Stable in vivo mapping and modulation of the same neurons and brain circuits over extended periods is critical to both neuroscience and medicine. Current electrical implants offer single-neuron spatiotemporal resolution but are limited by such factors as relative shear motion and chronic immune responses during long-term recording. To overcome these limitations, we developed a chronic in vivo recording and stimulation platform based on flexible mesh electronics, and we demonstrated stable multiplexed local field potentials and single-unit recordings in mouse brains for at least 8 months without probe repositioning. Properties of acquired signals suggest robust tracking of the same neurons over this period. This recording and stimulation platform allowed us to evoke stable single-neuron responses to chronic electrical stimulation and to carry out longitudinal studies of brain aging in freely behaving mice. Such advantages could open up future studies in mapping and modulating changes associated with learning, aging and neurodegenerative diseases.
Subscribe to Journal
Get full journal access for 1 year
only $20.17 per issue
All prices are NET prices.
VAT will be added later in the checkout.
Rent or Buy article
Get time limited or full article access on ReadCube.
All prices are NET prices.
Yuste, R. From the neuron doctrine to neural networks. Nat. Rev. Neurosci. 16, 487–497 (2015).
Stanley, G.B. Reading and writing the neural code. Nat. Neurosci. 16, 259–263 (2013).
Cash, S.S. & Hochberg, L.R. The emergence of single neurons in clinical neurology. Neuron 86, 79–91 (2015).
Hochberg, L.R. et al. Reach and grasp by people with tetraplegia using a neurally controlled robotic arm. Nature 485, 372–375 (2012).
Shenoy, K.V. & Carmena, J.M. Combining decoder design and neural adaptation in brain-machine interfaces. Neuron 84, 665–680 (2014).
Bensmaia, S.J. & Miller, L.E. Restoring sensorimotor function through intracortical interfaces: progress and looming challenges. Nat. Rev. Neurosci. 15, 313–325 (2014).
Aflalo, T. et al. Neurophysiology. Decoding motor imagery from the posterior parietal cortex of a tetraplegic human. Science 348, 906–910 (2015).
Perge, J.A. et al. Intra-day signal instabilities affect decoding performance in an intracortical neural interface system. J. Neural Eng. 10, 036004 (2013).
Grady, C. The cognitive neuroscience of ageing. Nat. Rev. Neurosci. 13, 491–505 (2012).
Wang, M. et al. Neuronal basis of age-related working memory decline. Nature 476, 210–213 (2011).
Yeoman, M., Scutt, G. & Faragher, R. Insights into CNS ageing from animal models of senescence. Nat. Rev. Neurosci. 13, 435–445 (2012).
Stam, C.J. Modern network science of neurological disorders. Nat. Rev. Neurosci. 15, 683–695 (2014).
Kornblum, H.I. et al. In vivo imaging of neuronal activation and plasticity in the rat brain by high resolution positron emission tomography (microPET). Nat. Biotechnol. 18, 655–660 (2000).
Poldrack, R.A. & Farah, M.J. Progress and challenges in probing the human brain. Nature 526, 371–379 (2015).
Chang, E.F. Towards large-scale, human-based, mesoscopic neurotechnologies. Neuron 86, 68–78 (2015).
Oxley, T.J. et al. Minimally invasive endovascular stent-electrode array for high-fidelity, chronic recordings of cortical neural activity. Nat. Biotechnol. 34, 320–327 (2016).
Hamel, E.J., Grewe, B.F., Parker, J.G. & Schnitzer, M.J. Cellular level brain imaging in behaving mammals: an engineering approach. Neuron 86, 140–159 (2015).
Berényi, A. et al. Large-scale, high-density (up to 512 channels) recording of local circuits in behaving animals. J. Neurophysiol. 111, 1132–1149 (2014).
Scholvin, J. et al. Close-packed silicon microelectrodes for scalable spatially oversampled neural recording. IEEE Trans. Biomed. Eng. 63, 120–130 (2016).
Yamamoto, J. & Wilson, M.A. Large-scale chronically implantable precision motorized microdrive array for freely behaving animals. J. Neurophysiol. 100, 2430–2440 (2008).
Schwarz, D.A. et al. Chronic, wireless recordings of large-scale brain activity in freely moving rhesus monkeys. Nat. Methods 11, 670–676 (2014).
Polikov, V.S., Tresco, P.A. & Reichert, W.M. Response of brain tissue to chronically implanted neural electrodes. J. Neurosci. Methods 148, 1–18 (2005).
