Flow cytometry has become a highly valuable method to monitor minimal residual disease (MRD) and evaluate the depth of complete response (CR) in bone marrow (BM) of multiple myeloma (MM) after therapy. However, current flow-MRD has lower sensitivity than molecular methods and lacks standardization. Here we report on a novel next generation flow (NGF) approach for highly sensitive and standardized MRD detection in MM. An optimized 2-tube 8-color antibody panel was constructed in five cycles of design-evaluation-redesign. In addition, a bulk-lysis procedure was established for acquisition of ⩾107 cells/sample, and novel software tools were constructed for automatic plasma cell gating. Multicenter evaluation of 110 follow-up BM from MM patients in very good partial response (VGPR) or CR showed a higher sensitivity for NGF-MRD vs conventional 8-color flow-MRD -MRD-positive rate of 47 vs 34% (P=0.003)-. Thus, 25% of patients classified as MRD-negative by conventional 8-color flow were MRD-positive by NGF, translating into a significantly longer progression-free survival for MRD-negative vs MRD-positive CR patients by NGF (75% progression-free survival not reached vs 7 months; P=0.02). This study establishes EuroFlow-based NGF as a highly sensitive, fully standardized approach for MRD detection in MM which overcomes the major limitations of conventional flow-MRD methods and is ready for implementation in routine diagnostics.
In recent years, the landscape of treatment of multiple myeloma (MM) has substantially changed1, 2, 3, 4 leading to significantly increased complete response (CR) rates and survival.5, 6, 7, 8, 9, 10, 11 Therefore, CR has become a major goal in newly diagnosed MM, even in aged patients.5, 12, 13, 14, 15 Despite these advances, most CR patients will ultimately relapse.7, 10 Consequently, better insight in depth of treatment response is required and more sensitive methods are needed for detection of minimal residual disease (MRD), particularly in cases that reached CR and stringent CR (sCR).16
Conventional 4–8-color flow cytometry,17, 18, 19, 20, 21, 22, 23, 24 and to a lesser extent also allele-specific oligonucleotide quantitative PCR (ASOqPCR) and next generation sequencing (NGS),25, 26, 27, 28, 29 are progressively being used to monitor MRD in bone marrow (BM) of MM after therapy.16, 30 These studies confirmed the relevance of MRD measurements for identification of MM patients at higher risk of relapse.16, 22 However, despite the greater sensitivity of the MRD approaches (vs classical CR/sCR criteria), identification among MRD− cases of patients that will eventually relapse vs those who are potentially cured still remains a challenge, implying that more sensitive MRD approaches are needed.16, 31
High sensitivity and broad applicability have both become mandatory requirements for MRD monitoring in MM.16 Early studies have shown that conventional 4–6-color flow-MRD is applicable in virtually all MM patients (⩾95%), whereas ASOqPCR and NGS have a more restricted applicability (50–90% of cases), mainly due to the high number of somatic hypermutations, which cause variable levels of primer annealing with unpredictable amplification/quantitation results.16, 25, 32, 33, 34 However, the sensitivity of conventional flow-MRD (<10−4) remains (systematically) lower than that of ASOqPCR and NGS (<10−4–10−6).25, 26, 29, 35 More recently, several studies in MM have shown that conventional 8-color flow-MRD assays have an increased sensitivity – limit of detection (LOD) of between <10−4 and <10−5-, leading to a significantly improved prediction of outcome.24, 36 However, current flow-MRD assays and NGS still suffer from a major limitation: lack of standardization.37 Actually, different markers and antibody panels, distinct numbers of cells measured, and highly variable criteria for MRD positivity are currently applied worldwide.38, 39 Therefore, standardization efforts have been made and consensus recommendations and guidelines have been recently proposed.40, 41, 42 However, such consensus recommendations still rely on subjective ‘expert-shared’ knowledge and experience, and do not completely solve the lack of technical standardization, whereas prospectively validated flow-MRD approaches are still missing.39
Here we report on the design of a EuroFlow-based next generation flow (NGF) approach for highly sensitive and standardized detection of MRD in MM, and the results of its validation vs a conventional 8-color flow-MRD method24 and NGS. The novel NGF-MRD approach takes advantage of innovative tools and procedures recently developed by the EuroFlow Consortium for sample preparation, antibody panel construction (including choice of type of antibody and fluorochrome), and automatic identification of plasma cells (PC) against reference databases of normal and patient BM.43, 44, 45, 46 Prospective validation of the whole procedure at two distinct centers confirmed its robustness and significantly greater sensitivity vs conventional 8-color flow-MRD approaches, comparable to current NGS methods, with an improved prediction of patient outcome.
Subjects and methods
Patients, controls and samples
A total of 375 BM and 10 peripheral blood EDTA-anticoagulated samples from 53 controls and 332 adult plasma cell disorder (PCD) patients, were studied (Supplementary Table 1) to design Version 1 of the MM-MRD antibody panel (n=94; 31 normal/reactive and 63 MM studied at diagnosis), to compare the performance of antibody reagents evaluated in Versions 2–5 of the NGF-MRD panel (19 BM diagnostic patient samples), to evaluate distinct sample preparation protocols (n=10 peripheral blood and 8 BM samples) and to validate NGF (Version 5) against conventional 8-color flow-MRD -n=244 consecutively recruited samples corresponding to 16 healthy donors, 66 PCD samples studied at diagnosis and 162 MM patient samples investigated during follow-up, including 110 follow-up BM samples from MM patients evaluated at ⩾very good partial response (VGPR): 39 VGPR; 52 CR; and 19 sCR cases-. Twenty-one ⩾VGPR BM samples and 10 additional BM samples -22/31 with very low MRD levels (⩽10−4)- were further used to blindly compare NGF vs NGS (Supplementary Table 1).
