The efficacy of wireless intracortical brain–computer interfaces (iBCIs) is limited in part by the number of recording channels, which is constrained by the power budget of the implantable system. Designing wireless iBCIs that provide the high-quality recordings of today’s wired neural interfaces may lead to inadvertent over-design at the expense of power consumption and scalability. Here, we report analyses of neural signals collected from experimental iBCI measurements in rhesus macaques and from a clinical-trial participant with implanted 96-channel Utah multielectrode arrays to understand the trade-offs between signal quality and decoder performance. Moreover, we propose an efficient hardware design for clinically viable iBCIs, and suggest that the circuit design parameters of current recording iBCIs can be relaxed considerably without loss of performance. The proposed design may allow for an order-of-magnitude power savings and lead to clinically viable iBCIs with a higher channel count.
Subscribe to Journal
Get full journal access for 1 year
only $8.25 per issue
All prices are NET prices.
VAT will be added later in the checkout.
Rent or Buy article
Get time limited or full article access on ReadCube.
All prices are NET prices.
The main data supporting the results in this study are available within the paper and its Supplementary Information. The raw human neural data are available on request to K.V.S. or J.M.H., yet owing to the potential sensitivity of the data and to respect the participant’s expectation of privacy, an agreement between the researcher’s institution and the BrainGate consortium is required to facilitate the sharing of these datasets. Processed data are available at https://shenoy.people.stanford.edu/data.
The custom code used to produce the figures is available at https://shenoy.people.stanford.edu/data.
Slutzky, M. W. Brain–machine interfaces: powerful tools for clinical treatment and neuroscientific investigations. Neuroscientist 25, 139–154 (2019).
Bouton, C. E. et al. Restoring cortical control of functional movement in a human with quadriplegia. Nature 533, 247–250 (2016).
Ajiboye, A. B. et al. Restoration of reaching and grasping movements through brain-controlled muscle stimulation in a person with tetraplegia: a proof-of-concept demonstration. Lancet 389, 1821–1830 (2017).
Hochberg, L. R. et al. Reach and grasp by people with tetraplegia using a neurally controlled robotic arm. Nature 485, 372–375 (2012).
Collinger, J. L. et al. Functional priorities, assistive technology, and brain-computer interfaces after spinal cord injury. J. Rehabil. Res. Dev. 50, 145–160 (2013).
Downey, J. E. et al. Blending of brain–machine interface and vision-guided autonomous robotics improves neuroprosthetic arm performance during grasping. J. Neuroeng. Rehabil. 13, 28 (2016).
Hochberg, L. R. et al. Neuronal ensemble control of prosthetic devices by a human with tetraplegia. Nature 442, 164–171 (2006).
Pandarinath, C. et al. High performance communication by people with paralysis using an intracortical brain-computer interface. eLife 6, 18554 (2017).
Jarosiewicz, B. et al. Virtual typing by people with tetraplegia using a self-calibrating intracortical brain-computer interface. Sci. Transl. Med. 7, 313ra179 (2015).
Blabe, C. H. et al. Assessment of brain-machine interfaces from the perspective of people with paralysis. J. Neural Eng. 12, 043002 (2015).
Gilja, V. et al. Clinical translation of a high-performance neural prosthesis. Nat. Med. 21, 1142–1145 (2015).
Homer, M. L., Nurmikko, A. V., Donoghue, J. P. & Hochberg, L. R. Sensors and decoding for intracortical brain computer interfaces. Annu. Rev. Biomed. Eng. 15, 383–405 (2013).
Lebedev, M. A. & Nicolelis, M. A. L. Brain-machine interfaces: from basic science to neuroprostheses and neurorehabilitation. Physiol. Rev. 97, 767–837 (2017).
Schwarz, D. A. et al. Chronic, wireless recordings of large-scale brain activity in freely moving rhesus monkeys. Nat. Methods 11, 670–676 (2014).
Carmena, J. M. et al. Learning to control a brain-machine interface for reaching and grasping by primates. PLoS Biol. 1, e42 (2003).
Gao, P. & Ganguli, S. On simplicity and complexity in the brave new world of large-scale neuroscience. Curr. Opin. Neurobiol. 32, 148–155 (2015).