Jackson, A. & Fetz, E.E. Compact movable microwire array for long-term chronic unit recording in cerebral cortex of primates. J. Neurophysiol. 98, 3109–3118 (2007).
Voigts, J., Siegle, J.H., Pritchett, D.L. & Moore, C.I. The flexDrive: an ultra-light implant for optical control and highly parallel chronic recording of neuronal ensembles in freely moving mice. Front. Syst. Neurosci. 7, 8 (2013).
Thompson, L.T. & Best, P.J. Long-term stability of the place-field activity of single units recorded from the dorsal hippocampus of freely behaving rats. Brain Res. 509, 299–308 (1990).
Krüger, J., Caruana, F., Volta, R.D. & Rizzolatti, G. Seven years of recording from monkey cortex with a chronically implanted multiple microelectrode. Front. Neuroeng. 3, 6 (2010).
Vaidya, M. et al. Ultra-long term stability of single units using chronically implanted multielectrode arrays. Conf. Proc. IEEE Eng. Med. Biol. Soc. 2014, 4872–4875 (2014).
Dickey, A.S., Suminski, A., Amit, Y. & Hatsopoulos, N.G. Single-unit stability using chronically implanted multielectrode arrays. J. Neurophysiol. 102, 1331–1339 (2009).
Karumbaiah, L. et al. Relationship between intracortical electrode design and chronic recording function. Biomaterials 34, 8061–8074 (2013).
Kozai, T.D.Y. et al. Ultrasmall implantable composite microelectrodes with bioactive surfaces for chronic neural interfaces. Nat. Mater. 11, 1065–1073 (2012).
Kim, J. et al. Next-generation flexible neural and cardiac electrode arrays. Biomed. Eng. Lett. 4, 95–108 (2014).
Liu, J. et al. Syringe-injectable electronics. Nat. Nanotechnol. 10, 629–636 (2015).
Hong, G. et al. Syringe injectable electronics: precise targeted delivery with quantitative input/output connectivity. Nano. Lett. 15, 6979–6984 (2015).
Xie, C. et al. Three-dimensional macroporous nanoelectronic networks as minimally invasive brain probes. Nat. Mater. 14, 1286–1292 (2015).
Tyler, W.J. The mechanobiology of brain function. Nat. Rev. Neurosci. 13, 867–878 (2012).
Schmitzer-Torbert, N. & Redish, A.D. Neuronal activity in the rodent dorsal striatum in sequential navigation: separation of spatial and reward responses on the multiple T task. J. Neurophysiol. 91, 2259–2272 (2004).
Shadlen, M.N. & Newsome, W.T. The variable discharge of cortical neurons: implications for connectivity, computation, and information coding. J. Neurosci. 18, 3870–3896 (1998).
Skaggs, W.E., McNaughton, B.L., Wilson, M.A. & Barnes, C.A. Theta phase precession in hippocampal neuronal populations and the compression of temporal sequences. Hippocampus 6, 149–172 (1996).
Siapas, A.G., Lubenov, E.V. & Wilson, M.A. Prefrontal phase locking to hippocampal theta oscillations. Neuron 46, 141–151 (2005).
Khodagholy, D. et al. NeuroGrid: recording action potentials from the surface of the brain. Nat. Neurosci. 18, 310–315 (2015).
Florian, M.C. et al. A canonical to non-canonical Wnt signalling switch in haematopoietic stem-cell ageing. Nature 503, 392–396 (2013).
Grion, N., Akrami, A., Zuo, Y., Stella, F. & Diamond, M.E. Coherence between rat sensorimotor system and hippocampus is enhanced during tactile discrimination. PLoS Biol. 14, e1002384 (2016).
Colgin, L.L. Rhythms of the hippocampal network. Nat. Rev. Neurosci. 17, 239–249 (2016).
Henze, D.A. et al. Intracellular features predicted by extracellular recordings in the hippocampus in vivo. J. Neurophysiol. 84, 390–400 (2000).
Bach, M.E. et al. Age-related defects in spatial memory are correlated with defects in the late phase of hippocampal long-term potentiation in vitro and are attenuated by drugs that enhance the cAMP signaling pathway. Proc. Natl. Acad. Sci. USA 96, 5280–5285 (1999).
Cohen, J. Statistical Power Analysis for the Behavioral Sciences 2nd edn (Routledge, 1988).
Cogan, S.F. Neural stimulation and recording electrodes. Annu. Rev. Biomed. Eng. 10, 275–309 (2008).
Quiroga, R.Q., Nadasdy, Z. & Ben-Shaul, Y. Unsupervised spike detection and sorting with wavelets and superparamagnetic clustering. Neural Comput. 16, 1661–1687 (2004).