The study was approved by the local ethics committees and written informed consent was given by each donor according to the Declaration of Helsinki. All samples were processed at each participating center (USAL/HUSAL, UNAV, EMC, UNIKIEL, IPOLFG, SUM) within 24 h after collection.
Immunophenotypic studies for selection of plasma cell (PC)-associated markers
BM samples used to establish Version 1 (Table 1) of the MM-MRD panel were stained with the EuroFlow PCD 8-color panel for a total of 12 different markers (CD38, CD138, CD45, CD19, CD27, CD28, CD56, CD81, CD117, CyIgκ, CyIgλ and β2-microglobulin), using previously described EuroFlow sample preparation protocols.44, 45 For data analysis, events from both 8-color tubes (per sample) were merged and the values of all parameters measured per tube were mathematically calculated for the individual PC events using the merge and calculation functions of the Infinicyt software (Cytognos SL, Salamanca, Spain).43, 44, 45 Subsequently, phenotypic data on normal PC (nPC) from 31 normal/reactive BM samples were merged in a reference database. PC were identified based on their unique pattern of expression of CD38, CD138, CD45 and light scatter features, following consensus recommendations.40, 41, 42 Principal component analysis44, 45 of single PC events was used to compare the reference database of nPC vs aberrant/clonal PC (aPC) from each of 63 MM BM studied at diagnosis, to identify the most discriminating markers that distinguished between (reference) nPCs and aPC from individual MM patients (Figure 1), and establish the applicability of the method. Additionally, CD200 expression was also evaluated on PC in a subset of 28 MM.
Results on the comparison between antibody reagents directed against the same marker (Supplementary Table 2), was based on their staining profiles on nPC vs aPC (vs other non-PC BM populations) as defined by median fluorescence intensity (range: 0–262,144 arbitrary units) and stain index values, as previously described47 (Supplementary text). The EuroFlow reagent evaluation criteria44 were used to discard or accept individual reagents: (i) increased background fluorescence; (ii) dim fluorescence of PCs vs other BM populations (CD38, CD138, CyIgκ+ or CyIgλ+) or the positive vs negative BM reference populations (CD19, CD27, CD56, CD45, CD81 vs CD117) and (iii) interaction with the staining pattern of other reagents.
Design and evaluation of sample preparation protocols
For the evaluation of different sample preparation protocols, eight BM and 10 peripheral blood samples were stained in parallel with the CD138-HV450 CD45-PacO CD56-PE CD5-PerCPCy5.5 CD19-PECy7 CD3-APC antibody combination under five different conditions: (i) the EuroFlow standard operating procedure (SOP) for staining of 50 μl of sample with cell surface markers,44 and (ii) four different ammonium chloride-based bulk-lysis procedures followed by staining of 5 × 106 cells in 100 μl/tube (final concentration of 5 × 104 cells/μl), as described in detail in the Supplementary text.
Validation of the NGF MM-MRD method
Overall, 228 MM diagnostic (n=66) and follow-up (n=162) BM samples (n=110 in VGPR or CR/sCR) were evaluated in parallel with the NGF MRD approach vs local routine flow-MRD methods (that is, conventional 8-color flow-MRD technique).24 Detailed description of these BM samples, related patient clinical data, disease status and time points at evaluation is provided in Supplementary Tables 1, 3 and 4. Briefly, conventional flow-MRD was based on staining of 300 μl of whole BM with a single 8-color antibody combination (CD45-PacB CD138-OC515 CD38-FITC CD56-PE CD27-PerCPCy5.5 CD19-PECy7 CD117-APC CD81-APCH7), as previously described.24 In turn, for the NGF approach a median volume of 1.5±1.3 ml (range: 0.1–5.3 ml) was employed adding up to a median total sample volume of 1.8 ml (maximum of 5.6 ml). PC populations that coexisted in individual BM samples were identified based on a combination of the CD38, CD138, CD45 PC-associated markers and light scatter characteristics, the presence vs absence of myeloma-associated phenotypes, plus CyIg light chain restriction in case of NGF, as described elsewhere.48 According to consensus recommendations,42 the limit of quantitation and the LOD of the NGF MRD method was calculated at <5 × 10−6 and <2 × 10−6 aPC, based on the identification of ⩾50 and ⩾20 aPC among 107 events, respectively. More detailed information about instrument conditions, data acquisition and analysis, and the specific reagents used in the present study is provided as Supplementary Material.
Automatic identification and enumeration of aPC was performed in 110 MM BM follow-up (VGPR or CR/sCR) samples using the automatic gating function of the Infinicyt software and previously described procedures,49, 50 and the results were compared against the conventional expert-guided PC-identification/gating approach. For automatic gating, a database consisting of a subset of 14 normal BM samples stained with Version 5 of the antibody panel was constructed and used.50
In a subset of 31 MM follow-up BM samples with low MRD levels (for example, ⩽10−4) in which enough DNA was available, MRD was also evaluated by NGS. For this purpose, patient-specific VDJH rearrangements were amplified and directly sequenced from DNA extracted from diagnostic samples using the DNAzol reagent (MRC, Cincinnati, OH, USA) and IGHV family-specific primers that covered framework regions 1 (FR1) and FR2, plus a JH consensus primer, as described elsewhere.26, 32 VDJH rearrangements identified at diagnosis were used as MRD-targeted sequences for subsequent follow-up samples. Follow-up DNA samples were amplified using the LymphoTrack IGH Assay (InVivoScribe Technologies, San Diego, CA, USA) and sequenced in an Illumina MiSeq platform (Illumina, San Diego, CA, USA). To all reactions, a known amount of DNA from the MWCL-1 cell line was added as reference control for cell enumeration. The Fastq files generated were analyzed with the LymphoTrack/MiSeq Software (InVivoScribe/Illumina). The number of MRD cells was calculated from the number of reads for the diagnostic VDJH target rearrangements and the number of reads of the reference rearrangement; percentage MRD was calculated upon dividing the number (× 100) of MRD cells by the total number of cells in the reaction.
