reduced (Fig. 7c,d). Moreover, one platelet and two microbes which were invisible by the single shear QWLSI could then be observed by the WSEIM. Clusters of acanthocytes and white blood cells, which contained more information on higher frequencies, were also imaged by the WSEIM to validate the robustness of our realtime phase retrieval algorithm, especially the performance of the DLPU algorithm (Fig. 7b). No obvious discontinuity was seen in the phase image and details was well presented in a resolution of 501 × 501 pixels without zero padding. Figure 8 shows the timeelapsed dynamic measurement results of the RBCs flowing in blood plasma in 15 seconds. To study the membrane fluctuation of the RBCs in the dashed box in Fig. 8a, their OPD fluctuation in a time interval of 3 seconds was measured as well (Fig. 8b).
Conclusion
We have presented wideband sensitivityenhanced interferometric microscopy (WSEIM) based on quadriwave lateral shearing interferometry (QWLSI) for realtime quantitative phase imaging. Theoretical framework that calculate the spectral relative sensitivities of a QWLSI is firstly proposed by introducing Fourier analysis into resolving the relationship between the shearing wavefronts and the original wavefront signal, and the sensitivity enhancement on a wide band of spatial frequencies can be achieved with the optimal selection of the lateral shear pair, which comes with the minimum standard deviation of spectral relative sensitivity. Agreeing with the theoretical prediction, the WSEIM, which combines two QWLSIs with the optimal lateral shear pair together, can successfully reduce the temporal standard deviation to its half and eliminate all the periodical error caused by spectral leaking problem.
To achieve the goal of realtime visualization, a fully vectorized phase retrieval algorithm, including the novel differential leveling phase unwrapping (DLPU) algorithm, is developed for the computation on parallel computing platforms. Utilizing the similar lateral shearing phase retrieval algorithm with a lateral shear of one pixel instead in the DLPU algorithm, no unwrapping path is ever needed and the processes will be only FFT, IFFT and other matrix operations, which have already been optimized by the Nvidia’s CUDA community. Therefore, our phase retrieval algorithm for the WSEIM is able to reach a framerate of 24.81 fps on a laptop GPU and a framerate of 47.85 fps on a highend desktop GPU. As the first three steps of our phase retrieval algorithm in WSEIM is similar to those of the DHMs, it will also benefit the acceleration of the phase retrieval process in conventional interferometric microscopes with external reference.
The WSEIM with realtime visualization provides an easy access to retrofitting existing brightfield microscopes into quantitative phase microscopes just by inserting two randomly encoded hybrid gratings (REHGs) in front of the CCD image planes. No scanning system for phase shifting is required and the WSEIM can be implemented without any isolation of environmental vibration, as its optical path for quadriwave lateral shearing interference is rigorously commonpath. Thanks to the sensitivity enhancement of WSEIM, a temporal OPD standard deviation of 1.32 nm can also be obtained even though the largest deviation between every two frames recorded by the CCD is as much as 20.5% relative to the maximum intensity due to the lack of photons. If the brightness can be further increased using a shortcoherence solidstate laser source, such as the Ti:Sapphire laser^{47}, and CCDs with larger pixel full well capacity^{24} are going to be employed at the same time, the WSEIM will then fully realize its potentials in sensitivity enhancement and offer the opportunity of new discoveries through realtime observation of the phase variations and fluctuations of living cells.
References
Popescu, G., Ikeda, T., Dasari, R. R. & Feld, M. S. Diffraction phase microscopy for quantifying cell structure and dynamics. Opt. Lett. 31, 775–777 (2006).
Chen, C. L. et al. Deep Learning in Labelfree Cell Classification. Sci. Rep. 6, 21471, doi:10.1038/srep21471 (2016).
Sugiyama, N. et al. Labelfree characterization of living human induced pluripotent stem cells by subcellular topographic imaging technique using fullfield quantitative phase microscopy coupled with interference reflection microscopy. Biomed. Opt. Express 3, 2175–2183, doi:10.1364/BOE.3.002175 (2012).
Shaked, N. T., Satterwhite, L. L., Bursac, N. & Wax, A. Wholecellanalysis of live cardiomyocytes using widefield interferometric phase microscopy. Biomed. Opt. Express 1, 706–719, doi:10.1364/BOE.1.000706 (2010).
Marrison, J., Räty, L., Marriott, P. & O’Toole, P. Ptychography–a label free, highcontrast imaging technique for live cells using quantitative phase information. Sci. Rep. 3, 2369, doi:10.1038/srep02369 (2013).