Schmitzer-Torbert, N., Jackson, J., Henze, D., Harris, K. & Redish, A.D. Quantitative measures of cluster quality for use in extracellular recordings. Neuroscience 131, 1–11 (2005).
Courtin, J. et al. Prefrontal parvalbumin interneurons shape neuronal activity to drive fear expression. Nature 505, 92–96 (2014).
We thank J. Huang for help with recording instrumentation, J. Liu and J. Tian for training in chronic surgery, S. Patel for help with data analysis and stimulation implementation, G. Guitchounts and S. Wolff for help in freely behaving mouse recording implementation and data interpretation, and C. Perez Estrada for helpful discussions. C.M.L. acknowledges support from the Air Force Office of Scientific Research, the Star Family and Fidelity Biosciences Funds, and the Harvard University Physical Sciences and Engineering Accelerator award. This work was performed in part at the Harvard University Center for Nanoscale Systems (CNS), a member of the National Nanotechnology Coordinated Infrastructure Network (NNCI), which is supported by the National Science Foundation.
The authors declare no competing financial interests.
Integrated supplementary information
Components include silicon wafer (light green), nickel relief layer (dark green), polymer ribbons (blue), metal interconnects (black) and exposed metal electrodes (red). For each step (a-g) both top and side views are shown, where the side view corresponds to a cross-section taken at the position indicated by the white horizontal dashed line in the top view image of (a). (h) Zoomed-in views of regions highlighted by black (exposed Pt electrodes) and red dashed boxes (fully passivated interconnects) in (g).
(a) Schematic of the injectable mesh electronics. Blue lines highlight the overall mesh structure and indicate the regions of supporting and passivating polymer layers, the ~horizontal orange lines indicate Au metal interconnects between input/output (I/O) pads (orange filled circles, indicated by the magenta arrow) and Pt recording electrodes (green filled circles), respectively. The solid-line black box at left highlights several of the recording electrodes, and the dashed-line black box in the middle highlights several of the metal interconnects of the mesh electronics. (b) A zoomed-in view of the recording electrodes highlighted by solid-line black box in (a). (c) A zoomed-in view of the mesh electronics highlighted by dashed-line black box in (a). (d) A zoomed-in view of a unit cell of the mesh (dashed-line green box in (c)) with the same color codes as in (a, c). Polymer ribbons (blue) with and without metal interconnects (orange) correspond to longitudinal and transverse elements, respectively. L1 and L2 show spacing between longitudinal and spacing between transverse elements, respectively; W1 and W2 are widths of the longitudinal and transverse mesh elements, respectively; and Wm is the width of metal interconnect lines.
Supplementary Figure 3 Schematics illustrating method used to load the mesh electronics into a glass capillary needle and subsequently inject into a medium.
(a) The injectable mesh electronics (dark blue) suspended in solution (light pink) was drawn into the glass needle such that the I/O pads (yellow dots, indicated by green arrows) of the mesh enter the tube first. (b) After the mesh electronics was fully loaded into the glass needle with mesh end at needle tip (magenta dashed box), the needle was removed from the solution. (c) The glass needle was mounted in an x-y-z manipulator for injection into solution, gel/polymer or tissue (light blue). Black arrows indicate the direction of the fluid flow during loading and injection.
(a) Image of a mouse fixed in a stereotaxic frame with scalp skin retracted, and a hole drilled through the skull plate. The glass needle (yellow arrow) loaded with mesh electronics is visible directly above the skull in this image. A flexible flat cable (FFC) is visible at the left of the image supported on a ceramic scaffold. A 0-80 grounding screw (white arrow) was positioned at the posterior side of the skull. (b) Image showing the relocation of the I/O end of the post-injected mesh electronics (red arrow) towards the FFC. (c) Image post-injection into the brain showing the input/output (I/O) region of the mesh electronics unfolded onto the FFC. Inset shows a zoom-in view of the red dashed box, which highlights the unfolded I/O part (red arrow) of the injected mesh electronics. (d) Image representing the unfolded I/O pads of the mesh electronics which were electrically-connected to the FFC using the conductive ink printing process described in ref. 33. The red dashed box highlighted by the inset shows the details of the bonding with the red and cyan arrows indicating the mesh electronics and conductive ink connections, respectively. (e) Image showing the FFC fixed on top of the mouse skull using dental cement. (f) Image of a fully awake but restrained mouse during chronic in vivo brain recording. The white and black arrows highlight the grounding screw and the connector between FFC and external recording setups, respectively. See Online Methods for all experimental details of surgery and injection. The width of FFC in all panels is ~8 mm.
