Tumour growth is accompanied by gene and/or protein changes that may lead to peroxidation of the cell membrane species and, hence, to the emission of volatile organic compounds (VOCs). In this study, we investigated the ability of a nanosensor array to discriminate between breath VOCs that characterise healthy states and the most widespread cancer states in the developed world: lung, breast, colorectal, and prostate cancers.
Exhaled alveolar breath was collected from 177 volunteers aged 20–75 years (patients with lung, colon, breast, and prostate cancers and healthy controls). Breath from cancerous subjects was collected before any treatment. The healthy population was healthy according to subjective patient's data. The breath of volunteers was examined by a tailor-made array of cross-reactive nanosensors based on organically functionalised gold nanoparticles and gas chromatography linked to the mass spectrometry technique (GC-MS).
The results showed that the nanosensor array could differentiate between ‘healthy’ and ‘cancerous’ breath, and, furthermore, between the breath of patients having different cancer types. Moreover, the nanosensor array could distinguish between the breath patterns of different cancers in the same statistical analysis, irrespective of age, gender, lifestyle, and other confounding factors. The GC-MS results showed that each cancer could have a unique pattern of VOCs, when compared with healthy states, but not when compared with other cancer types.
The reported results could lead to the development of an inexpensive, easy-to-use, portable, non-invasive tool that overcomes many of the deficiencies associated with the currently available diagnostic methods for cancer.
Cancer kills more than seven million people every year. The most widespread cancers in the developed world, accounting for half of the cancer deaths, are prostate, lung, and colorectal cancers for men, and breast, colorectal, and lung cancers for women (Culter, 2008; Jemal et al, 2008). The prognosis of a cancer patient improves considerably if the disease is discovered at an early stage, when the tumour is still localised (Hirsch et al, 2001; Kalogerakos et al, 2008). However, early detection is possible only through widespread and screening, because of the asymptomatic onset of the disease. The currently available techniques do not always fulfill the requirements for reliable discrimination between cancer patients and healthy subjects. For instance, the impact of computed tomography screening on lung cancer mortality and the benefit of the prostate-specific antigen blood test for prostate cancer screening are uncertain (Culter, 2008). Screening methods such as colonoscopy for colorectal cancer and mammography for breast cancer are very reliable, but these methods are far from ideal. Colonoscopy is invasive and extremely unpleasant to the patient (Culter, 2008; Jemal et al, 2008). Mammography uses X-rays which may cause radiation-induced mutations. In addition, image quality depends on the breast structure, so that it is not applicable to certain segments of the screening population, such as young at-risk women (young women with a family history of breast cancer and/or BRCA mutations) (Bermejo-Perez et al, 2008) and women with dense breast tissue (∼25% of women over the age of 40 years) (Mandana, 2003; Andrew and Srinivasan, 2008).
The analysis of volatile organic compounds (VOCs) that are linked to cancer is a new frontier in medical diagnostics because it is non-invasive and potentially inexpensive (Amann et al, 2007; Mazzone, 2008). The principle behind this approach is based on cell biology. In particular tumour growth is accompanied by gene and/or protein changes that may lead to peroxidation of the cell membrane species and, hence, to the emission of VOCs (Singer and Nicolson, 1972; Kneepkens et al, 1994; Alberts et al, 2002). These VOCs can be detected either directly from the headspace of cancer cells (Bajaj et al, 2009; Barash et al, 2009; Filipiak et al, 2010) or through exhaled breath (see Amann et al, 2007; Ligor et al, 2008; Mazzone, 2008; Bajtarevic et al, 2009; Horvath et al, 2009; Ligor et al, 2009 and bibliography therein), as cancer-related changes in the blood chemistry lead to measurable changes in the breath by exchange through the lung (Horvath et al, 2009). In this study, we report that a tailor-made array of cross-reactive sensors based on organically functionalised gold nanoparticles (GNPs) discriminates between breath VOCs of healthy controls and of patients suffering from lung, breast, colorectal, and prostate cancers. Moreover, we report on the ability of these GNP sensors to distinguish between the breath patterns of different cancer types in the same pattern analysis, irrespective of age, gender, lifestyle, and other confounding factors. We find that these results compare favourably with breath testing via a chemical analysis of the constituent compounds by gas chromatography linked with mass spectroscopy (GC-MS). The results reported in this study could lead to the development of a cost-effective, easy-to-use, portable, non-invasive tool that overcomes many of the deficiencies associated with the currently available techniques used for the diagnosis of cancer.
