Implantable neural interfaces advance the possibilities for neuroscientists to study the brain. They are also promising for use in a multitude of bioelectronic therapies. Electrode technology plays a central role in these developments, as the electrode surfaces form the physical interfaces between technology and the biological targets. Despite this, a common understanding of how electrodes should best be evaluated and compared with respect to their efficiency in recording and stimulation is currently lacking. Without broadly accepted performance tests, it is difficult to rank the many suggestions for electrode materials available in the literature, or to identify where efforts should be focused to advance the field most efficiently. This tutorial critically discusses the most relevant performance tests for characterization of neural interface electrodes and explains their implementation, interpretation and respective limitations. We propose a unified standard to facilitate transparent reporting on electrode performance, promote efficient scientific process and ultimately accelerate translation into clinical practice.
Subscribe to Journal
Get full journal access for 1 year
only $8.25 per issue
All prices are NET prices.
VAT will be added later in the checkout.
Tax calculation will be finalised during checkout.
Rent or Buy article
Get time limited or full article access on ReadCube.
All prices are NET prices.
Buzsaki, G. Large-scale recording of neuronal ensembles. Nat. Neurosci. 7, 446–451 (2004).
Fernández-Ruiz, A. et al. Long-duration hippocampal sharp wave ripples improve memory. Science 364, 1082 (2019). 189–197.
Henze, D. A. et al. Intracellular features predicted by extracellular recordings in the hippocampus in vivo. J. Neurophysiol. 84, 390–400 (2000).
Nakazono, T., Jun, H., Blurton-Jones, M., Green, K. N. & Igarashi, K. M. Gamma oscillations in the entorhinal-hippocampal circuit underlying memory and dementia. Neurosci. Res. 129, 40–46 (2018).
Hochberg, L. R. et al. Reach and grasp by people with tetraplegia using a neurally controlled robotic arm. Nature 485, 372–375 (2012).
Petrini, F. M. et al. Enhancing functional abilities and cognitive integration of the lower limb prosthesis. Sci. Transl. Med. https://doi.org/10.1126/scitranslmed.aav8939 (2019).
Simmons, F. B. et al. Auditory nerve: electrical stimulation in man. Science 148, 104–106 (1965).
Famm, K., Litt, B., Tracey, K. J., Boyden, E. S. & Slaoui, M. Drug discovery: a jump-start for electroceuticals. Nature 496, 159–161 (2013).
Wu, H., Gao, W. & Yin, Z. Materials, devices and systems of soft bioelectronics for precision therapy. Adv. Healthc. Mater. 6, 1700017 (2017).
Viswam, V., Obien, M. E. J., Franke, F., Frey, U. & Hierlemann, A. Optimal electrode size for multi-scale extracellular-potential recording from neuronal assemblies. Front. Neurosci. https://doi.org/10.3389/fnins.2019.00385 (2019).
Negi, S., Bhandari, R., Rieth, L. & Solzbacher, F. In vitro comparison of sputtered iridium oxide and platinum-coated neural implantable microelectrode arrays. Biomed. Mater. 5, 15007 (2010).
Boehler, C., Aqrawe, Z. & Asplund, M. Applications of PEDOT in bioelectronic medicine. Bioelectron. Med. 2, 89–99 (2019).
Carli, S. et al. Electrodeposited PEDOT:Nafion composite for neural recording and stimulation. Adv. Healthc. Mater. 8, 1900765 (2019).
Cogan, S. F. et al. Sputtered iridium oxide films for neural stimulation electrodes. J. Biomed. Mater. Res. 89, 353–361 (2009).
Gerwig, R. et al. PEDOT–CNT composite microelectrodes for recording and electrostimulation applications: fabrication, morphology, and electrical properties. Front. Neuroeng. https://doi.org/10.3389/fneng.2012.00008 (2012).
Ludwig, K. A., Uram, J. D., Yang, J., Martin, D. C. & Kipke, D. R. Chronic neural recordings using silicon microelectrode arrays electrochemically deposited with a poly(3,4-ethylenedioxythiophene) (PEDOT) film. J. Neural Eng. 3, 59–70 (2006).
Luo, X., Weaver, C. L., Zhou, D. D., Greenberg, R. & Cui, X. T. Highly stable carbon nanotube doped poly(3,4-ethylenedioxythiophene) for chronic neural stimulation. Biomaterials 32, 5551–5557 (2011).