Wodlinger, B. et al. Ten-dimensional anthropomorphic arm control in a human brain-machine interface: difficulties, solutions, and limitations. J. Neural Eng. 12, 016011 (2015).
Mitz, A. R. et al. High channel count single-unit recordings from nonhuman primate frontal cortex. J. Neurosci. Methods 289, 39–47 (2017).
Chen, X. et al. 3D printing and modelling of customized implants and surgical guides for non-human primates. J. Neurosci. Methods 286, 38–55 (2017).
Gao, H. et al. HermesE: a 96-channel full data rate direct neural interface in 0.13 μm CMOS. IEEE J. Solid-State Circuits 47, 1043–1055 (2012).
Miranda, H., Gilja, V., Chestek, C. A., Shenoy, K. V. & Meng, T. H. HermesD: a high-rate long-range wireless transmission system for simultaneous multichannel neural recording applications. IEEE Trans. Biomed. Circuits Syst. 4, 181–191 (2010).
Borton, D. A., Yin, M., Aceros, J. & Nurmikko, A. An implantable wireless neural interface for recording cortical circuit dynamics in moving primates. J. Neural Eng. 10, 026010 (2013).
Yin, M. et al. Wireless neurosensor for full-spectrum electrophysiology recordings during free behavior. Neuron 84, 1170–1182 (2014).
Kao, J. C., Stavisky, S. D., Sussillo, D., Nuyujukian, P. & Shenoy, K. V. Information systems opportunities in brain-machine interface decoders. Proc. IEEE 102, 666–682 (2014).
Fraser, G. W., Chase, S. M., Whitford, A. & Schwartz, A. B. Control of a brain-computer interface without spike sorting. J. Neural Eng. 6, 055004 (2009).
Perel, S. et al. Single-unit activity, threshold crossings, and local field potentials in motor cortex differentially encode reach kinematics. J. Neurophysiol. 114, 1500–1512 (2015).
Christie, B. P. et al. Comparison of spike sorting and thresholding of voltage waveforms for intracortical brain-machine interface performance. J. Neural Eng. 12, 016009 (2015).
Li, J. & Li, Z. Sums of spike waveform features for motor decoding. Front. Neurosci. 11, 406 (2017).
Trautmann, E. M. et al. Accurate estimation of neural population dynamics without spike sorting. Neuron 103, 292–308 (2019).
Han, D., Zheng, Y., Rajkumar, R., Dawe, G. & Je, M. A 0.45 v 100-channel neural-recording IC with sub-μW/channel consumption in 0.18 μm CMOS. In IEEE Int. Solid-State Circuits Conf. 291–292 (IEEE, 2013).
Irwin, Z. T. et al. Enabling low-power, multi-modal neural interfaces through a common, low-bandwidth feature space. IEEE Trans. Neural Syst. Rehabil. Eng. 24, 521–531 (2016).
Sodagar, A. M., Wise, K. D. & Najafi, K. A fully integrated mixed-signal neural processor for implantable multichannel cortical recording. IEEE Trans. Biomed. Eng. 54, 1075–1088 (2007).
Karkare, V., Gibson, S. & Marković, D. A 75-μw, 16-channel neural spike-sorting processor with unsupervised clustering. IEEE J. Solid-State Circuits 48, 2230–2238 (2013).
Muratore, D. G. et al. A data-compressive wired-OR readout for massively parallel neural recording. IEEE Trans. Biomed. Circuits Syst. 13, 1128–1140 (2019).
Aprile, C. et al. Adaptive learning-based compressive sampling for low-power wireless implants. IEEE Trans. Circuits Syst. I 65, 3929–3941 (2018).
Pagin, M. & Ortmanns, M. A neural data lossless compression scheme based on spatial and temporal prediction. In IEEE Biomedical Circuits and Systems Conf. 1–4 (IEEE, 2017).
Wu, T., Zhao, W., Keefer, E. & Yang, Z. Deep compressive autoencoder for action potential compression in large-scale neural recording. J. Neural Eng. 15, 066019 (2018).