Materials and methods
Breath samples were collected after signed consent from 96 volunteers aged 30–75 years, who had not ingested coffee or alcohol for at least 12 h. These were 30 primary lung cancer patients, 26 primary colon cancer patients, 22 primary breast cancer patients, and 18 primary prostate cancer patients and 22 healthy controls. All cancer patients were tested directly after being diagnosed by conventional clinical means (various imaging techniques to locate the tumour and biopsy for final diagnosis) and before chemotherapy and/or other cancer treatments. No breath collection was carried out in the 4 days after biopsy. The 22 healthy controls matched the tested cancer patients in age and lifestyle. The clinical characteristics of the study population are listed in Table 1. Additional breath samples were collected from 59 healthy volunteers, aged 20–75 years, for studying the effect of various confounding factors (Table 2). All experiments were approved by and performed according to the guidelines of Technion's committee for supervision of human experiments (Haifa, Israel).
Exhaled breath was collected in a controlled manner from individuals with cancer in the same room/atmosphere and from healthy subjects. The inhaled air was cleared of ambient contaminants by repeatedly inhaling to total lung capacity for 3–5 min through a mouthpiece (purchased from Eco Medics, Duerten, Switzerland) that contains a filter cartridge on the inspiratory port, thus removing >99.99% of exogenous VOCs from the air during inspiration. Immediately after the lung washout, subjects exhaled through a separate exhalation port of the mouthpiece against 10–15 cm H2O pressure to ensure closure of the vellum so that nasal entrainment of gas is excluded. Complementary experiments optimising the breath collection procedure have shown that the sampling methodology simply measures alveolar breath uncontaminated by upper airways release and exogenous compounds. Exhaled breath is a mixture of alveolar air and respiratory dead space air. The dead space was automatically filled into a separate bag and the alveolar breath into a 750 ml Mylar sampling bag (purchased from Eco Medics). It should be emphasised that the described breath collection is a single-step process that does not require the volunteer to take care of changing between the dead space and alveolar breath bags. The Mylar bags used in this study are made from polyvinyl fluoride, which is chemically inert with respect to most compounds in the breath. The Mylar bags were re-used and thoroughly cleaned before each use with flowing N2 (99.999% purity) gas for 5–8 min (Notably, GC-MS/solid-phase microextraction (SPME) has shown that this purification process eliminates >98% of contaminants and/or VOCs from the previous sample tested in a specific Mylar bag.). At least two bags were collected per test person for analysis with GC-MS and/or for analysis with a nanosensor array based on organically functionalised GNPs – see below. All bags were analysed within 4 days from the time of breath collection, much before the allowed 3-weeks storage period, after which, according to our control experiments, the samples might start to deteriorate.
Design and fabrication of the nanosensor array
Monolayer-capped 5 nm GNPs were synthesised by a modified two-phase method (Brust et al, 1995), combined, if necessary, with a ligand-exchange procedure. Dodecanethiol, 4-methoxy-toluenethiol, hexanethiol, 11-mercapto-1-undecanol, decanethiol, octadecanethiol, tert-dodecanethiol, 1-butanethiol, 2-ethyl-hexanethiol, 3-methyl-1-butanethiol, 2-mercaptobenzoxazole, 11-mercapto-1-undecanol, 2-mercapto-benzyl alcohol, and 3-methyl-1-butanethiol (all purchased from Sigma-Aldrich, Rehovot, Israel) were used as organic capping layers. The GNP synthesis was described in detail in the study by Peng et al (2009). In all, 10 pairs of circular inter-digitated gold electrodes were deposited by an electron-beam evaporator TFDS-870 (VST Ltd, Petah Tiqwa, Israel) on device quality silicon wafers capped with 300 nm thermal oxide (purchased from Silicon Quest International Inc., Santa Clara, CA, USA). The outer diameter of the circular electrode area was 3000 μm; the gaps between two adjacent electrodes and the width of each electrode were both 20 μm. The functionalised GNPs were dispersed in toluene by sonication and drop cast onto the electrodes. While still coated with solution, the substrate was blown dry with N2. This process was repeated several times to yield the desired resistance of ∼1 MΩ. The device was dried for 2 h at room temperature in an ambient atmosphere and then baked overnight at 50°C in a vacuum oven. Figure 1A shows a schematic representation of the device. Overall, 14 GNP sensors with different organic functionalities were mounted onto a custom PTFE circuit board to form the nanosensor array.