Nimbalkar, S. et al. Ultra-capacitive carbon neural probe allows simultaneous long-term electrical stimulations and high-resolution neurotransmitter detection. Sci. Rep. 8, 6958 (2018).
Vomero, M. et al. Highly stable glassy carbon interfaces for long-term neural stimulation and low-noise recording of brain activity. Sci. Rep. 7, 40332 (2017).
Boehler, C., Stieglitz, T. & Asplund, M. Nanostructured platinum grass enables superior impedance reduction for neural microelectrodes. Biomaterials 67, 346–353 (2015).
Boehler, C., Vieira, D. M., Egert, U. & Asplund, M. NanoPt – A nanostructured electrode coating for neural recording and microstimulation. ACS Appl. Mater. Interfaces https://doi.org/10.1021/acsami.9b22798 (2020).
Zhao, Z., Gong, R., Zheng, L. & Wang, J. In vivo neural recording and electrochemical performance of microelectrode arrays modified by rough-surfaced aupt alloy nanoparticles with nanoporosity. Sensors 16, 1851 (2016).
U. S. P. C. USP Class VI in USP Biological Reactivity Tests In Vivo. US Pharmacopeia and National Formulary.
International Organization for Standardization. 10993: Biological evaluation of medical devices.
Hill, D. N., Mehta, S. B. & Kleinfeld, D. Quality metrics to accompany spike sorting of extracellular signals. J. Neurosci. 31, 8699 (2011).
Kimura, J. Electrodiagnosis in Diseases of Nerve and Muscle: Principles and Practice (Oxford University Press, 2013).
McCann, H., Pisano, G. & Beltrachini, L. Variation in reported human head tissue electrical conductivity values. Brain Topogr. 32, 825–858 (2019).
Baumann, S. B., Wozny, D. R., Kelly, S. K. & Meno, F. M. The electrical conductivity of human cerebrospinal fluid at body temperature. IEEE Trans. Biomed. Eng. 44, 220–223 (1997).
Hladky, S. B. & Barrand, M. A. Mechanisms of fluid movement into, through and out of the brain: evaluation of the evidence. Fluids Barriers CNS 11, 26 (2014).
Levin, E., Muravchick, S. & Gold, M. I. Density of normal human cerebrospinal fluid and tetracaine solutions. Anesthesia Analgesia 60, 814–817 (1981).
Franks, W., Schenker, I., Schmutz, P. & Hierlemann, A. Impedance characterization and modeling of electrodes for biomedical applications. IEEE Trans. Biomed. Eng. 52, 1295–1302 (2005).
Woods, V. et al. Long-term recording reliability of liquid crystal polymer µECoG arrays. J. Neural Eng. 15, 066024 (2018).
Nelson, M. J., Pouget, P., Nilsen, E. A., Patten, C. D. & Schall, J. D. Review of signal distortion through metal microelectrode recording circuits and filters. J. Neurosci. Methods 169, 141–157 (2008).
Kumsa, D. W. et al. Electron transfer processes occurring on platinum neural stimulating electrodes: a tutorial on the i(V e) profile. J. Neural Eng. 13, 052001 (2016).
Ivanovskaya, A. N. et al. Electrochemical roughening of thin-film platinum for neural probe arrays and biosensing applications. J. Electrochem. Soc. 165, G3125–G3132 (2018).
Marrese, C. A. Preparation of strongly adherent platinum black coatings. Anal. Chem. 59, 217–218 (1987).
Young, A. T., Cornwell, N. & Daniele, M. A. Neuro interfaces: neuro-nano interfaces: utilizing nano-coatings and nanoparticles to enable next-generation electrophysiological recording, neural stimulation, and biochemical modulation. Adv. Funct. Mater. 28, 1870079 (2018).
Doña Rodríguez, J. M., Herrera Melián, J. A. & Pérez Peña, J. Determination of the real surface area of Pt electrodes by hydrogen adsorption using cyclic voltammetry. J. Chem. Educ. 77, 1195 (2000).
Łukaszewski, M., Soszko, M. & Czerwiński, A. Electrochemical methods of real surface area determination of noble metal electrodes – an overview. Int. J. Electrochem. Sci. 11, 4442–4469 (2016).