Number of samples required in the assay development and validation experiments were calculated using the hypothesis contrast strategy for the comparison of paired quantitative data (EPIDAT 4.0 software, Consellería de Sanidade, Xunta de Galicia, Spain). To evaluate the distribution of MRD data, the Kolmogorov–Smirnov test was used. The Wilcoxon or Friedman tests and the Mann–Whitney U or the Kruskal–Wallis tests were used to assess the (two-sided) statistical significance of differences observed between ⩾2 groups for paired and unpaired variables, respectively. For correlation studies, the (two-sided) Spearman’s rho (ρ) for non-parametric paired data was employed. The Kaplan–Meier method and the (two-sided) log-rank test were used to plot and compare progression-free survival (PFS) curves. PFS was defined as the time from MRD assessment to either disease progression or the last follow-up visit. Statistical significance was set at P<0.05. All samples evaluated were blindly analyzed during the experimental phase.
Antibody panel construction for optimal identification of MM plasma cells at MRD levels
Comparison of the whole immunophenotypic profile of aPC from individual MM patients (n=63) vs the normal/reactive BM PC database (n=31) showed multiple aberrant phenotypes in every case (Figure 1). Eight of the 12 markers evaluated contributed most frequently to the discrimination between aPC and nPC based on principal component analysis of single PC phenotypes: CD19 (97% of cases); CD45 (89%); CD56 (86%); CD81 (86%); CyIgλ (73%); CD27 (71%); CD117 (60%); and CyIgκ (56%). Re-evaluation of the utility of the combination of these eight top markers alone in the same 63 MM BM confirmed clear-cut distinction between aPC and nPC in the database, in every case. Consequently, the six surface membrane markers of this list, plus the CD138 and CD38 PC-identification markers, were selected to be combined in a single 8-color tube. In a second 8-color antibody combination, CyIgκ and CyIgλ were added to the CD138 and CD38, together with CD229 as an extra PC-identification antigen, plus the three most informative markers (CD19, CD45, CD56), for parallel confirmation of immunoglobulin light chain (κ vs λ) restriction of PC suspected to be (clonal) myeloma PC (Versions 2 and 3).
Optimization of the two 8-color MM MRD antibody combinations
Subsequently, evaluation of the same 63 MM diagnostic samples using the two 8-color antibody combinations selected above, but focusing now on the detection of minimal numbers (that is, 0.02–0.1%) of MM PC, was performed using virtual (software) dilution experiments of decreasing numbers of PC in the nPC reference database, as previously described.51 This revealed suboptimal reagent performance (for example dim staining) for two fluorochrome positions (that is, CD138-PacO and CD81/CyIgλ-APCH7).
From this initial Version 1, until the final version of the antibody panel (Version 5), four other versions of different fluorochrome-conjugated reagents of the same markers were tested (Table 1) as described in detail in Supplementary text. Briefly, in Version 2, the suboptimal CD138-PacO reagent was replaced by CD138-HV500C; in Version 3, suboptimal CD81-APCH7 and CyIgλ-APCH7 reagents were both replaced by CD81-APCC750 and CyIgλ-APCC750. For Version 4, the CD138-HV500C reagent found to be still suboptimal was replaced, together with the CD27-PerCPCy5.5 and CD45-PacB fluorochrome positions, by CD138-HV450, CD27-HV500C and CD45-PerCPCy5.5, respectively. Finally, in Version 5, CD138-HV450, CD27-HV500C and CD38-FITC were replaced by the optimized CD138-BV421 and CD27-BV510 conjugates, and the multi-epitope CD38-FITC antibody, respectively. The later CD38 reagent showed an equivalent performance in diagnostic MM samples to that of the original CD38 antibody clone, but a much better resolution in BM samples from MM patients who had received Daratumumab therapy (Supplementary Figure 1). Moreover, CD229 was excluded from tube 2, since this marker did not identify all aPC in 4/49 (8%) MM cases tested, and it was not specific for PC, being also (strongly) expressed on plasmacytoid dendritic cells and a subset of lymphocytes.52
Evaluation of sample preparation protocols
Overall, bulk-lysis procedures were systematically associated with acquisition of a significantly (P<0.05) greater number of cells vs the conventional BD FACS Lysing Solution (BD Biosciences, San Jose, CA, USA) based (FACS-lyse) SOP (Supplementary Figure 2A). However, all bulk-lysis conditions but that using low bovine serum albumin (0.5% bovine serum albumin) and a FACS-lysing-fixation step (protocol A1 in Supplementary Figure 2), showed a significantly higher proportion of debris and dead cells (P<0.05 vs FACS-lyse protocols) associated with similar numbers (P>0.05) of cell doublets (Supplementary Figure 2B). Of note, detailed analysis of the specific recovery of the major leukocyte populations (that is, eosinophils, neutrophils, monocytes, mature lymphocytes and nucleated red cells) as well as of nPC and aPC showed no significant differences between the conditions evaluated (P>0.05), except for higher nPC percentages for the bulk-lysis protocol B1 (Supplementary Figures 2C and D). Therefore, the bulk-lysis procedure including a FACS-lysing-fixation step and 0.5% bovine serum albumin (protocol A1 in Supplementary Figure 2) was selected as the reference SOP and used to further titrate the individual antibody reagents selected, for staining of ⩾107 cells/tube (Supplementary Table 5).