Breath testing using the GNP sensor array
The developed nanosensor array was placed inside a home-made stainless steel test chamber with a volume of <100 cm3. The sampling system first evacuated the chamber, then delivered a pulse of breath from the sample bags to the GNP sensors, and then evacuated the chamber again. In a typical experiment, signals of the GNP sensor array were collected for 5 min in vacuum, then for 5 min during breath exposure, and then again for 5 min in vacuum. The cycles were repeated 3–5 times to test reproducibility. Each sensor of the array underwent a reversible change in electrical resistance when exposed to the sample breath. The responses of the different sensors were unique because of the chemical diversity of the sensor materials. An Agilent Multifunction switch 34980 (Agilent Technologies Ltd, CA, USA) controlled by USB was used to select the active GNP sensor and measure the corresponding resistance at a given time. The electrical resistance of the GNP sensors was measured using a device analyser using integration times that span at least two power line cycles. The entire system was computer controlled.
Signals collected from all sensors in the array were analysed using standard principal component analysis (PCA) and cluster analysis (Peng et al, 2008, 2009; Roeck et al, 2008). Principal component analysis is an effective method to reduce multidimensional data space to its main components and, therefore improves the human perception ability of the data. It determines the linear combinations of sensor values so that the maximum variance between all data points can be obtained in mutually orthogonal dimensions. This results in the largest variance between sensor values from the first principal component and produces decreasing magnitudes of variance from the second to the third principle components and so on.
Breath testing by GC-MS
The collected breath was chemically analysed, that is, its constituent compounds were identified, using GC-MS (GC-6890N; MS-5975; Agilent Technologies Ltd), combined with SPME. The SPME was used for pre-concentrating VOCs in breath samples. A manual SPME holder with an extraction fibre was inserted into the Mylar bag for 20–30 min before being delivered to the GC-MS. We used a commercially available fibre with polydimethylsiloxane-divinylbenzene coating obtained from Sigma-Aldrich. The extracted fibre in the manual SPME holder was inserted into the injector of the GC (splitless mode). We used the following oven profile: 60°C, 2 min, 8°C per min to 100°C, 15°C per min to 120°C, 8°C per min to 180°C, 15°C per min to 200°C, 8°C per min to 225°C. A capillary column H5-5MS 5% phenyl methyl siloxane (30 m length, 0.25 mm i.d., 0.25 μm thickness, from Agilent Technologies Ltd) was used. The column pressure was set at 8.22 p.s.i., and the initial flow rate was 1.0 ml per min. The molecular structures of VOCs were determined by a spectral library match (and not by retention time using calibration mixtures) using the standard modular set. The VOCs common for >80% of the healthy and cancer samples, as well as their abundance with experimental error, were identified using the Automated Mass Spectral Deconvolution and Identification System (AMDIS) software. Out of the common VOCs, those that showed no overlap in abundance (taking into account their experimental errors) between the studied groups were determined. The PCA algorithm was applied to this choice of VOCs.
Results and Discussion
This study was conducted in four phases. In the first phase, exhaled alveolar breath was collected from 177 volunteers aged 20–75 years (lung, colon, breast, and prostate cancer patients and healthy controls; Tables 1 and 2), using an ‘offline’ method described by Peng et al (2009). This method effectively separates endogenous VOCs in the breath from exogenous VOCs, and excludes nasal entrainment. Endogenous VOCs are generated by cellular biochemical processes in the body and, thus, may provide insight into the body's function (Kneepkens et al, 1994; Baubach et al, 2005), whereas exogenous VOCs are adsorbed from the environment (Baubach et al, 2005). All cancer patients were tested immediately after being diagnosed by conventional clinical means, and before any treatment.