Humphry, M. J., Kraus, B., Hurst, A. C., Maiden, A. M. & Rodenburg, J. M. Ptychographic electron microscopy using highangle darkfield scattering for subnanometre resolution imaging. Nat. Commun. 3, 730, doi:10.1038/ncomms1733 (2012).
Thibault, P. & Menzel, A. Reconstructing state mixtures from diffraction measurements. Nature 494, 68–71 (2013).
Suman, R. et al. Labelfree imaging to study phenotypic behavioural traits of cells in complex cocultures. Sci. Rep. 6, 22032, doi:10.1038/srep22032 (2016).
Ou, X., Horstmeyer, R., Yang, C. & Zheng, G. Quantitative phase imaging via Fourier ptychographic microscopy. Opt. Lett. 38, 4845–4848 (2013).
Ou, X., Horstmeyer, R., Zheng, G. & Yang, C. High numerical aperture Fourier ptychography: principle, implementation and characterization. Opt. Express 23, 3472–3491, doi:10.1364/OE.23.003472 (2015).
Zuo, C., Chen, Q., Qu, W. & Asundi, A. Highspeed transportofintensity phase microscopy with an electrically tunable lens. Opt. Express 21, 24060–24075, doi:10.1364/OE.21.024060 (2013).
Rodrigo, J. A. & Alieva, T. Rapid quantitative phase imaging for partially coherent light microscopy. Opt. Express 22, 13472–13483, doi:10.1364/OE.22.013472 (2014).
Nguyen, T. H., Edwards, C., Goddard, L. L. & Popescu, G. Quantitative phase imaging with partially coherent illumination. Opt. Lett. 39, 5511–5514 (2014).
Pham, H. V., Edwards, C., Goddard, L. L. & Popescu, G. Fast phase reconstruction in white light diffraction phase microscopy. Appl. Opt. 52, A97–A101 (2013).
Bhaduri, B. et al. Diffraction phase microscopy: principles and applications in materials and life sciences. Adv. Opt. Photon. 6, 57–119 (2014).
Cuche, E., Bevilacqua, F. & Depeursinge, C. Digital holography for quantitative phasecontrast imaging. Opt. Lett. 24, 291–293 (1999).
Carl, D., Kemper, B., Wernicke, G. & von Bally, G. Parameteroptimized digital holographic microscope for highresolution livingcell analysis. Appl. Opt. 43, 6536–6544 (2004).
Mann, C. J., Yu, L., Lo, C.M. & Kim, M. K. Highresolution quantitative phasecontrast microscopy by digital holography. Opt. Express 13, 8693–8698, doi:10.1364/OPEX.13.008693 (2005).
Paturzo, M. et al. Superresolution in digital holography by a twodimensional dynamic phase grating. Opt. Express 16, 17107–17118, doi:10.1364/OE.16.017107 (2008).
Weijuan, Q., Yingjie, Y., Choo, C. O. & Asundi, A. Digital holographic microscopy with physical phase compensation. Opt. Lett. 34, 1276–1278 (2009).
Tsang, P. W. M., Cheung, K. W. K., Kim, T., Kim, Y. S. & Poon, T. C. Fast reconstruction of sectional images in digital holography. Opt. Lett. 36, 2650–2652 (2011).
Girshovitz, P. & Shaked, N. T. Compact and portable lowcoherence interferometer with offaxis geometry for quantitative phase microscopy and nanoscopy. Opt. Express 21, 5701–5714, doi:10.1364/OE.21.005701 (2013).
Girshovitz, P. & Shaked, N. T. Doubling the field of view in offaxis lowcoherence interferometric imaging. LightSci. Appl 3, 9, doi:10.1038/lsa.2014.32 (2014).
Hosseini, P. et al. Pushing phase and amplitude sensitivity limits in interferometric microscopy. Opt. Lett. 41, 1656–1659 (2016).
Backoach, O., Kariv, S., Girshovitz, P. & Shaked, N. T. Fast phase processing in offaxis holography by CUDA including parallel phase unwrapping. Opt. Express 24, 3177–3188, doi:10.1364/OE.24.003177 (2016).
Kim, Y. et al. Commonpath diffraction optical tomography for investigation of threedimensional structures and dynamics of biological cells. Opt. Express 22, 10398–10407, doi:10.1364/oe.22.010389 (2014).
Kim, Y. et al. Profiling individual human red blood cells using commonpath diffraction optical tomography. Sci. Rep. 4, doi:10.1038/srep06659 (2014).
Oh, S. et al. LabelFree Imaging of Membrane Potential Using Membrane Electromotility. Biophys. J. 103, 11–18 (2012).
Primot, J. & Guérineau, N. Extended Hartmann Test Based on the Pseudoguiding Property of a Hartmann Mask Completed by a Phase Chessboard. Appl. Opt. 39, 5715–5720 (2000).
Chanteloup, J. C. Multiplewave lateral shearing interferometry for wavefront sensing. Appl. Opt. 44, 1559–1571 (2005).
Bon, P., Maucort, G., Wattellier, B. & Monneret, S. Quadriwave lateral shearing interferometry for quantitative phase microscopy of living cells. Opt. Express 17, 13080–13094, doi:10.1364/OE.17.013080 (2009).
Bon, P. et al. Threedimensional nanometre localization of nanoparticles to enhance superresolution microscopy. Nat. Commun. 6, doi:10.1038/ncomms8764 (2015).
Ling, T. et al. Quadriwave lateral shearing interferometer based on a randomly encoded hybrid grating. Opt. Lett. 40, 2245–2248 (2015).
Li, J., Tang, F., Wang, X., Dai, F. & Zhang, H. Analysis of lateral shearing interferometry without selfimaging limitations. Appl. Opt. 54, 8070–8079 (2015).