Validation of the EuroFlow-based NGF MM-MRD method
The final NGF MM-MRD approach described above was validated against conventional 8-color flow-MRD in 228 MM BM samples studied at diagnosis (n=66) or after therapy (n=162), particularly focusing on 110 BM samples from patients in VGPR or CR/sCR. A strong correlation was found between both methods in diagnostic and follow-up samples with relatively high tumor burden from patients in partial response, stable disease and progressive disease (ρ=0.96; P<0.001; Figure 2a). Most importantly, a fairly good overall correlation was also observed among cases in VGPR and CR/sCR (ρ=0.77; P<0.001; Figure 2b), albeit significantly different rates of MRD+ samples were detected with both methods: 37/110 (34%) for conventional flow-MRD vs 52/110 (47%) for EuroFlow-NGF, respectively (P=0.003; Figure 2b). This was due to a relatively high number of discrepant cases, which were mostly MRD+ by EuroFlow-NGF but MRD− by conventional flow: 18/21 (86%) vs 3/21 (14%) discrepant samples, respectively. Of note, such discrepant NGF+ cases typically showed MRD levels <10−4 by EuroFlow-NGF with median (range) MRD levels of 0.001% (0.0001–0.03%) vs 0.02% (0.0008–1.79%) for MRD+ cases by both methods (P<0.001; Figures 2b and 3). Interestingly, in three MRD− cases by EuroFlow-NGF, MRD+ results at relatively high levels (median of 0.01%; range: 0.006–0.02%) were observed by conventional MRD-flow (Table 2). Evaluation of cytoplasmic κ/λ expression in the suspicious PC from these three patients by EuroFlow-NGF, demonstrated the polyclonal nature of the suspected aPC in 2/3 cases, indicating false-positivity in conventional flow-MRD; in contrast, no clear explanation was found for the discrepant results observed in the other patient. Overall, the frequency of aberrant expression profiles for individual markers on clonal PC by NGF, was as follows: CD45, 96%; CD19, 96%; CD56, 96%; CD27, 89%; CD81, 79%; CD38, 77%; CD117, 48% and CD138, 37%. In 8/52 MRD+ cases confirmation of light chain restriction among small numbers of suspicious PC carrying slightly aberrant phenotypes was required. No significant differences were observed in the validation phase between the participating centers with respect to rate and type of MRD discrepant cases (P=0.63). Importantly, in all but 7/110 cases, >7 × 106 cells were evaluated (median 10.4 × 106 cells) with an impact on the sensitivity of the method because of not reaching the LOD in only 2 cases (1.8%). Interestingly, an alarm for decreased percentage of CD117hi mast cells (⩽0.002%) suggesting blood contamination, was observed in 17/110 samples, 11/17 MRD− and 6/17 MRD+ samples (Table 3).
Automatic identification and enumeration of aPCs showed an excellent correlation with expert-based gating, also in VGPR and CR/sCR cases (ρ=0.96; P<0.001; Figure 2d). However, due to the minimum number of events required by the software algorithm, aPC were not identified in 3/110 cases (2.7%) with low MRD levels by NGF (and MRD− by conventional flow-MRD) -median % aPC (range): 0.0006% (0.0001–0.005%; Figure 2d)-.
Parallel assessment of MRD by NGF and NGS in a subset of 31 samples showed a higher applicability for the EuroFlow-NGF approach: 31/31 (100%) vs 27/31 (87%) cases, respectively (P<0.001). Among those 27 cases assessed by both methods (22 of them with MRD levels ⩽10−4), a good correlation was found between the percentage of residual aPC by NGF and NGS (ρ=0.62; P=0.001; Figure 2c). However, NGF showed a higher sensitivity than NGS with 19/27 (70%) vs 14/27 (52%) MRD+ samples (P=0.06) with higher MRD levels -mean percentage±s.d. MRD+ cells of 0.01±0.04% vs 0.006±0.02% (P=0.07), respectively-. This was due to 7/27 discrepant cases including six MRD+ by NGF and MRD− by NGS: median (range) percentage aPC of 0.001% (0.0002–0.07%) and one MRD+ by NGS (0.0004% aPC) and negative (LOD <0.0002% aPC) by NGF (Figures 2c and 3). In fact, 8/27 samples (30%) were NGF-positive (quantifiable), but NGS-negative or discrepantly low positive.
From the prognostic point of view, MM patients who were MRD− by NGF had a significantly (P=0.01) longer PFS vs MRD+ cases -75% PFS not reached (NR) vs 10 months; Figure 4a-, including also those that were MRD+ by NGF and MRD− by conventional 8-color flow (75% PFS of 10 months; Figure 4b); similar results were observed when the analysis was restricted to patients in CR/sCR (P=0.02; 75% PFS NR vs 7 and 5 months, respectively; Figures 4c and d). Of note 2/6 patients who were MRD+ by NGF and MRD− by NGS also showed disease progression, while the only NGF−/NGS+ discrepant patient remained in continuous CR after 14 months follow-up.