In the second phase, we designed an array of 14 cross-reactive nanosensors, based on 5-nm GNPs (Haick, 2007; Barash et al, 2009; Peng et al, 2009) with different organic functionalities (see the ‘Materials and Methods’ section), and tested its feasibility for simultaneous detection of primary lung, colon, breast, and prostate cancers. Figure 1A shows a schematic representation of the device. Arrays of broadly cross-reactive GNP sensors are ideally suited to trace the cancer odour prints directly, without identifying the constituent VOCs (Roeck et al, 2008). The GNP sensors combine the robustness and processing ease of inorganic materials with the sensitivity and chemical selectivity of certain organic molecules to cancer biomarker VOCs (Haick, 2007; Jain et al, 2007). Some of the GNP sensors have detection limits of 1–5 p.p.b. to typical cancer biomarkers (Peng et al, 2009), whereas the others have a detection limit down to ∼10 p.p.t. The apparent detection limit and sensitivity of these sensors were demonstrated for different compounds elsewhere (Barash et al, 2009; Peng et al, 2009). Every sensor of the array undergoes a rapid and fully reversible change in electrical resistance when exposed to the breath sample. Some of the GNP sensors show a positive resistance change, others show a negative resistance change (Figure 1B–D). The sensing mechanism in GNP thin films is still subject to controversy (Haick, 2007). In most cases, the responses are unique because of the chemical diversity of the sensor materials. Each sensor responds to various odourants, that is, to an individual subset of VOCs in a breath sample. This method distinguishes the odour print resulting from the constituent VOCs directly, without identifying the individual compounds. This black-box approach is essentially different from the chemical analysis of a breath sample by GC-MS or related methods (see below). The subtle differences in the chemical composition that distinguish the breath of healthy individuals, and lung, breast, colorectal, and prostate cancer patients from one another lead to measurable changes in the responses of the individual GNP chemiresistors (Figure 1B–D).
Signals collected from the 14-sensors-array after exposure to the breath of representative subjects from the 5 different test groups were analysed using standard PCA (Roeck et al, 2008; Peng et al, 2009). Principal component analysis is an effective method to improve human perception of experimental data by reducing the multidimensional data space to its main components (for more details, see the ‘Materials and Methods’ section). Figure 2 shows the maps of the first two principal components (PC1 and PC2) for each signal, which account for >88% variance. We analysed (1) lung cancer patients and healthy subjects only, (2) colon cancer patients and healthy subjects only, (3) breast cancer patients and healthy subjects only, (4) prostate cancer patients and healthy subjects only, and (5) lung, breast, colorectal, and prostate cancer patients and healthy subjects together (patient data in Table 1). A very good separation was achieved in the principal component space between the patterns of healthy controls and of patients suffering from lung, colon, and breast cancers, as seen in Figure 2A–C. Complementary analysis has not shown any correlation between the classification quality and the age of the sensors. The cluster of prostate cancer patients (Figure 2D) slightly overlaps with the healthy cluster, probably owing to the fact that many of the prostate cancer test groups have stage I cancer. Moreover, we were able to distinguish between the patterns of the four different cancers in the same principal component plot, with only minimal overlap between the prostate cancer and healthy clusters, as seen in Figure 2E. Notably, although we used the same study populations in Figure 2A–E, the clusters appear differently in principal component space, because the input data for the statistical algorithm is different. The unambiguous identification of the cancer site is a major step towards the development of a robust breath test for cancer. This study was conducted using a relatively small test population, and therefore constitutes only a proof of concept. A wider clinical study would be necessary to confirm the criteria for distinguishing different cancers, and to establish criteria for distinguishing cancer subtypes and stages (Rubin et al, 1997). It is reasonable to expect that clusters for a larger study population would be less well defined and some overlap would occur. In turn, cluster separation could be improved again by developing the GNP sensor array further while maintaining minimum sensitivity to potential confounding effects of common breath metabolites (such as ammonia (CAS: 7664-41-7), acetone (CAS: 67-64-1), methanol (CAS: 67-56-1), and ethanol (CAS: 64-17-5)), which, based on GC-MS/SPME, were found in approximately 20–30% of the examined patients. The iterative improvement in the sensor array while expanding the study population would eventually yield an optimised device. The use of organically functionalised GNPs allows tailoring the properties of the constituent sensors to tune their sensitivity to cancer VOCs.