Macdonald, J. R. Impedance spectroscopy. Ann. Biomed. Eng. 20, 289–305 (1992).
Bobacka, J., Lewenstam, A. & Ivaska, A. Electrochemical impedance spectroscopy of oxidized poly(3,4-ethylenedioxythiophene) film electrodes in aqueous solutions. J. Electroanal. Chem. 489, 17–27 (2000).
Barsoukov, E. & Macdonald, J. R. Impedance Spectroscopy: Theory, Experiment and Applications 3rd edn (John Wiley & Sons, 2018).
Scholtz, F. Electroanalytical Methods: Guide to Experiments and Applications 2nd edn (Springer, 2010).
Harris, K. D., Quiroga, R. Q., Freeman, J. & Smith, S. L. Improving data quality in neuronal population recordings. Nat. Neurosci. 19, 1165–1174 (2016).
Liu, X., Demosthenous, A. & Donaldson, N. Platinum electrode noise in the ENG spectrum. Med. Biol. Eng. Comput. 46, 997–1003 (2008).
Buzsaki, G. Rhythms of the Brain (Oxford University Press, 2006).
Neto, J. P. et al. Does impedance matter when recording spikes with polytrodes? Front. Neurosci. 12, 715 (2018).
Zijlmans, M. et al. How to record high-frequency oscillations in epilepsy: a practical guideline. Epilepsia 58, 1305–1315 (2017).
Brummer, S. B. & Turner, M. J. Electrochemical considerations for safe electrical stimulation of the nervous system with platinum electrodes. IEEE Trans. Bio-med. Eng. 24, 59–63 (1977).
Hung, A., Zhou, D., Greenberg, R., Goldberg, I. B. & Judy, J. W. Pulse-clamp technique for characterizing neural-stimulating electrodes. J. Electrochem. Soc. 154, C479–C486 (2007).
Merrill, D. R., Bikson, M. & Jefferys, J. G. Electrical stimulation of excitable tissue: design of efficacious and safe protocols. J. Neurosci. Methods 141, 171–198 (2005).
Rose, T. L. & Robblee, L. S. Electrical stimulation with Pt electrodes. VIII. Electrochemically safe charge injection limits with 0.2 ms pulses. IEEE Trans. Bio-med. Eng. 37, 1118–1120 (1990).
Janders, M., Egert, U., Stelzle, M. & Nisch, W. Novel thin film titanium nitride micro-electrodes with excellent charge transfer capability for cell stimulation and sensing applications. in Proceedings of 18th Annual International Conference of the IEEE Engineering in Medicine and Biology Society 245–247 (1996).
Weiland, J. D., Anderson, D. J. & Humayun, M. S. In vitro electrical properties for iridium oxide versus titanium nitride stimulating electrodes. IEEE Trans. Bio-med. Eng. 49, 1574–1579 (2002).
Erdey-Grúz, T. & Volmer, M. Zur. Theor. der Wasserst. Überspannung 150A, 203 (1930).
Morton, S. L. The role of oxygen reduction in electrical stimulation of neural tissue. J. Electrochem. Soc. 141, 122 (1994).
Cogan, S. F., Ehrlich, J., Plante, T. D., Gingerich, M. D. & Shire, D. B. Contribution of oxygen reduction to charge injection on platinum and sputtered iridium oxide neural stimulation electrodes. IEEE Trans. Bio-med. Eng. 57, 2313–2321 (2010).
Cogan, S. F., Plante, T. D. & Ehrlich, J. Sputtered iridium oxide films (SIROFs) for low-impedance neural stimulation and recording electrodes. Conf. Proc. IEEE Eng. Med. Biol. Soc. 6, 4153–4156 (2004).
Slavcheva, E., Vitushinsky, R., Mokwa, W. & Schnakenberg, U. Sputtered iridium oxide films as charge injection material for functional electrostimulation. J. Electrochem. Soc. 151, E226–E237 (2004).
van Ooyen, A., Zagolla, V. G., Ulrich, C. & Schnakenberg, U. Pulse-clamp technique for single neuron stimulation electrode characterization. Conf. Proc. IEEE Eng. Med. Biol. Soc. 2009, 1635–1638 (2009).