Okazawa, T. & Akita, I. A time-domain analog spatial compressed sensing encoder for multi-channel neural recording. Sensors 18, 184 (2018).
Shoaran, M., Lopez, M. M., Pasupureddi, V. S. R., Leblebici, Y. & Schmid, A. A low-power area-efficient compressive sensing approach for multi-channel neural recording. In IEEE Int. Symp. on Circuits and Systems 2191–2194 (IEEE, 2013).
Musk, E. Neuralink, an integrated brain-machine interface platform with thousands of channels. J. Med. Internet. Res. 21, e16194 (2019).
Jun, J. J. et al. Fully integrated silicon probes for high-density recording of neural activity. Nature 551, 232–236 (2017).
Lopez, C. M. et al. A neural probe with up to 966 electrodes and up to 384 configurable channels in 0.13 μm SOI CMOS. IEEE Trans. Biomed. Circuits Syst. 11, 510–522 (2017).
De Dorigo, D. et al. Fully immersible subcortical neural probes with modular architecture and a Delta-Sigma ADC integrated under each electrode for parallel readout of 144 recording sites. IEEE J. Solid-State Circuits 53, 3111–3125 (2018).
Lee, B. et al. An inductively-powered wireless neural recording and stimulation system for freely-behaving animals. IEEE Trans. Biomed. Circuits Syst. 13, 413–424 (2019).
Angotzi, G. N. et al. SiNAPS: an implantable active pixel sensor CMOS-probe for simultaneous large-scale neural recordings. Biosens. Bioelectron. 126, 355–364 (2019).
Fiáth, R. et al. A silicon-based neural probe with densely-packed low-impedance titanium nitride microelectrodes for ultrahigh-resolution in vivo recordings. Biosens. Bioelectron. 106, 86–92 (2018).
Einevoll, G. T., Kayser, C., Logothetis, N. K. & Panzeri, S. Modelling and analysis of local field potentials for studying the function of cortical circuits. Nat. Rev. Neurosci. 14, 770–785 (2013).
Belitski, A. et al. Low-frequency local field potentials and spikes in primary visual cortex convey independent visual information. J. Neurosci. 28, 5696–5709 (2008).
Buzsáki, G., Anastassiou, C. A. & Koch, C. The origin of extracellular fields and currents—EEG, ECoG, LFP and spikes. Nat. Rev. Neurosci. 13, 407–420 (2012).
Chestek, C. A. et al. Long-term stability of neural prosthetic control signals from silicon cortical arrays in rhesus macaque motor cortex. J. Neural Eng. 8, 045005 (2011).
Todorova, S., Sadtler, P., Batista, A., Chase, S. & Ventura, V. To sort or not to sort: the impact of spike-sorting on neural decoding performance. J. Neural Eng. 11, 056005 (2014).
Kao, J. C., Nuyujukian, P., Ryu, S. I. & Shenoy, K. V. A high-performance neural prosthesis incorporating discrete state selection with hidden Markov models. IEEE Trans. Biomed. Eng. 64, 935–945 (2017).
Shanechi, M. M. et al. Rapid control and feedback rates enhance neuroprosthetic control. Nat. Commun. 8, 13825 (2017).
Even-Chen, N., Stavisky, S. D., Kao, J. C., Ryu, S. I. & Shenoy, K. V. Augmenting intracortical brain-machine interface with neurally driven error detectors. J. Neural Eng. 14, 066007 (2017).
Collinger, J. L. et al. High-performance neuroprosthetic control by an individual with tetraplegia. Lancet 381, 557–564 (2013).
Muelling, K. et al. Autonomy infused teleoperation with application to BCI manipulation. Auton. Robots 41, 1401–1422 (2017).
Gilja, V. et al. A high-performance neural prosthesis enabled by control algorithm design. Nat. Neurosci. 15, 1752–1757 (2012).
Katyal, K. D. et al. A collaborative BCI approach to autonomous control of a prosthetic limb system. In 2014 IEEE Int. Conf. on Systems, Man, and Cybernetics 1479–1482 (IEEE, 2014).