In the third phase, we compared our proposed method of breath testing using a GNP sensor array with breath analysis through GC-MS. GC-MS is often used for identifying biomarker VOCs of diseases, including breast (Phillips et al, 2003) and lung cancers (Ligor et al, 2008, 2009; Mazzone, 2008; Bajtarevic et al, 2009; Peng et al, 2009), but has not been applied to distinguish between different types of cancer. In this study, we combined GC-MS with SPME. GC-MS identifies the VOCs that can serve as cancer biomarkers in breath samples and determines their relative compositions (Ouyang and Pawliszyn, 2006; Amorimb and Cardeal, 2007). SPME is a convenient method to pre-concentrate cancer biomarker VOCs in the breath samples, because GC-MS per se is not sensitive enough to concentrations below the p.p.m. (part-per-million) level. It must be noted that solid-phase extracts include only part of the constituent compounds, and, therefore selectively enhance the signals of certain biomarkers, whereas it potentially misses others (Barash et al, 2009). Hence, it is likely that not all of the cancer biomarker VOCs in the breath samples, to which the array of GNP sensors responded, were identified. For example, there is some evidence that formaldehyde is elevated in the urine (and hence blood/breath) of prostate cancer patients (Spanel et al, 1999) and nitrosamines in the flatulence (and perhaps in breath as is gut-produced methane) of bowel cancer patients (Amann et al, 2007). These may be the major compounds that are activating the sensors but they are not picked up by GC-MS/SPME.
Typically, our GC-MS/SPME analysis identifies 300–400 different VOCs per breath sample (Peng et al, 2009), with >87% reproducibility for a specific volunteer that was examined multiple times over a period of 6 months. Using forward stepwise discriminant analysis, we identified a number of VOCs, which are present in >80% of lung cancer patients and healthy subjects (33 VOCs) (Peng et al, 2009), colon cancer patients and healthy subjects (39 VOCs), breast cancer patients and healthy subjects (54 VOCs), and prostate cancer patients and healthy subjects (36 VOCs). Some of these common VOCs occur in distinctly different concentration ratios, so that the change in their overall composition can be used as a cancer biomarker (Peng et al, 2009). Owing to the large experimental scatter in their abundance, not all common VOCs are suitable for distinguishing between healthy controls and cancer patients. We selected the most suitable ones, for which there was no overlap in abundance between healthy controls and cancer patients – six VOCs for lung cancer, six VOCs for colon cancer, five VOCs for breast cancer, and four VOCs for prostate cancer (see Figure 3A–D and figure caption therein). To the best of our knowledge, most of the VOCs reported in this study appear for the first time in the literature. The exceptions are the two lung cancer VOCs: VOC 1 (1-methyl-4-(1-methylethyl)benzene; m/z=119; CAS: 99-87-6), which was reported earlier by Phillips et al (1999a, 1999b, Poli et al (2005), and Ligor et al (2009); and VOC 2 (toluene; m/z=91; CAS: 108-88-3), which was reported earlier by Poli et al (2005) and Ligor et al (2009).