MRD detection in BM has proven clinically meaningful for MM monitoring after therapy, particularly to predict outcome of patients that reach CR12, 13, 14, 15, 16, 29 independently of therapy.22, 53 Currently, several different flow- and PCR-based MRD approaches are available.30 Flow-MRD has clear advantages vs PCR-based methods because of its relative simplicity, high speed, greater clinical applicability and worldwide availability.16, 25 However, major concerns have been recently raised about the lack of standardization of flow-MRD in MM38 and its lower sensitivity vs PCR-based approaches, particularly NGS.25, 29, 53, 54, 55
Here we describe an innovative EuroFlow-based high-sensitive, standardized and validated NGF-MRD method which can be applied to virtually every MM patient for MRD monitoring in BM after therapy. Overall, our results show a similar applicability but a significantly increased sensitivity for the novel EuroFlow-NGF approach vs conventional flow-MRD, with around one fourth of all MRD-negative samples by conventional flow becoming MRD-positive by NGF. To the best of our knowledge, this is the first time that a validated high-sensitive flow-MRD assay is described with a LOD close to 10−6 (ability to identify down to 20 tumor PCs among 107 evaluated BM cells) and a limit of quantitation of <5 × 10−6 (ability to accurately quantity tumor PC percentages at levels down to five cells per million cells, that is, 0.0005%), calculated following consensus recomendations.42 Importantly, the EuroFlow-NGF approach provided similar results in different centers, which further confirms the high standardization level of the method. From the clinical point of view, despite the (still) limited follow-up time, the NGF approach already showed a significant prognostic impact on PFS of MM patients, even among those in CR/sCR, significantly improving the predictive clinical value of conventional flow-MRD.
The greater sensitivity of NGF vs conventional flow-MRD was mostly due to the use of both an optimized combination of fluorochromes and antibody reagents for increased specificity at very low MRD levels, and the 10-fold increase in the number of cells evaluated, in the context of fully standardized laboratory protocols. The two-tube approach proposed also allows confirmation in a second independent measurement, of the clonal nature of suspicious (low numbers of) PC, through evaluation of the cytoplasmic κ/λ restriction of phenotypically aberrant PC which proved to be required in a significant number of cases. Interestingly, the phenotypic markers selected as most informative did not differ from those considered to be essential by expert consensus.40, 41, 48 However, we proved here that selection of optimal fluorochrome-conjugated antibody clones per marker could not be predicted by pre-existing (shared) expertise. Thus, identification of the optimal marker combinations for the 2-tube 8-color antibody panel required five rounds of optimization of what we already considered initially to be a potentially ‘optimal’ panel (that is, Version 1). Major limitations of suboptimal reagents were: (i) increased non-specific and background fluorescence; (ii) too dim fluorescence intensity; (iii) specific interactions among mixed reagents and/or; (iv) suboptimal staining or reduced reactivity on nPC vs aPCs, particularly on Daratumumab-treated vs non-treated patients. Altogether, these results indicate that (extensive) prospective testing is mandatory to define optimal combinations of reagents for flow-MRD monitoring in MM, due to the problems encountered and the significantly different staining profiles obtained with distinct combinations of reagents of the same CD markers. As an example, only two of the many (n=9) CD38 antibody clones evaluated proved efficient for detecting CD38 on PC from Daratumumab-treated MM patients (Supplementary Figure 1); even more, only one of these two clones proved to be effective for detecting CD38 on MM PC treated in vitro with the Isatuximab antibody (Supplementary Figure 1).
Although the specific combination(s) of markers used is a key factor for optimal identification of BM PC and discrimination between nPC and aPC, another critical factor in building a sensitive flow-MRD technique is the number of cells analyzed.42, 56 To the best of our knowledge, this is the first report in which consistently >5 × 106 cells/patient sample (usually >107 cells) were investigated. For the most frequently used stain-lyse-and-then-wash sample preparation flow-MRD procedures, hundred thousand to 1–2 million cells in ⩽300 μl BM are analyzed. In contrast, the here described EuroFlow SOP assured acquisition of >107 events in most MM BM MRD samples, by means of staining a median of 1.5 ml of total BM sample, with a proven limit of quantitation for the NGF MRD method of <5 × 10−6 and a LOD of <2 cells per million. These features contribute to explain the higher sensitivity of EuroFlow-based NGF vs conventional flow-MRD, and NGS. The relatively high frequency of discrepant results (30%) in the NGF vs NGS comparison among cases with low (<10−4) MRD levels, might be caused by suboptimal annealing of the NGS-PCR primers due to high levels of somatic hypermutations in the IG genes of nPC and aPC,16, 32, 34 and deserves further investigation.
Independently of its potentially greater sensitivity, NGF has other additional advantages over NGS:37 it is faster (<4 h), standardized and reproducible, it has a greater applicability (⩾98%), and it does not require a diagnostic sample or patient-specific probes, which potentially lead to a lower variability in the sensitivity reached per patient.16 However, EuroFlow-based NGF required fresh material analyzed within 24 h after sampling; this is feasible in virtually all countries, since the standardized EuroFlow procedures have now been implemented worldwide. In addition, the costs of the NGF reagents (∼100 USD) and assay (∼350 USD), are estimated to be lower than those of NGS (∼350 USD and 700 USD, respectively).16
Importantly, NGF can also provide an overall assessment of the quality of the patient sample through identification of a significant decrease in non-PC BM cell populations (for example, CD117hi mast cells, CD45− sideward-scatterlo nucleated red blood cells, CD117+ myeloid precursors, CD19+CD38hiCD45lo B-cell precursors and CD19− nPC) in hemodiluted BM samples (Table 3) and through providing full insight in the normal B-cell compartment via the identification of normal BM residual B-cell precursors (CD19+CD38hiCD45lo and CD45int), immature B-lymphocytes (CD19+CD45hiCD38lo), naive B cells (CD19+CD38−CD27−), memory B-lymphocytes (CD19+CD38−CD27+), and nPC (CD19+CD56−, CD19−CD56− and CD19−CD56lo) in addition to myeloma PC (Supplementary Figure 3). Of note, decreased numbers of mast cells were found in 17/110 ⩾VGPR BM samples, particularly among MRD− cases, including the only two patients that showed disease progression (Table 3), pointing out the need for careful evaluation of MRD− cases for blood contamination. Whether evaluation of an additional sample from the same patient is required in the such cases, still remains to be established.