Oby, E. R. et al. Extracellular voltage threshold settings can be tuned for optimal encoding of movement and stimulus parameters. J. Neural Eng. 13, 036009 (2016).
Stavisky, S. D., Kao, J. C., Nuyujukian, P., Ryu, S. I. & Shenoy, K. V. A high performing brain-machine interface driven by low-frequency local field potentials alone and together with spikes. J. Neural Eng. 12, 036009 (2015).
Flint, R. D., Lindberg, E. W., Jordan, L. R., Miller, L. E. & Slutzky, M. W. Accurate decoding of reaching movements from field potentials in the absence of spikes. J. Neural Eng. 9, 046006 (2012).
Even-Chen, N. et al. Feasibility of automatic error detect-and-undo system in human intracortical brain-computer interfaces. IEEE Trans. Biomed. Eng. 65, 1771–1784 (2018).
Brandman, D. M. et al. Robust closed-loop control of a cursor in a person with tetraplegia using Gaussian process regression. Neural Comput. 30, 2986–3008 (2018).
Fernández, E. & Botella, P. Biotolerability of intracortical microelectrodes. Adv. Biosyst. 2, 1700115 (2018).
Zhai, S., Hunter, M. & Smith, B. A. Performance optimization of virtual keyboards. Hum. Comput. Interact. 17, 229–269 (2002).
Zumsteg, Z. S. et al. Power feasibility of implantable digital spike sorting circuits for neural prosthetic systems. IEEE Trans. Neural Syst. Rehabil. Eng. 13, 272–279 (2005).
Kao, J. C. et al. Single-trial dynamics of motor cortex and their applications to brain-machine interfaces. Nat. Commun. 6, 7759 (2015).
Shanechi, M. M. Brain-machine interface control algorithms. IEEE Trans. Neural Syst. Rehabil. Eng. 25, 1725–1734 (2017).
Sussillo, D., Stavisky, S. D., Kao, J. C., Ryu, S. I. & Shenoy, K. V. Making brain-machine interfaces robust to future neural variability. Nat. Commun. 7, 13749 (2016).
Glaser, J. I., Chowdhury, R. H., Perich, M. G., Miller, L. E. & Kording, K. P. Machine learning for neural decoding. Preprint at https://arxiv.org/abs/1708.00909 (2017).
Cunningham, J. P., Gilja, V., Ryu, S. I. & Shenoy, K. V. Methods for estimating neural firing rates, and their application to brain-machine interfaces. Neural Netw. 22, 1235–1246 (2009).
Perge, J. A. et al. Intra-day signal instabilities affect decoding performance in an intracortical neural interface system. J. Neural Eng. 10, 036004 (2013).
Cerebus: Instructions for Use (Blackrock Microsystems, 2020); https://www.blackrockmicro.com/wp-content/ifu/LB-0028-15.00-Cerebus-Instructions-for-Use.pdf
Bahrami, H., Mirbozorgi, S. A., Rusch, L. A. & Gosselin, B. BER performance of implant-to-air high-speed UWB data communications for neural recording systems. IEEE Proc. Eng. Med. Biol. Soc. Conf. 2014, 3961–3964 (2014).
Ebrazeh, A. & Mohseni, P. 30 pJ/b, 67 Mbps, centimeter-to-meter range data telemetry with an IR-UWB wireless link. IEEE Trans. Biomed. Circuits Syst. 9, 362–369 (2015).
Harrison, R. R. et al. Wireless neural recording with single low-power integrated circuit. IEEE Trans. Neural Syst. Rehabil. Eng. 17, 322–329 (2009).
Walden, R. H. Analog-to-digital converter survey and analysis. IEEE J. Sel. areas Commun. 17, 539–550 (1999).
Gibson, S., Chandler, R., Karkare, V., Markovic, D. & Judy, J. W. An efficiency comparison of analog and digital spike detection. In 2009 4th Int. IEEE/EMBS Conf. on Neural Engineering 423–428 (IEEE, 2009).
Gibson, S., Judy, J. W. & Marković, D. Technology-aware algorithm design for neural spike detection, feature extraction, and dimensionality reduction. IEEE Trans. Neural Syst. Rehabil. Eng. 18, 469–478 (2010).