Stefan, I. C. et al. Theoretical analysis of the pulse-clamp method as applied to neural stimulating electrodes. J. Electrochem. Soc. 148, E73–E78 (2001).
Suesserman, M. F., Spelman, F. A. & Rubinstein, J. T. In vitro measurement and characterization of current density profiles produced by non-recessed, simple recessed, and radially varying recessed stimulating electrodes. IEEE Trans. Bio-med. Eng. 38, 401–408 (1991).
Cogan, S. F., Troyk, P. R., Ehrlich, J., Gasbarro, C. M. & Plante, T. D. The influence of electrolyte composition on the in vitro charge-injection limits of activated iridium oxide (AIROF) stimulation electrodes. J. Neural Eng. 4, 79–86 (2007).
Anderson, D. N., Duffley, G., Vorwerk, J., Dorval, A. D. & Butson, C. R. Anodic stimulation misunderstood: preferential activation of fiber orientations with anodic waveforms in deep brain stimulation. J. Neural Eng. 16, 016026 (2019).
McIntyre, C. C. & Grill, W. M. Extracellular stimulation of central neurons: influence of stimulus waveform and frequency on neuronal output. J. Neurophysiol. 88, 1592–1604 (2002).
Tehovnik, E. J. & Slocum, W. M. Depth-dependent detection of microampere currents delivered to monkey V1. Eur. J. Neurosci. 29, 1477–1489 (2009).
Bonner, M. D., Daroux, M., Crish, T. & Mortimer, J. T. The pulse‐clamp method for analyzing the electrochemistry on neural stimulating electrodes. J. Electrochem. Soc. 140, 2740–2744 (1993).
Abd Hamid, A. I., Gall, C., Speck, O., Antal, A. & Sabel, B. A. Effects of alternating current stimulation on the healthy and diseased brain. Front. Neurosci. 9, 391 (2015).
Goodman, J. H., Berger, R. E. & Tcheng, T. K. Preemptive low-frequency stimulation decreases the incidence of amygdala-kindled seizures. Epilepsia 46, 1–7 (2005).
Boehler, C. & Asplund, M. PEDOT as a high charge injection material for low-frequency stimulation. Conf. Proc. IEEE Eng. Med. Biol. Soc. 2018, 2202–2205 (2018).
Ghazavi, A. & Cogan, S. F. Electrochemical characterization of high frequency stimulation electrodes: role of electrode material and stimulation parameters on electrode polarization. J. Neural Eng. 15, 036023 (2018).
Biegler, T., Rand, D. A. J. & Woods, R. Limiting oxygen coverage on platinized platinum; relevance to determination of real platinum area by hydrogen adsorption. J. Electroanal. Chem. Interfacial Electrochem. 29, 269–277 (1971).
Weltin, A. et al. New life for old wires: electrochemical sensor method for neural implants. J. Neural Eng. 17, 016007 (2019).
Beebe, X. & Rose, T. L. Charge injection limits of activated iridium oxide electrodes with 0.2 ms pulses in bicarbonate buffered saline. IEEE Trans. Bio-med. Eng. 35, 494–495 (1988).
Hudak, E. M., Kumsa, D. W., Martin, H. B. & Mortimer, J. T. Electron transfer processes occurring on platinum neural stimulating electrodes: calculated charge-storage capacities are inaccessible during applied stimulation. J. Neural Eng. 14, 046012 (2017).
Kozai, T. D., Jaquins-Gerstl, A. S., Vazquez, A. L., Michael, A. C. & Cui, X. T. Brain tissue responses to neural implants impact signal sensitivity and intervention strategies. ACS Chem. Neurosci. 6, 48–67 (2015).
Polikov, V. S., Tresco, P. A. & Reichert, W. M. Response of brain tissue to chronically implanted neural electrodes. J. Neurosci. Methods 148, 1–18 (2005).
Prodanov, D. & Delbeke, J. Mechanical and biological interactions of implants with the brain and their impact on implant design. Front. Neurosci. 10, 11 (2016).
Salatino, J. W., Ludwig, K. A., Kozai, T. D. Y. & Purcell, E. K. Glial responses to implanted electrodes in the brain. Nat. Biomed. Eng. 1, 862–877 (2017).
Johnson, M. D., Kao, O. E. & Kipke, D. R. Spatiotemporal pH dynamics following insertion of neural microelectrode arrays. J. Neurosci. methods 160, 276–287 (2007).