The relatively short time needed for the complete EuroFlow-NGF procedure (<4 h) can be further reduced by implementing automatic sample preparation procedures, pre-mixed and dried antibody cocktails, and software algorithms for automatic data analysis. Such improvements are ongoing and will further contribute to prevent diagnostic errors.
A major challenge we faced during the design phase was to determine whether the 2-tube 8-color NGF approach could be replaced by a single 10-color tube to decrease reagent costs and data acquisition time, as suggested by others.57 Direct comparison of Version 5 of the two 8-color tube antibody panel vs three different versions of a single 10-color antibody combination (Supplementary text) showed quite comparable results for both formats. However, the 2-tube 8-color method emerged as a more robust assay because (i) higher numbers of cells were measured; (ii) the confirmatory value of the second tube in case of small populations of suspicious PC found in the first tube; and (iii) the increased consistency and precision of replicate vs single measurements.
In summary, here we describe a novel validated EuroFlow-NGF assay for high-sensitive, fast and standardized quantification of MRD in MM that overcomes previous limitations of conventional flow-MRD approaches and improves prediction of patient outcome. This method is ready-to-use and well-suited for implementation in clinical trials to establish the diagnostic role of MRD in MM.
Ocio EM, Richardson PG, Rajkumar SV, Palumbo A, Mateos MV, Orlowski R et al. New drugs and novel mechanisms of action in multiple myeloma in 2013: a report from the International Myeloma Working Group (IMWG). Leukemia 2014; 28: 525–542.
Bianchi G, Richardson PG, Anderson KC . Promising therapies in multiple myeloma. Blood 2015; 126: 300–310.
Lonial S, Durie B, Palumbo A, San-Miguel J . Monoclonal antibodies in the treatment of multiple myeloma: current status and future perspectives. Leukemia 2016; 30: 526–535.
Rajkumar SV, Kumar S . Multiple myeloma: diagnosis and treatment. Mayo Clin Proc 2016; 91: 101–119.
Lahuerta JJ, Mateos MV, Martinez-Lopez J, Rosinol L, Sureda A, de la Rubia J et al. Influence of pre- and post-transplantation responses on outcome of patients with multiple myeloma: sequential improvement of response and achievement of complete response are associated with longer survival. J Clin Oncol 2008; 26: 5775–5782.
Sonneveld P, Goldschmidt H, Rosinol L, Blade J, Lahuerta JJ, Cavo M et al. Bortezomib-based versus nonbortezomib-based induction treatment before autologous stem-cell transplantation in patients with previously untreated multiple myeloma: a meta-analysis of phase III randomized, controlled trials. J Clin Oncol 2013; 31: 3279–3287.
Usmani SZ, Crowley J, Hoering A, Mitchell A, Waheed S, Nair B et al. Improvement in long-term outcomes with successive total therapy trials for multiple myeloma: are patients now being cured? Leukemia 2013; 27: 226–232.
Palumbo A, Bringhen S, Larocca A, Rossi D, Di Raimondo F, Magarotto V et al. Bortezomib-melphalan-prednisone-thalidomide followed by maintenance with bortezomib-thalidomide compared with bortezomib-melphalan-prednisone for initial treatment of multiple myeloma: updated follow-up and improved survival. J Clin Oncol 2014; 32: 634–640.
Mateos MV, Leleu X, Palumbo A, San Miguel JF . Initial treatment of transplant-ineligible patients in multiple myeloma. Expert Rev Hematol 2014; 7: 67–77.
Rollig C, Knop S, Bornhauser M . Multiple myeloma. Lancet 2015; 385: 2197–2208.
Mateos MV, Ocio EM, Paiva B, Rosinol L, Martinez-Lopez J, Blade J et al. Treatment for patients with newly diagnosed multiple myeloma in 2015. Blood Rev 2015; 29: 387–403.
Fulciniti M, Munshi NC, Martinez-Lopez J . Deep response in multiple myeloma: a critical review. BioMed Res Int 2015; 2015: Article ID 832049.
Lonial S, Anderson KC . Association of response endpoints with survival outcomes in multiple myeloma. Leukemia 2014; 28: 258–268.
Gay F, Larocca A, Wijermans P, Cavallo F, Rossi D, Schaafsma R et al. Complete response correlates with long-term progression-free and overall survival in elderly myeloma treated with novel agents: analysis of 1175 patients. Blood 2011; 117: 3025–3031.
Barlogie B, Mitchell A, van Rhee F, Epstein J, Morgan GJ, Crowley J . Curing myeloma at last: defining criteria and providing the evidence. Blood 2014; 124: 3043–3051.
Paiva B, van Dongen JJ, Orfao A . New criteria for response assessment: role of minimal residual disease in multiple myeloma. Blood 2015; 125: 3059–3068.
Paiva B, Gutierrez NC, Rosinol L, Vidriales MB, Montalban MA, Martinez-Lopez J et al. High-risk cytogenetics and persistent minimal residual disease by multiparameter flow cytometry predict unsustained complete response after autologous stem cell transplantation in multiple myeloma. Blood 2012; 119: 687–691.
Paiva B, Martinez-Lopez J, Vidriales MB, Mateos MV, Montalban MA, Fernandez-Redondo E et al. Comparison of immunofixation, serum free light chain, and immunophenotyping for response evaluation and prognostication in multiple myeloma. J Clin Oncol 2011; 29: 1627–1633.
Paiva B, Montalbán MA, Puig N, Cordon L, Martinez-Lopez J, Ocio EM et al. Clinical significance of sensitive Flow-MRD monitoring in elderly multiple myeloma patients on the Pethema/GEM2010MAS65 Trial. Blood 2014; 124: 3390.