Yang, Z., Zhao, Q., Keefer, E. & Liu, W. Noise characterization, modeling, and reduction for in vivo neural recording. Adv. Neural Inf. Process. Syst. 22, 2160–2168 (2009).
Chandrakumar, H. & Marković, D. An 80-mVpp linear-input range, 1.6-GΩ input impedance, low-power chopper amplifier for closed-loop neural recording that is tolerant to 650-mVpp common-mode interference. IEEE J. Solid-State Circuits 52, 2811–2828 (2017).
Mendrela, A. E. et al. A bidirectional neural interface circuit with active stimulation artifact cancellation and cross-channel common-mode noise suppression. IEEE J. Solid-State Circuits 51, 955–965 (2016).
Muller, R., Gambini, S. & Rabaey, J. M. A 0.013 mm2, 5 μW, DC-coupled neural signal acquisition IC with 0.5 V supply. IEEE J. Solid-State Circuits 47, 232–243 (2012).
Steyaert, M. S. & Sansen, W. M. A micropower low-noise monolithic instrumentation amplifier for medical purposes. IEEE J. Solid-State Circuits 22, 1163–1168 (1987).
Kim, S.-J. et al. A 0.5-V sub-μW/channel neural recording IC with delta-modulation-based spike detection. In IEEE Asian Solid-State Circuits Conference 189–192 (IEEE, 2014).
Dong, H., Yuanjin, Z., Rajkumar, R., Dawe, G. & Minkyu, J. 0.45 V 100-channel neural-recording IC with sub-mW/channel consumption in 0.18 mm CMOS. In IEEE Int. Solid-State Circuits Conf. 17–21 (IEEE, 2013).
Muller, R. Low power, scalable platforms for implantable neural recording. PhD dissertation, Univ. California Berkeley (2015); http://www2.eecs.berkeley.edu/Pubs/TechRpts/2015/EECS-2015-19.html
McCreary, J. L. & Gray, P. R. All-MOS charge redistribution analog-to-digital conversion techniques. IEEE J. Solid-State Circuits 10, 371–379 (1975).
Karkare, V., Chandrakumar, H., Rozgić, D. & Marković, D. Robust, reconfigurable, and power-efficient biosignal recording systems. In IEEE Custom Integrated Circuits Conf. 1–8 (IEEE, 2014).
Goldsmith, A. Wireless Communications (Cambridge University Press, 2005).
Miranda, H. & Meng, T. H. A programmable pulse UWB transmitter with 34 energy efficiency for multichannel neuro-recording systems. In IEEE Custom Integrated Circuits Conf. 1–4 (IEEE, 2010).
Obeid, I. & Wolf, P. D. Evaluation of spike-detection algorithms for a brain-machine interface application. IEEE Trans. Biomed. Eng. 51, 905–911 (2004).
Kaiser, J. F. On a simple algorithm to calculate the energy of a signal. Proc. IEEE Int. Conf. Acoust., Speech, Signal Process. 1, 381–384 (1990).
Chase, S. M., Schwartz, A. B. & Kass, R. E. Bias, optimal linear estimation, and the differences between open-loop simulation and closed-loop performance of spiking-based brain-computer interface algorithms. Neural Netw. 22, 1203–1213 (2009).
Willett, F. R. et al. Signal-independent noise in intracortical brain-computer interfaces causes movement time properties inconsistent with Fitts’ law. J. Neural Eng. 14, 026010 (2017).
Gao, P. et al. A theory of multineuronal dimensionality, dynamics and measurement. Preprint at https://www.biorxiv.org/content/10.1101/214262v2 (2017).
Liberti, W. A., Perkins, L. N., Leman, D. P. & Gardner, T. J. An open source, wireless capable miniature microscope system. J. Neural Eng. 14, 045001 (2017).
Foster, J. D. et al. A freely-moving monkey treadmill model. J. Neural Eng. 11, 046020 (2014).
Cunningham, J. P. et al. A closed-loop human simulator for investigating the role of feedback control in brain-machine interfaces. J. Neurophysiol. 105, 1932–1949 (2011).