(a,b) Correlation maps of 16-channel local field potential (LFP) recordings at two (a) and four (b) months post-injection. (c,d) Correlation maps of 16-channel extracellular action potential recordings at two (c) and four (d) months post-injection. Colors indicate the correlation coefficient between any two given channels according to the color bar shown on the far right. All the maps were calculated from 2 s long data traces at both time points. See Online Methods for details of correlation coefficient calculations.
(a) Representative extracellular action potential recordings from the same electrode (Mouse4-ChannelA shown in Fig. 2b) located in somatosensory cortex (CTX) at different time points post-injection. (b) Spike sorting results of the corresponding recordings shown in (a). All of the spikes from the corresponding time points in (a) are included with ca. 50 spikes at each point. (c) Autocorrelation histograms of average waveforms of sorted spikes from 30 1-min segments within a 30-min recording session for Mouse4-ChannelA at 3 months post-injection. (d) Autocorrelation histograms of average waveforms for each of the two identified clusters (color coded in magenta (neuron 1) and green (neuron 2)) for Mouse4-ChannelA at the 8 time points shown in (b). See Online Methods for details of waveform autocorrelation calculations. (e) Time evolution of normalized average peak-to-peak spike amplitudes across all channels with single-unit action potentials recorded (the average was done across 14, 5, 6 and 7 channels for Mouse 1 to 4, respectively) from four mice. The spike amplitude was normalized (value=1.0, gray dashed horizontal lines) against the average peak-to-peak amplitude values between 5 to 26 weeks post-injection for each channel. (f) Average impedance values at 1 kHz across all channels with recorded single-unit action potentials of the four mice shown in (e) plotted as a function of time over the same time period. The shaded areas in (e) and (f) indicate ±1 standard error of the mean (s.e.m.).
Supplementary Figure 7 Immunohistochemical images of horizontal sectioned mouse brains containing the mesh electronics from different times post-injection.
(a) GFAP stained images of representative horizontal brain slices used for normalized fluorescence intensity plots (Fig. 2e) showing the interfaces between mesh electronics and astrocytes, and (b) iba-1 stained images of representative horizontal slices used for normalized fluorescence intensity plots (Fig. 2e) showing the interfaces between mesh electronics and microglia at 2, 6 and 12 weeks post-injection. All samples were 10 μm-thick. Blue, cyan and purple correspond to mesh electronics, GFAP and iba-1, respectively. Scale bars: 100 μm.
Supplementary Figure 8 Autocorrelation and cross-correlation analyses of average spike waveforms across all recording time points.
(a) Auto- and cross-correlation histograms for three identified neurons of Mouse2-ChannelA from week 3 to 34 post-injection shown in Fig. 3b. (b) Auto- and cross-correlation histograms for two identified neurons of Mouse2-ChannelB from week 1 to 34 post-injection shown in Supplementary Fig. 10c. (c) Auto- and cross-correlation histograms for two identified neurons of Mouse3-ChannelA from week 1 to 26 post-injection shown in Supplementary Fig. 10d.
(a-c) Interspike interval (ISI) histograms of the data shown in Fig. 3c at 26 weeks post-injection but replotted with a bin size of 1 ms. The data show clearly a 2-3 ms refractory period (orange-shaded regions) for neuron 1 (a), neuron 2 (b) and neuron 3 (c).
(a,b) Time evolution of representative single-unit spikes of Mouse2-ChannelB (a) and Mouse3-ChannelA (b) shown in Fig. 2 clustered by principal component analysis (PCA) over eight and six months, respectively. In each plot, the x- and y-axis denote the first and second principal component, respectively, and the z-axis indicates post-injection time. (c,d) Time course analysis of average spike waveforms from each PCA cluster shown in (a) from 1 to 34 weeks post-injection (c), and for each PCA cluster shown in (b) from 1 to 26 weeks (d), respectively. (e,f) Time evolution of ISI histograms of each of the 2 neurons identified from the PCA clusters in (a) and (b) from 3 to 34 (e) and 3 to 26 weeks (f), respectively. Bin size: 20 ms. (g,h) Scatter plot with analysis of variance (ANOVA) (n=32 and 26 for (g) and (h), respectively) of the firing parameter, λ, obtained by fitting each ISI distribution profile shown in (e) and (f) to an exponential decay. All plots for Mouse3 are shown up to 26 weeks post-injection instead of 34 weeks shown for Mouse2, because Mouse3 was older than Mouse2 when the initial mesh electronics injection was carried out, and thus exhibited ageing-associated changes (shown in Fig. 5 and Supplementary Fig. 12) at an earlier post-injection time point.