Rawstron AC, Child JA, de Tute RM, Davies FE, Gregory WM, Bell SE et al. Minimal residual disease assessed by multiparameter flow cytometry in multiple myeloma: impact on outcome in the Medical Research Council Myeloma IX Study. J Clin Oncol 2013; 31: 2540–2547.
Rawstron AC, Davies FE, DasGupta R, Ashcroft AJ, Patmore R, Drayson MT et al. Flow cytometric disease monitoring in multiple myeloma: the relationship between normal and neoplastic plasma cells predicts outcome after transplantation. Blood 2002; 100: 3095–3100.
Rawstron AC, Gregory WM, de Tute RM, Davies FE, Bell SE, Drayson MT et al. Minimal residual disease in myeloma by flow cytometry: independent prediction of survival benefit per log reduction. Blood 2015; 125: 1932–1935.
Robillard N, Bene MC, Moreau P, Wuilleme S . A single-tube multiparameter seven-colour flow cytometry strategy for the detection of malignant plasma cells in multiple myeloma. Blood Cancer J 2013; 3: e134.
Paiva B, Cedena MT, Puig N, Arana P, Vidriales MB, Cordon L et al. Minimal residual disease monitoring and immune profiling in multiple myeloma in elderly patients. Blood 2016; 127: 3165–3174.
Puig N, Sarasquete ME, Balanzategui A, Martinez J, Paiva B, Garcia H et al. Critical evaluation of ASO RQ-PCR for minimal residual disease evaluation in multiple myeloma. A comparative analysis with flow cytometry. Leukemia 2014; 28: 391–397.
Sarasquete ME, Garcia-Sanz R, Gonzalez D, Martinez J, Mateo G, Martinez P et al. Minimal residual disease monitoring in multiple myeloma: a comparison between allelic-specific oligonucleotide real-time quantitative polymerase chain reaction and flow cytometry. Haematologica 2005; 90: 1365–1372.
Ladetto M, Donovan JW, Harig S, Trojan A, Poor C, Schlossnan R et al. Real-Time polymerase chain reaction of immunoglobulin rearrangements for quantitative evaluation of minimal residual disease in multiple myeloma. Biol Blood Marrow Transplant 2000; 6: 241–253.
Puig N, Sarasquete ME, Alcoceba M, Balanzategui A, Chillon MC, Sebastian E et al. Kappa deleting element as an alternative molecular target for minimal residual disease assessment by real-time quantitative PCR in patients with multiple myeloma. Eur J Haematol 2012; 89: 328–335.
Martinez-Lopez J, Lahuerta JJ, Pepin F, Gonzalez M, Barrio S, Ayala R et al. Prognostic value of deep sequencing method for minimal residual disease detection in multiple myeloma. Blood 2014; 123: 3073–3079.
Mailankody S, Korde N, Lesokhin AM, Lendvai N, Hassoun H, Stetler-Stevenson M et al. Minimal residual disease in multiple myeloma: bringing the bench to the bedside. Nat Rev Clin Oncol 2015; 12: 286–295.
Kumar S, Paiva B, Anderson KC, Durie B, Landgren O, Moreau P et al. International Myeloma Working Group consensus criteria for response and minimal residual disease assessment in multiple myeloma. Lancet Oncol 2016; 17: e328–e346.
van Dongen JJ, Langerak AW, Bruggemann M, Evans PA, Hummel M, Lavender FL et al. Design and standardization of PCR primers and protocols for detection of clonal immunoglobulin and T-cell receptor gene recombinations in suspect lymphoproliferations: report of the BIOMED-2 Concerted Action BMH4-CT98-3936. Leukemia 2003; 17: 2257–2317.
Ladetto M, Bruggemann M, Monitillo L, Ferrero S, Pepin F, Drandi D et al. Next-generation sequencing and real-time quantitative PCR for minimal residual disease detection in B-cell disorders. Leukemia 2014; 28: 1299–1307.
Bai Y, Wong KY, Fung TK, Chim CS . High applicability of ASO-RQPCR for detection of minimal residual disease in multiple myeloma by entirely patient-specific primers/probes. J Hematol Oncol 2016; 9: 107.
Martinez-Lopez J, Fernandez-Redondo E, Garcia-Sanz R, Montalban MA, Martinez-Sanchez P, Pavia B et al. Clinical applicability and prognostic significance of molecular response assessed by fluorescent-PCR of immunoglobulin genes in multiple myeloma. Results from a GEM/PETHEMA study. Br J Haematol 2013; 163: 581–589.
Rawstron AC, de Tute RM, Haughton J, Owen RG . Measuring disease levels in myeloma using flow cytometry in combination with other laboratory techniques: lessons from the past 20 years at the leeds haematological malignancy diagnostic service. Cytometry B Clin Cytom 2016; 90: 54–60.
van Dongen JJ, van der Velden VH, Bruggemann M, Orfao A . Minimal residual disease diagnostics in acute lymphoblastic leukemia: need for sensitive, fast, and standardized technologies. Blood 2015; 125: 3996–4009.
Flanders A, Stetler-Stevenson M, Landgren O . Minimal residual disease testing in multiple myeloma by flow cytometry: major heterogeneity. Blood 2013; 122: 1088–1089.
Salem D, Stetler-Stevenson M, Yuan C, Landgren O . Myeloma minimal residual disease testing in the United States: evidence of improved standardization. Am J Hematol 2016; 91: E502–E503.
Stetler-Stevenson M, Paiva B, Stoolman L, Lin P, Jorgensen JL, Orfao A et al. Consensus guidelines for myeloma minimal residual disease sample staining and data acquisition. Cytometry B Clin Cytom 2016; 90: 26–30.
Rawstron AC, Orfao A, Beksac M, Bezdickova L, Brooimans RA, Bumbea H et al. Report of the European Myeloma Network on multiparametric flow cytometry in multiple myeloma and related disorders. Haematologica 2008; 93: 431–438.