Halliwell, B. Reactive oxygen species and the central nervous system. J. Neurochem. 59, 1609–1623 (1992).
Magnotta, V. A. et al. Detecting activity-evoked pH changes in human brain. Proc. Natl Acad. Sci. USA 109, 8270–8273 (2012).
Hibbert, D. B., Weitzner, K., Tabor, B. & Carter, P. Mass changes and dissolution of platinum during electrical stimulation in artificial perilymph solution. Biomaterials 21, 2177–2182 (2000).
Robblee, L. S., McHardy, J., Marston, J. M. & Brummer, S. B. Electrical stimulation with Pt electrodes. V. The effect of protein on Pt dissolution. Biomaterials 1, 135–139 (1980).
Maas, A. I., Fleckenstein, W., de Jong, D. A. & van Santbrink, H. Monitoring cerebral oxygenation: experimental studies and preliminary clinical results of continuous monitoring of cerebrospinal fluid and brain tissue oxygen tension. Acta Neurochirurgica Supplementum 59, 50–57 (1993).
Patrick, E., Orazem, M. E., Sanchez, J. C. & Nishida, T. Corrosion of tungsten microelectrodes used in neural recording applications. J. Neurosci. Methods 198, 158–171 (2011).
Cogan, S. F., Jones, G. S., Hills, D. V., Walter, J. S. & Riedy, L. W. Comparison of 316LVM and MP35N alloys as charge injection electrodes. J. Biomed. Mater. Res. 28, 233–240 (1994).
Geddes, L. A. & Roeder, R. Criteria for the selection of materials for implanted electrodes. Ann. Biomed. Eng. 31, 879–890 (2003).
Shepherd, R. K., Carter, P. M., Enke, Y. L., Wise, A. K. & Fallon, J. B. Chronic intracochlear electrical stimulation at high charge densities results in platinum dissolution but not neural loss or functional changes in vivo. J. Neural Eng. 16, 026009 (2019).
Spiers, K. et al. An X-ray fluorescence microscopic analysis of the tissue surrounding the multi-channel cochlear implant electrode array. Cochl. Implants Int. 17, 129–131 (2016).
Percival, S. J., Dick, J. E. & Bard, A. J. Cathodically dissolved platinum resulting from the O2 and H2O2 reduction reactions on platinum ultramicroelectrodes. Anal. Chem. 89, 3087–3092 (2017).
Pourbaix, M. Atlas of Electrochemical Equilibria in Aqueous Solutions (National Association of Corrosion Engineers, 1966).
Williams, D. F. Corrosion of implant materials. Ann. Rev. Mater. Sci. 6, 237–266 (1976).
Asplund, M., Nyberg, T. & Inganäs, O. Electroactive polymers for neural interfaces. Polym. Chem. 1, 1374–1391 (2010).
Boehler, C., Oberueber, F., Schlabach, S., Stieglitz, T. & Asplund, M. Long-term stable adhesion for conducting polymers in biomedical applications: IrOx and nanostructured platinum solve the chronic challenge. ACS Appl. Mater. Interfaces 9, 189–197 (2016).
Schander, A. et al. In-vitro evaluation of the long-term stability of PEDOT:PSS coated microelectrodes for chronic recording and electrical stimulation of neurons. in 38th Annual International Conference of the IEEE Engineering in Medicine and Biology Society (EMBC) 6174–6177 (2016).
Thaning, E. M., Asplund, M. L., Nyberg, T. A., Inganas, O. W. & von Holst, H. Stability of poly(3,4-ethylene dioxythiophene) materials intended for implants. J. Biomed. Mater. Res. 93, 407–415 (2010).
Ordonez, J., Schuettler, M., Boehler, C., Boretius, T. & Stieglitz, T. Thin films and microelectrode arrays for neuroprosthetics. MRS Bull. 37, 590–598 (2012).
Čvančara, P. et al. On the reliability of chronically implanted thin-film electrodes in human arm nerves for neuroprosthetic applications. Preprint at bioRxiv https://doi.org/10.1101/653964 (2019).
Jorfi, M., Skousen, J. L., Weder, C. & Capadona, J. R. Progress towards biocompatible intracortical microelectrodes for neural interfacing applications. J. Neural Eng. 12, 011001–011001 (2015).