(a-c) Rayleigh Z-test of all recording data for neuron 1 (a), neuron 2 (b) and neuron 3 (c) identified for Mouse2-ChannelA from week 3 to 34 post-injection (n=32). Each trial presents the recording data at a given week. The null hypothesis is: a given neuron does not have phase-locking behavior at a given week. The majority of trials on the right side of the vertical dashed lines suggest a >95% confidence interval (p<0.05) to reject the null hypothesis.
(a,b) Time evolution of representative single-unit spikes of Mouse2-ChannelA (a) and Mouse3-ChannelA (b) shown in Fig. 2 clustered by PCA from 35 to 57 weeks of age. In each plot, the x- and y-axes denote the first and second principal components, respectively, and the z-axis indicates the corresponding mouse age. (c,d) Representative 2D (PC1-PC2 plane) plots of the PCA results shown in (a,b) at week 52 and 57 of mouse age. (e,f) Time evolution of ISI histograms of each of the neurons identified from the PCA clusters in (a) and (b) from 35 to 57 weeks of age. Bin size: 20 ms. (g) Impedance values at 1 kHz of the two channels shown in (a-f) plotted as a function of mouse age over the same period. (h) Immunohistochemical images of a 10 μm-thick horizontal CTX brain slice showing the mesh electronics/brain tissue interface at 1-year mouse age. Red, yellow and blue colors correspond to neurofilaments, NeuN and mesh electronics, respectively. Scale bar: 100 μm. (i) Neurofilament (red), NeuN (yellow), GFAP (cyan) and iba-1 (purple) fluorescence intensity normalized against background values (gray dashed horizontal lines) plotted versus distance from the interface (see Online Methods). The pink-shaded regions indicate the interior of the mesh electronics. All error bars in this figure reflect ±1 standard error of the mean (s.e.m.).
(a) Sorted single-unit action potentials from chronic freely behaving mouse recordings. Each column represents the sorted spikes from an individual channel shown in Fig. 6 at 5, 10 and 27 weeks post-injection. (b) Noise distributions of all red unit clusters of Channels A-E shown at week 10 in a (I), and the intrinsic recording noise of each corresponding electrode of the channel at the same week (II). See Online Methods for details of noise calculations. (c) Single-unit spikes of Channel D in (a) clustered by PCA when the mouse was actively whisking (I) and foraging (II) at week 5, 10 and 27.
Supplementary Figure 14 Autocorrelation analyses of average waveforms from week 1 to 8 post-injection.
(a) Autocorrelation histograms of average waveforms for neurons 1 (magenta) and 2 (green) for Mouse2-ChannelA from weeks 1 to 8 post-injection shown in Fig. 3b. (b) Autocorrelation histograms of average waveforms for neurons 1 (magenta) and 2 (green) for Mouse2-ChannelB from weeks 1 to 8 shown in Supplementary Fig. 10c. Inset: Cross-correlation histogram of all raw spikes for neuron 2 between weeks 1 and 8 (light green, dashed line) and autocorrelation histogram of all raw spikes for neuron 2 on week 8 only (dark green, solid line). (c) Autocorrelation histograms of average waveforms for neurons 1 (magenta) and 2 (green) for Mouse3-ChannelA from weeks 1 to 8 shown in Supplementary Fig. 10d. See Online Methods for details of autocorrelation calculations.
Supplementary Figures 1–14 and Supplementary Tables 1–3 (PDF 2325 kb)
This video shows a mouse with a head-mounted voltage-amplifier roaming freely in a cage with randomly positioned food pellets during recording. The frame rate is 25 frames per second (fps) and the video is played at 1× real time. The mesh electronics was injected into the somatosensory cortex and hippocampus, and bonded to the interface cable as described in the Online Methods. (MOV 4511 kb)
About this article
Cite this article
Fu, T., Hong, G., Zhou, T. et al. Stable long-term chronic brain mapping at the single-neuron level. Nat Methods 13, 875–882 (2016). https://doi.org/10.1038/nmeth.3969
Brain Structure and Function (2020)
Longitudinal neural and vascular structural dynamics produced by chronic microelectrode implantation
Journal of Neural Engineering (2020)
Surface Modifications of an Organic Polymer-Based Microwire Platform for Sustained Release of an Anti-Inflammatory Drug
ACS Applied Bio Materials (2020)
Nanoscale Advances (2020)