Arroz M, Came N, Lin P, Chen W, Yuan C, Lagoo A et al. Consensus guidelines on plasma cell myeloma minimal residual disease analysis and reporting. Cytometry B Clin Cytom 2016; 90: 31–39.
Pedreira CE, Costa ES, Barrena S, Lecrevisse Q, Almeida J, van Dongen JJ et al. Generation of flow cytometry data files with a potentially infinite number of dimensions. Cytometry A 2008; 73: 834–846.
Kalina T, Flores-Montero J, van der Velden VH, Martin-Ayuso M, Bottcher S, Ritgen M et al. EuroFlow standardization of flow cytometer instrument settings and immunophenotyping protocols. Leukemia 2012; 26: 1986–2010.
van Dongen JJ, Lhermitte L, Bottcher S, Almeida J, van der Velden VH, Flores-Montero J et al. EuroFlow antibody panels for standardized n-dimensional flow cytometric immunophenotyping of normal, reactive and malignant leukocytes. Leukemia 2012; 26: 1908–1975.
Pedreira CE, Costa ES, Lecrevisse Q, van Dongen JJ, Orfao A, EuroFlow C . Overview of clinical flow cytometry data analysis: recent advances and future challenges. Trends Biotechnol 2013; 31: 415–425.
Maecker HT, Frey T, Nomura LE, Trotter J . Selecting fluorochrome conjugates for maximum sensitivity. Cytometry A 2004; 62: 169–173.
Flores-Montero J, de Tute R, Paiva B, Perez JJ, Bottcher S, Wind H et al. Immunophenotype of normal vs. myeloma plasma cells: Toward antibody panel specifications for MRD detection in multiple myeloma. Cytometry B Clin Cytom 2016; 90: 61–72.
Orfao De Matos Correia, e Vale A, Sobral da Costa E, Pedreira CE Method for the multidimensional detection of aberrant phenotypes in neoplastic cells to be used to monitor minimal disease levels using flow cytometry, European patent 1724581B1 (2014).
Orfao De Matos A, Pedreira CE, Sobral Da Costa E Generation of flow cytometry datafiles with a potentially infinite number of dimensions derived from the fusion of a group of separate flow cytometry datafiles and their multidimensional reconstruction with both actually measured and estimated flow cytometry data, European patent EP1770387B1 (2011).
Pedreira CE, Costa ES, Almeida J, Fernandez C, Quijano S, Flores J et al. A probabilistic approach for the evaluation of minimal residual disease by multiparameter flow cytometry in leukemic B-cell chronic lymphoproliferative disorders. Cytometry A 2008; 73A: 1141–1150.
Pojero F, Flores-Montero J, Sanoja L, Perez JJ, Puig N, Paiva B et al. Utility of CD54, CD229, and CD319 for the identification of plasma cells in patients with clonal plasma cell diseases. Cytometry B Clin Cytom 2016; 90: 91–100.
Paiva B, Puig N, Garcia-Sanz R, San Miguel JF . Is this the time to introduce minimal residual disease in multiple myeloma clinical practice? Clin Cancer Res 2015; 21: 2001–2008.
Rawstron AC, Paiva B, Stetler-Stevenson M . Assessment of minimal residual disease in myeloma and the need for a consensus approach. Cytometry B Clin Cytom 2016; 90: 21–25.
Silvennoinen R, Lundan T, Kairisto V, Pelliniemi TT, Putkonen M, Anttila P et al. Comparative analysis of minimal residual disease detection by multiparameter flow cytometry and enhanced ASO RQ-PCR in multiple myeloma. Blood Cancer J 2014; 4: e250.
Wood BL . Principles of minimal residual disease detection for hematopoietic neoplasms by flow cytometry. Cytometry B Clin Cytom 2016; 90: 47–53.
Royston DJ, Gao Q, Nguyen N, Maslak P, Dogan A, Roshal M . Single-Tube 10-fluorochrome analysis for efficient flow cytometric evaluation of minimal residual disease in plasma cell myeloma. Am J Clin Pathol 2016; 146: 41–49.
This work has been supported by the International Myeloma Foundation-Black Swan Research Initiative, the Red Temática de Investigación Cooperativa en Cáncer (RTICC); grant SA079U14 from the Consejería de Educación, Junta de Castilla y León, Valladolid, Spain and; grant DTS15/00119 from Instituto de Salud Carlos III, Ministerio de Economía y Competitividad, Madrid, Spain.
G-EG and RF are employees of Cytognos SL, Salamanca, Spain. JJMvD received research support and traveling support from Roche, Amgen and Becton Dickinson. SB received research support from Roche, Celgene, Becton Dickinson and AbbVie as well as honoraria from Roche and AbbVie. JF-M, LS-F, OG-S, J-JP-M, AC-M, CJ, JM-L, M-VM, VvdV, JC, LS, NP, M-BV, RGS, MG, RP, M-CdC, JB, J-JL, CA, AB, AG-M, JL, PL, CA-S, JS-M, BD and AO declare no conflict of interest.
Supplementary Information accompanies this paper on the Leukemia website
About this article
Bone Marrow Transplantation (2019)
High-Risk Multiple Myeloma: Integrated Clinical and Omics Approach Dissects the Neoplastic Clone and the Tumor Microenvironment
Journal of Clinical Medicine (2019)
The Minimal Residual Disease in Non-Hodgkin's Lymphomas: From the Laboratory to the Clinical Practice
Frontiers in Oncology (2019)
EuroFlow Lymphoid Screening Tube (LST) data base for automated identification of blood lymphocyte subsets
Journal of Immunological Methods (2019)