From the data presented in Figure 3, we determined the characteristic odour prints of the studied cancer types through PCA and cluster analysis (see Supporting Information, Supplementary Figure S1A–D). As can be seen in the figure, good discrimination can be achieved between lung cancer patients and healthy subjects (Supporting Information, Supplementary Figure S1A) and between breast cancer patients and healthy subjects (Supporting Information, Supplementary Figure S1C). The clusters for prostate cancer patients and healthy subjects are closer together and the signature of one prostate cancer patient falls into the healthy cluster (Supporting Information, Supplementary Figure S1D). The sensitivity of this method to colon cancer is slightly lower: ∼30% of the colon cancer patient signatures are found in the healthy cluster (Supporting Information, Supplementary Figure S1B). It must be noted that the GNP array achieved the best separation for colon cancer patients and healthy subjects, illustrating the fundamental difference in the two approaches (Kushch et al, 2008). While achieving reasonable discrimination between healthy controls and patients suffering from one specific cancer, we were unable to determine suitable representative VOCs for distinguishing all cancers. This is because the GC-MS analysis did not yield compounds without overlap in abundance for lung, colon, breast, and prostate cancers and healthy test groups together. To achieve separation between the five study groups, we performed the PCA using all 16 VOCs used before for the successful separation of each cancer type from healthy controls, which were found in >80% of the breath samples in each study group (Figure 3E). It must be noted that one compound used to distinguish between breast cancer and healthy states (decane 2,3,4-trimethyl; CAS: 62238-15-7), was found in <80% of the breath samples in one or more of the other study groups. Hence, it was not considered in this study. Supplementary Figure S1E of the Supporting Information shows that the clusters of the five study groups are also strongly overlapping for the second set of selected VOCs. It must be noted that for any other arbitrary choice of representative VOCs, the abundances for the five study groups would overlap. Hence, it is not reasonable to expect a better discrimination through another choice of representative VOCs.
Several studies using GC-MS have shown that age, gender, lifestyle, nutrition, medication, smoking habits, and other confounding factors affect the chemical composition of the breath (Mendis et al, 1994; Lechner et al, 2006; Kushch et al, 2008) and, as a result, the sensitivity and specificity of the diagnosis. Therefore we tested, in the fourth phase of this study, whether the same applies for GNP sensors array as well. We examined breath samples from a heterogeneous group of 59 healthy volunteers aged 20–75 years (Table 2). We found that none of the following confounding factors affected the sensor array output: gender, age, ethnic origin, family cancer history, intake of food additives, drug treatment, exposure to environmental toxins, and smoking habits (Figure 4). In contrast, we could trace and identify by GC-MS/SPME a multitude of VOCs, which can be linked to these confounding factors. For example, 2,6,11-trimethyl-dodecane (CAS: 31295-56-4) was found in 80% of the males, but in none of the females participating in the test; 3,7-dimethyl-undecane (CAS: 17301-29-0) was found in 100% of the subjects with allergies, but only in 9% of the subjects without allergies. The presence of various non-cancer-related breath VOCs in a heterogeneous study population might obscure cancer biomarker VOCs, and might be in part responsible for the observed inferior discriminative power of GC-MS/SPME compared with the GNP sensor array.
Summary and Conclusions
Breath analysis with a GNP sensor array compared favourably with breath analysis using GC-MS because: (1) no pre-treatment of the breath samples, such as pre-concentration or de-humidification, is required; (2) the test itself is fast, easy to carry out, and does not necessarily require a trained operator; (3) no time-consuming pre-selection of the experimental data is necessary before the statistical analysis to achieve separation between the tested groups; (4) the different cancers form clusters well separated from one another and from the healthy clusters; and (5) the GNP sensor array is insensitive to various confounding factors, which strongly affect the chemical composition of the breath. The VOC signatures of the different types of cancer, with almost no overlapping among them, could lead to the development of a cost-effective, easy-to-use, portable, and non-invasive diagnostic tool for detecting lung, breast, colorectal, and prostate cancers through a single breath test, which could have a significant impact on cancer mortality through improved widespread screening. The same tool might also help to distinguish between primary cancer vs metastases, rapidly monitor therapy success and response to the administered treatment, as well as to detect early recurrence in an annual routine survey.