Girshovitz, P. & Shaked, N. T. Fast phase processing in offaxis holography using multiplexing with complex encoding and livecell fluctuation map calculation in realtime. Opt. Express 23, 8773–8787, doi:10.1364/OE.23.008773 (2015).
Ling, T. et al. General measurement of optical system aberrations with a continuously variable lateral shear ratio by a randomly encoded hybrid grating. Appl. Opt. 54, 8913–8920 (2015).
Malacara, D. Optical Shop Testing. 126 (John Wiley & Sons, 2007).
Elster, C. & Weingärtner, I. Exact wavefront reconstruction from two lateral shearing interferograms. J. Opt. Soc. Am. A 16, 2281–2285 (1999).
Elster, C. Exact twodimensional wavefront reconstruction from lateral shearing interferograms with large shears. Appl. Opt. 39, 5353–5359 (2000).
Guo, Y.f, Chen, H., Xu, J. & Ding, J. Twodimensional wavefront reconstruction from lateral multishear interferograms. Opt. Express 20, 15723–15733, doi:10.1364/OE.20.015723 (2012).
Ling, T., Yang, Y., Liu, D., Yue, X. & Jiang, J. Retrieval of phase distributions from the quadriwave lateral shearing interferogram obtained by randomly encoded hybrid grating. Proc. SPIE 9633, 96332G (2015).
Takeda, M., Ina, H. & Kobayashi, S. Fouriertransform method of fringepattern analysis for computerbased topography and interferometry. J. Opt. Soc. Am. 72, 156–160 (1982).
Goldstein, R. M., Zebker, H. A. & Werner, C. L. Satellite radar interferometry: Twodimensional phase unwrapping. Radio Sci. 23, 713–720 (1988).
Herráez, M. A., Burton, D. R., Lalor, M. J. & Gdeisat, M. A. Fast twodimensional phaseunwrapping algorithm based on sorting by reliability following a noncontinuous path. Appl. Opt. 41, 7437–7444 (2002).
Parkhurst, J., Price, G., Sharrock, P. & Moore, C. Phase unwrapping algorithms for use in a true realtime optical body sensor system for use during radiotherapy. Appl. Opt. 50, 6430–6439 (2011).
Dubra, A., Paterson, C. & Dainty, C. Wavefront reconstruction from shear phase maps by use of the discrete Fourier transform. Appl. Opt. 43, 1108–1113 (2004).
Witte, S. et al. Shortcoherence offaxis holographic phase microscopy of live cell dynamics. Biomed. Opt. Express 3, 2184–2189, doi:10.1364/BOE.3.002184 (2012).
Acknowledgements
We thank Prof. Daniel Palanker in Department of Ophthalmology at Stanford University for providing the initial inspiration of this research. This work was partially supported by National Natural Science Foundation of China (61627825, 11275172) and the State Key Lab. of MOI Innovation Program (MOI2015 B06).
Author information
Authors and Affiliations
Contributions
T.L. and Y.Y.Y. conceived the experiments, T.L., J.J. and R.Z. conducted the experiments, T.L. analysed the results, T.L. and Y.Y.Y. wrote the paper. All authors reviewed the manuscript.
Corresponding authors
Ethics declarations
Competing Interests
The authors declare that they have no competing interests.
Additional information
Publisher's note: Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Electronic supplementary material
Rights and permissions
This work is licensed under a Creative Commons Attribution 4.0 International License. The images or other third party material in this article are included in the article’s Creative Commons license, unless indicated otherwise in the credit line; if the material is not included under the Creative Commons license, users will need to obtain permission from the license holder to reproduce the material. To view a copy of this license, visit http://creativecommons.org/licenses/by/4.0/
About this article
Cite this article
Ling, T., Jiang, J., Zhang, R. et al. Quadriwave lateral shearing interferometric microscopy with wideband sensitivity enhancement for quantitative phase imaging in real time. Sci Rep 7, 9 (2017). https://doi.org/10.1038/s41598017000537
Received:
Accepted:
Published:
DOI: https://doi.org/10.1038/s41598017000537
This article is cited by

The fate of volcanic ash: premature or delayed sedimentation?
Nature Communications (2021)

Effect of nitenpyram on the control of Reticulitermes flaviceps
International Journal of Tropical Insect Science (2021)

Restoration at the landscape scale as a means of mitigation and adaptation to climate change
Current Landscape Ecology Reports (2020)

Elevated atmospheric CO2 concentrations alter grapevine (Vitis vinifera) systemic transcriptional response to European grapevine moth (Lobesia botrana) herbivory
Scientific Reports (2019)

Growing vulnerability in the smallscale fishing communities of Maio, Cape Verde
Maritime Studies (2019)
Comments
By submitting a comment you agree to abide by our Terms and Community Guidelines. If you find something abusive or that does not comply with our terms or guidelines please flag it as inappropriate.