Sohal, H. S., Vassilevski, K., Jackson, A., Baker, S. N. & O’Neill, A. Design and microfabrication considerations for reliable flexible intracortical implants. Front. Mech. Eng. https://doi.org/10.3389/fmech.2016.00005 (2016).
Yasuda, H., Yu, Q. S. & Chen, M. Interfacial factors in corrosion protection: an EIS study of model systems. Prog. Org. Coat. 41, 273–279 (2001).
Schmitt, G. et al. Passivation and corrosion of microelectrode arrays. Electrochim. Acta 44, 3865–3883 (1999).
Vanhoestenberghe, A. & Donaldson, N. Corrosion of silicon integrated circuits and lifetime predictions in implantable electronic devices. J. Neural Eng. 10, 031002 (2013).
Hemmerich, K. J. General aging theory and simplified protocol for accelerated aging of medical devices. Med. Plastics Biomater. 16–23 (1998).
Hukins, D. W., Mahomed, A. & Kukureka, S. N. Accelerated aging for testing polymeric biomaterials and medical devices. Med. Eng. Phys. 30, 1270–1274 (2008).
Mazzucco, D. C., Dumbleton, J. & Kurtz, S. M. Can accelerated aqueous aging simulate in vivo oxidation of gamma-sterilized UHMWPE? J. Biomed. Mater. Res. 79, 79–85 (2006).
Takmakov, P. et al. Rapid evaluation of the durability of cortical neural implants using accelerated aging with reactive oxygen species. J. Neural Eng. 12, 026003 (2015).
Street, M. G., Welle, C. G. & Takmakov, P. A. Automated reactive accelerated aging for rapid in vitro evaluation of neural implant performance. Rev. Sci. Instrum. 89, 094301 (2018).
Rubehn, B. & Stieglitz, T. In vitro evaluation of the long-term stability of polyimide as a material for neural implants. Biomaterials 31, 3449–3458 (2010).
Jerkiewicz, G., Vatankhah, G., Lessard, J., Soriaga, M. P. & Park, Y.-S. Surface-oxide growth at platinum electrodes in aqueous H2SO4: Reexamination of its mechanism through combined cyclic-voltammetry, electrochemical quartz-crystal nanobalance, and Auger electron spectroscopy measurements. Electrochim. Acta 49, 1451–1459 (2004).
Butz, N., Kuhl, M. & Manoli, Y. Charge balancing circuit, stimulator circuit and method for charge balancing. US Patent 20190111249 (2019).
This work was partially supported by BrainLinks-BrainTools, Cluster of Excellence funded by the German Research Foundation (DFG, EXC 1086). M Asplund and C Boehler were supported by the European Research Council (ERC) under the European Union’s Horizon 2020 research and innovation programme (grant agreement number 759655). L. Fadiga was supported by PRIN 2015. We acknowledge S. Shaner, University of Freiburg, for support in proofreading of the final manuscript.
The authors declare no competing interests.
Peer review information Nature Protocols thanks Andreas Hierlemann, Jonathan Viventi and the other, anonymous, reviewer(s) for their contribution to the peer review of this work.
Publisher’s note Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Key references using the performance tests described in this tutorial:
Boehler, C., Stieglitz, T. & Asplund, M. Biomaterials 67, 346–353 (2015): https://www.sciencedirect.com/science/article/pii/S0142961215006183
Boehler, C., Vieira, D. M., Egert, U. & Asplund, M. ACS Appl. Mater. Interfaces 12, 14855–14865 (2020): https://pubs.acs.org/doi/abs/10.1021/acsami.9b22798
Boehler, C., Oberueber, F., Schlabach, S., Stieglitz, T. & Asplund, M. ACS Appl. Mater. Interfaces 9, 189–197 (2017): https://pubs.acs.org/doi/abs/10.1021/acsami.6b13468
Vomero, M. et al. Materials 11, 2486 (2018): https://europepmc.org/article/med/30544545
About this article
Cite this article
Boehler, C., Carli, S., Fadiga, L. et al. Tutorial: guidelines for standardized performance tests for electrodes intended for neural interfaces and bioelectronics. Nat Protoc 15, 3557–3578 (2020). https://doi.org/10.1038/s41596-020-0389-2