Alberts B, Johnson A, Lewis J (2002) Molecular Biology of the Cell. Garland Publishing: New York
Amann A, Spanel P, Smith D (2007) Breath analysis: the approach towards clinical applications. Mini Rev Med Chem 7: 115–129
Amorimb LCA, Cardeal ZL (2007) Breath air analysis and its use as a biomarker in biological monitoring of occupational and environmental exposure to chemical agents. J Chromatogr B 853: 1–9
Andrew K, Srinivasan V (2008) Breast cancer imaging: a perspective for the next decade. Med Phys 35: 4878–4897
Bajaj A, Miranda OR, Kim I-B, Phillips RL, Jerry DJ, Bunz UHF, Rotello VM (2009) Detection and differentiation of normal, cancerous, and metastatic cells using nanoparticle-polymer sensor arrays. Proc Natl Acad Sci USA 106: 10912–10916
Bajtarevic A, Ager C, Pienz M, Klieber M, Schwarz K, Ligor M, Ligor T, Filipiak W, Denz H, Fiegl M, Hilbe W, Weiss W, Lukas P, Jamnig H, Hackl M, Haidenberger A, Buszewski B, Miekisch W, Schubert J, Amann A (2009) Noninvasive detection of lung cancer by analysis of exhaled breath. BMC Cancer 9: 348
Barash O, Peled N, Hirsch FR, Haick H (2009) Sniffing the unique ‘odor print’ of non-small-cell lung cancer with gold nanoparticles. Small 5: 2618–2624
Baubach JI, Vautz W, Ruzsanyi V (2005) Metabolites in Human Breath: Ion Mobility Spectrometers as Diagnostic Tools for Lung Diseases. Breath Analysis for Clinical Diagnosis and Therapeutic Monitoring. World Scientific Publishing Co. Pte. Ltd: Toh Tuck Link, Singapore
Bermejo-Perez MJ, Marquez-Calderon S, Llanos-Mendez A (2008) Cancer surveillance based on imaging techniques in carriers of BRCA1/2 gene mutations: a systematic review. Br J Rad 81: 172–179
Brust M, Fink J, Bethell D, Schiffrin DJ, Kiely CJ (1995) Synthesis and reactions of functionalised gold nanoparticles. J Chem Soc Chem Commun, 1655–1656
Culter DM (2008) Are we finally winning the war on cancer? J Eco Perspec 22: 3–26
Filipiak W, Sponring A, Filipiak A, Ager C, Schubert J, Miekisch W, Amann A, Troppmair J (2010) TD-GC-MS analysis of volatile metabolites of human lung cancer and normal cells in-vitro. Cancer Epidemiol Biomarkers Prev 19: 182–195
Haick H (2007) Chemical sensors based molecularly modified metallic nanoparticles. J Phys D 40: 7173–7186
Hirsch FR, Franklin WA, Gazdar AF, Bunn PAJ (2001) Early detection of lung cancer: clinical perspectives of recent advances in biology and radiology. Clin Cancer Res 7: 5–22
Horvath L, Lazar Z, Gyulai N, Kollai M, Losonczy G (2009) Exhaled biomarkers in lung cancer. Eur Respir J 34: 261–275
Jain PK, El-Sayed IH, El-Sayed MA (2007) Au nanoparticles target cancer. NanoToday 2: 18–29
Jemal A, Siegel R, Ward E, Hao Y, Xu J, Murray T, Thun MJ (2008) Cancer statistics, 2008. CA Cancer J Clin 58: 71–96
Kalogerakos K, Sofoudis C, Baltayiannis N (2008) Early breast cancer: a review. Cancer Ther 6: 463–476
Kneepkens CMF, Lepage G, Roy CC (1994) The potential of the hydrocarbon breath test as a measure of lipid peroxidation. Free Rad Biol Med 17: 127–160
Kushch I, Arendacka B, Stolc S, Mochalski P, Filipiak W, Schwarz K (2008) Breath isoprene–aspects of normal physiology related to age, gender and cholesterol profile as determined in a proton transfer reaction mass spectrometry study. Clin Chem Lab Med 46: 1011–1018
Lechner M, Moser B, Niederseer D, Karlseder A, Holzknecht B, Fuchs M (2006) Gender and age specific differences in exhaled isoprene levels. Respir Physiol Neurobiol 154: 478–483
Ligor M, Ligor T, Bajtarevic A, Ager C, Pienz M, Klieber M, Denz H, Fiegl M, Hilbe W, Weiss W, Lukas P, Jamnig H, Hackl M, Miekisch W, Schubert J, Amann A (2009) Determination of volatile organic compounds in exhaled breath of patients with lung cancer using solid phase microextraction and gas chromatography mass spectrometry. Clin Chem Lab Med 47: 550–560