Santhanam, G., Ryu, S. I., Yu, B. M., Afshar, A. & Shenoy, K. V. A high-performance brain-computer interface. Nature 442, 195–198 (2006).
Jindal, R. Compact noise models for MOSFETs. IEEE Trans. Electron Devices 53, 2051–2061 (2006).
Scholten, A. et al. Noise modeling for RF CMOS circuit simulation. IEEE Trans. Electron Devices 50, 618–632 (2003).
Hariprasath, V., Guerber, J., Lee, S.-H. & Moon, U.-K. Merged capacitor switching based SAR ADC with highest switching energy-efficiency. Electron. Lett. 46, 620–621 (2010).
Razavi, B. The strongarm latch [a circuit for all seasons]. IEEE Solid-State Circuits Mag. 7, 12–17 (2015).
Harpe, P., Gao, H., van Dommele, R., Cantatore, E. & van Roermund, A. H. A 0.20 mm2 3 nW signal acquisition IC for miniature sensor nodes in 65 nm CMOS. IEEE J. Solid-State Circuits 51, 240–248 (2016).
Chandrakumar, H. & Markovic, D. A 15.2-ENOB continuous-time ΣΔ ADC for a 200mV pp-linear-input-range neural recording front-end. In IEEE International Solid-State Circuits Conf. 232–234 (IEEE, 2018).
Murmann, B. The race for the extra decibel: a brief review of current ADC performance trajectories. IEEE Solid State Circuits Mag. 7, 58–66 (2015).
We thank S. I. Ryu for electrode array implantation surgical assistance, M. Risch, M. Wechsler and A. Craig for expert surgical assistance and veterinary care. We thank B. Davis, G. Bell and N. Lam for administrative assistance. We thank W. L. Gore Inc. for donating Preclude artificial dura, used as part of the chronic non-human-primate electrode array implantation procedure. This work was supported by a Stanford Bio-X Institute fellowship (to N.E.-C.), Wu Tsai Neurosciences Institute fellowships (to D.G.M. and S.D.S.), an ALS Association Milton Safenowitz Postdoctoral Fellowship (to S.D.S.), an A. P. Giannini Foundation Postdoctoral Research Fellowship in California (to S.D.S.), a Career Award at the Scientific Interface from the Burroughs Wellcome Fund (to S.D.S.), the Rehabilitation R&D Service, the Department of Veterans Affairs (grants N2864C, A2295R, N9288C and B6453R to L.R.H.), grant NINDS-UH2NS095548 (to L.R.H., J.M.H. and K.V.S.), grant NIDCD-R01DC009899 (to L.R.H., J.M.H. and K.V.S.), grant NIDCD RO1DC014034 (to J.M.H., K.V.S. and L.R.H.), grant NINDS-U01NS098968 (to L.R.H., J.M.H. and K.V.S.), the DARPA ‘NESD’ (to B.M. and K.V.S.), the DARPA ‘NeuroFast’ (grant W911NF-14-2-0013 to K.V.S.), NIH Director’s Pioneer Award 8DP1HD075623 (to K.V.S.), a Simons Foundation grant ‘Simons Collaboration on the Global Brain (SCGB)’ 543045 (to K.V.S.), the Office of Naval Research grant W911NF-14-2-0013 (to K.V.S.), and the Howard Hughes Medical Institute (to K.V.S.).
The MGH Translational Research Center has clinical research support agreements with Neuralink Inc., Paradromics Inc. and Synchron Medical, for which L.R.H. provides consultative input. K.V.S. and J.M.H. are consultants to Neuralink Inc. K.V.S. is on the Scientific Advisory Boards of CTRL-Labs Inc., Mind-X Inc., Inscopix Inc. and Heal Inc. These entities did not support this work.
Publisher’s note Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Even-Chen, N., Muratore, D.G., Stavisky, S.D. et al. Power-saving design opportunities for wireless intracortical brain–computer interfaces. Nat Biomed Eng 4, 984–996 (2020). https://doi.org/10.1038/s41551-020-0595-9
ACS Applied Electronic Materials (2021)
Nature Biomedical Engineering (2020)