Ligor T, Ligor M, Amann A, Ager CB, Pienz M, Dzien A, Buszewski B (2008) The analysis of healthy volunteers’ exhaled breath by the use of solid-phase microextraction and GC-MS. J Breath Res 2: 046006/1–046006/8
Mandana V (2003) Breast cancer screening methods: a review of the evidence. Health Care Women Int 24: 773–793
Mazzone PJ (2008) Analysis of volatile organic compounds in the exhaled breath for the diagnosis of lung cancer. J Thorac Oncol 3: 774–780
Mendis S, Sobotka PA, Euler DE (1994) Pentane and isoprene in expired air from humans: gas-chromatographic analysis of single breath. Clin Chem 40: 1485–1488
Ouyang G, Pawliszyn J (2006) SPME in environmental analysis. Anal Bioanal Chem 386: 1059–1073
Peng G, Tisch U, Adams U, Hakim M, Shehada N, Broza YY, Billan S, Abdah-Bortnyak R, Kuten A, Haick H (2009) Diagnosing lung cancer in exhaled breath using gold nanoparticles. Nature Nanotechnol 4: 669–673
Peng G, Trock E, Haick H (2008) Detecting simulated patterns of lung cancer biomarkers by random network of single-walled carbon nanotubes coated with nonpolymeric organic materials. Nano Lett 8 (11): 3631–3635
Phillips M, Cataneo RN, Ditkoff BA, Fisher P, Greenberg J, Gunawardena R, Kwon CS, Rahbari-Oskoui F, Wong C (2003) Volatile markers of breast cancer in the breath. Breast J 9: 184–191
Phillips M, Gleeson K, Hughes JMB, Greenberg J, Cataneo RN, Baker L, McVay WP (1999a) Volatile organic compounds in breath as markers of lung cancer: a cross-sectional study. Lancet 353: 1930–1933
Phillips M, Herrera J, Krishnan S, Zain M, Greenberg J, Cataneo RN (1999b) Variation in volatile organic compounds in the breath of normal humans. J Chrom B Biomed Sci Appl 729: 75–88
Poli D, Carbognani P, Corradi M, Goldoni M, Acampa O, Balbi B, Bianchi L, Rusca M, Mutti A (2005) Exhaled volatile organic compounds in patients with non-small cell lung cancer: cross sectional and nested short-term follow-up study. Respir Res 6: 71
Roeck F, Barsan N, Weimar U (2008) Electronic nose: current status and future trends. Chem Rev 108: 705–725
Rubin P, Chen Y, Brasacchio R (1997) Staging and classification of cancer: a unified approach. In Principles and Practice of Radiation Oncology. Perez CA, Brady LW (eds). Lippincott-Raven: Philadelphia
Singer SJ, Nicolson GL (1972) The fluid mosaic model of the structure of cell membranes. Science 175: 720–731
Spanel P, Smith D, Holland TA, Singary WA, Elder JB (1999) Analysis of formaldehyde in the headspace of urine from bladder and prostate cancer patients using selected ion flow tube mass spectrometry. Rapid Commun Mass Spect 13: 1354–1359
The research was funded by the Marie Curie Excellence Grant of the European Commission's FP6 program (grant no. 1006752). We thank Mrs Orna Barash, Ms Nisreen Shehada, Ms Carmelina Attalah, Ms Irena Dvorkind, and Mr Ophir Marom (all Technion-IIT) for assistance and Dr Nir Peled for useful comments. HH is a Knight of the Order of Academic Palms and holds the Horev Chair for Leaders in Science and Technology.
The authors declare no conflict of interest.
Supplementary Information accompanies the paper on British Journal of Cancer website
About this article
Cite this article
Peng, G., Hakim, M., Broza, Y. et al. Detection of lung, breast, colorectal, and prostate cancers from exhaled breath using a single array of nanosensors. Br J Cancer 103, 542–551 (2010). https://doi.org/10.1038/sj.bjc.6605810
- volatile biomarker
This article is cited by
Scientific Reports (2022)
Nano Research (2022)
Bioprospecting of indigenous biosurfactant-producing oleophilic bacteria for green remediation: an eco-sustainable approach for the management of petroleum contaminated soil
3 Biotech (2022)
Breath analysis based early gastric cancer classification from deep stacked sparse autoencoder neural network
Scientific Reports (2021)